DICOM PS3.16 2019b - Content Mapping Resource

Hemodynamics Report Templates (Current)
Prev | A Structured Reporting Templates (Normative) | Next

## Hemodynamics Report Templates

The Templates that comprise the Hemodynamic Report are interconnected as shown in Figure A-6.

Figure A-6. Hemodynamic Report Template Hierarchy

Figure A-6 shows only the use of Templates specific to the Hemodynamic Report; it does not show common structural Templates such as TID 300 “Measurement”.

### Note

Figure A-6 shows the relationship of Templates; it does not show the structural hierarchy of Content Items in the IOD. See Figure L-1 “Hemodynamics Report Structure” in PS3.17 .

### TID 3500 Hemodynamics Report

The Hemodynamic Report Template is the root structure for the representation of measurements acquired during a procedure in a cardiac catheterization lab.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 3500. Hemodynamics Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122120, DCM, "Hemodynamics Report") | 1 | M |  | Root node
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
3 | > |  | INCLUDE | DTID 3601 “Procedure Context” | 1 | M |  |
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | M |  |
5 | > | HAS ACQ CONTEXT | INCLUDE | DTID 3603 “Procedure Environmental Characteristics” | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3501 “Hemodynamics Measurement Group” | 1-n | M |  |
7 | > | CONTAINS | INCLUDE | DTID 3570 “Summary, Hemodynamics” | 1 | U |  |

Prev | Up | Next
TID 3320 Conclusions and Recommendations | Home | TID 3501 Hemodynamics Measurement Group

DICOM PS3.16 2019b - Content Mapping Resource