DICOM PS3.16 2019a - Content Mapping Resource

TID 4012 Mammography CAD Non-lesion (Current)
Prev | Mammography CAD SR IOD Templates | Next

### TID 4012 Mammography CAD Non-lesion

This Template provides the detail specific to a finding other than a lesion (see CID 6040 “Non-lesion Object Type”).

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4012. Mammography CAD Non-lesion

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111039, DCM, "Object type") | 1 | M |  | DCID 6040 “Non-lesion Object Type”
2 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1400 “Linear Measurement” shall be used.
3 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1401 “Area Measurement” shall be used.
4 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1402 “Volume Measurement” shall be used.

Prev | Up | Next
TID 4011 Mammography CAD Density | Home | TID 4013 Mammography CAD Selected Region

DICOM PS3.16 2019a - Content Mapping Resource