# PS3.16

## DICOM PS3.16 2017d - Content Mapping Resource

### DICOM Standards Committee

Copyright © 2017 NEMA

List of Figures

A-3. Quantitative Ventricular Analysis Report SR IOD Template Structure
A-4. Quantitative Arterial Analysis Report SR IOD Template Structure
A-4b. Direction of Blood Flow
A-5. IVUS Report Template Hierarchy
A-6. Hemodynamic Report Template Hierarchy
A-7. Cardiac Catheterization Report Template Hierarchy
A-8. Mammography CAD SR IOD Template Structure
A-8a. Example of Breast Outline Including Pectoral Muscle Tissue
A-8b. Example of Pectoral Muscle Outline
A-9. Chest CAD SR IOD Template Structure
A-9b. Colon CAD SR IOD Template Structure
A-10. Breast Imaging Report Template Structure
A-10b. Simplified Adult Echocardiography Template Structure
A-11. Implantation Plan SR Document IOD Template Structure
A-12. Implant Assembly and Components Terminology
A-13. References to Registration Objects
A-14. X-Ray Radiation Dose SR IOD Template Structure
A-15. CT Radiation Dose SR IOD Template Structure
A-16. Spiral Acquisition Parameters
A-17. Radiopharmaceutical Radiation Dose SR IOD Template Structure
A-18. Patient Radiation Dose Structured Report IOD Template Structure

