DICOM PS3.3 2020c - Information Object Definitions

A.50.3 RT Ion Beams Treatment Record IOD Modules (Current)
Prev | A.50 RT Ion Beams Treatment Record IOD | Next

### A.50.3 RT Ion Beams Treatment Record IOD Modules

Table A.50-1 identifies and defines the Modules that comprise this IOD. Modules listed are either mandatory or optional as specified in PS3.4. Mandatory Modules contain Attributes that are included in all SOP Instances employing this IOD.

Table A.50-1. RT Ion Beams Treatment Record IOD Modules

IE | Module | Reference | Usage
--- | --- | --- | ---
Patient | Patient | C.7.1.1 | M
Clinical Trial Subject | C.7.1.3 | U
Study | General Study | C.7.2.1 | M
Clinical Trial Study | C.7.2.3 | U
Patient Study | C.7.2.2 | U
Series | RT Series | C.8.8.1 | M
Clinical Trial Series | C.7.3.2 | U
Equipment | General Equipment | C.7.5.1 | M
Treatment Record | RT General Treatment Record | C.8.8.17 | M
RT Patient Setup | C.8.8.12 | U
RT Treatment Machine Record | C.8.8.18 | M
Measured Dose Reference Record | C.8.8.19 | U
Calculated Dose Reference Record | C.8.8.20 | U
RT Ion Beams Session Record | C.8.8.26 | M
RT Treatment Summary Record | C.8.8.23 | U
General Reference | C.12.4 | U
SOP Common | C.12.1 | M
Common Instance Reference | C.12.2 | U

Prev | Up | Next
A.50.2 RT Ion Beams Treatment Record IOD Entity-Relationship Model | Home | A.51 Segmentation IOD

DICOM PS3.3 2020c - Information Object Definitions