DICOM PS3.16 2020b - Content Mapping Resource

TID 3922 Absolute Values of Ventricular Measurements (Current)
Prev | CT/MR Cardiovascular Analysis Report Templates | Next

### TID 3922 Absolute Values of Ventricular Measurements

Ventricular measurement results related to the absolute volume of a ventricle.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3922. Absolute Values of Ventricular Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (122608, DCM, "Absolute Values Of Ventricular Measurements") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3833 “Cardiac Ejection Fraction” $ModType = DT (122670, DCM, "Papillary Muscle Included/Excluded") $ModValue = DCID 3821 “Papillary Muscle Included/Excluded” $Method = DCID 3807 “Volume Measurement Methods” $Units = DT (%, UCUM, "%")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3835 “Cardiac Volume Measurements” $ModType = DT (122670, DCM, "Papillary Muscle Included/Excluded") $ModValue = DCID 3821 “Papillary Muscle Included/Excluded” $Units = DT (ml, UCUM, "ml")
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (82799009, SCT, "Cardiac Output") $ModType = DT (122670, DCM, "Papillary Muscle Included/Excluded") $ModValue = DCID 3821 “Papillary Muscle Included/Excluded” $Units = DT (l/min, UCUM, "l/min")
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $ModType = DT (122670, DCM, "Papillary Muscle Included/Excluded") $ModValue = DCID 3821 “Papillary Muscle Included/Excluded” $Units = DT (g, UCUM, "g")
6 | >> | HAS CONCEPT MOD | CODE | EV (272518008, SCT, "Cardiac Cycle Point") | 1 | U |  | DCID 12233 “Cardiac Phase”
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122616, DCM, "Peak Ejection Rate") $Units = DT (ml/s, UCUM, "ml/s")
8 | > | CONTAINS | NUM | EV (122617, DCM, "Peak Ejection Time") | 1 | U |  | UNITS = EV (s, UCUM, "s")
9 | >> | HAS CONCEPT MOD | CODE | EV (122611, DCM, "Reference Point") | 1 | M |  | EV (416190007, SCT, "End-Diastolic")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122618, DCM, "Peak Filling Rate") $Units = DT (ml/s, UCUM, "ml/s")
11 | > | CONTAINS | NUM | EV (122619, DCM, "Peak Filling Time") | 1 | U |  | UNITS = DT (s, UCUM, "s")
12 | >> | HAS CONCEPT MOD | CODE | EV (122611, DCM, "Reference Point") | 1 | M |  | DT (109070, DCM, "End-Systolic")

Prev | Up | Next
TID 3921 Ventricular Measurements | Home | TID 3923 BSA-Normalized Ventricular Measurements

DICOM PS3.16 2020b - Content Mapping Resource