List of Tables

TID <#>. <SR Context Template Name>
TID <#>. <Acquisition Context Template Name>
TID <#>. <Protocol Context Template Name>
6.1.3-1. Syntax of Relationship Constraints
6.1.6-1. Permitted Values for VM
CID <#>. <Context Group Name>
7.2.1-1. Include Context Group Macro
8-1. Coding Schemes
8-2. HL7v3 Coding Schemes
TID 300. Parameters
TID 300. Measurement
TID 310. Parameters
TID 310. Measurement Properties
TID 311. Parameters
TID 311. Measurement Statistical Properties
TID 312. Parameters
TID 312. Normal Range Properties
TID 315. Parameters
TID 315. Equation or Table
TID 320. Parameters
TID 320. Image or Spatial Coordinates
TID 321. Parameters
TID 321. Waveform or Temporal Coordinates
TID 350. References to Supporting Evidence
TID 351. Previous Reports
TID 400. Reference Location
TID 1000. Quotation
TID 1001. Observation Context
TID 1002. Observer Context
TID 1003. Person Observer Identifying Attributes
TID 1004. Device Observer Identifying Attributes
TID 1005. Procedure Context
TID 1006. Subject Context
TID 1007. Subject Context, Patient
TID 1008. Subject Context, Fetus
TID 1009. Subject Context, Specimen
TID 1010. Subject Context, Device
TID 1020. Parameters
TID 1020. Person Participant
TID 1021. Parameters
TID 1021. Device Participant
TID 1200. Language Designation
TID 1201. Language of Value
TID 1202. Language of Name and Value
TID 1204. Language of Content Item and Descendants
TID 1210. Equivalent Meaning(s) of Concept Name
TID 1211. Equivalent Meaning(s) of Value
TID 1350. Negation Modifier, Presence of Finding
TID 1400. Linear Measurement
TID 1401. Area Measurement
TID 1402. Volume Measurement
TID 1404. Numeric Measurement
TID 1406. Three Dimensional Linear Measurement
TID 1410. Parameters
TID 1410. Planar ROI Measurements
TID 1411. Parameters
TID 1411. Volumetric ROI Measurements
TID 1419. Parameters
TID 1419. ROI Measurements
TID 1420. Parameters
TID 1420. Measurements Derived From Multiple ROI Measurements
TID 1500. Measurement Report
TID 1501. Parameters
TID 1501. Measurement Group
TID 1502. Time Point Context
TID 1600. Image Library
TID 1601. Image Library Entry
TID 1602. Image Library Entry Descriptors
TID 1603. Image Library Entry Descriptors for Projection Radiography
TID 1604. Image Library Entry Descriptors for Cross-Sectional Modalities
TID 1605. Image Library Entry Descriptors for CT
TID 1606. Image Library Entry Descriptors for MR
TID 1607. Image Library Entry Descriptors for PET
TID 2000. Basic Diagnostic Imaging Report
TID 2001. Basic Diagnostic Imaging Report Observations
TID 2002. Report Narrative
TID 2005. Transcribed Diagnostic Imaging Report
TID 2006. Imaging Report With Conditional Radiation Exposure and Protection Information
TID 2007. Imaging Procedure Description
TID 2008. Radiation Exposure and Protection Information
TID 2010. Key Object Selection
TID 2020. Spectacle Prescription Report
TID 2021. Spectacle Prescription Details
TID 2100. Macular Grid Thickness and Volume Report
TID 2101. Parameters
TID 2101. Macular Grid Thickness and Volume Measurement
TID 2102. Quality Rating Identification
TID 3001. Procedure Log
TID 3010. Log Entry Qualifiers
TID 3100. Procedure Action
TID 3101. Image Acquisition
TID 3102. Waveform Acquisition
TID 3103. Referenced Object
TID 3104. Consumables
TID 3105. Lesion Identification and Properties
TID 3106. Drugs/Contrast Administered
TID 3107. Device Used
TID 3108. Intervention
TID 3109. Measurements
TID 3110. Impressions or Findings
TID 3111. Percutaneous Entry
TID 3112. Specimen Obtained
TID 3113. Patient Support
TID 3114. Patient Assessment
TID 3115. ECG ST Assessment
TID 3202. Ventricular Analysis
TID 3205. Parameters
TID 3205. Calibration
TID 3206. VA Main Results
TID 3207. AA Main Results
TID 3208. Frame-To-Frame Result
TID 3209. Centerline Wall Motion
TID 3210. Radial Based Wall Motion
TID 3211. Landmark Based Wall Motion
TID 3213. Quantitative Arterial Analysis
TID 3214. Analyzed Segment
TID 3215. Angiographic Lesion Analysis
TID 3216. Stenotic Flow Reserve
TID 3217. Sub-Segmental Data
TID 3218. Position in Arterial Segment
TID 3219. Segment Values
TID 3250. IVUS Report
TID 3251. IVUS Vessel
TID 3252. IVUS Lesion
TID 3253. IVUS Measurements
TID 3254. IVUS Qualitative Assessments
TID 3255. IVUS Volume Measurement
TID 3300. Stress Testing Report
TID 3301. Stress Test Procedure Description
TID 3303. Stress Test Phase Data
TID 3304. Stress Test Measurement Group
TID 3307. NM/PET Perfusion Measurement Group
TID 3309. Stress Echo Measurement Group
TID 3311. Stress Test Summary
TID 3312. Physiological Summary
TID 3313. Stress ECG Summary
TID 3317. Stress Imaging Summary
TID 3318. Comparison to Prior Stress Exam
TID 3320. Conclusions and Recommendations
TID 3500. Hemodynamics Report
TID 3501. Hemodynamics Measurement Group
TID 3504. Arterial Pressure Measurement
TID 3505. Atrial Pressure Measurement
TID 3506. Venous Pressure Measurement
TID 3507. Ventricular Pressure Measurement
TID 3508. Gradient Measurement
TID 3509. Blood Velocity Measurement
TID 3510. Vital Signs
TID 3515. Cardiac Output Measurement By Indicator Dilution
TID 3516. Blood Lab Measurements
TID 3520. Hemodynamic Clinical Context
TID 3521. Parameters
TID 3521. Relative Time
TID 3530. Parameters
TID 3530. Hemodynamic Acquisition Context
TID 3550. Pressure Waveform Measurements
TID 3560. Derived Hemodynamic Measurements
TID 3570. Summary, Hemodynamics
TID 3601. Procedure Context
TID 3602. Cardiovascular Patient Characteristics
TID 3603. Procedure Environmental Characteristics
TID 3700. ECG Report
TID 3702. Prior ECG Exam
TID 3704. Patient Characteristics for ECG
TID 3708. ECG Waveform Information
TID 3713. ECG Global Measurements
TID 3714. ECG Lead Measurements
TID 3715. ECG Measurement Source
TID 3717. Qualitative Analysis, ECG
TID 3719. Summary, ECG
TID 3800. Cardiac Catheterization Report Root
TID 3802. Cardiovascular Patient History
TID 3803. Patient Presentation, Cath
TID 3806. Cath Procedure
TID 3807. Percutaneous Coronary Intervention Procedure
TID 3808. Lesion Intervention Information
TID 3809. Other Interventional Procedures
TID 3810. Cardiac Catheterization Findings
TID 3812. Hemodynamic Findings
TID 3814. Left Ventriculography Findings
TID 3815. Right Ventriculography Findings
TID 3816. Ventricular Assessment
TID 3817. Coronary Arteriography Findings
TID 3818. Other Cardiographic Findings
TID 3819. Parameters
TID 3819. Common Findings
TID 3820. Adverse Outcomes, Cath
TID 3824. Summary, Cath
TID 3828. Discharge Summary, Cath
TID 3829. Parameters
TID 3829. Problem Properties
TID 3830. Parameters
TID 3830. Procedure Properties
TID 3831. Parameters
TID 3831. Medical Device Use
TID 3900. CT/MR Cardiovascular Analysis Report
TID 3901. Procedure Summary
TID 3902. Parameters
TID 3902. Vascular Analysis
TID 3905. Calcium Scoring Results
TID 3906. Parameters
TID 3906. Vascular Section Measurements
TID 3907. Vessel Measurements
TID 3908. Vascular Lesion
TID 3909. Best Illustration of Findings
TID 3910. Flow Quantification
TID 3911. Plaque Properties
TID 3912. Stenosis Properties
TID 3913. Aneurysm Properties
TID 3914. Arterial Dissection Properties
TID 3915. Vascular Occlusion Properties
TID 3916. Stent Properties
TID 3917. Aneurysm Measurements
TID 3920. Ventricular Analysis
TID 3921. Parameters
TID 3921. Ventricular Measurements
TID 3922. Absolute Values of Ventricular Measurements
TID 3923. BSA-Normalized Ventricular Measurements
TID 3924. Heart Rate-Normalized Ventricular Measurements
TID 3925. Thickening Analysis
TID 3926. Myocardial Perfusion Analysis
TID 3927. Report Summary
TID 3929. Cardiovascular Analysis Observation Context
TID 3990. Parameters
TID 3990. Two Dimensional Measurement Graph
TID 4000. Mammography CAD Document Root
TID 4001. Mammography CAD Overall Impression/Recommendation
TID 4002. Mammography CAD Impression/Recommendation Body
TID 4003. Mammography CAD Individual Impression/Recommendation
TID 4004. Mammography CAD Composite Feature
TID 4005. Mammography CAD Composite Feature Body
TID 4006. Mammography CAD Single Image Finding
TID 4007. Mammography CAD Breast Composition
TID 4008. Mammography CAD Breast Geometry
TID 4009. Mammography CAD Individual Calcification
TID 4010. Mammography CAD Calcification Cluster
TID 4011. Mammography CAD Density
TID 4012. Mammography CAD Non-lesion
TID 4013. Mammography CAD Selected Region
TID 4014. Parameters
TID 4014. CAD Image Quality
TID 4015. Parameters
TID 4015. CAD Detections Performed
TID 4016. Parameters
TID 4016. CAD Analyses Performed
TID 4017. Parameters
TID 4017. CAD Detection Performed
TID 4018. Parameters
TID 4018. CAD Analysis Performed
TID 4019. CAD Algorithm Identification
TID 4020. Parameters
TID 4020. CAD Image Library Entry
TID 4021. Mammography CAD Geometry
TID 4022. CAD Observation Context
TID 4023. CAD Operating Points
TID 4100. Chest CAD Document Root
TID 4101. Chest CAD Findings Summary
TID 4102. Chest CAD Composite Feature
TID 4103. Chest CAD Composite Feature Body
TID 4104. Chest CAD Single Image Finding
TID 4105. Chest CAD Descriptors
TID 4106. Response Evaluation
TID 4107. Chest CAD Geometry
TID 4108. Tracking Identifier
TID 4120. Colon CAD Document Root
TID 4121. Colon CAD Findings Summary
TID 4122. CAD Common Image Properties Entry
TID 4125. Colon CAD Composite Feature
TID 4126. Colon CAD Composite Feature Body
TID 4127. Colon CAD Single Image Finding
TID 4128. Colon CAD Descriptors
TID 4129. Colon CAD Geometry
TID 4200. Breast Imaging Report
TID 4201. Breast Imaging Procedure Reported
TID 4202. Breast Imaging Report Narrative
TID 4203. Breast Imaging Assessment
TID 4204. Breast Imaging Report Intervention Section
TID 4205. Breast Composition Section
TID 4206. Breast Imaging Report Finding Section
TID 4207. Breast Imaging Pathology Results
TID 4208. Breast Imaging Report Supplementary Data
TID 5000. OB-GYN Ultrasound Procedure Report
TID 5001. OB-GYN Patient Characteristics
TID 5002. OB-GYN Procedure Summary Section
TID 5003. OB-GYN Procedure Fetus Summary
TID 5004. Fetal Biometry Ratio Section
TID 5005. Fetal Biometry Section
TID 5006. Fetal Long Bones Section
TID 5007. Fetal Cranium Section
TID 5008. Parameters
TID 5008. Fetal Biometry Group
TID 5009. Fetal Biophysical Profile Section
TID 5010. Amniotic Sac Section
TID 5011. Early Gestation Section
TID 5012. Ovaries Section
TID 5013. Parameters
TID 5013. Follicles Section
TID 5014. Follicle Measurement Group
TID 5015. Pelvis and Uterus Section
TID 5016. Parameters
TID 5016. LWH Volume Group
TID 5025. Parameters
TID 5025. OB-GYN Fetal Vascular Ultrasound Measurement Group
TID 5026. Parameters
TID 5026. OB-GYN Pelvic Vascular Ultrasound Measurement Group
TID 5100. Vascular Ultrasound Report
TID 5101. Vascular Patient Characteristics
TID 5102. Vascular Procedure Summary Section
TID 5103. Parameters
TID 5103. Vascular Ultrasound Section
TID 5104. Parameters
TID 5104. Vascular Ultrasound Measurement Group
TID 5105. Ultrasound Graft Section
TID 5200. Echocardiography Procedure Report
TID 5201. Echocardiography Patient Characteristics
TID 5202. Parameters
TID 5202. Echo Section
TID 5203. Parameters
TID 5203. Echo Measurement
TID 5204. Parameters
TID 5204. Wall Motion Analysis
5204-1. Numeric Score Assignment for Segmental Findings
TID 5220. Pediatric, Fetal and Congenital Cardiac Ultrasound Reports
TID 5221. Cardiac Ultrasound Pediatric Echo Measurement Section
TID 5222. Parameters
TID 5222. Pediatric, Fetal and Congenital Cardiac Ultrasound Section
TID 5223. Parameters
TID 5223. Pediatric, Fetal and Congenital Cardiac Ultrasound Measurement
TID 5225. Cardiac Ultrasound Fetal Characteristics
TID 5226. Cardiac Ultrasound Summary Section
TID 5227. Cardiac Ultrasound Fetal Summary Section
TID 5228. Cardiac Ultrasound Fetal Measurement Section
TID 5300. Simplified Echo Procedure Report
TID 5301. Parameters
TID 5301. Pre-coordinated Echo Measurement
TID 5302. Parameters
TID 5302. Post-coordinated Echo Measurement
TID 5303. Parameters
TID 5303. Adhoc Measurement
TID 7000. Implantation Plan
TID 7001. Related Implantation Reports
TID 8101. Preclinical Small Animal Image Acquisition Context
TID 8110. Biosafety Conditions
TID 8121. Animal Housing
TID 8122. Animal Feeding
TID 8130. Anesthesia
TID 8131. Medications and Mixture Medications
TID 8140. Heating Conditions
TID 8150. Circadian Effects
TID 8170. Physiological Monitoring Performed During Procedure
TID 8182. Parameters
TID 8182. Exogenous Substance Administration
TID 9000. Relevant Patient Information for Breast Imaging
TID 9001. Gynecological History
TID 9002. Parameters
TID 9002. Medication, Substance, Environmental Exposure
TID 9003. Parameters
TID 9003. Previous Procedure
TID 9004. Parameters
TID 9004. Indicated Problem
TID 9005. Parameters
TID 9005. Risk Factor
TID 9006. Obstetric History
TID 9007. General Relevant Patient Information
TID 10001. Projection X-Ray Radiation Dose
TID 10002. Parameters
TID 10002. Accumulated X-Ray Dose
TID 10003. Irradiation Event X-Ray Data
TID 10003A. Irradiation Event X-Ray Detector Data
TID 10003B. Irradiation Event X-Ray Source Data
TID 10003C. Irradiation Event X-Ray Mechanical Data
TID 10004. Accumulated Fluoroscopy and Acquisition Projection X-Ray Dose
TID 10005. Accumulated Mammography X-Ray Dose
TID 10006. Accumulated Cassette-Based Projection Radiography Dose
TID 10007. Accumulated Total Projection Radiography Dose
TID 10011. CT Radiation Dose
TID 10012. CT Accumulated Dose Data
TID 10013. CT Irradiation Event Data
TID 10014. Scanning Length
TID 10015. CT Dose Check Details
TID 10021. Radiopharmaceutical Radiation Dose
TID 10022. Radiopharmaceutical Administration Event Data
TID 10023. Organ Dose
TID 10024. Radiopharmaceutical Administration Patient Characteristics
TID 10030. Patient Radiation Dose
TID 10031. Radiation Dose Estimate
TID 10032. Radiation Dose Estimate Representation
TID 10033. Radiation Dose Estimate Methodology
TID 10034. Radiation Dose Estimate Parameters
CID 2. Anatomic Modifier
CID 4. Anatomic Region
CID 5. Transducer Approach
CID 6. Transducer Orientation
CID 7. Ultrasound Beam Path
CID 8. Angiographic Interventional Devices
CID 9. Image Guided Therapeutic Procedures
CID 10. Interventional Drug
CID 11. Route of Administration
CID 12. Radiographic Contrast Agent
CID 13. Radiographic Contrast Agent Ingredient
CID 18. Isotopes in Radiopharmaceuticals
CID 19. Patient Orientation
CID 20. Patient Orientation Modifier
CID 21. Patient Equipment Relationship
CID 23. Cranio-Caudad Angulation
CID 25. Radiopharmaceuticals
CID 26. Nuclear Medicine Projections
CID 27. Basic Cardiac Views
CID 29. Acquisition Modality
CID 30. DICOM Devices
CID 31. Abstract Priors
CID 42. Numeric Value Qualifier
CID 50. Instance Availability Status
CID 83. Units for Real World Value Mapping
CID 84. PET Units
CID 85. SUV Units
CID 91. Functional Condition Present During Acquisition
CID 92. Joint Position During Acquisition
CID 93. Joint Positioning Method
CID 94. Physical Force Applied During Acquisition
CID 100. Quantitative Diagnostic Imaging Procedures
CID 220. Level of Significance
CID 221. Measurement Range Concepts
CID 222. Normality Codes
CID 223. Normal Range Values
CID 224. Selection Method
CID 225. Measurement Uncertainty Concepts
CID 226. Population Statistical Descriptors
CID 227. Sample Statistical Descriptors
CID 228. Equation or Table
CID 230. Yes-No
CID 231. Yes-No Only
CID 240. Present-Absent
CID 241. Present-Absent Only
CID 242. Normal-Abnormal
CID 244. Laterality
CID 250. Positive-Negative
CID 251. Severity of Complication
CID 252. S-M-L Size Descriptor
CID 270. Observer Type
CID 271. Observation Subject Class
CID 280. Longitudinal Temporal Event Types
CID 400. Audit Event ID
CID 401. Audit Event Type Code
CID 402. Audit Active Participant Role ID Code
CID 403. Security Alert Type Code
CID 404. Audit Participant Object ID Type Code
CID 405. Media Type Code
CID 501. Volumetric View Description
CID 502. Volumetric View Modifier
CID 601. Biosafety Levels
CID 602. Biosafety Control Reasons
CID 603. Animal Room Types
CID 604. Device Reuse
CID 605. Animal Bedding Material
CID 606. Animal Shelter Types
CID 607. Animal Feed Types
CID 608. Animal Feed Sources
CID 609. Animal Feeding Methods
CID 610. Water Types
CID 611. Anesthesia Category Code Type for Small Animal Anesthesia
CID 612. Anesthesia Category Code Type from Anesthesia Quality Initiative (AQI)
CID 613. Anesthesia Induction Code Type for Small Animal Anesthesia
CID 614. Anesthesia Induction Code Type from Anesthesia Quality Initiative (AQI)
CID 615. Anesthesia Maintenance Code Type for Small Animal Anesthesia
CID 616. Anesthesia Maintenance Code Type from Anesthesia Quality Initiative (AQI)
CID 617. Airway Management Method Code Type for Small Animal Anesthesia
CID 618. Airway Management Method Code Type from Anesthesia Quality Initiative (AQI)
CID 619. Airway Management Sub-Method Code Type for Small Animal Anesthesia
CID 620. Airway Management Sub-Method Code Type from Anesthesia Quality Initiative (AQI)
CID 621. Medication Type Code Type for Small Animal Anesthesia
CID 622. Medication Type Code Type from Anesthesia Quality Initiative (AQI)
CID 623. Medication for Small Animal Anesthesia
CID 624. Inhalational Anesthesia Agents for Small Animal Anesthesia
CID 625. Injectable Anesthesia Agents for Small Animal Anesthesia
CID 626. Premedication Agents for Small Animal Anesthesia
CID 627. Neuromuscular Blocking Agents for Small Animal Anesthesia
CID 628. Ancillary Medications for Small Animal Anesthesia
CID 629. Carrier Gases for Small Animal Anesthesia
CID 630. Local Anesthetics for Small Animal Anesthesia
CID 631. Phase of Procedure Requiring Anesthesia
CID 632. Phase of Surgical Procedure Requiring Anesthesia
CID 633. Phase of Imaging Procedure Requiring Anesthesia
CID 634. Phase of Animal Handling
CID 635. Heating Method
CID 636. Temperature Sensor Device Component Type for Small Animal Procedures
CID 637. Exogenous Substance Types
CID 638. Exogenous Substance
CID 639. Tumor Graft Histologic Type
CID 640. Fibrils
CID 641. Viruses
CID 642. Cytokines
CID 643. Toxins
CID 644. Exogenous Substance Administration Sites
CID 645. Exogenous Substance Tissue of Origin
CID 646. Preclinical Small Animal Imaging Procedures
CID 647. Position Reference Indicator for Frame of Reference
CID 701. Content Assessment Types
CID 702. RT Content Assessment Types
CID 703. Basis of Assessment
CID 800. Protocol Assertion Codes
CID 1000. CT Transverse Plane Reference Basis
CID 1001. Anatomical Reference Basis
CID 1002. Anatomical Reference Basis - Head
CID 1003. Anatomical Reference Basis - Spine
CID 1004. Anatomical Reference Basis - Chest
CID 1005. Anatomical Reference Basis - Abdomen/Pelvis
CID 1006. Anatomical Reference Basis - Extremities
CID 1010. Reference Geometry - Planes
CID 1011. Reference Geometry - Points
CID 1015. Patient Alignment Methods
CID 1200. Contraindications For CT Imaging
CID 3000. Audio Channel Source
CID 3001. ECG Leads
CID 3003. Hemodynamic Waveform Sources
CID 3004. Arterial Pulse Waveform
CID 3005. Respiration Waveform
CID 3010. Cardiovascular Anatomic Locations
CID 3011. Electrophysiology Anatomic Locations
CID 3014. Coronary Artery Segments
CID 3015. Coronary Arteries
CID 3016. Major Coronary Arteries
CID 3019. Cardiovascular Anatomic Location Modifiers
CID 3083. Units of Radioactivity
CID 3090. Time Synchronization Channel Types
CID 3101. Cardiac Procedural State Values
CID 3102. Rest-Stress
CID 3104. Cardiac Synchronization Technique
CID 3106. PET Cardiology Protocols
CID 3107. PET Cardiology Radiopharmaceuticals
CID 3108. NM/PET Procedures
CID 3110. Nuclear Cardiology Protocols
CID 3111. Nuclear Cardiology Radiopharmaceuticals
CID 3112. Attenuation Correction
CID 3113. Types of Perfusion Defects
CID 3114. Study Quality
CID 3115. Stress Imaging Quality Issues
CID 3116. NM Extracardiac Findings
CID 3117. Attenuation Correction Methods
CID 3118. Level of Risk
CID 3119. LV Function
CID 3120. Perfusion Findings
CID 3121. Perfusion Morphology
CID 3122. Ventricular Enlargement
CID 3200. Stress Test Procedure
CID 3201. Indications for Stress Test
CID 3202. Chest Pain
CID 3203. Exerciser Device
CID 3204. Stress Agents
CID 3205. Indications for Pharmacological Stress Test
CID 3206. Non-invasive Cardiac Imaging Procedures
CID 3207. Stress Test Procedure Phases
CID 3208. Summary Codes Exercise ECG
CID 3209. Summary Codes Stress Imaging
CID 3210. Speed of Response
CID 3211. BP Response
CID 3212. Treadmill Speed
CID 3213. Stress Hemodynamic Findings
CID 3215. Perfusion Finding Method
CID 3217. Comparison Finding
CID 3220. Stress Symptoms
CID 3221. Stress Test Termination Reasons
CID 3227. QTc Measurements
CID 3228. ECG Timing Measurements
CID 3229. ECG Axis Measurements
CID 3230. ECG Findings
CID 3231. ST Segment Findings
CID 3232. ST Segment Location
CID 3233. ST Segment Morphology
CID 3234. Ectopic Beat Morphology
CID 3235. Perfusion Comparison Findings
CID 3236. Tolerance Comparison Findings
CID 3237. Wall Motion Comparison Findings
CID 3238. Stress Scoring Scales
CID 3239. Perceived Exertion Scales
CID 3240. Electrophysiology Measurement Functions and Techniques
CID 3241. Hemodynamic Measurement Techniques
CID 3250. Catheterization Procedure Phase
CID 3254. Electrophysiology Procedure Phase
CID 3261. Stress Protocols
CID 3262. ECG Patient State Values
CID 3263. Electrode Placement Values
CID 3271. Hemodynamic Physiological Challenges
CID 3335. ECG Annotations
CID 3337. Hemodynamic Annotations
CID 3339. Electrophysiology Annotations
CID 3400. Procedure Log Titles
CID 3401. Types of Log Notes
CID 3402. Patient Status and Events
CID 3403. Percutaneous Entry
CID 3404. Staff Actions
CID 3405. Procedure Action Values
CID 3406. Non-coronary Transcatheter Interventions
CID 3407. Purpose of Reference to Object
CID 3408. Actions With Consumables
CID 3409. Administration of Drugs/Contrast
CID 3410. Numeric Parameters of Drugs/Contrast
CID 3411. Intracoronary Devices
CID 3412. Intervention Actions and Status
CID 3413. Adverse Outcomes
CID 3414. Procedure Urgency
CID 3415. Cardiac Rhythms
CID 3416. Respiration Rhythms
CID 3418. Lesion Risk
CID 3419. Findings Titles
CID 3421. Procedure Action
CID 3422. Device Use Actions
CID 3423. Numeric Device Characteristics
CID 3425. Intervention Parameters
CID 3426. Consumables Parameters
CID 3427. Equipment Events
CID 3428. Imaging Procedures
CID 3429. Catheterization Devices
CID 3430. DateTime Qualifiers
CID 3440. Peripheral Pulse Locations
CID 3441. Patient Assessments
CID 3442. Peripheral Pulse Methods
CID 3446. Skin Condition
CID 3448. Airway Assessment
CID 3451. Calibration Objects
CID 3452. Calibration Methods
CID 3453. Cardiac Volume Methods
CID 3455. Index Methods
CID 3456. Sub-segment Methods
CID 3458. Contour Realignment
CID 3460. Circumferential Extent
CID 3461. Regional Extent
CID 3462. Chamber Identification
CID 3463. Ventricle Identification
CID 3465. QA Reference Methods
CID 3466. Plane Identification
CID 3467. Ejection Fraction
CID 3468. ED Volume
CID 3469. ES Volume
CID 3470. Vessel Lumen Cross-sectional Area Calculation Methods
CID 3471. Estimated Volumes
CID 3472. Cardiac Contraction Phase
CID 3480. IVUS Procedure Phases
CID 3481. IVUS Distance Measurements
CID 3482. IVUS Area Measurements
CID 3483. IVUS Longitudinal Measurements
CID 3484. IVUS Indices and Ratios
CID 3485. IVUS Volume Measurements
CID 3486. Vascular Measurement Sites
CID 3487. Intravascular Volumetric Regions
CID 3488. Min/Max/Mean
CID 3489. Calcium Distribution
CID 3491. IVUS Lesion Morphologies
CID 3492. Vascular Dissection Classifications
CID 3493. IVUS Relative Stenosis Severities
CID 3494. IVUS Non Morphological Findings
CID 3495. IVUS Plaque Composition
CID 3496. IVUS Fiducial Points
CID 3497. IVUS Arterial Morphology
CID 3500. Pressure Units
CID 3502. Hemodynamic Resistance Units
CID 3503. Indexed Hemodynamic Resistance Units
CID 3510. Catheter Size Units
CID 3515. Specimen Collection
CID 3520. Blood Source Type
CID 3524. Blood Gas Pressures
CID 3525. Blood Gas Content
CID 3526. Blood Gas Saturation
CID 3527. Blood Base Excess
CID 3528. Blood pH
CID 3529. Arterial / Venous Content
CID 3530. Oxygen Administration Actions
CID 3531. Oxygen Administration
CID 3550. Circulatory Support Actions
CID 3551. Ventilation Actions
CID 3552. Pacing Actions
CID 3553. Circulatory Support
CID 3554. Ventilation
CID 3555. Pacing
CID 3560. Blood Pressure Methods
CID 3600. Relative Times
CID 3602. Hemodynamic Patient State
CID 3604. Arterial Lesion Locations
CID 3606. Arterial Source Locations
CID 3607. Venous Source Locations
CID 3608. Atrial Source Locations
CID 3609. Ventricular Source Locations
CID 3610. Gradient Source Locations
CID 3611. Pressure Measurements
CID 3612. Blood Velocity Measurements
CID 3613. Hemodynamic Time Measurements
CID 3614. Valve Areas, Non-mitral
CID 3615. Valve Areas
CID 3616. Hemodynamic Period Measurements
CID 3617. Valve Flows
CID 3618. Hemodynamic Flows
CID 3619. Hemodynamic Resistance Measurements
CID 3620. Hemodynamic Ratios
CID 3621. Fractional Flow Reserve
CID 3627. Measurement Type
CID 3628. Cardiac Output Methods
CID 3629. Procedure Intent
CID 3630. Cardiovascular Anatomic Locations
CID 3640. Hypertension
CID 3641. Hemodynamic Assessments
CID 3642. Degree Findings
CID 3651. Hemodynamic Measurement Phase
CID 3663. Body Surface Area Equations
CID 3664. Oxygen Consumption Equations and Tables
CID 3666. P50 Equations
CID 3667. Framingham Scores
CID 3668. Framingham Tables
CID 3670. ECG Procedure Types
CID 3671. Reason for ECG Exam
CID 3676. Lead Measurement Technique
CID 3677. Summary Codes ECG
CID 3678. QT Correction Algorithms
CID 3680. ECG Lead Noise Descriptions
CID 3687. Electrophysiology Waveform Durations
CID 3688. Electrophysiology Waveform Voltages
CID 3689. ECG Global Waveform Durations
CID 3690. ECG Control Variables Numeric
CID 3691. ECG Control Variables Text
CID 3700. Cath Diagnosis
CID 3701. Cardiac Valves and Tracts
CID 3703. Wall Motion
CID 3704. Myocardium Wall Morphology Findings
CID 3705. Chamber Size
CID 3706. Overall Contractility
CID 3707. VSD Description
CID 3709. Aortic Root Description
CID 3710. Coronary Dominance
CID 3711. Valvular Abnormalities
CID 3712. Vessel Descriptors
CID 3713. TIMI Flow Characteristics
CID 3714. Thrombus
CID 3715. Lesion Margin
CID 3716. Severity
CID 3717. Myocardial Wall Segments
CID 3718. Myocardial Wall Segments in Projection
CID 3719. Canadian Clinical Classification
CID 3721. Cardiovascular Surgeries
CID 3722. Diabetic Therapy
CID 3723. MI Types
CID 3724. Smoking History
CID 3726. Indications for Coronary Intervention
CID 3727. Indications for Catheterization
CID 3728. Cath Findings
CID 3729. Admission Status
CID 3730. Insurance Payor
CID 3733. Primary Cause of Death
CID 3735. Acute Coronary Syndrome Time Period
CID 3736. NYHA Classification
CID 3737. Non-invasive Test - Ischemia
CID 3738. Pre-Cath Angina Type
CID 3739. Cath Procedure Type
CID 3740. Thrombolytic Administration
CID 3741. Medication Administration, Lab Visit
CID 3742. Medication Administration, PCI
CID 3743. Clopidogrel/Ticlopidine Administration
CID 3744. EF Testing Method
CID 3745. Calculation Method
CID 3746. Percutaneous Entry Site
CID 3747. Percutaneous Closure
CID 3748. Angiographic EF Testing Method
CID 3749. PCI Procedure Result
CID 3750. Previously Dilated Lesion
CID 3752. Guidewire Crossing
CID 3754. Vascular Complications
CID 3755. Cath Complications
CID 3756. Cardiac Patient Risk Factors
CID 3757. Cardiac Diagnostic Procedures
CID 3758. Cardiovascular Family History
CID 3760. Hypertension Therapy
CID 3761. Antilipemic Agents
CID 3762. Antiarrhythmic Agents
CID 3764. Myocardial Infarction Therapies
CID 3769. Concern Types
CID 3770. Problem Status
CID 3772. Health Status
CID 3773. Use Status
CID 3774. Social History
CID 3777. Implanted Devices
CID 3778. Stages
CID 3802. Plaque Structures
CID 3804. Stenosis Measurement Methods
CID 3805. Stenosis Types
CID 3806. Stenosis Shape
CID 3807. Volume Measurement Methods
CID 3808. Aneurysm Types
CID 3809. Associated Conditions
CID 3810. Vascular Morphology
CID 3813. Stent Findings
CID 3814. Stent Composition
CID 3815. Source of Vascular Finding
CID 3817. Vascular Sclerosis Types
CID 3820. Non-invasive Vascular Procedures
CID 3821. Papillary Muscle Included/Excluded
CID 3823. Respiratory Status
CID 3826. Heart Rhythm
CID 3827. Vessel Segments
CID 3829. Pulmonary Arteries
CID 3831. Stenosis Length
CID 3832. Stenosis Grade
CID 3833. Cardiac Ejection Fraction
CID 3835. Cardiac Volume Measurements
CID 3836. Time-based Perfusion Measurements
CID 3837. Fiducial Feature
CID 3838. Diameter Derivation
CID 3839. Coronary Veins
CID 3840. Pulmonary Veins
CID 3843. Myocardial Subsegment
CID 3850. Intravascular OCT Flush Agent
CID 4005. Partial View Section for Mammography
CID 4009. DX Anatomy Imaged
CID 4010. DX View
CID 4011. DX View Modifier
CID 4012. Projection Eponymous Name
CID 4013. Anatomic Region for Mammography
CID 4014. View for Mammography
CID 4015. View Modifier for Mammography
CID 4016. Anatomic Region for Intra-oral Radiography
CID 4017. Anatomic Region Modifier for Intra-oral Radiography
CID 4018. Primary Anatomic Structure for Intra-oral Radiography (Permanent Dentition - Designation of Teeth)
CID 4019. Primary Anatomic Structure for Intra-oral Radiography (Deciduous Dentition - Designation of Teeth)
CID 4020. PET Radionuclide
CID 4021. PET Radiopharmaceutical
CID 4025. Primary Anatomic Structure for Intra-oral Radiography (Supernumerary Dentition - Designation of Teeth)
CID 4026. Primary Anatomic Structure for Intra-oral and Craniofacial Radiography - Teeth
CID 4028. Craniofacial Anatomic Regions
CID 4030. CT, MR and PET Anatomy Imaged
CID 4031. Common Anatomic Regions
CID 4032. MR Spectroscopy Metabolites
CID 4033. MR Proton Spectroscopy Metabolites
CID 4040. Endoscopy Anatomic Regions
CID 4042. XA/XRF Anatomy Imaged
CID 4050. Drug or Contrast Agent Characteristics
CID 4051. General Devices
CID 4052. Phantom Devices
CID 4100. T1 Measurement Methods
CID 4101. Tracer Kinetic Models
CID 4102. Perfusion Measurement Methods
CID 4103. Arterial Input Function Measurement Methods
CID 4104. Bolus Arrival Time Derivation Methods
CID 4105. Perfusion Analysis Methods
CID 4106. Quantitative Methods used for Perfusion And Tracer Kinetic Models
CID 4107. Tracer Kinetic Model Parameters
CID 4108. Perfusion Model Parameters
CID 4109. Model-Independent Dynamic Contrast Analysis Parameters
CID 4110. Tracer Kinetic Modeling Covariates
CID 4111. Contrast Characteristics
CID 4200. Ophthalmic Imaging Agent
CID 4201. Patient Eye Movement Command
CID 4202. Ophthalmic Photography Acquisition Device
CID 4203. Ophthalmic Photography Illumination
CID 4204. Ophthalmic Filter
CID 4205. Ophthalmic Lens
CID 4206. Ophthalmic Channel Description
CID 4207. Ophthalmic Image Position
CID 4208. Mydriatic Agent
CID 4209. Ophthalmic Anatomic Structure Imaged
CID 4210. Ophthalmic Tomography Acquisition Device
CID 4211. Ophthalmic OCT Anatomic Structure Imaged
CID 4214. Ophthalmic Horizontal Directions
CID 4215. Ophthalmic Vertical Directions
CID 4216. Ophthalmic Visual Acuity Type
CID 4220. Visual Fixation Quality During Acquisition
CID 4221. Visual Fixation Quality Problem
CID 4222. Ophthalmic Macular Grid Problem
CID 4230. Ophthalmic Ultrasound Axial Measurements Type
CID 4231. Lens Status
CID 4232. Vitreous Status
CID 4233. Ophthalmic Axial Length Measurements Segment Names
CID 4234. Refractive Surgery Types
CID 4235. Keratometry Descriptors
CID 4236. IOL Calculation Formula
CID 4237. Lens Constant Type
CID 4238. Refractive Error Types
CID 4239. Anterior Chamber Depth Definition
CID 4240. Ophthalmic Measurement or Calculation Data Source
CID 4241. Ophthalmic Axial Length Selection Method
CID 4243. Ophthalmic Quality Metric Type
CID 4244. Ophthalmic Agent Concentration Units
CID 4245. Wide Field Ophthalmic Photography Transformation Method
CID 4250. Visual Field Static Perimetry Test Patterns
CID 4251. Visual Field Static Perimetry Test Strategies
CID 4252. Visual Field Static Perimetry Screening Test Modes
CID 4253. Visual Field Static Perimetry Fixation Strategy
CID 4254. Visual Field Static Perimetry Test Analysis Results
CID 4255. Visual Field Illumination Color
CID 4256. Visual Field Procedure Modifier
CID 4257. Visual Field Global Index Name
CID 4260. Ophthalmic Mapping Units for Real World Value Mapping
CID 4261. Ophthalmic Mapping Acquisition Method
CID 4262. Retinal Thickness Definition
CID 4263. Ophthalmic Thickness Map Value Type
CID 4264. Ophthalmic Map Purposes of Reference
CID 4265. Ophthalmic Thickness Deviation Categories
CID 4266. Ophthalmic Anatomic Structure Reference Point
CID 4267. Corneal Topography Mapping Units for Real World Value Mapping
CID 4268. Corneal Topography Map Value Type
CID 4270. OCT-A Processing Algorithm Families
CID 4271. En Face Image Types
CID 4272. OPT Scan Pattern Types
CID 4273. Retinal Segmentation Surfaces
CID 6000. Overall Breast Composition
CID 6001. Overall Breast Composition from BI-RADS®
CID 6002. Change Since Last Mammogram or Prior Surgery
CID 6003. Change Since Last Mammogram or Prior Surgery from BI-RADS®
CID 6004. Mammography Characteristics of Shape
CID 6005. Characteristics of Shape from BI-RADS®
CID 6006. Mammography Characteristics of Margin
CID 6007. Characteristics of Margin from BI-RADS®
CID 6008. Density Modifier
CID 6009. Density Modifier from BI-RADS®
CID 6010. Mammography Calcification Types
CID 6011. Calcification Types from BI-RADS®
CID 6012. Calcification Distribution Modifier
CID 6013. Calcification Distribution Modifier from BI-RADS®
CID 6014. Mammography Single Image Finding
CID 6015. Single Image Finding from BI-RADS®
CID 6016. Mammography Composite Feature
CID 6017. Composite Feature from BI-RADS®
CID 6018. Clockface Location or Region
CID 6019. Clockface Location or Region from BI-RADS®
CID 6020. Quadrant Location
CID 6021. Quadrant Location from BI-RADS®
CID 6022. Side
CID 6023. Side from BI-RADS®
CID 6024. Depth
CID 6025. Depth from BI-RADS®
CID 6026. Mammography Assessment
CID 6027. Assessment from BI-RADS®
CID 6028. Mammography Recommended Follow-up
CID 6029. Recommended Follow-up from BI-RADS®
CID 6030. Mammography Pathology Codes
CID 6031. Benign Pathology Codes from BI-RADS®
CID 6032. High Risk Lesions Pathology Codes from BI-RADS®
CID 6033. Malignant Pathology Codes from BI-RADS®
CID 6034. Intended Use of CAD Output
CID 6035. Composite Feature Relations
CID 6036. Scope of Feature
CID 6037. Mammography Quantitative Temporal Difference Type
CID 6038. Mammography Qualitative Temporal Difference Type
CID 6039. Nipple Characteristic
CID 6040. Non-lesion Object Type
CID 6041. Mammography Image Quality Finding
CID 6042. Status of Results
CID 6043. Types of Mammography CAD Analysis
CID 6044. Types of Image Quality Assessment
CID 6045. Mammography Types of Quality Control Standard
CID 6046. Units of Follow-up Interval
CID 6047. CAD Processing and Findings Summary
CID 6048. CAD Operating Point Axis Label
CID 6050. Breast Procedure Reported
CID 6051. Breast Procedure Reason
CID 6052. Breast Imaging Report Section Title
CID 6053. Breast Imaging Report Elements
CID 6054. Breast Imaging Findings
CID 6055. Breast Clinical Finding or Indicated Problem
CID 6056. Associated Findings for Breast
CID 6057. Ductography Findings for Breast
CID 6058. Procedure Modifiers for Breast
CID 6059. Breast Implant Types
CID 6060. Breast Biopsy Techniques
CID 6061. Breast Imaging Procedure Modifiers
CID 6062. Interventional Procedure Complications
CID 6063. Interventional Procedure Results
CID 6064. Ultrasound Findings for Breast
CID 6065. Instrument Approach
CID 6066. Target Confirmation
CID 6067. Fluid Color
CID 6068. Tumor Stages From AJCC
CID 6069. Nottingham Combined Histologic Grade
CID 6070. Bloom-Richardson Histologic Grade
CID 6071. Histologic Grading Method
CID 6072. Breast Implant Findings
CID 6080. Gynecological Hormones
CID 6081. Breast Cancer Risk Factors
CID 6082. Gynecological Procedures
CID 6083. Procedures for Breast
CID 6084. Mammoplasty Procedures
CID 6085. Therapies for Breast
CID 6086. Menopausal Phase
CID 6087. General Risk Factors
CID 6088. OB-GYN Maternal Risk Factors
CID 6089. Substances
CID 6090. Relative Usage, Exposure Amount
CID 6091. Relative Frequency of Event Values
CID 6092. Quantitative Concepts for Usage, Exposure
CID 6093. Qualitative Concepts for Usage, Exposure Amount
CID 6094. Qualitative Concepts for Usage, Exposure Frequency
CID 6095. Numeric Properties of Procedures
CID 6096. Pregnancy Status
CID 6097. Side of Family
CID 6100. Chest Component Categories
CID 6101. Chest Finding or Feature
CID 6102. Chest Finding or Feature Modifier
CID 6103. Abnormal Lines Finding or Feature
CID 6104. Abnormal Opacity Finding or Feature
CID 6105. Abnormal Lucency Finding or Feature
CID 6106. Abnormal Texture Finding or Feature
CID 6107. Width Descriptor
CID 6108. Chest Anatomic Structure Abnormal Distribution
CID 6109. Radiographic Anatomy Finding or Feature
CID 6110. Lung Anatomy Finding or Feature
CID 6111. Bronchovascular Anatomy Finding or Feature
CID 6112. Pleura Anatomy Finding or Feature
CID 6113. Mediastinum Anatomy Finding or Feature
CID 6114. Osseous Anatomy Finding or Feature
CID 6115. Osseous Anatomy Modifiers
CID 6116. Muscular Anatomy
CID 6117. Vascular Anatomy
CID 6118. Size Descriptor
CID 6119. Chest Border Shape
CID 6120. Chest Border Definition
CID 6121. Chest Orientation Descriptor
CID 6122. Chest Content Descriptor
CID 6123. Chest Opacity Descriptor
CID 6124. Location in Chest
CID 6125. General Chest Location
CID 6126. Location in Lung
CID 6127. Segment Location in Lung
CID 6128. Chest Distribution Descriptor
CID 6129. Chest Site Involvement
CID 6130. Severity Descriptor
CID 6131. Chest Texture Descriptor
CID 6132. Chest Calcification Descriptor
CID 6133. Chest Quantitative Temporal Difference Type
CID 6134. Chest Qualitative Temporal Difference Type
CID 6135. Image Quality Finding
CID 6136. Chest Types of Quality Control Standard
CID 6137. Types of CAD Analysis
CID 6138. Chest Non-lesion Object Type
CID 6139. Non-lesion Modifiers
CID 6140. Calculation Methods
CID 6141. Attenuation Coefficient Measurements
CID 6142. Calculated Value
CID 6143. Lesion Response
CID 6144. RECIST Defined Lesion Response
CID 6145. Baseline Category
CID 6146. Time Point Types
CID 6147. Response Criteria
CID 6151. Background Echotexture
CID 6152. Orientation
CID 6153. Lesion Boundary
CID 6154. Echo Pattern
CID 6155. Posterior Acoustic Features
CID 6157. Vascularity
CID 6158. Correlation to Other Findings
CID 6159. Malignancy Type
CID 6160. Breast Primary Tumor Assessment From AJCC
CID 6161. Clinical Regional Lymph Node Assessment for Breast
CID 6162. Assessment of Metastasis for Breast
CID 6163. Menstrual Cycle Phase
CID 6164. Time Intervals
CID 6165. Breast Linear Measurements
CID 6166. CAD Geometry Secondary Graphical Representation
CID 6200. Colon Overall Assessment
CID 6201. Colon Finding or Feature
CID 6202. Colon Finding or Feature Modifier
CID 6203. Colon Non-lesion Object Type
CID 6204. Anatomic Non-colon Findings
CID 6205. Clockface Location for Colon
CID 6206. Recumbent Patient Orientation for Colon
CID 6207. Colon Quantitative Temporal Difference Type
CID 6208. Colon Types of Quality Control Standard
CID 6209. Colon Morphology Descriptor
CID 6210. Location in Intestinal Tract
CID 6211. Colon CAD Material Description
CID 6212. Calculated Value for Colon Findings
CID 6300. Prostate Sector Anatomy
CID 6301. Prostate Sector Anatomy from PI-RADS v2
CID 6302. Prostate Sector Anatomy from European Concensus 16 Sector (Minimal) Model
CID 6303. Prostate Sector Anatomy from European Concensus 27 Sector (Optimal) Model
CID 6401. Non-lesion Object Type - Physical Objects
CID 6402. Non-lesion Object Type - Substances
CID 6403. Non-lesion Object Type - Tissues
CID 6404. Chest Non-lesion Object Type - Physical Objects
CID 6405. Chest Non-lesion Object Type - Tissues
CID 7000. Diagnostic Imaging Report Document Titles
CID 7001. Diagnostic Imaging Report Headings
CID 7002. Diagnostic Imaging Report Elements
CID 7003. Diagnostic Imaging Report Purposes of Reference
CID 7004. Waveform Purposes of Reference
CID 7005. Contributing Equipment Purposes of Reference
CID 7006. SR Document Purposes of Reference
CID 7008. Media Import
CID 7009. Purpose of Reference to Predecessor Report
CID 7010. Key Object Selection Document Title
CID 7011. Rejected for Quality Reasons
CID 7012. Best in Set
CID 7013. Non-Image Source Instance Purposes of Reference
CID 7014. Export Additional Information Document Titles
CID 7015. Export Delay Reasons
CID 7016. Level of Difficulty
CID 7017. Category of Teaching Material - Imaging
CID 7018. Miscellaneous Document Titles
CID 7019. Segmentation Non-Image Source Purposes of Reference
CID 7021. Measurement Report Document Titles
CID 7022. Radiotherapy Purposes of Reference
CID 7023. RT Process Output
CID 7024. RT Process Input
CID 7025. RT Process Input Used
CID 7026. Radiotherapeutic Dose Measurement Devices
CID 7030. Institutional Departments, Units and Services
CID 7035. Actionable Finding Classification
CID 7036. Image Quality Assessment
CID 7039. Pediatric Size Categories
CID 7040. Broselow-Luten Pediatric Size Categories
CID 7041. Calcium Scoring Patient Size Categories
CID 7042. CMDCTECC Calcium Scoring Patient Size Categories
CID 7050. De-identification Method
CID 7100. RCS Registration Method Type
CID 7101. Brain Atlas Fiducials
CID 7110. Fiducials Categories
CID 7111. Fiducials
CID 7140. Brain Structures for Volumetric Measurements
CID 7150. Segmentation Property Categories
CID 7151. Segmentation Property Types
CID 7152. Cardiac Structure Segmentation Types
CID 7153. CNS Segmentation Types
CID 7154. Abdominal Segmentation Types
CID 7155. Thoracic Segmentation Types
CID 7156. Vascular Segmentation Types
CID 7157. Device Segmentation Types
CID 7158. Artifact Segmentation Types
CID 7159. Lesion Segmentation Types
CID 7160. Pelvic Organ Segmentation Types
CID 7161. Physiology Segmentation Types
CID 7162. Surface Processing Algorithm Families
CID 7165. Abstract Segmentation Types
CID 7166. Common Tissue Segmentation Types
CID 7167. Peripheral Nervous System Segmentation Types
CID 7180. Abstract Multi-dimensional Image Model Component Semantics
CID 7181. Abstract Multi-dimensional Image Model Component Units
CID 7182. Abstract Multi-dimensional Image Model Dimension Semantics
CID 7183. Abstract Multi-dimensional Image Model Dimension Units
CID 7184. Abstract Multi-dimensional Image Model Axis Direction
CID 7185. Abstract Multi-dimensional Image Model Axis Orientation
CID 7186. Abstract Multi-dimensional Image Model Qualitative Dimension Sample Semantics
CID 7191. Tissue Segmentation Property Types
CID 7192. Anatomical Structure Segmentation Property Types
CID 7193. Physical Object Segmentation Property Types
CID 7194. Morphological Abnormal Structure Segmentation Property Types
CID 7195. Function Segmentation Property Types
CID 7196. Spatial and Relational Concept Segmentation Property Types
CID 7197. Body Substance Segmentation Property Types
CID 7198. Substance Segmentation Property Types
CID 7201. Referenced Image Purposes of Reference
CID 7202. Source Image Purposes of Reference
CID 7203. Image Derivation
CID 7205. Purpose of Reference to Alternate Representation
CID 7210. Related Series Purposes of Reference
CID 7215. Spectroscopy Purpose of Reference
CID 7220. RT Dose Derivation
CID 7221. RT Dose Purpose of Reference
CID 7222. Parametric Map Derivation Image Purpose of Reference
CID 7250. Multi-Frame Subset Type
CID 7260. Diffusion Acquisition Value Types
CID 7261. Diffusion Model Value Types
CID 7262. Diffusion Tractography Algorithm Families
CID 7263. Diffusion Tractography Measurement Types
CID 7270. MR Diffusion Component Semantics
CID 7271. MR Diffusion Anisotropy Indices
CID 7272. MR Diffusion Model Parameters
CID 7273. MR Diffusion Models
CID 7274. MR Diffusion Model Fitting Methods
CID 7275. MR Diffusion Model Specific Methods
CID 7276. MR Diffusion Model Inputs
CID 7277. Units of Diffusion Rate Area Over Time
CID 7300. Implant Materials
CID 7301. Intervention Types
CID 7302. Implant Templates View Orientations
CID 7303. Implant Templates Modified View Orientations
CID 7304. Implant Target Anatomy
CID 7305. Implant Planning Landmarks
CID 7306. Human Hip Implant Planning Landmarks
CID 7307. Implant Component Types
CID 7308. Human Hip Implant Component Types
CID 7309. Human Trauma Implant Component Types
CID 7310. Implant Fixation Method
CID 7320. Planning Methods
CID 7445. Device Participating Roles
CID 7449. Reader Specialty
CID 7450. Person Roles
CID 7451. Family Member
CID 7452. Organizational Roles
CID 7453. Performing Roles
CID 7454. Animal Taxonomic Rank Values
CID 7455. Sex
CID 7456. Units of Measure for Age
CID 7457. Sex - Male Female or Both
CID 7460. Units of Linear Measurement
CID 7461. Units of Area Measurement
CID 7462. Units of Volume Measurement
CID 7464. General Region of Interest Measurement Modifiers
CID 7465. Measurements Derived From Multiple ROI Measurements
CID 7466. PET Region of Interest Measurements
CID 7467. Grey Level Co-occurrence Matrix Measurements
CID 7468. Texture Measurements
CID 7469. Generic Intensity and Size Measurements
CID 7470. Linear Measurements
CID 7471. Area Measurements
CID 7472. Volume Measurements
CID 7473. General Area Calculation Methods
CID 7474. General Volume Calculation Methods
CID 7480. Breed
CID 7481. Breed Registry
CID 7482. DX Anatomy Imaged for Animals
CID 7483. Common Anatomic Regions for Animals
CID 7484. DX View for Animals
CID 7486. Mixed Breeds
CID 7490. Research Animal Source Registries
CID 7600. Lymph Node Anatomic Sites
CID 7601. Head and Neck Cancer Anatomic Sites
CID 7701. Fiber Tracts In Brainstem
CID 7702. Projection and Thalamic Fibers
CID 7703. Association Fibers
CID 7704. Limbic System Tracts
CID 7705. Commissural Fibers
CID 7706. Cranial Nerves
CID 7707. Spinal Cord Fibers
CID 7710. Tractography Anatomic Sites
CID 8101. Container Types
CID 8102. Container Component Types
CID 8103. Anatomic Pathology Specimen Types
CID 8104. Breast Tissue Specimen Types
CID 8109. Specimen Collection Procedure
CID 8110. Specimen Sampling Procedure
CID 8111. Specimen Preparation Procedure
CID 8112. Specimen Stains
CID 8113. Specimen Preparation Steps
CID 8114. Specimen Fixatives
CID 8115. Specimen Embedding Media
CID 8120. WSI Referenced Image Purposes of Reference
CID 8121. Microscopy Lens Type
CID 8122. Microscopy Illuminator and Sensor Color
CID 8123. Microscopy Illumination Method
CID 8124. Microscopy Filter
CID 8125. Microscopy Illuminator Type
CID 8130. Staining Protocols
CID 8131. Pathology Imaging Protocols
CID 8132. Magnification Selection
CID 8133. Tissue Selection
CID 8201. Surface Scan Acquisition Types
CID 8202. Surface Scan Mode Types
CID 8203. Surface Scan Registration Method Types
CID 8300. Visual Evaluation Methods
CID 8301. Test Pattern Codes
CID 8302. Measurement Pattern Codes
CID 8303. Display Device Type
CID 9000. Physical Quantity Descriptors
CID 9231. Workitem Definition
CID 9233. Requested Report Types
CID 9241. Radiotherapy General Workitem Definition
CID 9242. Radiotherapy Acquisition Workitem Definition
CID 9243. Radiotherapy Registration Workitem Definition
CID 9250. Scheduled Processing Parameter Concept Codes for RT Treatment
CID 9300. Procedure Discontinuation Reasons
CID 9301. Modality PPS Discontinuation Reasons
CID 9302. Media Import PPS Discontinuation Reasons
CID 9401. IEC61217 Device Position Parameters
CID 9402. IEC61217 Gantry Position Parameters
CID 9403. IEC61217 Patient Support Position Parameters
CID 10000. Scope of Accumulation
CID 10001. UID Types
CID 10002. Irradiation Event Types
CID 10003. Equipment Plane Identification
CID 10004. Fluoro Modes
CID 10006. X-Ray Filter Materials
CID 10007. X-Ray Filter Types
CID 10008. Dose Related Distance Measurements
CID 10009. Measured/Calculated
CID 10010. Dose Measurement Devices
CID 10011. Effective Dose Evaluation Method
CID 10013. CT Acquisition Type
CID 10014. Contrast Imaging Technique
CID 10015. CT Dose Reference Authorities
CID 10016. Anode Target Material
CID 10017. X-Ray Grid
CID 10020. Source of Projection X-Ray Dose Information
CID 10021. Source of CT Dose Information
CID 10022. Label Types
CID 10023. Size Specific Dose Estimation Method for CT
CID 10024. Water Equivalent Diameter Method
CID 10025. Radiation Dose Reference Points
CID 10030. Detector Types
CID 10031. CR/DR Mechanical Configuration
CID 10032. Projection X-Ray Acquisition Device Types
CID 10033. CT Reconstruction Algorithm
CID 10034. Reason for Repeating Acquisition
CID 10040. Radiopharmaceutical Organ Dose Reference Authority
CID 10041. Source of Radioisotope Activity Information
CID 10043. Intravenous Extravasation Symptoms
CID 10044. Radiosensitive Organs
CID 10045. Radiopharmaceutical Patient State
CID 10046. GFR Measurements
CID 10047. GFR Measurement Methods
CID 10050. Summary Radiation Exposure Quantities
CID 10060. Organs for Radiation Dose Estimates
CID 10061. Absorbed Radiation Dose Types
CID 10062. Equivalent Radiation Dose Types
CID 10063. Radiation Dose Estimate Distribution Representation
CID 10064. Patient Model Type
CID 10065. Radiation Transport Model Type
CID 10066. Attenuator Category
CID 10067. Radiation Attenuator Materials
CID 10068. Estimate Method Types
CID 10069. Radiation Dose Estimation Parameter
CID 10070. Radiation Dose Types
CID 12001. Ultrasound Protocol Types
CID 12002. Ultrasound Protocol Stage Types
CID 12003. OB-GYN Dates
CID 12004. Fetal Biometry Ratios
CID 12005. Fetal Biometry Measurements
CID 12006. Fetal Long Bones Biometry Measurements
CID 12007. Fetal Cranium
CID 12008. OB-GYN Amniotic Sac
CID 12009. Early Gestation Biometry Measurements
CID 12011. Ultrasound Pelvis and Uterus
CID 12012. OB Equations and Tables
CID 12013. Gestational Age Equations and Tables
CID 12014. OB Fetal Body Weight Equations and Tables
CID 12015. Fetal Growth Equations and Tables
CID 12016. Estimated Fetal Weight Percentile Equations and Tables
CID 12017. Growth Distribution Rank
CID 12018. OB-GYN Summary
CID 12019. OB-GYN Fetus Summary
CID 12020. Fetal Biometry Anatomic Sites
CID 12021. Fetal Long Bone Anatomic Sites
CID 12022. Fetal Cranium Anatomic Sites
CID 12023. Pelvis and Uterus Anatomic Sites
CID 12030. Ultrasound Contrast/Bolus Agents
CID 12031. Protocol Interval Events
CID 12032. Transducer Scan Pattern
CID 12033. Ultrasound Transducer Geometry
CID 12034. Ultrasound Transducer Beam Steering
CID 12035. Ultrasound Transducer Application
CID 12101. Vascular Summary
CID 12102. Temporal Periods Relating to Procedure or Therapy
CID 12103. Vascular Ultrasound Anatomic Location
CID 12104. Extracranial Arteries
CID 12105. Intracranial Cerebral Vessels
CID 12106. Intracranial Cerebral Vessels (Unilateral)
CID 12107. Upper Extremity Arteries
CID 12108. Upper Extremity Veins
CID 12109. Lower Extremity Arteries
CID 12110. Lower Extremity Veins
CID 12111. Abdominal Arteries (Lateral)
CID 12112. Abdominal Arteries (Unilateral)
CID 12113. Abdominal Veins (Lateral)
CID 12114. Abdominal Veins (Unilateral)
CID 12115. Renal Vessels
CID 12116. Vessel Segment Modifiers
CID 12117. Vessel Branch Modifiers
CID 12118. Measurement Orientation
CID 12119. Vascular Ultrasound Property
CID 12120. Blood Velocity Measurements by Ultrasound
CID 12121. Vascular Indices and Ratios
CID 12122. Other Vascular Properties
CID 12123. Carotid Ratios
CID 12124. Renal Ratios
CID 12140. Pelvic Vasculature Anatomical Location
CID 12141. Fetal Vasculature Anatomical Location
CID 12200. Echocardiography Left Ventricle
CID 12201. Left Ventricle Linear
CID 12202. Left Ventricle Volume
CID 12203. Left Ventricle Other
CID 12204. Echocardiography Right Ventricle
CID 12205. Echocardiography Left Atrium
CID 12206. Echocardiography Right Atrium
CID 12207. Echocardiography Mitral Valve
CID 12208. Echocardiography Tricuspid Valve
CID 12209. Echocardiography Pulmonic Valve
CID 12210. Echocardiography Pulmonary Artery
CID 12211. Echocardiography Aortic Valve
CID 12212. Echocardiography Aorta
CID 12214. Echocardiography Pulmonary Veins
CID 12215. Echocardiography Vena Cavae
CID 12216. Echocardiography Hepatic Veins
CID 12217. Echocardiography Cardiac Shunt
CID 12218. Echocardiography Congenital
CID 12219. Pulmonary Vein Modifiers
CID 12220. Echocardiography Common Measurements
CID 12221. Flow Direction
CID 12222. Orifice Flow Properties
CID 12223. Echocardiography Stroke Volume Origin
CID 12224. Ultrasound Image Modes
CID 12226. Echocardiography Image View
CID 12227. Echocardiography Measurement Method
CID 12228. Echocardiography Volume Methods
CID 12229. Echocardiography Area Methods
CID 12230. Gradient Methods
CID 12231. Volume Flow Methods
CID 12232. Myocardium Mass Methods
CID 12233. Cardiac Phase
CID 12234. Respiration State
CID 12235. Mitral Valve Anatomic Sites
CID 12236. Echo Anatomic Sites
CID 12237. Echocardiography Anatomic Site Modifiers
CID 12238. Wall Motion Scoring Schemes
CID 12239. Cardiac Output Properties
CID 12240. Left Ventricle Area
CID 12241. Tricuspid Valve Finding Sites
CID 12242. Aortic Valve Finding Sites
CID 12243. Left Ventricle Finding Sites
CID 12244. Congenital Finding Sites
CID 12245. Cardiac Ultrasound Report Titles
CID 12246. Cardiac Ultrasound Indication for Study
CID 12247. Pediatric, Fetal and Congenital Cardiac Surgical Interventions
CID 12248. Cardiac Ultrasound Summary Codes
CID 12249. Cardiac Ultrasound Fetal Summary Codes
CID 12250. Cardiac Ultrasound Common Linear Measurements
CID 12251. Cardiac Ultrasound Linear Valve Measurements
CID 12252. Cardiac Ultrasound Cardiac Function
CID 12253. Cardiac Ultrasound Area Measurements
CID 12254. Cardiac Ultrasound Hemodynamic Measurements
CID 12255. Cardiac Ultrasound Myocardium Measurements
CID 12257. Cardiac Ultrasound Left Ventricle
CID 12258. Cardiac Ultrasound Right Ventricle
CID 12259. Cardiac Ultrasound Ventricles Measurements
CID 12260. Cardiac Ultrasound Pulmonary Artery
CID 12261. Cardiac Ultrasound Pulmonary Vein
CID 12262. Cardiac Ultrasound Pulmonary Valve
CID 12263. Cardiac Ultrasound Venous Return Pulmonary Measurements
CID 12264. Cardiac Ultrasound Venous Return Systemic Measurements
CID 12265. Cardiac Ultrasound Atria and Atrial Septum Measurements
CID 12266. Cardiac Ultrasound Mitral Valve
CID 12267. Cardiac Ultrasound Tricuspid Valve
CID 12268. Cardiac Ultrasound Atrioventricular Valves Measurements
CID 12269. Cardiac Ultrasound Interventricular Septum Measurements
CID 12270. Cardiac Ultrasound Aortic Valve
CID 12271. Cardiac Ultrasound Outflow Tracts Measurements
CID 12272. Cardiac Ultrasound Semilunar Valves, Annulate and Sinuses Measurements
CID 12273. Cardiac Ultrasound Aortic Sinotubular Junction
CID 12274. Cardiac Ultrasound Aorta Measurements
CID 12275. Cardiac Ultrasound Coronary Arteries Measurements
CID 12276. Cardiac Ultrasound Aorto Pulmonary Connections Measurements
CID 12277. Cardiac Ultrasound Pericardium and Pleura Measurements
CID 12279. Cardiac Ultrasound Fetal General Measurements
CID 12280. Cardiac Ultrasound Target Sites
CID 12281. Cardiac Ultrasound Target Site Modifiers
CID 12282. Cardiac Ultrasound Venous Return Systemic Finding Sites
CID 12283. Cardiac Ultrasound Venous Return Pulmonary Finding Sites
CID 12284. Cardiac Ultrasound Atria and Atrial Septum Finding Sites
CID 12285. Cardiac Ultrasound Atrioventricular Valves Finding Sites
CID 12286. Cardiac Ultrasound Interventricular Septum Finding Sites
CID 12287. Cardiac Ultrasound Ventricles Finding Sites
CID 12288. Cardiac Ultrasound Outflow Tracts Finding Sites
CID 12289. Cardiac Ultrasound Semilunar Valves, Annulus and Sinuses Finding Sites
CID 12290. Cardiac Ultrasound Pulmonary Arteries Finding Sites
CID 12291. Cardiac Ultrasound Aorta Finding Sites
CID 12292. Cardiac Ultrasound Coronary Arteries Finding Sites
CID 12293. Cardiac Ultrasound Aortopulmonary Connections Finding Sites
CID 12294. Cardiac Ultrasound Pericardium and Pleura Finding Sites
CID 12300. Core Echo Measurements
CID 12301. Measurement Selection Reasons
CID 12302. Echo Finding Observation Types
CID 12303. Echo Measurement Types
CID 12304. Echo Measured Properties
CID 12305. Basic Echo Anatomic Sites
CID 12306. Echo Flow Directions
CID 12307. Cardiac Phases and Time Points
TID 3401. ECG Acquisition Context
TID 3403. Catheterization Acquisition Context
TID 3450. Cardiac Electrophysiology Acquisition Context
TID 3460. Projection Radiography Acquisition Context
TID 3470. NM/PET Acquisition Context
TID 3471. PET Covariates Acquisition Context
TID 8001. Specimen Preparation
TID 8002. Specimen Sampling
TID 8003. Specimen Staining
TID 8004. Specimen Localization
TID 8010. Slide Imaging Parameters
TID 8200. Radiology Reading Task Parameters
TID 15100. Contrast Agent/Pre-Medication Protocol Context
TID 15101. NM/PET Protocol Context
TID 15200. JJ1017 Protocol Context
D-1. DICOM Controlled Terminology Definitions
E-1. French Language Meanings of Selected Codes
E-2. Mapping of Pathology Codes used in DICOM to ADICAP
F-1. Japanese Language Meanings of Selected Codes
G-1. English Code Meanings of Selected Codes
H-1. Code Meanings of LOINC Codes
I-1. Examples of the Common Nomenclature for the Type of Endoscopy Performed
J-1. SNOMED Codes Retired from DICOM Use
J-2. SNOMED Synonyms Retired from DICOM Use
L-1. Corresponding SNOMED Terms for Human Use
L-2. Corresponding SNOMED Terms for Large Animals
L-3. Corresponding Codes for Small Animal Use
L-4. Correspondence between Animal-specific and Generic NCI Thesaurus Codes
L-5. Pairedness of Anatomic Concepts
M-1. German Language Meanings of Selected Codes
N.1-1. HL7 Value Sets
N.1.1-1. ActPriority Value Set
N.1.2-1. AdministrativeGender Value Set
N.1.3-1. ImageMediaType Value Set
N.1.4-1. NullFlavor Value Set
N.1.5-1. ObservationInterpretation Value Set
N.1.6-1. x_BasicConfidentialityKind Value Set
N.1.7-1. x_serviceEventPerformer Value Set
N.2-1. LOINC Value Sets
N.2.1-1. LOINC Imaging Document Codes (examples)
N.2.2-1. LOINC Y/N/NA

