DICOM PS3.16 2019b - Content Mapping Resource

TID 3550 Pressure Waveform Measurements (Current)
Prev | Hemodynamics Report Templates | Next

### TID 3550 Pressure Waveform Measurements

The Pressure Waveform Measurements Template is invoked by the various hemodynamic measurement Templates for recording general measurements made in conjunction with the specific required measurements for that anatomic location.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3550. Pressure Waveform Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3611 “Pressure Measurements” $Units = DCID 3500 “Pressure Units”
2 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3612 “Blood Velocity Measurements” $Units = EV (mm/s, UCUM, "mm/s")
3 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3613 “Hemodynamic Time Measurements” $Units = DT (ms, UCUM, "ms")
4 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (82799009, SCT, "Cardiac Output") $Units = EV (l/min, UCUM, "l/min")
5 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (90096001, SCT, "Stroke Volume") $Units = DT (ml, UCUM, "ml")
6 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (10230-1, LN, "LV Ejection Fraction") $Units = EV (%, UCUM, "%")
7 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8867-4, LN, "Heart rate") $Units = DT ({H.B.}/min, UCUM, "BPM")
8 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (86290005, SCT, "Respiration rate") $Units = DT (/min, UCUM, "breaths/min")
9 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (109025, DCM, "Max dp/dt") $Units = DT (mm[Hg]/s, UCUM, "mmHg/s")
10 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (109026, DCM, "Max neg dp/dt") $Units = DT (mm[Hg]/s, UCUM, "mmHg/s")
11 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122190, DCM, "Max dp/dt/P") $Units = DT (/s, UCUM, "/s")
12 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122192, DCM, "Indicator appearance time") $Units = DT (s, UCUM, "s")
13 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122193, DCM, "Maximum pressure acceleration") $Units = DT (mm[Hg]/s2, UCUM, "mmHg/s/s")
14 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3617 “Valve Flows” $Units = DT (ml/min, UCUM, "ml/min")

Prev | Up | Next
TID 3530 Hemodynamic Acquisition Context | Home | TID 3560 Derived Hemodynamic Measurements

DICOM PS3.16 2019b - Content Mapping Resource