DICOM PS3.3 2019e - Information Object Definitions

C.8.8.18 RT Treatment Machine Record Module (Current)
Prev | C.8.8 Radiotherapy Modules | Next

#### C.8.8.18 RT Treatment Machine Record Module

Table C.8-54. RT Treatment Machine Record Module Attributes

Attribute Name | Tag | Type | Attribute Description
--- | --- | --- | ---
Treatment Machine Sequence | (300A,0206) | 1 | Sequence describing treatment machine used for treatment delivery. Only a single Item shall be included in this Sequence.
>Treatment Machine Name | (300A,00B2) | 2 | User-defined name identifying treatment machine used for treatment delivery.
>Manufacturer | (0008,0070) | 2 | Manufacturer of the equipment used for treatment delivery.
>Institution Name | (0008,0080) | 2 | Institution where the equipment is located that was used for treatment delivery.
>Institution Address | (0008,0081) | 3 | Mailing address of the institution where the equipment is located that was used for treatment delivery.
>Institutional Department Name | (0008,1040) | 3 | Department in the institution where the equipment is located that was used for treatment delivery.
>Institutional Department Type Code Sequence | (0008,1041) | 3 | A coded description of the type of Department or Service within the healthcare facility. Only a single Item is permitted in this Sequence.
>>Include Table 8.8-1 “Code Sequence Macro Attributes” | BCID 7030 “Institutional Departments, Units and Services”.
>Manufacturer's Model Name | (0008,1090) | 2 | Manufacturer's model name of the equipment used for treatment delivery.
>Device Serial Number | (0018,1000) | 2 | Manufacturer's serial number of the equipment used for treatment delivery.

Prev | Up | Next
C.8.8.17 RT General Treatment Record Module | Home | C.8.8.19 Measured Dose Reference Record Module

DICOM PS3.3 2019e - Information Object Definitions