DICOM PS3.16 2018d - Content Mapping Resource

TID 3206 VA Main Results (Current)
Prev | Quantitative Ventricular Analysis Report SR IOD Templates | Next

### TID 3206 VA Main Results

The VA Main Results Template consists of a CONTAINER with a structure for reporting the main ventricular analysis measurements.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3206. VA Main Results

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3462 “Chamber Identification”
3 | > | CONTAINS | CODE | EV (122429, DCM, "Volume Method") | 1 | M |  | DCID 3453 “Cardiac Volume Methods”
4 | > | CONTAINS | NUM | EV (122435, DCM, "Regression Volume Exponent") | 1 | U |  | Unit = DT (1, UCUM, "no units")
5 | > | CONTAINS | NUM | EV (122431, DCM, "Regression Slope ED") | 1 | U |  | Unit = DT ({ratio}, UCUM, "ratio")
6 | > | CONTAINS | NUM | EV (122432, DCM, "Regression Offset ED") | 1 | U |  | Unit = DT (ml, UCUM, "ml")
7 | > | CONTAINS | NUM | EV (122433, DCM, "Regression Slope ES") | 1 | U |  | Unit = DT ({ratio}, UCUM, "ratio")
8 | > | CONTAINS | NUM | EV (122434, DCM, "Regression Offset ES") | 1 | U |  | Unit = DT (ml, UCUM, "ml")
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = DCID 3467 “Ejection Fraction” $Unit = DT (%, UCUM, "%")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3468 “ED Volume” $Unit = DT (ml, UCUM, "ml")
11 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3469 “ES Volume” $Unit = DT (ml, UCUM, "ml")
12 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $Unit = DT (ml, UCUM, "ml")
13 | > | CONTAINS | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  | Unit = DT ({H.B.}/min, UCUM, "BPM")
14 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3468 “ED Volume” $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (ml/m2, UCUM, "ml/m2")
15 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3468 “ED Volume” $ModType = EV (121425, DCM, "Index") $ModValue = EV (29463-7, LN, "Patient Weight") $Unit = DT (ml/kg, UCUM, "ml/kg")
16 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3469 “ES Volume” $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (ml/m2, UCUM, "ml/m2")
17 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3469 “ES Volume” $ModType = EV (121425, DCM, "Index") $ModValue = EV (29463-7, LN, "Patient Weight") $Unit = DT (ml/kg, UCUM, "ml/kg")
18 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (ml/m2, UCUM, "ml/m2")
19 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $ModType = EV (121425, DCM, "Index") $ModValue = EV (29463-7, LN, "Patient Weight") $Unit = DT (ml/kg, UCUM, "ml/kg")
20 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32100, SRT, "Cardiac Output") $Unit = DT (l/min, UCUM, "l/min")
21 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32110, SRT, "Cardiac Index") $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (l/min/m2, UCUM, "l/min/m2")
22 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122445, DCM, "Wall Thickness") $Unit = DT (mm, UCUM, "mm")
23 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122446, DCM, "Wall Volume") $Unit = DT (ml, UCUM, "ml")
24 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $Unit = DT (g, UCUM, "gram")
25 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (g/m2, UCUM, "gram/m2")
26 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $ModType = EV (121425, DCM, "Index") $ModValue = EV (29463-7, LN, "Patient Weight") $Unit = DT (g/kg, UCUM, "gram/kg")
27 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122448, DCM, "Wall Stress") $Unit = DT (dyn/cm2, UCUM, "dynes/cm2")
28 | > | CONTAINS | IMAGE | No purpose of reference | 1-n | U |  |

Content Item Descriptions

Row 28 | Secondary Capture image with ED and/or ES contours

Prev | Up | Next
TID 3205 Calibration | Home | TID 3207 AA Main Results

DICOM PS3.16 2018d - Content Mapping Resource