DICOM PS3.16 2019e - Content Mapping Resource

TID 3530 Hemodynamic Acquisition Context (Current)
Prev | Hemodynamics Report Templates | Next

### TID 3530 Hemodynamic Acquisition Context

The Hemodynamic Acquisition Context Template is invoked by the various hemodynamic measurement Templates.

Table TID 3530. Parameters

Parameter Name | Parameter Usage
--- | ---
$LocationName | Coded term for Concept Name of measurement location
$LocationValue | Context Group for appropriate anatomic locations

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3530. Hemodynamic Acquisition Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | CODE | $LocationName | 1 | M |  | $LocationValue
2 | > | HAS CONCEPT MOD | CODE | EV (106233006, SCT, "Topographical modifier") | 1 | U |  | BCID 3019 “Cardiovascular Anatomic Location Modifiers”
3 |  | HAS ACQ CONTEXT | CODE | EV (370129005, SCT, "Measurement Method ") | 1 | U |  | BCID 3241 “Hemodynamic Measurement Techniques”
4 |  | HAS ACQ CONTEXT | WAVEFORM | DT (121112, DCM, "Source of Measurement") | 1 | UC | XOR Row 5 |
5 |  | HAS ACQ CONTEXT | TCOORD | DT (121112, DCM, "Source of Measurement") | 1 | UC | XOR Row 4 |
6 | > | SELECTED FROM | WAVEFORM |  | 1 | M |  |

Prev | Up | Next
TID 3521 Relative Time | Home | TID 3550 Pressure Waveform Measurements

DICOM PS3.16 2019e - Content Mapping Resource