DICOM PS3.3 2021b - Information Object Definitions

A.53.4 X-Ray 3D Angiographic Image Functional Group Macros (Current)
Prev | A.53 X-Ray 3D Angiographic Image IOD | Next

### A.53.4 X-Ray 3D Angiographic Image Functional Group Macros

Table A.53-2 specifies the use of the Functional Group Macros used in the Multi-frame Functional Groups Module for the X-Ray 3D Angiographic Image IOD.

Table A.53-2. X-Ray 3D Angiographic Image Functional Group Macros

Functional Group Macro | Section | Usage
--- | --- | ---
Pixel Measures | C.7.6.16.2.1 | M
Frame Content | C.7.6.16.2.2 | M - May not be used as a Shared Functional Group.
Plane Position (Patient) | C.7.6.16.2.3 | M
Plane Orientation (Patient) | C.7.6.16.2.4 | M
Referenced Image | C.7.6.16.2.5 | U
Derivation Image | C.7.6.16.2.6 | C - Required if Image Type (0008,0008) Value 1 equals DERIVED.
Cardiac Synchronization | C.7.6.16.2.7 | U
Frame Anatomy | C.7.6.16.2.8 | M
Pixel Value Transformation | C.7.6.16.2.9 | U
Frame VOI LUT | C.7.6.16.2.10 | M
Real World Value Mapping | C.7.6.16.2.11 | U
Contrast/Bolus Usage | C.7.6.16.2.12 | C - Required if the Enhanced Contrast/Bolus Module is present.
Respiratory Synchronization | C.7.6.16.2.17 | U
X-Ray 3D Frame Type | C.8.21.5.1 | M

#### A.53.4.1 X-Ray 3D Angiographic Image Functional Group Macros Content Constraints

##### A.53.4.1.1 Frame Anatomy Functional Group Macro

For Anatomic Region Sequence (0008,2218) DCID 4042 “XA/XRF Anatomy Imaged” shall be used.

Prev | Up | Next
A.53.3 X-Ray 3D Angiographic Image IOD Module Table | Home | A.54 X-Ray 3D Craniofacial Image IOD

DICOM PS3.3 2021b - Information Object Definitions