# Notice and Disclaimer

The information in this publication was considered technically sound by the consensus of persons engaged in the development and approval of the document at the time it was developed. Consensus does not necessarily mean that there is unanimous agreement among every person participating in the development of this document.

NEMA standards and guideline publications, of which the document contained herein is one, are developed through a voluntary consensus standards development process. This process brings together volunteers and/or seeks out the views of persons who have an interest in the topic covered by this publication. While NEMA administers the process and establishes rules to promote fairness in the development of consensus, it does not write the document and it does not independently test, evaluate, or verify the accuracy or completeness of any information or the soundness of any judgments contained in its standards and guideline publications.

NEMA disclaims liability for any personal injury, property, or other damages of any nature whatsoever, whether special, indirect, consequential, or compensatory, directly or indirectly resulting from the publication, use of, application, or reliance on this document. NEMA disclaims and makes no guaranty or warranty, expressed or implied, as to the accuracy or completeness of any information published herein, and disclaims and makes no warranty that the information in this document will fulfill any of your particular purposes or needs. NEMA does not undertake to guarantee the performance of any individual manufacturer or seller's products or services by virtue of this standard or guide.

In publishing and making this document available, NEMA is not undertaking to render professional or other services for or on behalf of any person or entity, nor is NEMA undertaking to perform any duty owed by any person or entity to someone else. Anyone using this document should rely on his or her own independent judgment or, as appropriate, seek the advice of a competent professional in determining the exercise of reasonable care in any given circumstances. Information and other standards on the topic covered by this publication may be available from other sources, which the user may wish to consult for additional views or information not covered by this publication.

NEMA has no power, nor does it undertake to police or enforce compliance with the contents of this document. NEMA does not certify, test, or inspect products, designs, or installations for safety or health purposes. Any certification or other statement of compliance with any health or safety-related information in this document shall not be attributable to NEMA and is solely the responsibility of the certifier or maker of the statement.

# Foreword

This DICOM Standard was developed according to the procedures of the DICOM Standards Committee.

The DICOM Standard is structured as a multi-part document using the guidelines established in [ISO/IEC Directives, Part 2].

PS3.1 should be used as the base reference for the current parts of this standard.

DICOM® is the registered trademark of the National Electrical Manufacturers Association for its standards publications relating to digital communications of medical information, all rights reserved.

HL7® and CDA® are the registered trademarks of Health Level Seven International, all rights reserved.

SNOMED®, SNOMED Clinical Terms®, SNOMED CT® are the registered trademarks of the International Health Terminology Standards Development Organisation (IHTSDO), all rights reserved.

LOINC® is the registered trademark of Regenstrief Institute, Inc, all rights reserved.

# 1 Scope and Field of Application

This part of the DICOM Standard specifies the DICOM Content Mapping Resource (DCMR), which defines the Templates and context groups used elsewhere in the standard.

# 2 Normative References

The following standards contain provisions that, through reference in this text, constitute provisions of this Standard. At the time of publication, the editions indicated were valid. All standards are subject to revision, and parties to agreements based on this Standard are encouraged to investigate the possibilities of applying the most recent editions of the standards indicated below.

## 2.1 General

[Alderman 1992] Coronary Artery Disease. Alderman EL and Stadius M. 1992. 3. 12. 1189–208. “The angiographic definitions of the bypass angioplasty revascularization investigation”. http://journals.lww.com/coronary-artery/Abstract/1992/12000/The_angiographie_definitions_of_the_Bypass.12.aspx .

[ASTM E 1762-04] ASTM. . Standard Guide for Electronic Authentication of Health Care Information, ASTM International.

[ASTM E 2084-00] ASTM. . Standard Specification for Authentication of Healthcare Information Using Digital Signatures, ASTM International.

[AAPM Report 204] American Association of Physicists in Medicine, College Park, Maryland. 2011. Size-Specific Dose Estimates (SSDE) in Pediatric and Adult Body CT Examinations. http://www.aapm.org/pubs/reports/RPT_204.pdf .

[AAPM Report 220] American Association of Physicists in Medicine. September 2014. Report of AAPM Task Group 220 - Use of Water Equivalent Diameter for Calculating Patient Size and Size-Specific Dose Estimates (SSDE) in CT. http://www.aapm.org/pubs/reports/rpt_220.pdf .

[AAPM OR 03] American Association of Physicists in Medicine. 2005. Assessment of Display Performance for Medical Imaging Systems. http://www.aapm.org/pubs/reports/OR_03.pdf .

[DIN 6868-57] Deutsches Institut für Normung. 2001. Image quality assurance in diagnostic X-ray departments - Acceptance testing for image display devices.

[ETDRS Report Number 10] ETDRS. . Report Number 10, Grading Diabetic Retinopathy from Stereoscopic Color Fundus Photographs- An Extension of the Modified Airlie House Classification. Ophthalmology, May 1991, vol98 (p786-805), Supplement.

[HL7 v3 CMET] HL7. . Version 3 Standard: Common Message Element Types.

[IEC 60601-2-44] IEC. . Medical Electrical Equipment - Part 2-44: Particular Requirements for the Safety of X-Ray Equipment for Computed Tomography.

[IEC 61217] IEC. 2011. Ed 2.0. Radiotherapy Equipment - Coordinates, Movements and Scales.

[IEC 62563-1] IEC. 2009. Ed 1.0. Medical Electrical Equipment - Medical image display systems - Part 1: Evaluation methods.

[ISO/IEC Directives, Part 2] ISO/IEC. 2016/05. 7.0. Rules for the structure and drafting of International Standards. http://www.iec.ch/members_experts/refdocs/iec/isoiecdir-2%7Bed7.0%7Den.pdf .

[ISO 639] ISO. . Codes for the representation of names of languages.

[ISO 639-1] ISO. 2002. Codes for the representation of names of languages — Part 1: Alpha-2 code.

[ISO 639-2] ISO. 1998. Codes for the representation of names of languages — Part 2: Alpha-3 code.

[ISO 639-3] ISO. 2007. Codes for the representation of names of languages — Part 3: Alpha-3 code for comprehensive coverage of languages.

[ISO 3166] ISO. . Codes for the representation of names of countries.

[ISO 3166-1] ISO. . Codes for the representation of names of countries — Part 1: Country codes.

[ISO 8824-1] ISO. 2015. Information Technology - Abstract Syntax 1 (ASN.1): Specification of Basic Notation.

[ISO 9834-1] ISO. 2012. Information technology - Open Systems Interconnection - Procedures for the operation of OSI Registration Authorities: General procedures and top arcs of the ASN.1 Object Identifier tree.

[ISO 15924] ISO. 2004. Codes for the representation of names of scripts.

[JJ1017] Japan Medical Imaging and Radiological Systems Industries Association (JIRA) and Japanese Association of Healthcare Information Systems Industry (JAHIS). Guidelines for HIS, RIS, PACS - Modality Data Communication on Scheduling, Billing, and Examination Records. http://www.jira-net.or.jp/commission/system/04_information/files/JJ1017VER3_20051005.doc .

[NEMA XR 25-2010] National Electrical Manufacturers Association, Rosslyn, Virginia. 2010. Computed Tomography Dose Check Standard. http://www.nema.org/stds/xr25.cfm .

[RadLex] RSNA, Chicago. 2006. A Lexicon for Uniform Indexing and Retrieval of Radiology Information Resources. http://www.radlex.org/ .

[RFC 1766] IETF. March 1995. Tags for the Identification of Languages.

[RFC 3066] IETF. January 2001. Tags for the Identification of Languages.

[RFC 4646] IETF. September 2006. Tags for Identifying Languages.

[RFC 4647] IETF. September 2006. Matching of Language Tags.

[RFC 5646] IETF. September 2009. Tags for Identifying Languages.

[Scanlon 1999] Journal of the American College of Cardiology. Scanlon PJ and et al. May 1999. 33. 6. 1756–824. “ACC/AHA guidelines for coronary angiography - A report of the American College of Cardiology/American Heart Association Task Force on Practice Guidelines (Committee on Coronary Angiography) developed in collaboration with the Society for Cardiac Angiography and Interventions”. doi:10.1016/S0735-1097(99)00126-6 http://www.sciencedirect.com/science/article/pii/S0735109799001266 .

[SMPTE RP133] Society of Motion Picture and Television Engineers (SMPTE). 1991. Specifications for Medical Diagnostic Imaging Test Pattern for Television Monitors and Hard-copy Recording Cameras.

[Stout et al 2013] Molecular Imaging. Stout D, Berr SS, LeBlanc A, Kalen JD, Osborne D, Price J, Schiffer W, Kuntner C, and Wall J. 2013. 12. 7. 1-15. “Guidance for Methods Descriptions Used in Preclinical Imaging Papers”. http://journals.sagepub.com/doi/pdf/10.2310/7290.2013.00055 .

## 2.2 BI-RADS® Terminology and Nomenclature

A portion of the terminology used within the Mammography CAD SR SOP Class and the Breast Imaging Report and Relevant Patient Information for Breast Imaging Templates is derived from BI-RADS®, a copyrighted lexicon of breast imaging terminology and nomenclature licensed by DICOM from the American College of Radiology.

[BI-RADS®] American College of Radiology, Reston, Virginia. 1998. 3.0. Breast Imaging Reporting and Data System Atlas. http://www.acr.org/Quality-Safety/Resources/BIRADS .

## 2.3 MQCM 1999 Terminology and Nomenclature

References to MQCM 1999 are made in the description of the Mammography CAD SR SOP Class. In this MQCM 1999 refers to the Mammography Quality Control Manual 1999, available from the American College of Radiology. This document describes a standardized approach to mammographic acquisition standards, patient positioning, and so on. The DICOM standard does not require Mammography CAD SR SOP Class implementations to adhere to MQCM 1999.

[MQCM] American College of Radiology, Reston, Virginia. 1999. Mammography Quality Control Manual. http://www.acr.org/Education/Education-Catalog/Products/639 .

## 2.4 MQSA Terminology and Nomenclature

References to MQSA are made in the description of the Mammography CAD SR SOP Class. In this MQSA refers to the Mammography Quality Standards Act final rules. While MQSA is a federal regulation of the United States government, it provides the only widely published standards for mammographic quality and is incorporated in this document for that reason. The DICOM standard does not require Mammography CAD SR SOP Class implementations to adhere to MQSA.

[MQSA] U.S. Food and Drug Administration, Silver Spring, Maryland. 2002. Mammography Quality Standards Act Regulations. http://www.fda.gov/Radiation-EmittingProducts/MammographyQualityStandardsActandProgram/Regulations/ucm110906.htm .

## 2.5 ACR Position Statement

[ACR Position Statement] American College of Radiology, Reston, Virginia. 2001. Quality Control and Improvement, Safety, Infection Control, and Patient Education. http://www.acr.org/Quality-Safety/Radiology-Safety .

## 2.6 Chest Radiology and CT

References are made in the description of the Chest CAD SR Templates and context groups.

[Fraser and Pare] Fraser, Muller, Colman, and Pare. 1999. 4th. I. xvii-xxxi. Diagnosis of Diseases of the Chest. Terms Used in Chest Radiology.

[Fraser and Pare] Fraser, Muller, Colman, and Pare. 1999. 4th. I. xxxiii-xxxvi. Diagnosis of Diseases of the Chest. Terms for CT of the Lungs.

[ACR CT PE] American College of Radiology, Reston, Virginia. 2001. 109-113. ACR Standards. ACR Standard for the Performance of Computed Tomography for the Detection of Pulmonary Embolism in Adults.

[ACR HR CT] American College of Radiology, Reston, Virginia. 2001. 115-118. ACR Standards. ACR Standard for the Performance of High-Resolution Computed Tomography (HRCT) of the Lungs in Adults.

[ACR Radiography] American College of Radiology, Reston, Virginia. 2001. 95-98. ACR Standards. ACR Standard for the Performance of Pediatric and Adult Chest Radiography.

[ACR Thoracic CT] American College of Radiology, Reston, Virginia. 2001. 103-107. ACR Standards. ACR Standard for the Performance of Pediatric and Adult Thoracic Computed Tomography (CT).

## 2.7 Response Evaluation Criteria

References to Response Evaluation Criteria are made from the Chest CAD SR Templates and context groups

[RECIST] Journal of the National Cancer Institute. Therasse. February 2, 2000. 92. 3. 205-216. “New Guidelines to Evaluate the Response to Treatment in Solid Tumors”. http://www.eortc.be/recist/ .

[WHO] World Health Organization, Geneva. 1979. WHO Handbook for Reporting Results for Cancer Treatment. WHO Offset Publication No. 48. http://whqlibdoc.who.int/publications/9241700483.pdf .

## 2.8 Myocardial Segmentation

[Cerqueira 2002] Circulation. Cerqueira MD. 2002. 105. 4. 539. “AHA Scientific Statement: Standardized Myocardial Segmentation and Nomenclature for Tomographic Imaging of the Heart”. doi:10.1161/hc0402.102975

## 2.9 Quantitation of the Left Ventricle

[Schiller 1989] Journal of the American Society of Echocardiography. Schiller. Oct 1989. 2. 5. 358-367. “Recommendations for Quantitation of the Left Ventricle by Two-Dimensional Echocardiography”.

## 2.10 Cancer Staging

[AJCC] American Joint Committee on Cancer. 2002. Sixth. AJCC Cancer Staging Handbook: From the AJCC Cancer Staging Manual. Springer-Verlag. 0-387-95270-5.

## 2.11 Quantitative Arteriography and Ventriculography

[Sheehan, 1986] Circulation. Sheehan FH, Bolson EL, Dodge HT, Mathey DG, Schofer J, and Woo HW. 1986. 74. 2. 293-305. “Advantages and applications of the centerline method for characterizing regional ventricular function”. doi:10.1161/01.CIR.74.2.293

[Slager, 1986] J Am Coll Cardiol.. Slager CJ, Hooghoudt TE, Serruys PW, Schuurbiers JC, Reiber JH, Meester GT, Verdouw PD, and Hugenholtz PG. 1986. 6. 2. 317-26. “Quantitative assessment of regional left ventricular motion using endocardial landmarks”. doi:10.1016/S0735-1097(86)80498-3

[Kennedy, 1970] Am Heart J. Kennedy JW, Trenholme SE, and Kasser IS. 1970. 80. 3. 343. “Left ventricular volume and mass from single-plane cineangiocardiogram. A comparison of anteroposterior and right anterior oblique methods”.

[Dodge, 1960] Am Heart J. Dodge HT, Sandler H, Ballew DW, and Lord JD. 1960. 60. 5. 762. “The use of biplane angiocardiography for the measurement of left ventricular volume in man”. http://www.sciencedirect.com/science/article/pii/0002870360903598 .

[Wynne, 1978] Am J Cardiol. Wynne J, Green LH, Mann T, Levin D, and Grossman W. 1978. 41. 4. 726. “Estimation of left ventricular volumes in man from biplane cineangiograms filmed in oblique projections”.

