DICOM PS3.3 2020c - Information Object Definitions

A.34.5.3 Ambulatory ECG IOD Module Table (Current)
Prev | A.34.5 Ambulatory ECG IOD | Next

#### A.34.5.3 Ambulatory ECG IOD Module Table

Table A.34.5-1 specifies the Modules of the Ambulatory ECG IOD.

Table A.34.5-1. Ambulatory ECG IOD Modules

IE | Module | Reference | Usage
--- | --- | --- | ---
Patient | Patient | C.7.1.1 | M
Clinical Trial Subject | C.7.1.3 | U
Study | General Study | C.7.2.1 | M
Patient Study | C.7.2.2 | U
Clinical Trial Study | C.7.2.3 | U
Series | General Series | C.7.3.1 | M
Clinical Trial Series | C.7.3.2 | U
Frame of Reference | Synchronization | C.7.4.2 | U
Equipment | General Equipment | C.7.5.1 | M
Waveform | Waveform Identification | C.10.8 | M
Waveform | C.10.9 | M
Acquisition Context | C.7.6.14 | U
Waveform Annotation | C.10.10 | C - Required if annotation is present.
SOP Common | C.12.1 | M

Prev | Up | Next
A.34.5.2 Ambulatory ECG IOD Entity-Relationship Model | Home | A.34.5.4 Ambulatory ECG IOD Content Constraints

DICOM PS3.3 2020c - Information Object Definitions