DICOM PS3.3 2020e - Information Object Definitions

A.68.3 Surface Scan Mesh IOD Module Table (Current)
Prev | A.68 Surface Scan Mesh IOD | Next

### A.68.3 Surface Scan Mesh IOD Module Table

Table A.68-1 specifies the Modules of the Surface Scan Mesh IOD.

Table A.68-1. Surface Scan Mesh IOD Modules

IE | Module | Reference | Usage
--- | --- | --- | ---
Patient | Patient | C.7.1.1 | M
Clinical Trial Subject | C.7.1.3 | U
Study | General Study | C.7.2.1 | M
Patient Study | C.7.2.2 | U
Clinical Trial Study | C.7.2.3 | U
Series | General Series | C.7.3.1 | M
Clinical Trial Series | C.7.3.2 | U
Optical Surface Scanner Series | C.8.29.1 | M
Frame of Reference | Frame of Reference | C.7.4.1 | M
Equipment | General Equipment | C.7.5.1 | M
Enhanced General Equipment | C.7.5.2 | M
Surface | Surface Mesh | C.27.1 | M
UV Mapping | C.27.6 | U
Scan Procedure | C.8.29.2 | M
Specimen | C.7.6.22 | U
SOP Common | C.12.1 | M

Prev | Up | Next
A.68.2 Surface Scan Mesh IOD Entity-Relationship Model | Home | A.69 Surface Scan Point Cloud IOD

DICOM PS3.3 2020e - Information Object Definitions