DICOM PS3.16 2020a - Content Mapping Resource

TID 4019 Algorithm Identification (Current)
Prev | Mammography CAD SR IOD Templates | Next

### TID 4019 Algorithm Identification

This Template details the algorithm unambiguously. Re-state the software identification from the General Equipment Module of the SR IOD if all algorithms are unambiguously defined by that module.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4019. Algorithm Identification

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (111001, DCM, "Algorithm Name") | 1 | M |  |
1b |  |  | CODE | EV (111001, DCM, "Algorithm Name") | 1 | U |  |
2 |  |  | TEXT | EV (111003, DCM, "Algorithm Version") | 1 | M |  |
3 |  |  | TEXT | EV (111002, DCM, "Algorithm Parameters") | 1-n | U |  |
4 |  |  | CODE | EV (111000, DCM, "Algorithm Family") | 1 | U |  |

Content Item Descriptions

Row 1 | May be the same as the Manufacturer's Model Name (0008,1090) in General Equipment Module, if the Algorithm is not distinguishable from the body of software that makes up the Equipment.
Row 2 | May be the same as Software Versions (0018,1020) in General Equipment Module, if the latter is a single Value, or its multiple Values are combined into a single TEXT Content Item Value.

Prev | Up | Next
TID 4018 CAD Analysis Performed | Home | TID 4020 CAD Image Library Entry

DICOM PS3.16 2020a - Content Mapping Resource