[Boak, 1977] Cathet Cardiovasc Diagn. Boak, JG, Bove AA, Kreulen T, and Spann JF. 1977. 3. 3. 217-30. “A geometric basis for calculation of right ventricular volume in man”. doi:10.1002/ccd.1810030305

[Ferlinz, 1977] Am Heart J. Ferlinz J. 1977. 94. 1. 87-90. “Measurements of right ventricular volumes in man from single plane cineangiograms. A comparison to the biplane approach”. http://www.sciencedirect.com/science/article/pii/S0002870377803487 .

[Graham, 1973] Circulation. Graham TP, Jarmakani JM, Atwood GF, and Canent RV. 1973. 47. 1. 144-53. “Right ventricular volume determinations in children. Normal values and observations with volume or pressure overload”. doi:10.1161/01.CIR.47.1.144

[Arcilla, 1971] Chest. Arcilla RA, Tsai P, Thilenius O, and Ranniger K. 1971. 60. 5. 446. “Angiographic method for volume estimation of right and left ventricles”. doi:10.1378/chest.60.5.446

## 2.12 IVUS

[Mintz, 2001] Journal of the American College of Cardiology. Mintz GS. 2001. 37. 5. 1478-1492. “American College of Cardiology Clinical Expert Consensus Document on Standards for Acquisition, Measurement and Reporting of Intravascular Ultrasound Studies (IVUS)”. doi:10.1016/S0735-1097(01)01175-5

[Di Mario, 1998] European Heart Journal. Di Mario C. 1998. 19. 2. 207-229. “Clinical Application and Image Interpretation in Intravascular Ultrasound”. doi:10.1053/euhj.1996.0433

[Tobis and Yock] Tobis JM and Yock PC. 1992. Intravascular Ultrasound Imaging. 0443088098.

## 2.13 C-RADS CT Colonography Reporting and Data System

[Zalis, 2005] Radiology. Zalis ME. 2005. 236. 1. 3-9. “CT Colonography Reporting and Data System: A Consensus Proposal”. doi:10.1148/radiol.2361041926

## 2.14 Implants

[Eggli, 1998] J Bone Joint Surg Br. Eggli S, Pisan M, and Müller ME. 1998. 80-B. 3. 382-390. “The value of preoperative planning for total hip arthroplasty”. http://www.bjj.boneandjoint.org.uk/content/80-B/3/382 .

## 2.15 LOINC

[LOINC] Regenstrief Institute, Indianapolis. 2014. Logical Observation Identifier Names and Codes. http://loinc.org/ .

This product includes all or a portion of the LOINC® table, LOINC panels and forms file, LOINC document ontology file, and/or LOINC hierarchies file, or is derived from one or more of the foregoing, subject to a license from Regenstrief Institute, Inc. Your use of the LOINC table, LOINC codes, LOINC panels and forms file, LOINC document ontology file, and LOINC hierarchies file also is subject to this license, a copy of which is available at http://loinc.org/terms-of-use. The current complete LOINC table, LOINC Users' Guide, LOINC panels and forms file, LOINC document ontology file, and LOINC hierarchies file are available for download at http://loinc.org/. The LOINC table and LOINC codes are copyright © 1995-2014, Regenstrief Institute, Inc. and the Logical Observation Identifiers Names and Codes (LOINC) Committee. The LOINC panels and forms file, LOINC document ontology file, and LOINC hierarchies file are copyright © 1995-2014, Regenstrief Institute, Inc. All rights reserved.

The LOINC table (in all formats), LOINC panels and forms file, LOINC document ontology file, and LOINC hierarchies are provided "as is." Any express or implied warranties are disclaimed, including, but not limited to, the implied warranties of merchantability and fitness for a particular purpose.

A small portion of the LOINC table may include content (e.g., survey instruments) that is subject to copyrights owned by third parties. Such content has been mapped to LOINC terms under applicable copyright and terms of use. Notice of such third party copyright and license terms would need to be included if such content is included.

## 2.16 UCUM

[UCUM] Regenstrief Institute, Indianapolis. 2013. Unified Code for Units of Measure. http://unitsofmeasure.org/ .

This product includes all or a portion of the UCUM table, UCUM codes, and UCUM definitions or is derived from it, subject to a license from Regenstrief Institute, Inc. and The UCUM Organization. Your use of the UCUM table, UCUM codes, UCUM definitions also is subject to this license, a copy of which is available at http://unitsofmeasure.org/. The current complete UCUM table, UCUM Specification are available for download at http://unitsofmeasure.org/. The UCUM table and UCUM codes are copyright © 1995-2013, Regenstrief Institute, Inc. and the Unified Codes for Units of Measures (UCUM) Organization. All rights reserved.

The UCUM table (in all formats), UCUM definitions, and specification are provided "as is." Any express or implied warranties are disclaimed, including, but not limited to, the implied warranties of merchantability and fitness for a particular purpose.

## 2.17 Anesthesia Quality Institute Schema

[AQI Schema] Anesthesia Quality Institute, Schaumburg, IL. 2015/07/30. Anesthesia Quality Institute Schema. http://www.aqihq.org/aqischdoc/default.html .

Used by permission of the Anesthesia Quality Institute (AQI) (http://www.aqihq.org/), established by the American Society of Anesthesiologists (ASA).

## 2.18 Point-of-Care Medical Device Nomenclature

Extracts of ISO/IEEE 11073 reprinted by permission of IEEE, 3 Park Avenue, New York, NY 10016-5997 USA. Copyright by IEEE. http://standards.ieee.org/.

Under license from IEEE, the term codes and descriptions of the ISO/IEEE 11073 Nomenclature are available at no cost through the Rosetta Terminology Mapping Management System of the U.S. National Institute of Standards and Technology. http://hit-testing.nist.gov:13110/rtmms/index.htm

[ISO/IEEE 11073-10101] ISO/IEEE. 2004. . Health informatics - Point-of-care medical device communication - Nomenclature.

[ISO/IEEE 11073-10101a] ISO/IEEE. 2015. . Health informatics - Point-of-care medical device communication - Nomenclature Amendment 1: Additional Definitions.

[ISO/IEEE 11073-10102] ISO/IEEE. 2015. . Health informatics - Point-of-care medical device communication - Annotated ECG - Nomenclature.

## 2.19 SNOMED Clinical Terms

This DICOM Standard incorporates SNOMED CT®, used by permission of the International Health Terminology Standards Development Organisation (IHTSDO). SNOMED CT®, was originally created by The College of American Pathologists (CAP). SNOMED CT® is a registered trademark of the International Health Terminology Standards Development Organisation , all rights reserved.

The SNOMED CT terms used in this Standard (the SNOMED CT DICOM Subset) are the subject of a licensing agreement between NEMA and IHTSDO that allows the use of this defined subset in DICOM conformant applications without further license or payment of fee. Any use of SNOMED CT beyond the terms published in the DICOM Standard is subject to SNOMED CT licensing rules, which may include a fee. For further information about SNOMED CT licensing, go to http://www.ihtsdo.org/snomed-ct/get-snomed-ct or contact IHTSDO at mailto:info@ihtsdo.org.

This DICOM Standard incorporates various veterinary terms from the SNOMED CT VetSCT extension, used by permission of the Veterinary Terminology Services Laboratory (VTSL) (http://vtsl.vetmed.vt.edu/). These terms were previously included in SNOMED CT but have since been inactivated as moved elsewhere.

[SNOMED] International Health Terminology Standards Development Organisation (IHTSDO). . SNOMED CT Systematized Nomenclature of Medicine - Clinical Terms.

## 2.20 Prostate Reporting Terminology and Nomenclature

The Prostate Imaging and Report and Data System Version 2 (PI-RADS v2) is a joint effort of the European Society of Urogenital Radiology, the American College of Radiology and the AdMetech Foundation.

[PI-RADS v2] Eur Urol. Weinreb JC, Barentsz JO, Choyke PL, Cornud F, Haider MA, Macura KJ, Margolis, D, Schnall, MD, Shtern, F, Tempany, CM., Thoeny, HC, and Verma, S. 2016/01. 69. 1. 16-40. “PI-RADS Prostate Imaging - Reporting and Data System: 2015, Version 2”. doi:10.1016/j.eururo.2015.08.052 http://www.europeanurology.com/article/S0302-2838%2815%2900848-9/ .

### Note

PI-RADS is also available from the following sources:

- American College of Radiology: http://www.acr.org/~/media/ACR/Documents/PDF/QualitySafety/Resources/PIRADS/PIRADS%20V2.pdf

[Prostate Eu Concensus] Eur Urol. Dickinson L, Ahmed HU., Allen C, Barentsz JO, Carey B, Futterer JJ, Heijmink SW, Hoskin PJ, Kirkham A, and Padhani AR. 2011. 59. 4. 477–94. “Magnetic resonance imaging for the detection, localisation, and characterisation of prostate cancer: recommendations from a European consensus meeting”. doi:10.1016/j.eururo.2010.12.009 http://www.europeanurology.com/article/S0302-2838(10)01187-5/ .

[ESUR Guidelines] Eur Radiol. Barentsz JO, Richenberg J, Clements R, Choyke P, Verma S, Villeirs G, Rouviere O, Logager V, and Fütterer JJ. 2012/04. 22. 4. 746–57. “ESUR prostate MR guidelines 2012”. doi:10.1007/s00330-011-2377-y http://www.ncbi.nlm.nih.gov/pmc/articles/PMC3297750/ .

# 3 Definitions

For the purposes of this Standard the following definitions apply.

## 3.1 Codes and Controlled Terminology Definitions:

The following definitions are commonly used in this Part of the DICOM Standard:

Baseline Context Group Identifier (BCID)

  Identifier that specifies the suggested Context Group for a Code Sequence Attribute.

  See Table 5.6-1 “Conventions for Specification of Context Groups” in PS3.3 .

Defined Context Group Identifier (DCID)

  Identifier that specifies the Context Group for a Code Sequence Attribute that shall be used.

  See Table 5.6-1 “Conventions for Specification of Context Groups” in PS3.3 .

Extensible Context Group

  Context Group that may be extended by a particular application by inclusion of additional concepts.

  See Table 5.6-1 “Conventions for Specification of Context Groups” in PS3.3 .

Non-Extensible Context Group

  Context Group whose defined set of concepts shall not be extended by an application.

  See Table 5.6-1 “Conventions for Specification of Context Groups” in PS3.3 .

Context Group

  A set of coded concepts defined by a Mapping Resource forming a set appropriate to use in a particular context.

Context Group Version

  Version of a Context Group.

Context ID (CID)

  Identifier of a Context Group.

Mapping Resource

  A resource that defines context-dependent usage constraints (i.e., Value Set or Relationship Type restrictions) for Attributes. A resource that specifies the mapping of the content of an external controlled terminology to the components of a message standard.

Relationship Type

  The association between two Concepts. Examples: "HAS PROPERTIES", "CONTAINS", "INFERRED FROM".

DICOM Content Mapping Resource (DCMR)

  A Mapping Resource that defines Templates and Context Groups for use in DICOM IODs.

Template

  A pattern that describes the Content Items, Value Types, Relationship Types and Value Sets that may be used in part of a Structured Report content tree, or in other Content Item constructs, such as Acquisition Context or Protocol Context. Analogous to a Module of an Information Object Definition.

Template ID (TID)

  Identifier of a Template.

Value Set

  The allowed values of a Code Sequence Attribute in a given context. Specified either as one or more individual values or by reference to a Context Group.

Baseline Template Identifier (BTID)

  Identifier that specifies a Template suggested to be used in the creation of a set of Content Items.

Defined Template Identifier (DTID)

  Identifier that specifies a Template that shall be used in the creation of a set of Content Items.

Extensible Template

  A Template that may be extended by a particular application by inclusion of additional Content Items beyond those specified in the Template.

Non-Extensible Template

  A Template that specifies the exact set of Content Items and corresponding Value Sets that shall be used and that shall not be extended by an application.

Root Template

  A Template whose first content item is a CONTAINER content item intended to be encoded in the top level Data Set of a SOP Instance. I.e., the "root node" of the "content tree".

Coding schemes

  Dictionaries (lexicons) of concepts (terms) with assigned codes and well defined meanings.

  ### Note

  Examples of coding schemes include SNOMED and LOINC.

## 3.2 Information Object Definitions:

This Part of the Standard makes use of the following terms defined in PS3.3:

Code Sequence Attribute

  See PS3.3.

# 4 Symbols and Abbreviations

The following symbols and abbreviations are used in this Part of the Standard.

Mammography CAD

  Computer-Aided Detection and/or Computer-Aided Diagnosis for Mammography

Chest CAD

  Computer-Aided Detection and/or Computer-Aided Diagnosis for chest radiography

Colon CAD

  Computer-Aided Detection and/or Computer-Aided Diagnosis for colon radiography

ACR

  American College of Radiology

ASE

  American Society of Echocardiography

CAP

  College of American Pathologists

DCMR

  DICOM Content Mapping Resource

NEMA

  National Electrical Manufacturers Association

RECIST

  Response Evaluation Criteria In Solid Tumors

SNOMED

  Systematized Nomenclature of Medicine

UCUM

  Unified Code for Units of Measure

WHO

  World Health Organization

EV

  Enumerated Value

DT

  Defined Term

CNAME

  Context Group Name

TNAME

  Template Name

BCID

  Baseline Context Group ID

DCID

  Defined Context Group ID

ECID

  Enumerated Context Group ID

BTID

  Baseline Template ID

DTID

  Defined Template ID

ETID

  Enumerated Template ID

The following upper-case abbreviations represent specific Attributes:

CV

  Code Value (0008,0100) or Long Code Value (0008,0119) or URN Code Value (0008,0120)

CSD

  Coding Scheme Designator (0008,0102)

CM

  Code Meaning (0008,0104)

CSV

  Coding Scheme Version (0008,0103)

# 5 Conventions

Terms listed in Section 3 are capitalized throughout the document.

# 6 Form of Template Specifications

Templates are patterns that specify the Concept Names, Requirements, Conditions, Value Types, Value Multiplicity, Value Set restrictions, Relationship Types and other attributes of Content Items for a particular application.

An IOD may specify that particular Standard Templates shall be used or may be used to define or constrain the content of a Content Item construct. A Content Item construct includes a coded concept name and one of several types of coded values. Content Item constructs are used in:

- the main Data Set and recursively nested Content Sequences (0040,A730) of the SR Document Content Module

- the Acquisition Context Sequence(0040,0555) of the Acquisition Context Module,

- the Protocol Context Sequence (0040,0440) and Content Item Modifier Sequence (0040,0441) of the Scheduled Procedure Step Module, Image Acquisition Results Module, and others.

- the Specimen Preparation Step Content Item Sequence (0040,0612) of the Specimen Module.

Annexes A and C of this Part define Standard Templates.

### Note

Standard Extended and Private Templates may be defined by implementers of the Standard. The rules for definition of Standard Extended and Private SR Templates are similar to the rules for definition of Standard Extended and Private SOP Classes. One row of a Template definition table corresponds to one row of a Module table.

Each Standard Template is specified by a Template table in this Part. Each Template table specifies exactly one Template, corresponding to a pattern of content within a Content Item construct.

Each Template table identifies whether the order of Content Items is significant or not significant. SOP Instances whose content is based on a Template where the order is significant shall encode the top level Content Items in the order they are specified in the Template, and the subsidiary Content Items under each parent item in the order they are specified, and so on for each Nesting Level. The significance of the order applies only to the Template itself; subsidiary included Templates may have a different order significance.

### Note

Even if a Template specifies that the order is not significant, there may be significance to the order in which Content Items are encoded in a SOP Instance. For example, CONTAINER Content Items with attribute Continuity of Content (0040,A050) value CONTINUOUS encode Content Items in narrative sequence, and procedure logs encode Content Items in time order.

The Content Items from subsidiary Templates may be intermingled if and only if the parent and subsidiary all specify that the order is not significant. This permits later refactoring into reusable Templates.

The range of concepts and the options that are permitted in a family of SR Documents vary inversely with the level of constraint that is applied by the corresponding SR Template. The more narrow the range of concepts and the more restricted the options permitted by a Template, the more predictable the content of the SR Documents will be.

### Note

1. A very specific Template defines a family of SR Documents that are very similar to each other. They have a narrow range of content options (e.g., high level of constraint of Content Item values; use of CODE or NUM with Enumerated Context Groups) and their content is therefore highly predictable. A very general (e.g., permissive or broad) Template defines a family of SR Documents that may differ considerably from one another. They have a broader range of content options (e.g., low level of constraint of Content Item values; use of TEXT and relatively little restriction of Content Item values) and their content is less predictable.

2. The degree of interoperability that may be achieved with a family of SR Documents generated from a Template may be determined intentionally and precisely at a desired level by appropriate Template design to achieve the necessary degree of predictability of SR Document contents.

## 6.1 Template Table Field Definition

SR Templates are described using tables of the following form:

Type:

  (Non-) Extensible

Order:

  (Non-) Significant

Root:

  Yes or No

Table TID <#>. <SR Context Template Name>

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  |  |  |  |  |  |
2 |  |  |  |  |  |  |  |
3 |  |  |  |  |  |  |  |

Acquisition Context Templates are described using tables of the following form:

Type:

  (Non-) Extensible

Order:

  (Non-) Significant

Table TID <#>. <Acquisition Context Template Name>

| VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | ---
1 |  |  |  |  |  |
2 |  |  |  |  |  |
3 |  |  |  |  |  |

Protocol Context Templates are described using tables of the following form:

Type:

  (Non-) Extensible

Order:

  (Non-) Significant

Table TID <#>. <Protocol Context Template Name>

| NL | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | ---
1 |  |  |  |  |  |  |
2 |  |  |  |  |  |  |
3 |  |  |  |  |  |  |

The semantics of the fields (columns) of Template tables are defined by subsections of this Section. A row of a Template table specifies either one Content Item or inclusion of another Template that may specify any number of Content Items (see Section 6.2.3 for definition of Included Templates). Each Template table is named by a title, identified by a TID number and further explained by a description such as explanation of Template contents, purpose and use cases.

The following conventions are defined for the form of references to coded concepts, Context Groups and Templates.

Code Meanings are enclosed in quotation marks (for example "cm"). Code Values and Coding Scheme Designators are not enclosed in quotation marks unless a comma occurs in the string.

References to coded concepts take the following form:

- EV or DT (CV, CSD, "CM")

  e.g., an Enumerated Value with only CV, CSD, and CM defined is represented as follows: EV (CV, CSD, "CM"), for example EV (T-04000, SRT, "Breast").

- MemberOf { BCID or DCID (CID) CNAME } MemberOf selects one term from the specified context group.

If reference to a specific coding scheme version is required, it takes the following form:

- EV or DT (CV, CSD [CSV], "CM")

  e.g., DT (D3-81922, SRT [V1], "Aortic fistula").

References to Context Groups take the following form:

- BCID or DCID (CID) CNAME

  e.g., Defined Context Group 5000 is represented as follows: DCID (5000) Language.

References to Templates take the following form:

- BTID or DTID (TID) TNAME

  e.g., Baseline Template 1000 is represented as follows: BTID (1000) Quotation.

### 6.1.1 Row Number

Each row of a Template Table is denoted by a row number. The first row is numbered 1 and subsequent rows are numbered in ascending order with increments of 1. This number denotes a row for convenient description as well as reference in conditions. The Row Number of a Content Item in a Template may or may not be the same as the ordinal position of the corresponding Sequence Item (representing the Content Item) in a Content Sequence (0040,A730), depending on the number of times the Content Item is repeated.

The Content Item specified in the first row of a Template table may be of any Value Type. Specifically, it is not constrained to be a CONTAINER.

### 6.1.2 Nesting Level (NL)

The nesting level of Content Items is denoted by ">" symbols, one per level of nesting below the initial Source Content Item (of the Template) in a manner similar to the depiction of nested Sequences of Items in Modules Tables in PS3.3. When it is necessary to specify the Target Content Item(s) of a relationship, they are specified in the row(s) immediately following the corresponding Source Content Item. The Nesting Level of a Target Content Item is one greater than the Nesting Level of the corresponding (parent) Source Content Item. The Content Item specified in row 1 of a Template Table is at the top level (i.e., no ">" symbol is ever present in the NL field for the first Content Item in the table).

Acquisition Context Templates have no Nesting Level field. Protocol Context and UPS Processing Parameter Templates allow a single Nesting Level implemented through the Content Item Modifier Sequence (see PS3.3).

### 6.1.3 Relationship With Source Content Item (Parent)

Relationship Type and Relationship Mode (i.e., By-value or By-reference) constraints, if defined, are specified in this field, as described Table 6.1.3-1.

Relationship Type and Mode are specified for each row that specifies a target Content Item.

Relationship Type and Mode may also be specified when another Template is included, either "top-down" or "bottom-up" or both (i.e., in the "INCLUDE Template" row of the calling Template, or in all rows of the included Template, or in both places). There shall be no conflict between the Relationship Type and Mode of a row that includes another Template and the Relationship Type and Mode of the rows of the included Template.

### Note

SR IODs specify Enumerated Values for Relationship Types. If a Relationship Type other than one of the Defined Terms for Relationship Type (0040,A010) is specified in a Private SOP Class, there is a significant risk to interoperability. Documentation accompanying Templates for Private SOP Classes should define any Relationship-type extensions in the manner that the Standard Relationship Types are defined in PS3.3.

Acquisition Context and Protocol Context Templates have no Relationship field.

Table 6.1.3-1. Syntax of Relationship Constraints

Expression | Definition
--- | ---
RTYPE | Relationship Mode is By-value and Relationship Type is RTYPE. For example, "INFERRED FROM".
R-RTYPE | Relationship Mode is By-reference and Relationship Type is RTYPE. For example, "R-INFERRED FROM".

### 6.1.4 Value Type (VT)

The Value Type field specifies the SR Value Type of the Content Item or conveys the word "INCLUDE" to indicate that another Template is to be included (substituted for the row). See Section 6.2.3 for further description of "Included Templates".

### 6.1.5 Concept Name

Any constraints on Concept Name are specified in this field as defined or enumerated coded entries, or as baseline or defined context groups. Alternatively, when the VT field is "INCLUDE", the Concept Name field specifies the Template to be included.

### 6.1.6 Value Multiplicity (VM)

The VM field indicates the number of times that either a Content Item of the specified pattern or an included Template may appear in this position. Table 6.1.6-1 specifies the values that are permitted in this field.

Table 6.1.6-1. Permitted Values for VM

Expression | Definition
--- | ---
i (where 'i' represents an integer) | Exactly i occurrences, where i>=1. E.g., when i=1 there shall be one occurrence of the Content Item in this position.
i-j (where 'i' and 'j' represent integers) | From i to j occurrences, where i and j are >=1 and j>i.
i-n (where 'i' and 'n' represent integers) | i or more occurrences, where i>=1.

### 6.1.7 Requirement Type

The Requirement Type field specifies the requirements on the presence or absence of the Content Item or included Template.

### Note

There is typically no need to specify Requirement Type separately for SCU and SCP of the Basic SR SOP Classes, because the SCP is required to support the entire content of any SR Document it receives. Therefore, for Basic SR SOP Classes, Requirement Type effectively only applies to the SCU.

The following symbols are used:

M

  Mandatory. Shall be present.

MC

  Mandatory Conditional. Shall be present if the specified condition is satisfied.

U

  User Option. May or may not be present.

UC

  User Option Conditional. May not be present. May be present according to the specified condition.

### Note

There is an interaction between the VM and the Requirement Type with respect to the number of times that a Content Item (or included Template) may actually be present, as follows:

Req Type | VM | Actual number of occurrences in the content tree
--- | --- | ---
M | 1 | 1
M | 1-n | 1 to n
U | 1 | 0 or 1
U | 1-n | 0 to n

### 6.1.8 Condition

The Condition field specifies any conditions upon which presence or absence of the Content Item or its values depends. This field specifies any Concept Name(s) or Values upon which there are dependencies.

References in Condition statements to coded concepts or values, whether to select a Content Item to test or to specify a value to test against, are of the form (CV, CSD, "CM"). As is always the case for coded entries, the matching is performed against CV and CSD, irrespective of the string value of CM.

References may also be made to row numbers (e.g., to specify exclusive OR conditions that span multiple rows of a Template table).

The following abbreviations are used:

XOR

  Exclusive OR. One and only one row shall be selected from mutually-exclusive options.

  ### Note

  For example, if one of rows 1, 2, 3 or 4 may be included, then for row 2, the abbreviation "XOR rows 1, 3, 4" is specified for the condition.

IF

  Shall be present if the condition is TRUE; may be present otherwise.

IFF

  If and only if. Shall be present if the condition is TRUE; shall not be present otherwise.

### 6.1.9 Value Set Constraint

Value Set Constraints, if any, are specified in this field as defined or enumerated coded entries, or as baseline or defined context groups.

The Value Set Constraint column may specify a default value for the Content Item if the Content Item is not present, either as a fixed value, or by reference to another Content Item, or by reference to an Attribute from the data set other than within the Content Sequence (0040,A730).

#### 6.1.9.1 NUM Units Constraint

Constraints on units of measurement, if any, are specified in the Value Set Constraint field if and only if the Value Type is NUM. The constraints are specified either as defined or enumerated coded entries, or as baseline or defined context groups.

#### 6.1.9.2 CONTAINER Continuation Flag Constraint

The value of the Continuity of Content Flag (0040,A050) may be specified in the Value Set Constraint field if and only if the Value Type is CONTAINER.

### Note

The SR Document Content Module specifies "SEPARATE" and "CONTINUOUS" as the Enumerated Values for Continuity of Content Flag (0040,A050).

#### 6.1.9.3 SCOORD Graphic Type Constraint

Constraints on the value of the Graphic Type(0070,0023) may be specified in the Value Set Constraint field if and only if the Value Type is SCOORD. The constraint may specify a set of allowed values, or a set of disallowed values. For example:

- GRAPHIC TYPE = {POINT}

- GRAPHIC TYPE = {CIRCLE, ELLIPSE}

- GRAPHIC TYPE = not {MULTIPOINT}

## 6.2 Special Conventions for Template Tables

### 6.2.1 Multiple Value Sets Depending On Different Conditions

When a Content Item may have different value sets, each depending on different conditions, the description of each different case begins in a separate row of the Template Table.

### 6.2.2 Target Content Items of Relationships

When it is necessary to specify the Target Content Item(s) of a relationship, they are specified in the row(s) immediately following the Source Content Item. The Nesting level of a Target Content Item (or set of Target Content Items specified indirectly via an 'include Template' macro) is one greater than the Nesting Level of the corresponding Source Content Item, as indicated by an increase in the number of ">" characters in the nesting level.

When a Content Item may be the Source of multiple relationships having different Relationship Types and/or different Relationship Modes and/or different patterns of Target Content Item(s), the description of each different case begins in a separate row of the Template Table.

When the Source Content Item of a relationship has VM of greater than 1, the specified pattern of Target Content Items applies to all instantiations of the Source Content Item.

### Note

For example, if a Template specifies that the VM of a Source Content Item is 1-n and specifies a By-value relationship to two CODE Content Items with particular value set constraints, then each instantiation of the Source Content Item has a By-value relationship to two CODE Content Items with the specified value constraints.

When a Source Content Item that has a Requirement Type of U, UC or MC is not present (is not instantiated), no Target Content Items of that Source Content Item are present, even if one or more of the Target Content Items is designated with a Requirement Type of M or MC.

### Note

In other words, potential children are not present when there is no parent.

### 6.2.3 Inclusion of Templates

A Template may specify another Template to be included by specifying "INCLUDE" in the Value Type field and the identifier of the included Template in the Concept Name field. All of the rows of the specified Template are in included in the invoking Template, effectively substituting the specified Template for the row where the inclusion is invoked. Whether or not the inclusion is user optional, mandatory or conditional is specified in the Requirement and Condition fields. The number of times the included Template may be repeated is specified in the VM field.

#### 6.2.3.1 Template Parameters

A Template that is included by another Template may include parameters that are replaced by values defined in the invoking Template. Parameters may be used to specify coded concepts or Context Groups in the Concept Name, Condition, or Value Set Constraint fields of a Template.

An included Template that accepts parameters shall be introduced by a table listing those parameters of the form:

Parameter Name | Parameter Usage
--- | ---
... | ...
... | ...

Parameters are indicated by a name beginning with the character "$".

