DICOM PS3.16 2018d - Content Mapping Resource

TID 3915 Vascular Occlusion Properties (Current)
Prev | CT/MR Cardiovascular Analysis Report Templates | Next

### TID 3915 Vascular Occlusion Properties

Properties of vascular occlusion finding

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3915. Vascular Occlusion Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | CODE | EV (G-D775, SRT, "Type of Stenosis") | 1 | M |  | DCID 3805 “Stenosis Types”
2 |  | HAS PROPERTIES | CODE | EV (G-C002, SRT, "Associated with") | 1 | U |  | DCID 3815 “Source of Vascular Finding”
3 |  | HAS PROPERTIES | CODE | EV (G-C2FE, SRT, "Shape") | 1 | U |  | DCID 3806 “Stenosis Shape”
4 |  | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (R-101BC, SRT, "Stenotic Lesion Length") $Method = DCID 3804 “Stenosis Measurement Methods” $Units = DT (mm, UCUM, "mm")

Prev | Up | Next
TID 3914 Arterial Dissection Properties | Home | TID 3916 Stent Properties

DICOM PS3.16 2018d - Content Mapping Resource