DICOM PS3.3 2018e - Information Object Definitions

C.8.15.4 Enhanced Multi-energy CT Acquisition Module (Current)
Prev | C.8.15 Enhanced CT Image | Next

#### C.8.15.4 Enhanced Multi-energy CT Acquisition Module

Table C.8.15.4-1 specifies the Attributes that describe the Multi-energy CT Acquisition technique in the Enhanced CT Image.

Table C.8.15.4-1. Enhanced Multi-energy CT Acquisition Module Attributes

Attribute Name | Tag | Type | Attribute Description
--- | --- | --- | ---
Include Table C.8.2.2-2 “Multi-energy CT X-Ray Source Macro Attributes” | See Section C.8.2.2.1.
Include Table C.8.2.2-3 “Multi-energy CT X-Ray Detector Macro Attributes” | See Section C.8.2.2.2.
Include Table C.8.2.2-4 “Multi-energy CT Path Macro Attributes” | See Section C.8.2.2.3.

Prev | Up | Next
C.8.15.3.13 Multi-energy CT Processing Macro | Home | C.8.16 Common CT, MR and US Descriptions

DICOM PS3.3 2018e - Information Object Definitions