The invoking Template may specify the value of the parameters in the included Template by name in the Value Set Constraint field of the INCLUDE row. The parameter in the included Template shall be replaced by the specified parameter value. Specification of a parameter value shall be of one of the following forms:

Notation | Definition
--- | ---
$parametername = EV or DT (CV, CSD, "CM") | The parameter passed to the Template is the specified coded term.
$parametername = (CV, CSD, "CM") | The parameter passed to the Template is the specified coded term, used as a parameter in a Condition field of the included Template.
$parametername = BCID or DCID (CID) CNAME | The parameter passed to the Template is the Context Group.
$parametername = MemberOf {BCID or DCID (CID) CNAME} | The parameter passed to the Template is a single coded term from the Context Group in curly braces.

The specification of a parameter value is valid only for the directly included Template. Therefore, it needs to be explicitly respecified in Templates intermediate between the originally specifying Template and the target Template. The intermediate Template may use the same parameter name as used by the Template it invokes; in such a case, the intermediate Template would invoke the subsidiary Template with a specification in the Value Set Constraint field such as:

$parametername = $parametername

### Note

In this case, the left hand instance of $parametername is the name in the subsidiary Template, and the right hand instance is the (parametrized) value passed into the current Template.

The invoking Template is not required to specify all parameters of included Templates. If not specified, the value set (term or context group) for that parameter is unconstrained. An unconstrained value in a Condition will cause the condition to fail.

### 6.2.4 Post-coordinated Codes and Has Concept Modifier Relationship

Though it may not be explicitly shown in a particular Template, the use of any coded Concept Name in any Content Item may be defined in a post-coordinated rather than pre-coordinated manner, unless explicitly forbidden by the IOD or the Template.

Accordingly, any such Content Item may have any number of Target Content Items via a "HAS CONCEPT MOD" relationship, even if not explicitly specified in a Template. Each Target Content Item of such a relationship may be more complicated than a single Content Item if the IOD permits (i.e., the post-coordinated concept may potentially be defined by a complex sub-tree).

### 6.2.5 Extension of Templates

An Extensible Template may be extended in an Application generating SOP Instances to include additional Content Items in its definition. Such Content Items shall not duplicate concepts for which an encoding is defined in the Template. I.e., if a method is provided for the encoding of a concept in the Template, that concept shall not be encoded using a different Content Item in an extension to the Template.

### Note

There is no requirement that the included additional Content Items in a Template extension be placed at the end of the Template. The additional Content Items may be included at any semantically appropriate location in the Template, regardless of whether the order of Content Items in the Template is significant.

A Non-extensible Template shall not be modified in an Application by the addition of Content Items to its definition.

### Note

The set of Content Items in either an Extensible or a Non-extensible Template may be changed in subsequent editions of the Standard, in accordance with the procedures of the DICOM Standards Committee.

A Non-Extensible Template may include a Template that is Extensible. In invoking such a Template, the content structure of SOP Instances created from the Non-Extensible Template may vary according to the varying content structure allowed by the extension of the included Template.

### Note

Specification of such extensible content in a Non-Extensible Template may be desirable if the Template defines, e.g., a fixed top level structure into which a variety of lower level structures may be "plugged".

# 7 DCMR Context Group Specifications

Context Groups specify Value Set restrictions for Code Value (0008,0100) (or Long Code Value (0008,0119) or URN Code Value (0008,0120)) and Code Meaning (0008,0104) of Code Sequence Attributes for given functional or operational contexts. This Section specifies the semantics of DCMR Context Group Tables.

## 7.1 Context Group Table Field Definition

Context Groups are described using tables of the following form (optional columns are shown with italic column titles):

Type:

  (Non-) Extensible

Version:

  <yyyymmdd>

Table CID <#>. <Context Group Name>

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning | <Reference Terminology> Equivalent Value
--- | --- | --- | --- | ---
... | ... | ... | ... | ...
... | ... | ... | ... | ...

A row of a Context Group table specifies one coded concept. Each Context Group table is named by a title and identified by a CID number and version.

The columns of the tables consist of:

- Coding Scheme Designator: the value of Coding Scheme Designator (0008,0102)

- Code Value: the value of Code Value (0008,0100) or Long Code Value (0008,0119) or URN Code Value (0008,0120)

- Coding Meaning: the value of Code Meaning (0008,0104)

In those cases where it is necessary, Coding Scheme Version (the value of Coding Scheme Version (0008,0103)) may also be specified. This column may be absent if Coding Scheme Version is not required for any of the coded concepts in the Context Group.

The value specified in the Code Meaning field is an acceptable value for the specified code value, but does not preclude the use of other synonymous text in the same or other language.

### Note

Some coding schemes such as LOINC do not specify the equivalent of a Code Meaning.

If further description of the concept represented by the code is required in the DCMR (rather than referring to an external coding scheme), it is included in a separate table.

An optional column may provide an informative mapping from the coded concepts of the Context Group to a reference terminology specified in the column heading.

A Context Group may alternatively be defined by narrative reference to an externally defined coding scheme.

### Note

See for instance CID 82 “Units of Measurement”.

## 7.2 Special Conventions for Context Group Tables

### 7.2.1 Include Context Group

The 'Include Context Group' macro is a concise mechanism for including (by-reference) all of the rows of a specified Context Group in the invoking Context Group, effectively substituting the specified Context Group for the row where the macro is invoked. If an 'Include Context Group' is specified, it shall be specified in the Concept Name column of a Context Group Table. Table 7.2.1-1 specifies the syntax of the 'Include Context Group macro. Inclusion may be nested, in that included Context Groups may themselves include other Context Groups. This gives rise to the possibility of circular inclusion and multiple inclusion, in which case the Context Group shall consist of the transitive closure of the set of all coded concepts within the included Context Groups.

### Note

For example, it is reasonable to have the following definitions for context groups:

- Context ID 1, includes Context IDs 2 and 3

- Context ID 2, includes Context IDs 4 and 5

- Context ID 3, includes Context IDs 5 and 6

- Context ID 4 contains a, b, c

- Context ID 5 contains e, f, g

- Context ID 6 contains a, h, i

The contents of Context ID 1 will be a, b, c, e, f, g, h, i.

Table 7.2.1-1. Include Context Group Macro

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
… | … | …
Include CID nnn
… | … | …

### 7.2.2 Units of Measurement

Context Group 82 is defined to include all units of measurement relevant to DICOM IODs. In the past it was envisaged that an extensible list of pre-coordinated codes would be included in the mapping resource.

DICOM has now adopted the Unified Codes for Units of Measurement (UCUM) standard for all units of measurement. This coding scheme allows for the "construction" of pre-coordinated codes from atomic components.

The specialization of the UCUM standard as it is used in DICOM involves the following rules:

- the Coding Scheme Designator is specified as "UCUM"

- the version of UCUM from which a code is constructed is not required, as it is not needed to resolve ambiguity in the Code Value or Code Meaning; however, there is no restriction on the version being specified in Coding Scheme Version

- the Code Value will be constructed from UCUM and make use of the "case-sensitive" form of UCUM code (e.g., "ml/s")

- the Code Meaning for other than UCUM unity may be one of the following:

  - the "print" value specified in UCUM (e.g., "mmHg" for Code Value mm[Hg])

  - the same string as sent in the Code Value (e.g., "ml/s")

  - constructed from the "names" of individual components using the Americanized form of name (e.g., "milliliters/second")

  - constructed from the "names" of individual components using the European form of name (e.g., " millilitres /second")

- In the case of UCUM unity ("1", or curly braces expression) it is forbidden to use "1" as a Code Meaning. Annex G provides Code Meanings for a Code Value (0008,0100) of 1. A Template or Context Group may constrain the Code Meaning according to the following rules:

  - UCUM default unit 1 shall use one of the Code Meaning synonyms specified in Annex G

  - ratios of identically dimensioned values may use ({ratio}, UCUM, "ratio")

  - unitless numeric scores may use ({M:N}, UCUM, "range: M:N") to specify the minimum and maximum value, for example, ({0:10}, UCUM, "range: 0:10")

  - counts using UCUM annotation shall always use the text within the curly braces as the Code Meaning, for example, ({masses}, UCUM, "masses")

  - compositions of a curly braces expression with other UCUM values may use a conventional clinical representation, for example, ({H.B.}/min, UCUM, "BPM")

The UCUM standard states that the preferred display values for codes deg (degrees of plane angle) and Cel (degrees Celsius) are "°" and "°C". However, the character ° does not have a representation in the DICOM default character set (ASCII, ISO-IR 6). The Code Meaning specified in this Part therefore uses "deg" and "C". SOP Instances that specify a Specific Character Set that allows the character ° may use Code Meanings "°" and "°C".

### Note

1. Code Meaning "C" formally conflicts with the Code Meaning for Coulomb. In the context of DICOM use, the possibility of confusion to a user based on the display of the Code Meaning is considered remote, as there is little use of Coulomb in imaging, and the context of the displayed item Concept Name would resolve between temperature and electric charge. Automated processing based on the Code Values should not face an issue as the Code Values differ.

2. The character ° has Unicode code point U+00B0, and is represented as 0xB0 in ISO-IR 100 (Latin-1), ISO-IR 101 (Latin-2), ISO-IR 109 (Latin-3), ISO-IR 110 (Latin-4), ISO-IR 126 (Greek), ISO-IR 138 (Hebrew), and ISO-IR 148 (Latin-5). It is not encodable in ISO-IR 13 (Katakana), ISO-IR 144 (Cyrillic), ISO-IR 127 (Arabic), or ISO-IR 166 (Thai).

### 7.2.3 Extension of Context Groups

An Application may extend an Extensible Context Group by adding terms for new concepts. Applications may not substitute other terms of the same concept in the Context Group. Applications may not add a term that means "unspecified" or "missing" or "unknown" similar; if such a concept is intended to be permitted then the Standard will include it in the Context Group already. Such extension may be made without a change in Context Group Identifier, but with the specification of Context Group Extensions (see PS3.3).

Non-extensible Context Groups shall not be modified in an Application.

### Note

The set of concepts in either an Extensible or a Non-extensible Context Group may be changed in subsequent editions of the Standard, in accordance with the procedures of the DICOM Standards Committee.

# 8 Coding Schemes

Table 8-1 lists the coding schemes (and their designators) defined for use in DICOM; Table 8-2 lists the HL7v3 coding schemes referenced for use in DICOM. Additionally, any coding scheme may be used that has an entry in the HL7 Registry of Coding Schemes (HL7 v2 Table 0396, or the equivalent online registry), in which case the HL7 Symbolic Name shall be used as the value for the Coding Scheme Designator in DICOM, as long as it does not conflict with an entry Table 8-1 and fits within the Value Representation of the DICOM Coding Scheme Designator (0008,0102) attribute. As specified in the HL7 v2 Table 0396, local or private coding schemes shall be identified by an alphanumeric identifier beginning with the characters "99".

### Note

1. An earlier version of this table was formerly contained in Annex D of PS3.3.

2. See Section 8.2 “Coding Scheme Designator and Coding Scheme Version” in PS3.3 for further description.

3. The Coding Scheme UIDs are provided for reference only; the normative specification of UIDs and their associated meaning is the responsibility of the coding scheme developer and/or HL7.

4. The HL7 registration of Coding Schemes is available at http://www.hl7.org/oid/index.cfm.

5. Publication of codes or references to coding schemes within DICOM does not constitute a grant of intellectual property rights to implementers. Use of some Coding Schemes may require a license, or purchase of the relevant coding scheme publication. Implementers should consult the relevant coding scheme publisher; see also Section 2.

6. The values of Coding Scheme Name (0008,0115), Coding Scheme Responsible Organization (0008,0116) and Coding Scheme Resources Sequence (0008,0109), if available, may be used to fill the corresponding optional attributes of the Coding Scheme Identification Sequence (0008,0110) in the Section C.12.1 “SOP Common Module” in PS3.3 .

Table 8-1. Coding Schemes

Coding Scheme Designator (0008,0102) | Coding Scheme UID (0008,010C) | Coding Scheme Name (0008,0115) | Coding Scheme Responsible Organization (0008,0116) | Coding Scheme Resources Sequence (0008,0109) Type: URL | Description
--- | --- | --- | --- | --- | ---
ACR | 2.16.840.1.​113883.6.76 | ACR Index | ACR |  | ACR Index for Radiological Diagnosis Revised, 3rd Edition 1986
ASTM-sigpurpose | 1.2.840.10065.​1.12 | ASTM E 2084 | ASTM |  | [ASTM E 2084-00] Signature Purpose codes (see Annex A1 of ASTM E 2084), ASTM Subcommittee E 31.20 Data and System Security for Health Information
BARI |  | BARI |  |  | Bypass Angioplasty Revascularization Investigation[Alderman 1992]; endorsed by ACC/AHA Guidelines for Coronary Angiography[Scanlon 1999].
BI |  | BI-RADS | ACR |  | ACR Breast Imaging Reporting and Data System [BI-RADS®], Coding Scheme Version (0008,0103) is required; code values are section and paragraph identifiers within the publication where the code meaning is defined (e.g., "I.D.1", where I = Breast Imaging Lexicon, D = Special Cases, 1 = Tubular Density, as the code value for "Tubular Density"). ### Note In the HL7 registry, the abbreviation BI is assigned to a different coding scheme, specifically the Beth Israel problem list.
C4 | 2.16.840.1.​113883.6.12 | CPT-4 | AMA |  | American Medical Association's Current Procedure Terminology 4 (CPT-4)
C5 | 2.16.840.1.​113883.6.82 | CPT-5 | AMA |  | American Medical Association's Current Procedure Terminology 5 (CPT-5)
caDSR | 2.16.840.1.113883.3.26.2 | Cancer Data Standard Repository | NCI |  | The Public ID is used as the Code Value. These can be looked up as in the following example (the version is required): https://​cdebrowser.​nci.​nih.​gov/​CDEBrowser/​search?​dataElementDetails​=​9/​&cdeId=2178693&​version=2.1​&​PageId​=DataElementsGroup
CD2 | 2.16.840.1.​113883.6.13 | CDT-2 | ADA |  | American Dental Association's (ADA) Current Dental Terminology 2 (CDT-2)
CTV3 | 2.16.840.1.​113883.6.6 | Clinical Terms Version 3 | UK NHS |  | Read Codes
DC | 1.2.840.10008.​2.​16.​10 | Dublin Core | W3C | DOC: http://​dublincore.​org/​documents/​1998/​09/​dces/ DOC: https://​www.​ietf.​org/​rfc/​rfc2413.txt | Dublin Code Metadata for Resource Discovery. The code value is the Label field, e.g., "Creator" (capitalization significant).
DCM | 1.2.840.10008.​2.16.4 | DICOM Controlled Terminology | DICOM | DOC: http://​dicom.​nema.​org/​medical/​dicom/​current/​output/​chtml/​part16/​chapter_D.html OWL: ftp://​medical.​nema.​org/​medical/​dicom/​current/​ontology/​dcm.owl.zip | PS3.16 Content Mapping Resource, Annex D (Note that HL7 also specifies an OID of 2.16.840.1.​113883.6.31, but deprecates it in favor of 1.2.840.10008.​2.16.4).
DCMUID | 1.2.840.10008.​2.6.1 | DICOM UID Registry | DICOM | DOC: http://​dicom.​nema.​org/​medical/​dicom/​current/​output/​chtml/​part06/​chapter_A.html |
FMA | 2.16.840.1.​113883.6.119 | FMA | University of Washington, Seattle | DOC: http://​sig.​biostr.​washington.​edu/​projects/​fm/​AboutFM.​html OWL: http://​sig.​biostr.​washington.​edu/​share/​downloads/​fma/​release/​latest/​fma.​zip | Digital Anatomist Foundational Model of Anatomy
HPC | 2.16.840.1.​113883.6.14 |  |  |  | Healthcare Financing Administration (HCFA) Common Procedure Coding System (HCPCS)
I10 | 2.16.840.1.​113883.6.3 | ICD-10 | WHO |  | International Classification of Diseases revision 10 (ICD-10)
I10P | 2.16.840.1.​113883.6.4 | ICD-10-PCS | US DHHS CMS |  | ICD-10 Procedure Coding System (ICD 10 PCS)
I9 | 2.16.840.1.​113883.6.42 | ICD-9 | WHO |  | International Classification of Diseases revision 9 (ICD-9)
I9C | 2.16.840.1.​113883.6.2 | ICD-9-CM |  |  | International Classification of Diseases revision 9, with Clinical Modifications (ICD-9-CM)
IETF4646 |  | RFC 4646 | IETF | DOC: http://​tools.​ietf.​org/​html/​rfc4646 | [RFC 4646], Tags for Identifying Languages, The Internet Society (2005) [RFC 4646] has been superceded by [RFC 5646].
ISO639_1 | 2.16.840.1.​113883.6.99 | ISO 639-1 | ISO |  | [ISO 639-1] Two-letter language codes ### Note HL7 uses "ISO639-1" for the symbolic name, with a hyphen rather than an underscore
ISO639_2 | 2.16.840.1.​113883.6.100 | ISO 639-2 | ISO |  | [ISO 639-2] Three-letter language codes ### Note HL7 uses "ISO639-2" for the symbolic name, with a hyphen rather than an underscore
ISO3166_1 | 2.16.1 | ISO 3166-1 | ISO |  | [ISO 3166-1] alpha-2 Country Codes ### Note HL7 uses "ISO3166-1" for the symbolic name, with a hyphen rather than an underscore
ISO5218_1 |  | ISO 5218-1 | ISO |  | Representation of Human Sexes (not used) ISO5218_1, which uses numeric codes, was improperly specified in CID 7455 Sex in earlier editions of the standard. The alphabetic codes improperly attributed to that coding scheme have been added to the DICOM Controlled Terminology, and thus all references to coding scheme ISO5218_1 should be considered equivalent to coding scheme DCM.
ISO_OID |  | ISO OID | ISO |  | [ISO 8824-1] ISO/IEC 8824-1- Information Technology - Abstract Syntax 1 (ASN.1): Specification of Basic Notation, and [ISO 9834-1] - Information technology - Open Systems Interconnection - Procedures for the operation of OSI Registration Authorities: General procedures and top arcs of the ASN.1 Object Identifier tree
ITIS_TSN | 1.2.840.10008.​2.​16.​7 | ITIS TSN | ITIS | DOC: http://​www.​itis.​gov | A Taxonomic Serial Number (TSN) is a unique, persistent, non-intelligent identifier for a scientific name in the context of the Integrated Taxonomic Information System (ITIS).
LN | 2.16.840.1.​113883.6.1 | LOINC | Regenstrief Institute | DOC: http://​loinc.​org/ | [LOINC] Logical Observation Identifier Names and Codes
MA | 1.2.840.10008.​2.​16.​5 | Adult Mouse Anatomy Ontology | The Jackson Laboratory | DOC: http://​www.​informatics.​jax.​org/​searches/​AMA.​cgi?​id=MA:0002405 | Hayamizu TF, Mangan M, Corradi JP, Kadin JA, Ringwald M. The Adult Mouse Anatomical Dictionary: a tool for annotating and integrating data. Genome Biology 2005;6(3):R29. doi:10.1186/gb-2005-6-3-r29. http://​www.​ncbi.​nlm.​nih.​gov/​pmc/​articles/​PMC1088948/
MDC | 2.16.840.1.​113883.6.24 |  |  |  | ISO/IEEE 11073 Medical Device Nomenclature, including all its subsections ([ISO/IEEE 11073-10101], [ISO/IEEE 11073-10101a], [ISO/IEEE 11073-10102], etc.), encoded as decimal strings <partition>:<element>
MDNS |  |  |  |  | Universal Medical Device (UMD) Nomenclature System
MGI | 1.2.840.10008.​2.​16.​8 | MGI | The Jackson Laboratory | DOC: http://​www.​informatics.​jax.​org/​mgihome/​nomen/ | The MGI ID from the Mouse Genome Initiative (MGI) nomenclature.
MSH | 2.16.840.1.​113883.6.177 | MeSH | NLM | DOC: http://​www.​nlm.​nih.​gov/​mesh/​meshhome.​html | US National Library of Medicine (NLM) Medical Subject Headings (MeSH)
NBD | 2.16.840.1.​113883.15.2 |  |  |  | NASPE/BPEG Defibrillator Code Bernstein AD, et al."The NASPE/BPEG Defibrillator Code" PACE, 16:1776-1780, 1993
NBG | 2.16.840.1.​113883.15.3 |  |  | DOC: http://www.​hrsonline.​org/​Practice-​Guidance/​Clinical-​Guidelines-​Documents/​2002-​The-​Revised-​NASPE-​BPEG​-​Generic-​Code-​for-​Antibradycardia-​​AdaptiveRate-​and-​Multisite-​Pacing | NASPE/BPEG Generic Pacemaker Code (2000) Bernstein AD, et al."The Revised NASPE/BPEG Generic Code for antibradycardia, adaptive-rate, and multisite pacing." Pacing Clin Electrophysiol., 25:260-264, 2002 See http://​www.​hrsonline.​org/​Practice-​Guidance/​Clinical-​Guidelines-​Documents/​2002-​The-​Revised-​NASPE-​BPEG​-​Generic-​Code-​for-​Antibradycardia-​​AdaptiveRate-​and-​Multisite-​Pacing.
NCDR |  |  |  |  | American College of Cardiology National Cardiovascular Data Registry™ Cath Lab Module Version 1.1, 1997; Version 2.0b, 1999
NCIt | 2.16.840.1.​113883.3.26.1.1 | NCI Thesaurus | NCI | DOC: http://​ncit.​nci.​nih.​gov/ |
NEU | 2.16.840.1.​113883.6.210 | NeuroNames |  | DOC: http://​braininfo.​rprc.​washington.​edu​/aboutBrainInfo.​aspx​#NeuroNames | The numeric brainInfoID is used as the code value. See
NICIP | 2.16.840.1.​113883.2.1.​3.2.4.21 | NICIP | UK NHS | DOC: https://​digital.​nhs.​uk/​article/​1108/​National-​Interim-​Clinical-​Imaging-​Procedure-​NICIP-​Code-​Set | UK National Health Service National Interim Clinical Imaging Procedures (NICIP) Short Code (e.g., "CCHAPC" for CT Thorax abdomen pelvis with contrast)
NPI |  |  |  |  | HCFA National Provider Identifier
POS | 2.16.840.1.​113883.6.50 |  |  |  | HCFA Place of Service (POS) Codes for Professional Claims
PUBCHEM_CID | 1.2.840.10008.​2.​16.​9 | PubChem | NCBI | DOC: https://​pubchem.​ncbi.​nlm.​nih.​gov/ | US National Center for Biotechnology Information (NCBI) PubChem Compound CID.
RADLEX | 2.16.840.1.​113883.6.256 | RadLex | RSNA | DOC: http://​www.radlex.org/ | [RadLex]
RFC3066 | 2.16.840.1.​113883.6.121 | RFC 3066 | IETF | DOC: http://​tools.ietf.org/​html/​rfc3066 | [RFC 3066], Tags for the Identification of Languages, Internet Engineering Task Force ### Note HL7 uses "IETF3066" for the symbolic name. [RFC 3066] has been superseded by [RFC 4646], which in turn has been superceded by [RFC 5646].
RFC5646 | 2.16.840.1.​113883.6.316 | RFC 5646 | IETF | DOC: http://​tools.ietf.org/​html/​rfc5646 | [RFC 5646], Tags for Identifying Languages, The Internet Society (2009) ### Note The HL7 OID Registry specifies "rfc5646", not "ietf5646", as the Desired Symbolic Name (inconsistent with the pattern used for [RFC 4646]). [RFC 5646] constitutes one part of IETF Best Current Practice BCP 47 Tags for Identifying Languages, which also includes [RFC 4647] Matching of Language Tags; [RFC 4647] is not relevant in this context.
99SDM | 2.16.840.1.​113883.6.53 | SDM | DICOM |  | SNOMED DICOM Microglossary (Retired) (see Section 8.1)
SCPECG |  |  |  |  | Standard Communications Protocol for Computer-Assisted Electrocardiography, Draft proposal for ISO Standard, AAMI, Revision 1.3
SNM3 | 2.16.840.1.​113883.6.51 | SNOMED V3 | SNOMED International | DOC: http://​www.snomed.org/ | SNOMED International Version 3 (see Section 8.1)
SCT | 2.16.840.1.​113883.6.96 | SNOMED CT | SNOMED International | DOC: http://​www.snomed.org/ | [SNOMED], using the CT code values
SRT | 2.16.840.1.​113883.6.96 | SNOMED CT | SNOMED International | DOC: http://​www.snomed.org/ | [SNOMED], using the "SNOMED-RT style" code values (see Section 8.1) ### Note HL7 uses "SNM" for the symbolic name.
UBERON | 1.2.840.10008.​2.​16.​6 | UBERON |  | DOC: http://​uberon.org/ | The Uberon ID from the Uberon integrated cross-species ontology covering anatomical structures in animals.
UCUM | 2.16.840.1.​113883.6.8 | UCUM | Regenstrief Institute | DOC: http://​unitsofmeasure.org/​ucum.html | [UCUM] Unified Code for Units of Measure
UMLS | 2.16.840.1.​113883.6.86 | UMLS | NLM | DOC: http://​www.nlm.nih.gov/​research/​umls/ | UMLS codes as CUIs making up the values in a coding system
UPC | 2.16.840.1.​113883.6.55 |  |  |  | Universal Product Code - Universal Code Council

Table 8-2. HL7v3 Coding Schemes

Coding Scheme Designator | Coding Scheme UID | Description
--- | --- | ---
ActCode | 2.16.840.1.113883.5.4 |
ActPriority | 2.16.840.1.113883.5.7 |
AdministrativeGender | 2.16.840.1.113883.5.1 |
mediaType | 2.16.840.1.113883.5.79 | RFC2046
NullFlavor | 2.16.840.1.113883.5.1008 |
ObservationInterpretation | 2.16.840.1.113883.5.83 |
Confidentiality | 2.16.840.1.113883.5.25 |
ParticipationType | 2.16.840.1.113883.5.90 |

## 8.1 SNOMED

SNOMED (the Systematized Nomenclature of Medicine) is the preferred coding system within DICOM for anatomy, clinical findings, procedures, pharmaceutical/biologic products (including contrast agents), and other clinical terms.

SNOMED has had various versions, including SNOMED International (Version 3), which was issued in 1993 and revised through 1998, SNOMED Reference Terminology, the successor to SNOMED 3 that was published between 1999 and 2001, and SNOMED Clinical Terms, which has been the name since 2002. The coding scheme is fully backward-compatible across SNOMED 3, SNOMED-RT, and SNOMED CT. SNOMED CT introduced a solely numeric set of codes (ConceptID) in addition to the former alphanumeric codes (SnomedID), but all SNOMED terminology concepts have both a numeric and an alphanumeric code.

In previous editions of the DICOM Standard, the following Coding Scheme Designators were used for SNOMED codes in DICOM:

- "99SDM", denoting the provisional SNOMED DICOM Microglossary

- "SNM3", denoting SNOMED International (Version 3)

- "SRT", originally denoting SNOMED-RT

All uses of SNOMED coded terms in DICOM are now indicated by the Coding Scheme Designator "SRT", identifying them as SNOMED CT terms using the "SNOMED-RT style" alphanumeric code values, with some exceptions:

- The Section A.5 “Nuclear Medicine Image IOD” in PS3.3 and Section A.21 “Positron Emission Tomography Image IOD” in PS3.3 in some code sequences require the Coding Scheme Designator "99SDM" as an Enumerated Value (see PS3.3).

- The Mammography View Codes of CID 4014 “View for Mammography” and CID 4015 “View Modifier for Mammography” may use the Coding Scheme Designator "SNM3" for implementation adherence to regulatory approvals.

Consequently, when a Coding Scheme Designator of "99SDM" or "SNM3" is encountered, it shall be treated as equivalent to "SRT" for the purpose of interpreting the Code Value.

### Note

"SRT" as a coding scheme designator is used only in the DICOM Standard. HL7v2 did not standardize a coding scheme designator for SNOMED-RT.

When interoperating with systems that use SNOMED CT codes obtained from a source other than the DICOM Standard, Application Entities may receive Code Sequences with a Coding Scheme Designator of "SNOMED-CT" and a numeric ConceptID code. It is the responsibility of such Application Entities to convert any such codes to the alphanumeric SnomedID with Coding Scheme Designator "SRT" for use in DICOM objects and services.

