DICOM PS3.16 2019c - Content Mapping Resource

Quantitative Ventricular Analysis Report SR IOD Templates (Current)
Prev | A Structured Reporting Templates (Normative) | Next

## Quantitative Ventricular Analysis Report SR IOD Templates

The Templates that comprise the Quantitative Ventricular Analysis SR are interconnected as in Figure A-3:

Figure A-3. Quantitative Ventricular Analysis Report SR IOD Template Structure

### TID 3202 Ventricular Analysis

The Ventricular Analysis Template provides a CONTAINER with a structure for reporting the result of the ventricular analysis.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3202. Ventricular Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122292, DCM, "Quantitative Ventriculography Report") | 1 | M |  |
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
4 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | U |  |
5 | > | CONTAINS | CONTAINER | EV (122144, DCM, "Quantitative Analysis") | 1-n | M |  |
6 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
7 | >> | HAS OBS CONTEXT | TEXT | EV (111001, DCM, "Algorithm Name") | 1 | M |  |
8 | >> | HAS OBS CONTEXT | TEXT | EV (111003, DCM, "Algorithm Version") | 1 | M |  |
9 | >> | HAS OBS CONTEXT | TEXT | EV (122405, DCM, "Algorithm Manufacturer") | 1 | M |  |
10 | >> | CONTAINS | IMAGE | EV (121112, DCM, "Source of Measurement") | 1-n | M |  |
11 | >>> | HAS CONCEPT MOD | CODE | EV (246092007, SCT, "Cardiac Phase") | 1 | M |  | DCID 12233 “Cardiac Phase”
12 | >>> | HAS CONCEPT MOD | CODE | EV (111031, DCM, "Image View") | 1 | MC | If Biplane Analysis | DCID 3466 “Plane Identification”
13 | >> | HAS ACQ CONTEXT | INCLUDE | DTID 3205 “Calibration” | 1-2 | U | VM = 1: Single plane analysis, VM = 2: Biplane analysis | $CalibrationPlane = DCID 3466 “Plane Identification”
14 | >> | HAS ACQ CONTEXT | INCLUDE | DTID 3520 “Hemodynamic Clinical Context” | 1 | U |  |
15 | >> | CONTAINS | INCLUDE | DTID 3206 “VA Main Results” | 1 | M |  |
16 | >> | CONTAINS | INCLUDE | DTID 3207 “AA Main Results” | 1 | U |  |
17 | >> | CONTAINS | INCLUDE | DTID 3208 “Frame-to-Frame Results” | 1 | U |  |
18 | >> | CONTAINS | INCLUDE | DTID 3209 “Centerline Wall Motion” | 1-2 | U | VM = 1: Single plane analysis, VM = 2: Biplane analysis |
19 | >> | CONTAINS | INCLUDE | DTID 3210 “Radial Based Wall Motion” | 1-2 | U | VM = 1: Single plane analysis, VM = 2: Biplane analysis |
20 | >> | CONTAINS | INCLUDE | DTID 3211 “Landmark Based Wall Motion” | 1-2 | U | VM = 1: Single plane analysis, VM = 2: Biplane analysis |

Content Item Descriptions

Row 7 | Identifies the Ventricular Analysis program
Row 8 | Identifies the Ventricular Analysis program version
Row 9 | Identifies the Ventricular Analysis program manufacturer
Row 10 | Identifies the ES and ED images on which the analysis is based, for frame by frame analysis the analyzed image are specified in the frame by frame results (3208) Template

Prev | Up | Next
TID 3115 ECG ST Assessment | Home | TID 3205 Calibration

DICOM PS3.16 2019c - Content Mapping Resource