DICOM PS3.16 2021c - Content Mapping Resource

TID 3708 ECG Waveform Information (Current)
Prev | ECG Report Templates | Next

### TID 3708 ECG Waveform Information

The ECG Waveform Information Template provides reference to, and important parameters of, the analyzed waveform.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3708. ECG Waveform Information

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (55111-9, LN, "Current Procedure Descriptions") | 1 | M |  |
2 | > | CONTAINS | WAVEFORM | EV (121112, DCM, "Source of Measurement") | 1 | U |  |
3 | > | CONTAINS | CODE | EV (10:11345, MDC, "Lead System") | 1 | U |  | DCID 3263 “Electrode Placement Values”
4 | > | CONTAINS | TEXT | EV (122142, DCM, "Acquisition Device Type") | 1 | U |  |
5 | > | CONTAINS | TEXT | EV (121122, DCM, "Equipment Identification") | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 1003 “Person Observer Identifying Attributes” | 1 | U |  | Person performing the ECG acquisition
7 | > | CONTAINS | TEXT | EV (121121, DCM, "Room Identification") | 1 | U |  |
8 | > | CONTAINS | DATETIME | EV (122146, DCM, "Procedure DateTime") | 1 | M |  |
9 | > | CONTAINS | NUM | DCID 3690 “ECG Control Variables Numeric” | 1-n | U |  |
10 | > | CONTAINS | TEXT | DCID 3691 “ECG Control Variables Text” | 1-n | U |  |
11 | > | CONTAINS | INCLUDE | DTID 4019 “Algorithm Identification” | 1 | U |  |

Prev | Up | Next
TID 3704 Patient Characteristics for ECG | Home | TID 3713 ECG Global Measurements

DICOM PS3.16 2021c - Content Mapping Resource