### 8.1.1 Use of SNOMED Anatomic Concepts

In general, DICOM uses the anatomic concepts with the term "structure", rather than with the term "entire". This is an important distinction in SNOMED. "Entire" is a child concept to "structure", has a more restricted meaning, and typically is used in conjunction with treatments (e.g., "excision of entireright kidney"). It is used in distinction to other sibling children of the parent concept that may identify parts of the parent anatomic feature. Since imaging typically targets both the anatomic feature and the area around it, or sometimes just part of the anatomic feature, DICOM usually uses "structure" concepts that are more inclusive than the "entire" concepts.

## 8.2 ISO_OID

[ISO 8824-1] and [ISO 9834-1] are the standards defined for the generation of object identifiers that are used as DICOM Unique Identifiers (see PS3.5), can also serve as a general mechanism for identifying organizations and objects defined by those organizations.

When the Coding Scheme Designator is ISO_OID, the Code Value shall be the numeric (dot delimited) form of a valid object identifier.

A repository of known existing object identifiers can be found at http://www.oid-info.com/index.htm. For example:

- the ISO 9834-1 assigned numeric object identifier for the country France, is "1.0.3166.2.2.1.250" (since ISO 3166 defines a means for maintaining country codes using object identifiers)

- the object identifier for the RIPEMD-160 cryptographic hash function is "1.0.10118.3.0.49"

- the object identifier for the HL7 V2 table of codes for marital status is "2.16.840.1.113883.12.2"

The re-use of object identifiers for existing concepts that do not have an alternative more appropriate coding scheme compatible with DICOM provides a mechanism to avoid defining new codes. For example, HL7 assigned object identifiers can be found at http://www.hl7.org/oid/index.cfm.

Though the intent of ISO_OID is to define organizational roots for the hierarchical assignment of object identifiers, and not specifically to identify organizations per se, the organizational root values can be construed as identifying the organization. For example, the DICOM Standards Organization itself can be identified by the value "1.2.840.10008". See also CID 5002 “Organizations”.

# A Structured Reporting Templates (Normative)

This Annex specifies the content of Standard Templates that may be used by DICOM SR IODs.

## General Templates

### TID 300 Measurement

This Template provides a general structure for a numeric measurement, together with evaluations of its normality and/or significance, and the inference source(s) for its value. This structure is instantiated by inclusion of this Template with specific contextual parameters from a parent Template.

Table TID 300. Parameters

Parameter Name | Parameter Usage
--- | ---
$Measurement | Coded term or Context Group for Concept Name of measurement
$Units | Units of Measurement
$ModType | Modifier Name for Concept Name of measurement
$ModValue | Modifier Value for Concept Name of measurement
$Method | Value for Measurement Method
$Derivation | Value for Measurement Derivation
$TargetSite | Value(s) for Anatomic Location of measurement
$TargetSiteMod | Modifier Value for Anatomic Location of measurement
$Equation | Coded term or Context Group for the equation or table from which the measurement was derived or computed
$ImagePurpose | Purpose of Reference for an image used as a source of the measurement
$WavePurpose | Purpose of Reference for a waveform used as a source of the measurement
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$DerivationParameter | Coded term or Context Group for Concept Name of a derivation parameter
$DerivationParameterUnits | Units of derivation parameter

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 300. Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | $Measurement | 1 | M |  | UNITS = $Units
2 | > | HAS CONCEPT MOD | CODE | $ModType | 1-n | U |  | $ModValue
3 | > | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | $Method
4 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | U |  | $Derivation
5 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | U |  | $TargetSite
6 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
7 | >> | HAS CONCEPT MOD | CODE | DT (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | $TargetSiteMod
8 | > | HAS PROPERTIES | INCLUDE | DTID 310 “Measurement Properties” | 1 | U |  | $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority
9 | > | INFERRED FROM | NUM | $DerivationParameter | 1-n | UC | XOR Row 10 | UNITS = $DerivationParameterUnits
10 | > | R-INFERRED FROM | NUM | $DerivationParameter | 1-n | UC | XOR Row 9 | UNITS = $DerivationParameterUnits
11 | > | INFERRED FROM | INCLUDE | DTID 315 “Equation or Table” | 1 | UC | XOR Row 12 | $Equation = $Equation
12 | > | INFERRED FROM | TEXT | DCID 228 “Equation or Table” | 1 | UC | XOR Row 11 |
13 | > |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1-n | U |  | $Purpose = $ImagePurpose
14 | > |  | INCLUDE | DTID 321 “Waveform or Temporal Coordinates” | 1-n | U |  | $Purpose = $WavePurpose
15 | > |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
16 | > | HAS CONCEPT MOD | TEXT | EV (121050, DCM, "Equivalent Meaning of Concept Name") | 1 | U |  |
17 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
18 | > | INFERRED FROM | COMPOSITE | EV (126100, DCM, "Real World Value Map used for measurement") | 1 | U |  | SOP Class UID shall be Real World Value Mapping Storage ("1.2.840.10008.5.1.4.1.1.67")

Content Item Descriptions

Rows 2, 3, 4, 5 | The HAS CONCEPT MOD items allow the explicit definition of terms for post-coordination of the measurement concept name. Additional post-coordinated modifier terms may be included in a SOP Instance based on this Template, in accordance with section 6.2.4, or as defined by Templates that invoke this Template and explicitly define additional post-coordinated modifiers (e.g., TID 5203).
Row 5 | Finding site may be multiple when a region of interest spans multiple anatomical locations and there is not a single pre-coordinated code describing the combination of locations. E.g., when a maligant, inflammatory or traumatic process spans actual or defined anatomical boundaries. There is no requirement that the multiple locations be contiguous.
Rows 9, 10 | The INFERRED FROM items allow the specification (by-value or by-reference) of numeric values that were used in the derivation of the numeric measurement of Row 1. The nature of the inference is not explicitly conveyed; it may be implicit in the Concept Names of the measurements. Inference by-reference is valid only in SOP Classes that permit the INFERRED FROM relationship by-reference.
Row 16 | Equivalent Meaning of Concept Name allows the creating application to specify the preferred composed concept name representing the measurement and the associated post-coordinated concept modifiers. The concept modifiers may include those specified in this Template, in a Template that invokes this Template, or at the option of the creating application in accordance with section 6.2.4. This composed concept name may be rendered by a display application.
Row 18 | Row 18 is a reference to an RWV that describes how measurements were made in units that differ from the stored pixel values in the images referenced in Row 13. E.g., for a PET SUVbw measurement, the mapping from activity/concentration units in the referenced image that was used (and which may be reused for measurements in the future) may be encoded in a referenced RWV instance. This reference overrides any reference in an including Template (such as for a Measurement Group).

### TID 310 Measurement Properties

This Template provides the properties of a numeric measurement, including evaluations of its normality and/or significance, its relationship to a reference population, and an indication of its selection from a set of measurements.

Table TID 310. Parameters

Parameter Name | Parameter Usage
--- | ---
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 310. Measurement Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121402, DCM, "Normality") | 1 | U |  | DCID 222 “Normality Codes”
2 |  |  | INCLUDE | DTID 311 “Measurement Statistical Properties” | 1 | U |  | $RefAuthority = $RefAuthority
3 |  |  | INCLUDE | DTID 312 “Normal Range Properties” | 1 | U |  | $RangeAuthority = $RangeAuthority
4 |  |  | CODE | EV (121403, DCM, "Level of Significance") | 1 | U |  | DCID 220 “Level of Significance”
5 |  |  | NUM | DCID 225 “Measurement Uncertainty Concepts” | 1-n | U |  |
6 |  |  | CODE | EV (121404, DCM, "Selection Status") | 1 | U |  | DCID 224 “Selection Method”

### TID 311 Measurement Statistical Properties

This Template provides the statistical properties of a reference population for a numeric measurement, and/or the position of a measurement in such a reference population.

Table TID 311. Parameters

Parameter Name | Parameter Usage
--- | ---
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 311. Measurement Statistical Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 221 “Measurement Range Concepts” | 1-n | M |  |
2 |  |  | TEXT | EV (121405, DCM, "Population description") | 1 | U |  |
3 |  |  | TEXT | EV (121406, DCM, "Reference Authority") | 1 | UC | XOR row 4 |
4 |  |  | CODE | EV (121406, DCM, "Reference Authority") | 1 | UC | XOR row 3 | $RefAuthority

### TID 312 Normal Range Properties

This Template provides the normal range of values for a numeric measurement.

Table TID 312. Parameters

Parameter Name | Parameter Usage
--- | ---
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 312. Normal Range Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 223 “Normal Range Values” | 1-n | M |  |
2 |  |  | TEXT | EV (121407, DCM, "Normal Range description") | 1 | U |  |
3 |  |  | TEXT | EV (121408, DCM, "Normal Range Authority") | 1 | UC | XOR row 4 |
4 |  |  | CODE | EV (121408, DCM, "Normal Range Authority") | 1 | UC | XOR row 3 | $RangeAuthority

### TID 315 Equation or Table

Table TID 315. Parameters

Parameter Name | Parameter Usage
--- | ---
$Equation | Coded term or Context Group for the equation or table from which a measurement was derived or computed

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 315. Equation or Table

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DCID 228 “Equation or Table” | 1 | M |  | $Equation
2 | > | HAS PROPERTIES | NUM |  | 1-n | U |  |
3 | > | R-HAS PROPERTIES | NUM |  | 1-n | U |  |

Content Item Descriptions

Row 2 | The HAS PROPERTIES allows the specification of the numeric values used as input to the equation or table identified in Row 1.
Row 3 | The HAS PROPERTIES allows the specification by-reference of the numeric values used as input to the equation or table. This row is valid only in SOP Classes that permit the HAS PROPERTIES relationship by-reference.

### Note

For example, if Row 1 identifies a specific Body Surface Area equation, Rows 2 and 3 can be used to convey (by-value or by-reference) the Patient Height and Patient Weight numeric measurements used in the BSA computation.

### TID 320 Image or Spatial Coordinates

This Template provides a general structure for inference from an image, either as a whole, or with specific spatial coordinates, as a single included Template in the invoking Template. If allowed by the IOD, the Image Content Item may be included by-reference.

Table TID 320. Parameters

Parameter Name | Parameter Usage
--- | ---
$Purpose | Purpose of Reference for an image used as a source of the measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 320. Image or Spatial Coordinates

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | INFERRED FROM | IMAGE | $Purpose | 1 | MC | XOR Rows 2, 3 |
2 |  | R-INFERRED FROM | IMAGE |  | 1 | MC | XOR Rows 1, 3 |
3 |  | INFERRED FROM | SCOORD | $Purpose | 1 | MC | XOR Rows 1, 2 |
4 | > | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 5 |
5 | > | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 4 |

### TID 321 Waveform or Temporal Coordinates

This Template provides a general structure for referencing a waveform, either as a whole, or with specific temporal coordinates, as a single included Template in the invoking Template. If allowed by the IOD, the Waveform Content Item may be included by-reference.

Table TID 321. Parameters

Parameter Name | Parameter Usage
--- | ---
$Purpose | Purpose of Reference for a waveform used as a source of the measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 321. Waveform or Temporal Coordinates

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | INFERRED FROM | WAVEFORM | $Purpose | 1 | MC | XOR Rows 2, 3 |
2 |  | R-INFERRED FROM | WAVEFORM |  | 1 | MC | XOR Rows 1, 3 |
3 |  | INFERRED FROM | TCOORD | $Purpose | 1 | MC | XOR Rows 1, 2 |
4 | > | SELECTED FROM | WAVEFORM |  | 1 | MC | XOR Row 5 |
5 | > | R-SELECTED FROM | WAVEFORM |  | 1 | MC | XOR Row 4 |

### TID 350 References to Supporting Evidence

This Template provides references to supporting evidence in the form of DICOM composite objects. This includes references to images, spatial coordinates on images, and other composite objects, such as Structured Reports.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 350. References to Supporting Evidence

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | IMAGE | BCID 7003 “Diagnostic Imaging Report Purposes of Reference” | 1-n | U |  |
2 |  |  | SCOORD | BCID 7003 “Diagnostic Imaging Report Purposes of Reference” | 1-n | U |  |
3 | > | SELECTED FROM | IMAGE |  | 1 | M |  |
4 |  |  | COMPOSITE | DT (122073, DCM, "Current procedure evidence") | 1-n | U |  |
5 | > | HAS CONCEPT MOD | CODE | EV (121144, DCM, "Document Title") | 1 | U |  |

### TID 351 Previous Reports

This general Template provides a means to reference previous structured reporting composite object instances.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 351. Previous Reports

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111549, DCM, "Previous Reports") | 1 | M |  |
2 | > | CONTAINS | COMPOSITE |  | 1-n | M |  |

Content Item Descriptions

Row 2 | Concept Name may be the Root Concept Name (title) of a Structured Report composite object instance.

### TID 400 Reference Location

This TID is a subset of the Reference Location Macro. See Section 10.27 “Reference Location Macro” in PS3.3 .

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 400. Reference Location

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (128772, DCM, "Reference Basis") | 1 | M |  | BCID 1001 “Anatomical Reference Basis”
2 |  |  | CODE | EV (128773, DCM, "Reference Geometry") | 1 | M |  | BCID 1010 “Reference Geometry - Planes”

### TID 1000 Quotation

Unless otherwise specified, content in an SR tree is "directly" observed. When material is quoted (from a source that is either a document or something spoken), then it is necessary to specify:

- the fact that one is quoting

- who is doing the quoting

- the source of the quote

- who is being quoted, and who and what the quote is about

This Template establishes a mechanism for quoting by specifying:

- the fact that one is quoting, by the presence of the contents of the Template in the tree

- that the "observer context" above the invocation of this Template establishes who is doing the quoting

- the source of the quote, by the values of the Content Items in this Template

- who is being quoted, and who and what the quote is about, by the observation context that is established at the start of the quoted material

This Template may be invoked recursively, to nest quotes within quotes. In essence, the chain of who is quoting whom can be established by maintaining a "stack" of observer context.

If a dimension of observation context is the same in the quoted material as in the enclosing tree, then the observation context does not need to be respecified (e.g., the quote may be about the same subject or procedure). Typically, the observer context would change (unless one were quoting oneself).

In the case of quoting something that was spoken, the "observer" is the person speaking.

TID 1000 is attached using HAS OBS CONTEXT relationships to the top node of the material that is being quoted. The presence of the Quoted Source concept signals the fact that the material is quoted rather than directly observed.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1000. Quotation

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS OBS CONTEXT | CODE | EV (121001, DCM, "Quotation Mode") | 1 | M |  | EV (121003, DCM, "Document") EV (121004, DCM, "Verbal")
2 |  | HAS OBS CONTEXT | COMPOSITE | EV (121002, DCM, "Quoted Source") | 1 | MC | Required if quoted material source is a DICOM composite object |
3 |  | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |

### TID 1001 Observation Context

Specifies attributes of observation context that may be defined, extended or replaced at any location in the SR tree.

This includes attributes that specify:

- who or what the observation is about ("subject context")

- what procedure the observation is about ("procedure context")

- who or what is making the observation ("observer context")

Establishing context includes two aspects of each dimension: identification and description (e.g., patient name and ID vs. patient's age, height or weight).

Whenever one dimension of context is changed or an attribute is added, all attributes of that dimension of context are "flushed", that is they need to be repeated in their entirety. For example, when the subject is changed from patient (name, id) to fetus (number), then the parameters of the patient are discarded. E.g., the patient's ID does not apply to the fetus.

"Extending" the same class and dimension of observation context isn't feasible, since one cannot "null out" or remove a previously set attribute. Any time a dimension of observation context is "replaced", any attributes that are unspecified remain unspecified (i.e., they are not inherited).

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 1001. Observation Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | MC | Required if all aspects of observer context are not inherited. | Defaults to the attributes of the Author Observer Sequence (0040,A078), or the Verifying Observer Sequence (0040,A073) if the Author Observer Sequence is not present
2 |  | HAS OBS CONTEXT | INCLUDE | DTID 1005 “Procedure Context” | 1 | MC | Required if all aspects of procedure context are not inherited. |
3 |  | HAS OBS CONTEXT | INCLUDE | DTID 1006 “Subject Context” | 1 | MC | Required if all aspects of observation subject context are not inherited. |

### TID 1002 Observer Context

The observer (person or device) that created the Content Items to which this context applies.

Whenever this Template is invoked, all previously inherited attributes of Observer Context are discarded and replaced.

There may be more than one observer, as this Template may be invoked with a VM 1-n, and both person and device observers. In such a case, the Content Items of TID 1003 “Person Observer Identifying Attributes” and TID 1004 “Device Observer Identifying Attributes” shall be included in the order in which the values of Observer Type are specified. Since TID 1003 “Person Observer Identifying Attributes” and TID 1004 “Device Observer Identifying Attributes” both include a single mandatory Content Item as their first Content Item, which observer is being described can be determined

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 1002. Observer Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS OBS CONTEXT | CODE | EV (121005, DCM, "Observer Type") | 1 | MC | IF Observer type is device | DCID 270 “Observer Type” Defaults to (121006, DCM, "Person")
2 |  | HAS OBS CONTEXT | INCLUDE | DTID 1003 “Person Observer Identifying Attributes” | 1 | MC | IFF Row 1 value = (121006, DCM, "Person") or Row 1 is absent |
3 |  | HAS OBS CONTEXT | INCLUDE | DTID 1004 “Device Observer Identifying Attributes” | 1 | MC | IFF Row 1 value = (121007, DCM, "Device") |

### TID 1003 Person Observer Identifying Attributes

This Template contains identifying (and optionally descriptive) attributes of persons that are observers.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1003. Person Observer Identifying Attributes

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | PNAME | EV (121008, DCM, "Person Observer Name") | 1 | M |  |
2 |  |  | TEXT | EV (121009, DCM, "Person Observer's Organization Name") | 1 | U |  | Defaults to Institution Name (0008,0080) of the General Equipment Module
3 |  |  | CODE | EV (121010, DCM, "Person Observer's Role in the Organization") | 1 | U |  | BCID 7452 “Organizational Roles”
4 |  |  | CODE | EV (121011, DCM, "Person Observer's Role in this Procedure") | 1 | U |  | BCID 7453 “Performing Roles”

### TID 1004 Device Observer Identifying Attributes

This Template (derived from the DICOM General Equipment Module of PS3.3) contains identifying (and optionally descriptive) attributes of devices that are observers.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1004. Device Observer Identifying Attributes

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | UIDREF | EV (121012, DCM, "Device Observer UID") | 1 | M |  |
2 |  |  | TEXT | EV (121013, DCM, "Device Observer Name") | 1 | U |  | Defaults to value of Station Name (0008,1010) in General Equipment Module
3 |  |  | TEXT | EV (121014, DCM, "Device Observer Manufacturer") | 1 | U |  | Defaults to value of Manufacturer (0008,0070) in General Equipment Module
4 |  |  | TEXT | EV (121015, DCM, "Device Observer Model Name") | 1 | U |  | Defaults to value of Manufacturer's Model Name (0008,1090) in General Equipment Module
5 |  |  | TEXT | EV (121016, DCM, "Device Observer Serial Number") | 1 | U |  | Defaults to value of Device Serial Number (0018,1000) in General Equipment Module
6 |  |  | TEXT | EV (121017, DCM, "Device Observer Physical Location During Observation") | 1 | U |  |
7 |  |  | CODE | EV (113876, DCM, "Device Role in Procedure") | 1-n | U |  | BCID 7445 “Device Participating Roles”
8 |  |  | TEXT | EV (110119, DCM, "Station AE Title") | 1 | U |  |

Content Item Descriptions

Row 7 | If the device performing the observations has other roles, e.g., as the irradiating device in a dose report, this may be recorded here, if not implicit.

### TID 1005 Procedure Context

This Template contains identifying (and optionally descriptive) attributes of the procedure that is the source of evidence being interpreted.

Whenever this Template is invoked, all previously inherited attributes of Procedure Context are discarded and replaced.

### Note

If an observed digital image is identified by other than a DICOM UID, a Study Instance UID must be generated for the non-DICOM evidence. The same must be done to document interpretation of hard-copy radiographs generated outside of the scope of the DICOM system.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 1005. Procedure Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | UIDREF | EV (121018, DCM, "Procedure Study Instance UID") | 1 | U |  | Defaults to Study Instance UID (0020, 000D) of General Study Module
2 |  |  | UIDREF | EV (121019, DCM, "Procedure Study Component UID") | 1-n | U |  |
3 |  |  | TEXT | EV (121020, DCM, "Placer Number") | 1 | U |  |
4 | > | HAS CONCEPT MOD | TEXT | EV (110190, DCM, "Issuer of Identifier") | 1 | U |  | See note
5 |  |  | TEXT | EV (121021, DCM, "Filler Number") | 1 | U |  |
6 | > | HAS CONCEPT MOD | TEXT | EV (110190, DCM, "Issuer of Identifier") | 1 | U |  | See note
7 |  |  | TEXT | EV (121022, DCM, "Accession Number") | 1 | U |  | Defaults to (0008,0050)
8 | > | HAS CONCEPT MOD | TEXT | EV (110190, DCM, "Issuer of Identifier") | 1 | U |  | See note
9 |  |  | CODE | EV (121023, DCM, "Procedure Code") | 1-n | U |  | Defaults to Procedure Code Sequence (0008,1032) of General Study Module

Content Item Descriptions

Rows 5, 6 | The issuer shall be formatted in accordance with the HL7v2 Hierarchic Designator Data Type. That format is [ Namespace ID]^[ Universal ID^ Universal ID Type], where Namespace IDidentifies an entity within the local namespace or domain, Universal IDis a universal or unique identifier for an entity, and Universal ID Typespecifies the standard format of the Universal ID (see HL7 v2 Section 2.A.33).

### TID 1006 Subject Context

This Template contains identifying (and optionally descriptive) attributes of the subject of the observation.

Subject context identifies (and optionally) describes the subject of the observation, whether it be a patient (human or animal), a fetus (human or animal), a specimen, or a device.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 1006. Subject Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121024, DCM, "Subject Class") | 1 | MC | IF subject is not the Patient | DCID 271 “Observation Subject Class” Defaults to (121025, DCM, "Patient")
2 |  |  | INCLUDE | DTID 1007 “Subject Context, Patient” | 1 | UC | IFF Row 1 value = (121025, DCM, "Patient") or Row 1 is absent | May be used for human or animal patients
3 |  |  | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | UC | IFF Row 1 value = (121026, DCM, "Fetus") | May be used for human or animal fetuses
4 |  |  | INCLUDE | DTID 1009 “Subject Context, Specimen” | 1 | UC | IFF Row 1 value = (121027, DCM, "Specimen") |
5 |  |  | INCLUDE | DTID 1010 “Subject Context, Device” | 1 | UC | IFF Row 1 value = (121192, DCM, "Device Subject") |

### TID 1007 Subject Context, Patient

Identifies (and optionally describes) a patient who is the subject.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1007. Subject Context, Patient

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | UIDREF | EV (121028, DCM, "Subject UID") | 1 | U |  | E.g., SOP Instance UID of Detached Patient Instance
2 |  |  | PNAME | EV (121029, DCM, "Subject Name") | 1 | MC | Required if not inherited. | Defaults to value of Patient's Name (0010,0010) in Patient Module
3 |  |  | CODE | EV (121030, DCM, "Subject ID") | 1 | MC | Required if not inherited. | Defaults to value of Patient ID (0010,0020) in Patient Module
4 |  |  | DATE | EV (121031, DCM, "Subject Birth Date") | 1 | U |  | Defaults to value of Patient's Birth Date (0010,0030) in Patient Module
5 |  |  | CODE | EV (121032, DCM, "Subject Sex") | 1 | U |  | Defaults to value equivalent to Patient's Sex (0010,0040) in Patient Module DCID 7455 “Sex”
6 |  |  | NUM | EV (121033, DCM, "Subject Age") | 1 | U |  | Defaults to value of Patient's Age (0010,1010) in Patient Study Module UNITS = DCID 7456 “Units of Measure for Age”
7 |  |  | CODE | EV (121034, DCM, "Subject Species") | 1 | MC | Required if not inherited. | DCID 7454 “Animal Taxonomic Rank Values” Defaults to value of Patient Species Code Sequence (0010,2202) in Patient Module, or if absent, (L-85003, SRT, "homo sapiens").
8 |  |  | CODE | EV (121035, DCM, "Subject Breed") | 1 | U |  | Defaults to value of Patient Breed Code Sequence (0010,2293) in Patient Module DCID 7480 “Breed”

### TID 1008 Subject Context, Fetus

Identifies (and optionally describes) a fetus who is the subject.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1008. Subject Context, Fetus

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | PNAME | EV (121036, DCM, "Mother of fetus") | 1 | U |  | Defaults to an observation subject that is a patient prior to replacing the Observation Subject Class with Fetus.
2 |  |  | UIDREF | EV (121028, DCM, "Subject UID") | 1 | U |  | For longitudinal tracking of individual fetuses
3 |  |  | TEXT | EV (121030, DCM, "Subject ID") | 1 | MC | IF row 4 is absent | For longitudinal tracking of individual fetuses (human readable value e.g., "A" or "1")
4 |  |  | TEXT | EV (11951-1, LN, "Fetus ID") | 1 | MC | IF row 3 is absent | For separation of multiple fetuses during this procedure e.g., fetus '1" of '2" … not for longitudinal comparisons.; i.e., the "m" of fetus "m" of "n"
5 |  |  | NUM | EV (11878-6, LN, "Number of Fetuses by US") | 1 | U | XOR Row 6 | UNITS = EV (1, UCUM, "no units")
6 |  |  | NUM | EV (55281-0, LN, "Number of Fetuses") | 1 | UC | XOR Row 5 | UNITS = EV (1, UCUM, "no units")

Content Item Descriptions

Row 5, 6 | The "n" of fetus "m" of "n"; either the code for the ultrasound method (Row 5) or for the non-specific method (Row 6) may be used

### TID 1009 Subject Context, Specimen

Identifies (and optionally describes) a specimen that is the subject.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1009. Subject Context, Specimen

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | UIDREF | EV (121039, DCM, "Specimen UID") | 1 | U |  |
2 |  |  | INCLUDE | DTID 1007 “Subject Context, Patient” | 1 | UC | IFF the source of the specimen is a human or animal patient |
3 |  |  | TEXT | EV (121041, DCM, "Specimen Identifier") | 1 | U |  |
4 |  |  | TEXT | EV (111724, DCM, "Issuer of Specimen Identifier") | 1 | U |  | See Content Item descriptions
5 |  |  | CODE | EV (R-00254, SRT, "Specimen Type") | 1 | U |  | DCID 8103 “Anatomic Pathology Specimen Types”
6 |  |  | TEXT | EV (111700, DCM, "Specimen Container Identifier") | 1 | U |  |

Content Item Descriptions

Rows 5, 6 | The Issuer of Specimen Identifier shall be formatted in accordance with the HL7 v2 Hierarchic Designator data type (see HL7 v2.6 Section 2.A.33), i.e., [ Namespace ID]^[ Universal ID^Universal ID Type]

### TID 1010 Subject Context, Device

Identifies (and optionally describes) a device that is the subject of observations.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1010. Subject Context, Device

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (121193, DCM, "Device Subject Name") | 1 | M |  |
2 |  |  | UIDREF | EV (121198, DCM, "Device Subject UID") | 1 | U |  |
3 |  |  | TEXT | EV (121194, DCM, "Device Subject Manufacturer") | 1 | U |  |
4 |  |  | TEXT | EV (121195, DCM, "Device Subject Model Name") | 1 | U |  |
5 |  |  | TEXT | EV (121196, DCM, "Device Subject Serial Number") | 1 | U |  |
6 |  |  | TEXT | EV (121197, DCM, "Device Subject Physical Location during observation") | 1 | U |  |

### TID 1020 Person Participant

This Template describes a person participating in an activity as other than an observer or subject. E.g., for a dose report documenting an irradiating procedure, participants include the person administering the irradiation and the person authorizing the irradiation.

This Template is included with specific contextual parameters from a parent Template.

