DICOM PS3.17 2020e - Explanatory Information

II.3 Device Module (Current)
Prev | II Use of Product Characteristics Attributes in Composite SOP Instances (Informative) | Next

## II.3 Device Module

Table II-3. Device Module Attribute Mapping

Device Module Attribute Name | Tag | Product Characteristics Module Source
--- | --- | ---
Device Sequence | (0050,0010) | --
>Include Table 8.8-1 “Code Sequence Macro Attributes” in PS3.3 | Product Type Code Sequence (0044,0007) > 'Code Sequence Macro'
>Device Length | (0050,0014) | Product Parameter Sequence (0044,0013) > Numeric Value (0040,A30A), where: Product Parameter Sequence > Concept Name Code Sequence (0040,A043) value is (410668003, SCT, "Length") Product Parameter Sequence > Measurement Units Code Sequence (0040,08EA) is (mm, UCUM, "mm")
>Device Diameter | (0050,0016) | Product Parameter Sequence (0044,0013) > Numeric Value (0040,A30A), where: Product Parameter Sequence > Concept Name Code Sequence (0040,A043) value is (81827009, SCT, "Diameter")
>Device Diameter Units | (0050,0017) | Product Parameter Sequence (0044,0013) > Measurement Units Code Sequence (0040,08EA) > Code Meaning (0008,0104), where: Product Parameter Sequence > Concept Name Code Sequence (0040,A043) value is (81827009, SCT, "Diameter") ### Note Device Diameter Units is a CS VR (16 characters max, upper case), so a conversion from the LO VR is required.
>Device Volume | (0050,0018) | Product Parameter Sequence (0044,0013) > Numeric Value (0040,A30A), where: Product Parameter Sequence > Concept Name Code Sequence (0040,A043) value is (118565006, SCT, "Volume") Product Parameter Sequence > Measurement Units Code Sequence (0040,08EA) is (ml, UCUM, "ml")
>Inter-Marker Distance | (0050,0019) | Product Parameter Sequence (0044,0013) > Numeric Value (0040,A30A), where: Product Parameter Sequence > Concept Name Code Sequence (0040,A043) value is (121208, DCM, "Inter-Marker Distance") Product Parameter Sequence > Measurement Units Code Sequence (0040,08EA) is (mm, UCUM, "mm")
>Device Description | (0050,0020) | Product Name (0044,0008) and/or Product Description (0044,0009)

Prev | Up | Next
II.2 Enhanced Contrast/bolus Module | Home | II.4 Intervention Module

DICOM PS3.17 2020e - Explanatory Information