Table TID 1020. Parameters

Parameter Name | Parameter Usage
--- | ---
$PersonProcedureRole | Coded term or Context Group for the Concept Name that describes the nature of the person's participation in this procedure.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1020. Person Participant

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | PNAME | EV (113870, DCM, "Person Name") | 1 | M |  |
2 | > | HAS PROPERTIES | CODE | EV (113875, DCM, "Person Role in Procedure") | 1 | M |  | $PersonProcedureRole
3 | > | HAS PROPERTIES | TEXT | EV (113871, DCM, "Person ID") | 1 | U |  |
4 | > | HAS PROPERTIES | TEXT | EV (113872, DCM, "Person ID Issuer") | 1 | U |  |
5 | > | HAS PROPERTIES | TEXT | EV (113873, DCM, "Organization Name") | 1 | U |  |
6 | > | HAS PROPERTIES | CODE | EV (113874, DCM, "Person Role in Organization") | 1 | U |  | BCID 7452 “Organizational Roles”

Content Item Descriptions

Row 1 | The name of the person participating in the role identified in Row 2.
Row 2 | The procedural role played by the person in this procedure.
Row 6 | The organizational role played by the person in the organization.

### TID 1021 Device Participant

This Template describes a device participating in an activity as other than an observer or subject. E.g., for a dose report documenting an irradiating procedure, participants include the irradiating device.

This Template is included with specific contextual parameters from a parent Template.

Table TID 1021. Parameters

Parameter Name | Parameter Usage
--- | ---
$DeviceProcedureRole | Coded term or Context Group for the Concept Name that describes the nature of the device's participation in this procedure.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1021. Device Participant

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (113876, DCM, "Device Role in Procedure") | 1 | M |  | $DeviceProcedureRole
2 | > | HAS PROPERTIES | TEXT | EV (113877, DCM, "Device Name") | 1 | U |  |
3 | > | HAS PROPERTIES | TEXT | EV (113878, DCM, "Device Manufacturer") | 1 | M |  |
4 | > | HAS PROPERTIES | TEXT | EV (113879, DCM, "Device Model Name") | 1 | M |  |
5 | > | HAS PROPERTIES | TEXT | EV (113880, DCM, "Device Serial Number") | 1 | M |  |
6 | > | HAS PROPERTIES | UIDREF | EV (121012, DCM, "Device Observer UID") | 1 | M |  |

Content Item Descriptions

Row 1 | If no Device Procedure Role is provided, BCID 7445 “Device Participating Roles” may be used.
Row 2 | This may be used for the name by which the organization manages the device.

### TID 1200 Language Designation

Defines a mechanism for specifying a language, optionally with designation of the country in which that language applies.

### Note

1. For example, the French language could be specified unmodified, or French as written in France or Canada could be distinguished.

2. The language codes specified in CID 5000 “Languages” optionally allow the encoding of the country of language in the code value for the language. Encoding of the country of language in a separate subsidiary Concept Modifier Content Item is allowed for backward compatibility with previous editions of the Standard.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1200. Language Designation

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121045, DCM, "Language") | 1 | M |  | DCID 5000 “Languages”
2 | > | HAS CONCEPT MOD | CODE | EV (121046, DCM, "Country of Language") | 1 | U |  | DCID 5001 “Countries”

### TID 1201 Language of Value

Defines a mechanism for specifying the language in which the value of the parent Content Item (only) is written. Does not specify the language of the Concept Name of the parent Content Item, nor of any other descendants of the parent Content Item.

### Note

The language codes specified in CID 5000 “Languages” optionally allow the encoding of the country of language in the code value for the language. Encoding of the country of language in a separate subsidiary Concept Modifier Content Item is allowed for backward compatibility with previous editions of the Standard.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 1201. Language of Value

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | CODE | EV (121047, DCM, "Language of Value") | 1 | M |  | DCID 5000 “Languages”
2 | > | HAS CONCEPT MOD | CODE | EV (121046, DCM, "Country of Language") | 1 | U |  | DCID 5001 “Countries”

### TID 1202 Language of Name and Value

Defines a mechanism for specifying the language in which the value and the Concept Name of the parent Content Item (only) is written. Does not specify the language of any other descendants of the parent Content Item.

### Note

The language codes specified in CID 5000 “Languages” optionally allow the encoding of the country of language in the code value for the language. Encoding of the country of language in a separate subsidiary Concept Modifier Content Item is allowed for backward compatibility with previous editions of the Standard.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1202. Language of Name and Value

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | CODE | EV (121048, DCM, "Language of Name and Value") | 1 | M |  | DCID 5000 “Languages”
2 | > | HAS CONCEPT MOD | CODE | EV (121046, DCM, "Country of Language") | 1 | U |  | DCID 5001 “Countries”

### TID 1204 Language of Content Item and Descendants

Defines a mechanism for specifying the language in which the value and the Concept Name of the parent Content Item and any other descendants of the parent Content Item is written.

### Note

The language codes specified in CID 5000 “Languages” optionally allow the encoding of the country of language in the code value for the language. Encoding of the country of language in a separate subsidiary Concept Modifier Content Item is allowed for backward compatibility with previous editions of the Standard.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 1204. Language of Content Item and Descendants

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | CODE | EV (121049, DCM, "Language of Content Item and Descendants") | 1 | M |  | DCID 5000 “Languages”
2 | > | HAS CONCEPT MOD | CODE | EV (121046, DCM, "Country of Language") | 1 | U |  | DCID 5001 “Countries”

### TID 1210 Equivalent Meaning(s) of Concept Name

Defines a mechanism for specifying one or more equivalent meanings for the Concept Name of the parent Content Item.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 1210. Equivalent Meaning(s) of Concept Name

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | TEXT | EV (121050, DCM, "Equivalent Meaning of Concept Name") | 1-n | MC | XOR Row 3 | Plain text equivalent of code meaning of the concept name of the Content Item being modified, in the specified language from the specified country, using the default character set or a character set selected from Specified Character Set
2 | > |  | INCLUDE | DTID 1201 “Language of Value” | 1 | U |  |
3 |  | HAS CONCEPT MOD | CODE | EV (121050, DCM, "Equivalent Meaning of Concept Name") | 1-n | MC | XOR Row 1 |
4 | > |  | INCLUDE | DTID 1201 “Language of Value” | 1 | U |  |

### Note

A coded equivalent meaning for the Concept Name can also be included using the attribute Equivalent Code Sequence (0008,0121) in the Concept Name Code Sequence (0040,A043) (see Section 8.9 “Equivalent Code Sequence” in PS3.3 ), though the equivalent code(s) in the Equivalent Code Sequence (0008,0121) need not be the same as those in TID 1210.

### TID 1211 Equivalent Meaning(s) of Value

Defines a mechanism for specifying one or more equivalent meanings for the Value of the parent Content Item.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1211. Equivalent Meaning(s) of Value

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | TEXT | EV (121051, DCM, "Equivalent Meaning of Value") | 1-n | MC | XOR Row 3 | Plain text equivalent of code meaning of the value of the Content Item being modified, in the specified language from the specified country, using the default character set or a character set selected from Specified Character Set
2 | > |  | INCLUDE | DTID 1201 “Language of Value” | 1 | U |  |
3 |  | HAS CONCEPT MOD | CODE | EV (121051, DCM, "Equivalent Meaning of Value") | 1-n | MC | XOR Row 1 |
4 | > |  | INCLUDE | DTID 1201 “Language of Value” | 1 | U |  |

### Note

1. For example, to describe a longer, more meaningful equivalent (in the same language) for a procedure code than is defined in a coding scheme:

   CODE: (121023, DCM, "Procedure Code") = (50291CC, ICD10PCS, "IMAGING:CNS:CT:SELLA:LOWOSMOLAR:IT, U, E:2PLANE3D")

   > HAS CONCEPT MOD TEXT: (121051, DCM, "Equivalent meaning of value") = "imaging study central nervous system of the sella turcica/pituitary gland with low osmolar contrast intrathecal, unenhanced and enhanced, in two planes with 3D reconstructions"

2. For example, to specify a concept name and value in both French and English in Canada:

   CODE:(T-D0005, SRT, "Anatomical structure") = (T-04000, SRT, "Breast")

   > HAS CONCEPT MOD CODE: (121048, DCM, "Language of name and value") = (en-CA, RFC3066, "English, Canada")

   > HAS CONCEPT MOD CODE: (121050, DCM, "Equivalent meaning of concept name") = (T-D0005, SRT, " Structure de l'anatomie ")

   >> HAS CONCEPT MOD CODE: (121047, DCM, " Langue de la valeur ") = (fr-CA, RFC3066, " Français, Canadien ")

   > HAS CONCEPT MOD CODE: (121051, DCM, "Equivalent meaning of value") = (T-04000, SRT, " Sein ")

   >> HAS CONCEPT MOD CODE: (121047, DCM, " Langue de la valeur ") = (fr-CA, RFC3066, " Français, Candie ")

3. 3. A coded equivalent meaning for the Concept Value of a CODE Content Item can also be included using the attribute Equivalent Code Sequence (0008,0121) in the Concept Code Sequence (0040, A168) (see Section 8.9 “Equivalent Code Sequence” in PS3.3 ).

### TID 1350 Negation Modifier, Presence of Finding

Concept Name Modifier for negation of the presence of a finding represented by a post-coordinated concept.

### Note

1. For example, negation modifier applied to "distention" in the post-coordinated structure:

   CODE: "anatomic location" = "bile duct"

   > HAS PROPERTY CODE: "morphology" = "distention"

   >> HAS CONCEPT MOD CODE: "presence of property" = "absent"

   means: "bile duct distention not present"

2. The presence-negation modifier modifies the entire post-coordinated concept, not just the Source Content Item of the HAS CONCEPT MOD relationship. The entire branch of the tree from the Content Item is included in the post-coordinated structure that is negated.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1350. Negation Modifier, Presence of Finding

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | CODE | EV (121052, DCM, "Presence of property") | 1 | M |  | DCID 240 “Present-Absent”

### TID 1400 Linear Measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1400. Linear Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 7470 “Linear Measurements” | 1 | M |  | UNITS = DCID 7460 “Units of Linear Measurement”
2 | > | INFERRED FROM | SCOORD | EV (121055, DCM, "Path") | 1 | UC | XOR Row 5 | GRAPHIC TYPE = {POLYLINE, CIRCLE, ELLIPSE}
3 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 4 |
4 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 3 |
5 | > | INFERRED FROM | SCOORD | EV (121230, DCM, "Path Vertex") | 2-n | UC | XOR Row 2 | GRAPHIC TYPE = {POINT}
6 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 7 |
7 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 6 |
8 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |

Content Item Descriptions

Row 2 | Path shall represent the measured path or a projection of the measured path in the image. The Graphic Type (0070,0023) of the Path SCOORD shall be: an open POLYLINE with two different points (to measure length, diameter, distance, proximity, etc), a CIRCLE or ELLIPSE (to measure circumference) or an open or closed POLYLINE (closed polygon) to measure path length (open) or perimeter (closed).
Row 5 "Path Vertex" | A measured path that traverses two or more images (e.g., the ends of the path are in different cross-sectional plane images) shall be identified by vertices along the path. The Path Vertices shall be ordered by the order of their SCOORD Content Items to identify the measured path. The Graphic Type (0070,0023) of each SCOORD shall be POINT

### TID 1401 Area Measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1401. Area Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 7471 “Area Measurements” | 1 | M |  | Value shall be > 0 UNITS = DCID 7461 “Units of Area Measurement”
2 | > | INFERRED FROM | SCOORD | EV (121056, DCM, "Area Outline") | 1 | MC | IF concept name of Row 1 is (G-A16A, SRT, "Area of defined region"), and IFF Row 5 or 6 not present. | GRAPHIC TYPE = not {MULTIPOINT}
3 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 4 |
4 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 3 |
5 | > | INFERRED FROM | IMAGE | EV (121214, DCM, "Referenced Segmentation Frame") | 1 | MC | IF concept name of Row 1 is (G-A16A, SRT, "Area of defined region"), and IFF Row 2 or 6 not present. | Reference shall be to a Segmentation Image, with a single value specified in Referenced Frame Number
6 | > | R- INFERRED FROM | IMAGE |  | 1 | MC | IF concept name of Row 1 is (G-A16A, SRT, "Area of defined region"), and IFF Row 2 or 5 not present. | Reference shall be to a Segmentation Image, with a single value specified in Referenced Frame Number
7 | > | HAS PROPERTIES | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | DCID 7473 “General Area Calculation Methods”
8 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |

Content Item Descriptions

Row 2 "Area Outline" | A Graphic Type of POINT implies that the object is a single pixel and the object's area is the area of the pixel. Otherwise the type shall be a closed POLYLINE (start and end point the same) or a CIRCLE or an ELLIPSE.
Rows 5, 6 | Referenced Frame Number (0008,1160) is an attribute of the IMAGE Content Item. If the Referenced Segmentation SOP Instance has Segmentation Type (0062,0001) value BINARY, it identifies the area of defined (measured) region by pixel values in the referenced frame with value 1. For Segmentation Type value FRACTIONAL, the area is computed by an implementation dependent method. Frame number shall be specified even if the Segmentation SOP Instance has only a single frame.
Row 8 | The values of (112039, DCM, "Tracking Identifier") and (112040, DCM, "Tracking Unique Identifier"), if present, shall match the corresponding values of Tracking ID (0062,0020) and Tracking UID (0062,0021), if present, in the corresponding Segment of any Segmentation instance referenced in Row 5.

### TID 1402 Volume Measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1402. Volume Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 7472 “Volume Measurements” | 1 | M |  | Value shall be > 0 UNITS = DCID 7462 “Units of Volume Measurement”
2 | > | INFERRED FROM | SCOORD | EV (121057, DCM, "Perimeter Outline") | 1-n | UC | XOR row 5, 6 | GRAPHIC TYPE = not {MULTIPOINT}
3 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 4 |
4 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 3 |
5 | > | INFERRED FROM | IMAGE | EV (121191, DCM, "Referenced Segment") | 1 | UC | XOR row 2, 6 | Reference shall be to a Segmentation Image, with a value specified in Referenced Segment Number
6 | > | R- INFERRED FROM | IMAGE |  | 1 | UC | XOR row 2, 5 | Reference shall be to a Segmentation Image, with a value specified in Referenced Segment Number
7 | > | HAS PROPERTIES | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | DCID 7474 “General Volume Calculation Methods”
8 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |

Content Item Descriptions

Row 2 "Perimeter Outline" | The two dimensional perimeter of the volume's intersection with or projection into the image. A Graphic Type of POINT implies that the volume's intersection or projection in a plane is a single pixel. A single pixel projection perimeter cannot cause a volume calculation to become 0. Otherwise the type shall be a closed POLYLINE (start and end point the same) or a CIRCLE or an ELLIPSE.
Rows 5, 6 | Referenced Segment Number (0062,000B) is an attribute of the IMAGE Content Item. If the Referenced Segmentation SOP Instance has Segmentation Type (0062,0001) value BINARY, it identifies the defined (measured) volume by pixel/voxel values in the frames of the referenced segment with value 1. For Segmentation Type value FRACTIONAL, the volume is computed by an implementation dependent method. Segment number shall be specified even if the Segmentation SOP Instance has only a single segment.
Row 8 | The values of (112039, DCM, "Tracking Identifier") and (112040, DCM, "Tracking Unique Identifier"), if present, shall match the corresponding values of Tracking ID (0062,0020) and Tracking UID (0062,0021), if present, in the corresponding Segment of any Segmentation instance referenced in Row 5.

### TID 1404 Numeric Measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1404. Numeric Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | No baseline CID | 1 | M |  | UNITS = DCID 82 “Units of Measurement”
2 | > | INFERRED FROM | SCOORD | No baseline CID | 1-n | UC | XOR Row 5, 6 |
3 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 4 |
4 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 3 |
5 | > | R-INFERRED FROM | IMAGE | BCID 7003 “Diagnostic Imaging Report Purposes of Reference” | 1-n | UC | XOR Row 2, 6 |
6 | > | INFERRED FROM | IMAGE | BCID 7003 “Diagnostic Imaging Report Purposes of Reference” | 1-n | UC | XOR Row 2, 5 |
7 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |

Content Item Descriptions

Row 2 | The SCOORD may indicate the points or area where the measurement was taken (e.g., a POINT showing the pixel location of a density measurement, or an open POLYLINE of three points showing the calculation of an angle).
Rows 3, 5 | Valid only in IODs that permit relationships by-reference.

### TID 1406 Three Dimensional Linear Measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 1406. Three Dimensional Linear Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 7470 “Linear Measurements” | 1 | M |  | UNITS = DCID 7460 “Units of Linear Measurement”
2 | > | INFERRED FROM | SCOORD3D | EV (121055, DCM, "Path") | 1 | M |  | GRAPHIC TYPE = {POLYLINE, ELLIPSE, POLYGON}
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |

Content Item Descriptions

Rows 5, 6 | Path shall represent the measured path in a reference coordinate space. The Graphic Type (0070,0023) of the Path SCOORD3D shall be: an open POLYLINE with two or more different (x,y,z) triplets (to measure length, diameter, distance, proximity, etc.), an ELLIPSE (to measure circumference) or a closed POLYGON to measure perimeter, where the (x,y,z) triplets are coplanar.

### TID 1410 Planar ROI Measurements

This Template provides a general structure to report one or more measurements for some metric, e.g., density, flow, or concentration, over a planar region of interest in an image. The ROI may be specified by an SCOORD on an image, or by a Segmentation Image.

Table TID 1410. Parameters

$Measurement | Coded term or Context Group for Concept Name of measurement
$Units | Units for the measurement
$ModType | Modifier Name for Concept Name of measurement
$ModValue | Modifier Value for Concept Name of measurement
$Method | Value for Measurement Method
$Derivation | Value for Measurement Derivation
$TargetSite | Value for Anatomic Location of measurement
$TargetSiteMod | Modifier Value for Anatomic Location of measurement
$Equation | Coded term or Context Group for the equation or table from which the measurement was derived or computed
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$DerivationParameter | Coded term or Context Group for Concept Name of a derivation parameter
$DerivationParameterUnits | Units of derivation parameter
$QualitativeEvaluations | Evaluations encoded with code or text responses
$FindingType | Type of the finding

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1410. Planar ROI Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125007, DCM, "Measurement Group") | 1 | M |  |
1b | > | HAS OBS CONTEXT | TEXT | EV (C67447, NCIt, "Activity Session") | 1 | U |  |
2 | > | HAS OBS CONTEXT | TEXT | DT (112039, DCM, "Tracking Identifier") | 1 | M |  |
3 | > | HAS OBS CONTEXT | UIDREF | EV (112040, DCM, "Tracking Unique Identifier") | 1 | M |  |
3b | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1 | U |  | $FindingType
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 1502 “Time Point Context” | 1 | U |  |
5 | > | CONTAINS | SCOORD | EV (111030, DCM, "Image Region") | 1 | MC | XOR Row 7 | GRAPHIC TYPE = not {MULTIPOINT}
6 | >> | SELECTED FROM | IMAGE |  | 1 | M |  |
7 | > | CONTAINS | IMAGE | EV (121214, DCM, "Referenced Segmentation Frame") | 1 | MC | XOR Row 5 | Reference shall be to a Segmentation Image, with a single value specified in Referenced Frame Number, and with a single value specified in Referenced Segment Number
8 | > | CONTAINS | IMAGE | EV (121233, DCM, "Source image for segmentation") | 1 | MC | IFF Row 7 |
9 | > | CONTAINS | IMAGE | EV (121200, DCM, "Illustration of ROI") | 1 | U |  |
10 | > | CONTAINS | COMPOSITE | EV (126100, DCM, "Real World Value Map used for measurement") | 1 | U |  | SOP Class UID shall be Real World Value Mapping Storage ("1.2.840.10008.5.1.4.1.1.67")
11 | > | CONTAINS | INCLUDE | DTID 1419 “ROI Measurements” | 1 | M |  | $Measurement = $Measurement $Units = $Units $ModType = $ModType $ModValue = $ModValue $Method = $Method $Derivation = $Derivation $TargetSite = $TargetSite $TargetSiteMod = $TargetSiteMod $Equation = $Equation $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority $DerivationParameter = $DerivationParameter $DerivationParameterUnits = $DerivationParameterUnits
12 | > | CONTAINS | CODE | $QualitativeEvaluations | 1-n | U |  |
13 | > | CONTAINS | TEXT | $QualitativeEvaluations | 1-n | U |  |

Content Item Descriptions

Row 1b | Identifies the session during which the measurements were made. The NCI Thesaurus definition is "time, period, or term devoted to some activity".
Rows 2, 3 | The Tracking Identifier and Tracking Unique Identifier are defined as a text label or unique identifier (respectively) used for tracking a finding or feature, potentially across multiple reporting objects, over time. As such, they are distinct from the Observation UID (0040,A171), which is unique identifier of the specific Content Item and its subsidiary Content Items that constitute an individual observation, and would be different for different observations on different occasions of the same finding or feature. The values of these content items shall match the corresponding values of Tracking ID (0062,0020) and Tracking UID (0062,0021), if present, in the corresponding Segment of any Segmentation instance referenced in Row 7.
Row 3b | The type of the finding describes whatever entity (finding or feature) is identified by Rows 2 and 3. E.g., a finding might be a lesion, a tumor, or a reference region (as distinct from its anatomical location, which is encoded in a different content item (Finding Site).
Rows 6, 7 | Referenced Frame Number (0008,1160) is an attribute of the IMAGE Content Item, and shall be present with a single value. If the Referenced Segmentation SOP Instance has Segmentation Type (0062,0001) value BINARY, it identifies the area of defined (measured) region of interest by pixel values in the referenced frame with value 1. For Segmentation Type value FRACTIONAL, the area is computed by an implementation dependent method. Frame number shall be specified even if the Segmentation SOP Instance has only a single frame.
Row 8 | Identifies the source image that was segmented to identify the ROI, and whose properties are described in this container.
Row 9 | This referenced image may contain a "screen shot" illustrating a rendered version of the ROI.
Row 10 | The reference to an RWV in Row 10 allows measurements to be made in units that differ from the stored pixel values in the images referenced elsewhere in the Template. E.g., for a PET SUVbw measurement, the mapping from activity/concentration units in the referenced image that was used (and which may be reused for measurements in the future) may be encoded in a referenced RWV instance. This reference applies to any measurements in included Templates, unless overridden).
Rows 12, 13 | Allows encoding a flat list of name-value pairs that are coded questions with coded or text answers, for example, to record categorical observations related to the subject of the measurement group.

### TID 1411 Volumetric ROI Measurements

This Template provides a general structure to report one or more measurements for some metric, e.g., density, flow, or concentration, over a volumetric region of interest in a set of images or a Frame of Reference. The volumetric ROI may be specified by a set of SCOORDs on an image set representing a volume, by a volumetric Segmentation Image, by a volume defined in a Surface Segmentation, or by a SCOORD3D.

Table TID 1411. Parameters

$Measurement | Coded term or Context Group for Concept Name of measurement
$Units | Units for the measurement
$ModType | Modifier Name for Concept Name of measurement
$ModValue | Modifier Value for Concept Name of measurement
$Method | Value for Measurement Method
$Derivation | Value for Measurement Derivation
$TargetSite | Value for Anatomic Location of measurement
$TargetSiteMod | Modifier Value for Anatomic Location of measurement
$Equation | Coded term or Context Group for the equation or table from which the measurement was derived or computed
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$DerivationParameter | Coded term or Context Group for Concept Name of a derivation parameter
$DerivationParameterUnits | Units of derivation parameter
$QualitativeEvaluations | Evaluations encoded with code or text responses
$FindingType | Type of the finding

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1411. Volumetric ROI Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125007, DCM, "Measurement Group") | 1 | M |  |
1b | > | HAS OBS CONTEXT | TEXT | EV (C67447, NCIt, "Activity Session") | 1 | U |  |
2 | > | HAS OBS CONTEXT | TEXT | DT (112039, DCM, "Tracking Identifier") | 1 | M |  |
3 | > | HAS OBS CONTEXT | UIDREF | EV (112040, DCM, "Tracking Unique Identifier") | 1 | M |  |
3b | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1 | U |  | $FindingType
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 1502 “Time Point Context” | 1 | U |  |
5 | > | CONTAINS | SCOORD | EV (111030, DCM, "Image Region") | 1-n | MC | XOR Rows 7, 10 | GRAPHIC TYPE = not {MULTIPOINT}
6 | >> | SELECTED FROM | IMAGE |  | 1 | M |  |
7 | > | CONTAINS | IMAGE | EV (121191, DCM, "Referenced Segment") | 1 | MC | XOR Rows 5, 10 | Reference shall be to a Segmentation Image or Surface Segmentation object, with a single value specified in Referenced Segment Number
10 | > | CONTAINS | SCOORD3D | EV (121231, DCM, "Volume Surface") | 1 | MC | XOR Rows 5, 7 | GRAPHIC TYPE = {ELLIPSOID}
11 | > | CONTAINS | IMAGE | EV (121233, DCM, "Source image for segmentation") | 1-n | MC | XOR Row 12 and IFF (Row 7 or Row 10) |
12 | > | CONTAINS | UIDREF | EV (121232, DCM, "Source series for segmentation") | 1 | MC | XOR Row 11 and IFF ((Row 7 or Row 10) |
13 | > | CONTAINS | IMAGE | EV (121200, DCM, "Illustration of ROI") | 1-n | U |  |
14 | > | CONTAINS | COMPOSITE | EV (126100, DCM, "Real World Value Map used for measurement") | 1 | U |  | SOP Class UID shall be Real World Value Mapping Storage ("1.2.840.10008.5.1.4.1.1.67")
15 | > | CONTAINS | INCLUDE | DTID 1419 “ROI Measurements” | 1 | M |  | $Measurement = $Measurement $Units = $Units $ModType = $ModType $ModValue = $ModValue $Method = $Method $Derivation = $Derivation $TargetSite = $TargetSite $TargetSiteMod = $TargetSiteMod $Equation = $Equation $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority $DerivationParameter = $DerivationParameter $DerivationParameterUnits = $DerivationParameterUnits
16 | > | CONTAINS | CODE | $QualitativeEvaluations | 1-n | U |  |
17 | > | CONTAINS | TEXT | $QualitativeEvaluations | 1-n | U |  |

Content Item Descriptions

Row 1b | Identifies the session during which the measurements were made. The NCI Thesaurus definition is "time, period, or term devoted to some activity".
Rows 2, 3 | The Tracking Identifier and Tracking Unique Identifier are defined as a text label or unique identifier (respectively) used for tracking a finding or feature, potentially across multiple reporting objects, over time. As such, they are distinct from the Observation UID (0040,A171), which is unique identifier of the specific Content Item and its subsidiary Content Items that constitute an individual observation, and would be different for different observations on different occasions of the same finding or feature. The values of these content items shall match the corresponding values of Tracking ID (0062,0020) and Tracking UID (0062,0021), if present, in the corresponding Segment of any Segmentation instance referenced in Row 7.
Row 3b | The type of the finding describes whatever entity (finding or feature) is identified by Rows 2 and 3. E.g., a finding might be a lesion, a tumor, or a reference region (as distinct from its anatomical location, which is encoded in a different content item (Finding Site).
Rows 6, 7 | Referenced Segment Number (0062,000B) is an attribute of the IMAGE Content Item, and shall be present with a single value. If the Referenced SOP Instance is a Segmentation Image, it shall have a defined Frame of Reference. If it has Segmentation Type (0062,0001) value BINARY, it identifies the volume of defined (measured) region of interest by voxel values in the referenced segment with value 1. If it has Segmentation Type value FRACTIONAL, the volume is defined by an implementation dependent method. If the referenced SOP Instance is a Surface Segmentation, the referenced segment shall constitute a finite volume. It identifies the volume of the defined (measured) region of interest by the interior of the finite volume. Segment number shall be specified even if the Segmentation SOP Instance has only a single segment.
Row 11 | Identifies the source images that were segmented to identify the ROI, when, for example a subset of images in a series was used.
Row 12 | Identifies the source series of images that were segmented to identify the ROI, when, for example an entire set of images in a series was used.
Row 13 | These referenced images may contain "screen shot" illustrating rendered versions of the ROI.
Row 14 | The reference to an RWV in Row 14 allows measurements to be made in units that differ from the stored pixel values in the images referenced elsewhere in the Template. E.g., for a PET SUVbw measurement, the mapping from activity/concentration units in the referenced image that was used (and which may be reused for measurements in the future) may be encoded in a referenced RWV instance. This reference applies to any measurements in included Templates, unless overridden).
Rows 16, 17 | Allows encoding a flat list of name-value pairs that are coded questions with coded or text answers, for example, to record categorical observations related to the subject of the measurement group.

### TID 1419 ROI Measurements

This Template encodes measurements for some metric, e.g., density, flow, or concentration.

Table TID 1419. Parameters

$Measurement | Coded term or Context Group for Concept Name of measurement
$Units | Units for the measurement
$ModType | Modifier Name for Concept Name of measurement
$ModValue | Modifier Value for Concept Name of measurement
$Method | Value for Measurement Method
$Derivation | Value for Measurement Derivation
$TargetSite | Value(s) for Anatomic Location of measurement
$TargetSiteMod | Modifier Value for Anatomic Location of measurement
$Equation | Coded term or Context Group for the equation or table from which the measurement was derived or computed
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$DerivationParameter | Coded term or Context Group for Concept Name of a derivation parameter
$DerivationParameterUnits | Units of derivation parameter

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1419. ROI Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | $Method
2 |  | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | U |  | $TargetSite
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
4 | > | HAS CONCEPT MOD | CODE | DT (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | $TargetSiteMod
5 |  |  | NUM | $Measurement | 1-n | M |  | UNITS = $Units
6 | > | HAS CONCEPT MOD | CODE | $ModType | 1-n | U |  | $ModValue
7 | > | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | $Method
8 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | U |  | $Derivation
9 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | U |  | $TargetSite
10 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
11 | >> | HAS CONCEPT MOD | CODE | DT (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | $TargetSiteMod
12 | > | HAS PROPERTIES | INCLUDE | DTID 310 “Measurement Properties” | 1 | U |  | $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority
13 | > | INFERRED FROM | NUM | $DerivationParameter | 1-n | UC | XOR Row 14 | $DerivationParameterUnits
14 | > | R-INFERRED FROM | NUM | $DerivationParameter | 1-n | UC | XOR Row 13 | $DerivationParameterUnits
15 | > | INFERRED FROM | INCLUDE | DTID 315 “Equation or Table” | 1 | UC | XOR Row 16 | $Equation = $Equation
16 | > | INFERRED FROM | TEXT | DCID 228 “Equation or Table” | 1 | UC | XOR Row 15 |
17 | > |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
18 | > | HAS CONCEPT MOD | TEXT | EV (121050, DCM, "Equivalent Meaning of Concept Name") | 1 | U |  |
19 | > | CONTAINS | COMPOSITE | EV (126100, DCM, "Real World Value Map used for measurement") | 1 | U |  | SOP Class UID shall be Real World Value Mapping Storage ("1.2.840.10008.5.1.4.1.1.67")

Content Item Descriptions

Rows 2, 9 | Finding site may be multiple when a region of interest spans multiple anatomical locations and there is not a single pre-coordinated code describing the combination of locations. E.g., when a maligant, inflammatory or traumatic process spans actual or defined anatomical boundaries. There is no requirement that the multiple locations be contiguous.
Row 5 | Specifies the metric for which measurements are reported, e.g., density, flow, or concentration. This metric is computed at each sample point (e.g., pixel or voxel) in an ROI (defined in the invoking Template), but those individual point measurements are not encoded. Instead, just the summary measurements for the ROI are encoded, and the means of computing a single value is defined in Row 8 Derivation (e.g., mean).
Rows 1, 2, 3, 4, 6, 7, 8, 9, 10, 11 | The HAS CONCEPT MOD items allow the explicit definition of terms for post-coordination of the measurement concept name. Additional post-coordinated modifier terms may be included in a SOP Instance based on this Template, in accordance with section 6.2.4, or as defined by Templates that invoke this Template and explicitly define additional post-coordinated modifiers.
Rows 13, 14 | The INFERRED FROM items allow the specification (by-value or by-reference) of numeric values that were used in the derivation of the numeric measurement of Row 1. The nature of the inference is not explicitly conveyed; it may be implicit in the Concept Names of the measurements. Inference by-reference is valid only in SOP Classes that permit the INFERRED FROM relationship by-reference.
Row 18 | Equivalent Meaning of Concept Name allows the creating application to specify the preferred composed concept name representing the measurement and the associated post-coordinated concept modifiers. The concept modifiers may include those specified in this Template, in a Template that invokes this Template, or at the option of the creating application in accordance with section 6.2.4. This composed concept name may be rendered by a display application.
Row 19 | The reference to an RWV in Row 19 allows measurements to be made in units that differ from the stored pixel values in the images referenced in the parent Template. E.g., for a PET SUVbw measurement, the mapping from activity/concentration units in the referenced image that was used (and which may be reused for measurements in the future) may be encoded in a referenced RWV instance. This reference overrides any reference in an including Template (such as for a Measurement Group).

### TID 1420 Measurements Derived From Multiple ROI Measurements

This Template encodes measurements for some metric, e.g., density, flow, or concentration, which are acquired over some defined sampling (e.g., over successive time slots in a dynamic contrast enhanced acquisition).

Table TID 1420. Parameters

$Measurement | Coded term or Context Group for Concept Name of measurement
$MeasurementUnits | Units for the measurement
$ModType | Modifier Name for Concept Name of measurement
$ModValue | Modifier Value for Concept Name of measurement
$Method | Value for Measurement Method
$Derivation | Value for Measurement Derivation
$TargetSite | Value for Anatomic Location of measurement
$TargetSiteMod | Modifier Value for Anatomic Location of measurement
$Equation | Coded term or Context Group for the equation or table from which the measurement was derived or computed
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$StatisticalRefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$StatisticalRangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$DerivationParameter | Coded term or Context Group for Concept Name of a derivation parameter
$DerivationParameterUnits | Units of derivation parameter

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1420. Measurements Derived From Multiple ROI Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 7465 “Measurements Derived From Multiple ROI Measurements” | 1-n | M |  |
2 | > | R-INFERRED FROM | INCLUDE | DTID 1410 “Planar ROI Measurements” | 1-n | MC | XOR Row 3 | $Measurement = $Measurement $Units = $MeasurementUnits $ModType = $ModType $ModValue = $ModValue $Method = $Method $Derivation = $Derivation $TargetSite = $TargetSite $TargetSiteMod = $TargetSiteMod $Equation = $Equation $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority $DerivationParameter = $DerivationParameter $DerivationParameterUnits = $DerivationParameterUnits
3 | > | R-INFERRED FROM | INCLUDE | DTID 1411 “Volumetric ROI Measurements” | 1-n | MC | XOR Row 2 | $Measurement = $Measurement $Units = $MeasurementUnits $ModType = $ModType $ModValue = $ModValue $Method = $Method $Derivation = $Derivation $TargetSite = $TargetSite $TargetSiteMod = $TargetSiteMod $Equation = $Equation $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority $DerivationParameter = $DerivationParameter $DerivationParameterUnits = $DerivationParameterUnits
4 | > | HAS PROPERTIES | INCLUDE | DTID 310 “Measurement Properties” | 1 | U |  | $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority

Content Item Descriptions

Row 1 | Specifies the type of derived measurement reported, e.g., the mean of the individual ROI mean density values. Note that the units may be different from the units in the ROI measurements
Rows 2, 3 | The measurement values of each ROI that contributes to the derived measurement, e.g., the mean density within an ROI. These are specified by reference, so as to not have to repeat the ROI information when it contributes to multiple derived measurements (e.g., if both mean and SD of ROI mean density values were specified).

### TID 1500 Measurement Report

This Root Template encodes a list of Measurement Groups each containing lists of measurements, together with any derived measurements.

Each Measurement Group is identified by Tracking ID and UIDs.

An image library is available to describe characteristics of the images referenced by the measurements, if any.

Type:

  Extensible

Order:

  Non-Significant

Root:

  Yes

Table TID 1500. Measurement Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DCID 7021 “Measurement Report Document Titles” | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
4 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1-n | M |  | BCID 100 “Quantitative Diagnostic Imaging Procedures”
5 | > | CONTAINS | INCLUDE | DTID 1600 “Image Library” | 1 | M |  |
6 | > | CONTAINS | CONTAINER | EV (126010, DCM, "Imaging Measurements") | 1 | C | IF row 10 and 12 are absent |
7 | >> | CONTAINS | INCLUDE | DTID 1410 “Planar ROI Measurements” | 1-n | U |  | $Measurement = BCID 7469 “Generic Intensity and Size Measurements” $Units = BCID 7181 “Abstract Multi-dimensional Image Model Component Units” $Derivation = BCID 7464 “General Region of Interest Measurement Modifiers” $Method = BCID 6147 “Response Criteria”
8 | >> | CONTAINS | INCLUDE | DTID 1411 “Volumetric ROI Measurements” | 1-n | U |  | $Measurement = BCID 7469 “Generic Intensity and Size Measurements” $Units = BCID 7181 “Abstract Multi-dimensional Image Model Component Units” $Derivation = BCID 7464 “General Region of Interest Measurement Modifiers” $Method = BCID 6147 “Response Criteria”
9 | >> | CONTAINS | INCLUDE | DTID 1501 “Measurement Group” | 1-n | U |  | $Measurement = BCID 7469 “Generic Intensity and Size Measurements” $Units = BCID 7181 “Abstract Multi-dimensional Image Model Component Units” $Derivation = BCID 7464 “General Region of Interest Measurement Modifiers” $Method = BCID 6147 “Response Criteria”
10 | > | CONTAINS | CONTAINER | EV (126011, DCM, "Derived Imaging Measurements") | 1 | C | IF row 6 and 12 are absent |
11 | >> | CONTAINS | INCLUDE | DTID 1420 “Measurements Derived From Multiple ROI Measurements” | 1-n | U |  |
12 | > | CONTAINS | CONTAINER | EV (C0034375, UMLS, "Qualitative Evaluations") | 1 | C | IF row 6 and 10 are absent |
13 | >> | CONTAINS | CODE |  | 1-n | U |  |
14 | >> | CONTAINS | TEXT |  | 1-n | U |  |

Content Item Descriptions

Rows 6, 10, 12 | The conditions require that at least one of the "heading" containers be present, though any of them may be present but empty.
Rows 7, 8, 9 | The baseline context groups defined allow for generic intensity and size measurements, regardless of the geometry of the ROI (e.g., linear distance can be measured on volumes, or volume can be estimated from a linear distance), and being baseline, do not constrain the invoker from using other appropriate concepts specific to the application.
Row 9 | A Measurement Group is used to contain one or more individual measurements that are invocations of TID 300, consistent with TIDs 1410 and 1411, which both already have Measurement Group containers as their roots.
Rows 12, 13, 14 | These Content Items allow encoding a flat list of name-value pairs that are coded questions with coded or text answers, for example, to record categorical observations related to the entire subject of the report rather than specific measurement groups.

### TID 1501 Measurement Group

This Template groups Measurements into a Measurement Group.

Each Measurement Group is identified by Tracking ID and UIDs, and may be described as having being made at a particular time point.

Measurement groups may contain various common measurement modifiers that are shared by all measurements in the group, such as method and finding site.

Table TID 1501. Parameters

Parameter Name | Parameter Usage
--- | ---
$Measurement | Coded term or Context Group for Concept Name of measurement
$Units | Units of Measurement
$ModType | Modifier Name for Concept Name of measurement
$ModValue | Modifier Value for Concept Name of measurement
$Method | Value for Measurement Method
$Derivation | Value for Measurement Derivation
$TargetSite | Value(s) for Anatomic Location of measurement
$TargetSiteMod | Modifier Value for Anatomic Location of measurement
$Equation | Coded term or Context Group for the equation or table from which the measurement was derived or computed
$ImagePurpose | Purpose of Reference for an image used as a source of the measurement
$WavePurpose | Purpose of Reference for a waveform used as a source of the measurement
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$DerivationParameter | Coded term or Context Group for Concept Name of a derivation parameter
$DerivationParameterUnits | Units of derivation parameter
$QualitativeEvaluations | Evaluations encoded with code or text responses
$FindingType | Type of the finding

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1501. Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (125007, DCM, "Measurement Group") | 1 | M |  |
1b | > | HAS OBS CONTEXT | TEXT | EV (C67447, NCIt, "Activity Session") | 1 | U |  |
2 | > | HAS OBS CONTEXT | TEXT | DT (112039, DCM, "Tracking Identifier") | 1 | M |  |
3 | > | HAS OBS CONTEXT | UIDREF | EV (112040, DCM, "Tracking Unique Identifier") | 1 | M |  |
3b | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1 | U |  | $FindingType
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 1502 “Time Point Context” | 1 | U |  |
5 | > | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | $Method
6 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | U |  | $TargetSite
7 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
8 | >> | HAS CONCEPT MOD | CODE | DT (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | $TargetSiteMod
9 | > | CONTAINS | COMPOSITE | EV (126100, DCM, "Real World Value Map used for measurement") | 1 | U |  | SOP Class UID shall be Real World Value Mapping Storage ("1.2.840.10008.5.1.4.1.1.67")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = $Measurement $Units = $Units $ModType = $ModType $ModValue = $ModValue $Method = $Method $Derivation = $Derivation $TargetSite = $TargetSite $TargetSiteMod = $TargetSiteMod $Equation = $Equation $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority $DerivationParameter = $DerivationParameter $DerivationParameterUnits = $DerivationParameterUnits
11 | > | CONTAINS | CODE | $QualitativeEvaluations | 1-n | U |  |
12 | > | CONTAINS | TEXT | $QualitativeEvaluations | 1-n | U |  |

Content Item Descriptions

Row 1b | Identifies the session during which the measurements were made. The NCI Thesaurus definition is "time, period, or term devoted to some activity".
Rows 2, 3, 10 | The included TID 300 already contains an optional inclusion of TID 1408 Tracking Identifier, which in turns allows for either or both Tracking Identifier and Tracking Unique Identifier Content Items; the intent of Rows 2 and 3 is not to send these Content Items twice, but rather to specialize their use such that their presence is mandatory, consistent with TIDs 1410 and 1411.
Row 3b | The type of the finding describes whatever entity (finding or feature) is identified by Rows 2 and 3. E.g., a finding might be a lesion, a tumor, or a reference region (as distinct from its anatomical location, which is encoded in a different content item (Finding Site).
Row 6 | Finding site may be multiple when a region of interest spans multiple anatomical locations and there is not a single pre-coordinated code describing the combination of locations. E.g., when a maligant, inflammatory or traumatic process spans actual or defined anatomical boundaries. There is no requirement that the multiple locations be contiguous.
Row 9 | The reference to an RWV in Row 9 allows measurements to be made in units that differ from the stored pixel values in the images referenced elsewhere in the Template. E.g., for a PET SUVbw measurement, the mapping from activity/concentration units in the referenced image that was used (and which may be reused for measurements in the future) may be encoded in a referenced RWV instance. This reference applies to any measurements in included Templates, unless overridden).
Rows 11, 12 | Allows encoding a flat list of name-value pairs that are coded questions with coded or text answers, for example, to record categorical observations related to the subject of the measurement group.

### TID 1502 Time Point Context

This Template describes information about the time point, for example, at which a measurement was obtained.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1502. Time Point Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS OBS CONTEXT | TEXT | EV (126070, DCM, "Subject Time Point Identifier") | 1 | U |  |
2 |  | HAS OBS CONTEXT | TEXT | EV (126071, DCM, "Protocol Time Point Identifier") | 1 | U |  |
3 |  | HAS OBS CONTEXT | TEXT | EV (C2348792, UMLS, "Time Point") | 1 | M |  |
4 |  | HAS OBS CONTEXT | CODE | EV (126072, DCM, "Time Point Type") | 1-n | U |  | BCID 6146 “Time Point Types”
5 |  | HAS OBS CONTEXT | NUM | EV (126073, DCM, "Time Point Order") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
6 |  | HAS OBS CONTEXT | NUM | EV (128740, DCM, "Longitudinal Temporal Offset from Event") | 1 | U |  | UNITS = DT (d, UCUM, "days")
7 | > | HAS CONCEPT MOD | CODE | EV (128741, DCM, "Longitudinal Temporal Event Type") | 1 | M |  | DCID 280 “Longitudinal Temporal Event Types”

Content Item Descriptions

Row 1 | Usually the same value as the Clinical Trial Time Point ID (0012,0050) attribute in the Clinical Trial Study Module, though not confined to clinical trial use. May or may not be human readable, and not required to be a DICOM UID.
Row 2 | All of the subjects within a treatment protocol that are examined at a particular scheduled time point (e.g., "baseline", "pre-treatment", "first post-treatment") will have the same Protocol Time Point Identifier, but different Subject Time Point Identifiers. However, in different protocols, the Protocol Time Point Identifiers for the same conceptual "time point" will be different. E.g., the "baseline" time point will have different Protocol Time Point Identifiers in different protocols. May or may not be human readable, and not required to be a DICOM UID.
Row 3 | Typically a short pre-defined label that has the same scope as Protocol Time Point Identifier (i.e., same conceptual time point within a treatment protocol) but is human-readable and understandable, e.g., "BASELINE" or "TP0", "TP1", etc. Usually the same value as Clinical Trial Time Point Description (0012,0051) attribute in the Clinical Trial Study Module, though not confined to clinical trial use. The Concept Name is selected as (C2348792, UMLS, "Time Point") (which is (C68568, NCIt, "Time Point"), defined as "a specific point in the time continuum, including those established relative to an event") in order to be compatible with external terminologies.
Row 4 | More than one type is permitted, e.g., a time point may be "posttreatment" as well as "unscheduled" or "nadir", etc.
Row 5 | The order is expected to be monotonically increasing within a particular scope of usage, but is not required to start at 0 or 1, nor required to increase in increments of 1 or even the same increment (e.g., to allow for retrospective insertion of unscheduled time points). In clinical usage, the Time Point Order would be expected to be temporally increasing, but in a clinical trial may be a randomized reading order rather than a temporal order.
Rows 6, 7 | Longitudinal temporal information may be inherited from Longitudinal Temporal Offset from Event (0012,0052) and Longitudinal Temporal Event Type (0012,0053) in the PS3.3 Section C.7.2.3 Clinical Trial Study Module, or may be specified or overridden within this template (e.g., if different measurements in the same SR Instance were measured on different time points).

### TID 1600 Image Library

The Image Library contains references to images and selected attributes describing them that facilitate analysis without having to retrieve the entire set of referenced images.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1600. Image Library

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111028, DCM, "Image Library") | 1 | M |  |
2 | > | CONTAINS | CONTAINER | EV (126200, DCM, "Image Library Group") | 1-n | U |  |
3 | >> | HAS ACQ CONTEXT | INCLUDE | DTID 1602 “Image Library Entry Descriptors” | 1 | U |  |
4 | >> | CONTAINS | INCLUDE | DTID 1601 “Image Library Entry” | 1-n | U |  |

### TID 1601 Image Library Entry

Each instance of the Image Library Entry Template contains the Image SOP Class and Instance UIDs, and selected attributes for an image that facilitate analysis without having to retrieve the entire set of referenced images.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1601. Image Library Entry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | IMAGE |  | 1 | M |  |
2 | > | HAS ACQ CONTEXT | INCLUDE | DTID 1602 “Image Library Entry Descriptors” | 1 | U |  |

### TID 1602 Image Library Entry Descriptors

This Template contains selected attributes for an image or group of images. The descriptive information may be copied from images or derived.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1602. Image Library Entry Descriptors

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | CODE | EV (121139, DCM, "Modality") | 1 | U |  | DCID 29 “Acquisition Modality”
2 |  | HAS ACQ CONTEXT | CODE | EV (123014, DCM, "Target Region") | 1 | U |  | DCID 4031 “Common Anatomic Regions”
3 |  | HAS ACQ CONTEXT | CODE | EV (111027, DCM, "Image Laterality") | 1 | U |  | DCID 244 “Laterality”
4 |  | HAS ACQ CONTEXT | DATE | EV (111060, DCM, "Study Date") | 1 | U |  |
5 |  | HAS ACQ CONTEXT | TIME | EV (111061, DCM, "Study Time") | 1 | U |  |
6 |  | HAS ACQ CONTEXT | DATE | EV (111018, DCM, "Content Date") | 1 | U |  |
7 |  | HAS ACQ CONTEXT | TIME | EV (111019, DCM, "Content Time") | 1 | U |  |
8 |  | HAS ACQ CONTEXT | DATE | EV (126201, DCM, "Acquisition Date") | 1 | U |  |
9 |  | HAS ACQ CONTEXT | TIME | EV (126202, DCM, "Acquisition Time") | 1 | U |  |
10 |  | HAS ACQ CONTEXT | UIDREF | EV (112227, DCM, "Frame of Reference UID") | 1 | U |  |
11 |  | HAS ACQ CONTEXT | NUM | EV (110910, DCM, "Pixel Data Rows") | 1 | U |  | UNITS = EV ({pixels}, UCUM, "pixels")
12 |  | HAS ACQ CONTEXT | NUM | EV (110911, DCM, "Pixel Data Columns") | 1 | U |  | UNITS = EV ({pixels}, UCUM, "pixels")
13 |  | HAS ACQ CONTEXT | INCLUDE | DTID 1603 “Image Library Entry Descriptors for Projection Radiography” | 1 | U |  |
14 |  | HAS ACQ CONTEXT | INCLUDE | DTID 1604 “Image Library Entry Descriptors for Cross-Sectional Modalities” | 1 | U |  |
15 |  | HAS ACQ CONTEXT | INCLUDE | DTID 1605 “Image Library Entry Descriptors for CT” | 1 | U |  |
16 |  | HAS ACQ CONTEXT | INCLUDE | DTID 1606 “Image Library Entry Descriptors for MR” | 1 | U |  |
17 |  | HAS ACQ CONTEXT | INCLUDE | DTID 1607 “Image Library Entry Descriptors for PET” | 1 | U |  |

Content Item Descriptions

Target Region | The value of Anatomic Region Sequence (0008,2218) in the Image IOD, or a code derived from Body Part Examined (0018,0015) using the mapping described in Annex L.

### TID 1603 Image Library Entry Descriptors for Projection Radiography

This Template contains selected attributes for a projection radiography image or group of such images. The descriptive information may be copied from images or derived.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1603. Image Library Entry Descriptors for Projection Radiography

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | CODE | EV (111031, DCM, "Image View") | 1 | U |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111032, DCM, "Image View Modifier") | 1-n | U |  |
3 |  | HAS ACQ CONTEXT | TEXT | EV (111044, DCM, "Patient Orientation Row") | 1 | U |  |
4 |  | HAS ACQ CONTEXT | TEXT | EV (111043, DCM, "Patient Orientation Column") | 1 | U |  |
5 |  | HAS ACQ CONTEXT | NUM | EV (111026, DCM, "Horizontal Pixel Spacing") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
6 |  | HAS ACQ CONTEXT | NUM | EV (111066, DCM, "Vertical Pixel Spacing") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
7 |  | HAS ACQ CONTEXT | NUM | EV (112011, DCM, "Positioner Primary Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
8 |  | HAS ACQ CONTEXT | NUM | EV (112012, DCM, "Positioner Secondary Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")

Content Item Descriptions

Patient Orientation Row | First (row) and second (column) components of Patient Orientation (0020,0020) in the Image IOD. See Section C.7.6.1.1.1 “Patient Orientation” in PS3.3 .
Patient Orientation Column
Horizontal Imager Pixel Spacing | The second component of Imager Pixel Spacing (0018,1164) in the Image IOD. See Section C.8.11.4 “DX Detector Module” in PS3.3 .
Vertical Imager Pixel Spacing | The first component of Imager Pixel Spacing (0018,1164) in the Image IOD. See Section C.8.11.4 “DX Detector Module” in PS3.3 .

### TID 1604 Image Library Entry Descriptors for Cross-Sectional Modalities

This Template contains selected attributes for a cross-sectional image or group of such images. The descriptive information may be copied from images or derived.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1604. Image Library Entry Descriptors for Cross-Sectional Modalities

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | NUM | EV (111026, DCM, "Horizontal Pixel Spacing") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
2 |  | HAS ACQ CONTEXT | NUM | EV (111066, DCM, "Vertical Pixel Spacing") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
3 |  | HAS ACQ CONTEXT | NUM | EV (112226, DCM, "Spacing between slices") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
4 |  | HAS ACQ CONTEXT | NUM | EV (112225, DCM, "Slice Thickness") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
5 |  | HAS ACQ CONTEXT | NUM | EV (110901, DCM, "Image Position (Patient) X") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
6 |  | HAS ACQ CONTEXT | NUM | EV (110902, DCM, "Image Position (Patient) Y") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
7 |  | HAS ACQ CONTEXT | NUM | EV (110903, DCM, "Image Position (Patient) Z") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
8 |  | HAS ACQ CONTEXT | NUM | EV (110904, DCM, "Image Orientation (Patient) Row X") | 1 | U |  | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
9 |  | HAS ACQ CONTEXT | NUM | EV (110905, DCM, "Image Orientation (Patient) Row Y") | 1 | U |  | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
10 |  | HAS ACQ CONTEXT | NUM | EV (110906, DCM, "Image Orientation (Patient) Row Z") | 1 | U |  | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
11 |  | HAS ACQ CONTEXT | NUM | EV (110907, DCM, "Image Orientation (Patient) Column X") | 1 | U |  | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
12 |  | HAS ACQ CONTEXT | NUM | EV (110908, DCM, "Image Orientation (Patient) Column Y") | 1 | U |  | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
13 |  | HAS ACQ CONTEXT | NUM | EV (110909, DCM, "Image Orientation (Patient) Column Z") | 1 | U |  | UNITS = EV ({-1:1}, UCUM, "{-1:1}")

Content Item Descriptions

Horizontal Imager Pixel Spacing | The second component of Pixel Spacing (0028,0030) in the Image IOD. See Section 10.7.1.1 “Pixel Spacing” in PS3.3 and Section C.7.6.2 “Image Plane Module” in PS3.3 .
Vertical Imager Pixel Spacing | The first component of Pixel Spacing (0028,0030) in the Image IOD. See Section 10.7.1.1 “Pixel Spacing” in PS3.3 and Section C.7.6.2 “Image Plane Module” in PS3.3 .

### TID 1605 Image Library Entry Descriptors for CT

This Template contains selected attributes for a CT image or group of such images. The descriptive information may be copied from images or derived.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1605. Image Library Entry Descriptors for CT

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | CODE | EV (113820, DCM, "CT Acquisition Type") | 1 | U |  | DCID 10013 “CT Acquisition Type”
2 |  | HAS ACQ CONTEXT | CODE | EV (113961, DCM, "Reconstruction Algorithm") | 1 | U |  | DCID 10033 “CT Reconstruction Algorithm”

Content Item Descriptions

CT Acquisition Type | A code derived from the value of Acquisition Type (0018,9302) in the Image IOD. See Section C.8.15.3.2 “CT Acquisition Type Macro” in PS3.3 .
Reconstruction Algorithm | A code derived from the value of Reconstruction Algorithm (0018,9315) in the Image IOD. See Section C.8.15.3.7 “CT Reconstruction Macro” in PS3.3 .

### TID 1606 Image Library Entry Descriptors for MR

This Template contains selected attributes for a MR image or group of such images. The descriptive information may be copied from images or derived.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1606. Image Library Entry Descriptors for MR

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | TEXT | EV (128230, DCM, "Pulse Sequence Name") | 1 | U |  |

Content Item Descriptions

Pulse Sequence Name | The value of Pulse Sequence Name (0018,9005) or Sequence Name (0018,0024) in the Image IOD. See Section C.8.13.4 “MR Pulse Sequence Module” in PS3.3 .

### TID 1607 Image Library Entry Descriptors for PET

This Template contains selected attributes for a PET image or group of such images. The descriptive information may be copied from images or derived.

### Note

The content of this Template is similar to that in TID 15101 NM/PET Protocol Context, but is in the form of an SR Template rather than a Protocol Context Template, and the content items are not nested as modifiers. There is also some similarity to TID 3307 NM/PET Perfusion Measurement Group.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1607. Image Library Entry Descriptors for PET

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | CODE | EV (C-10072, SRT, "Radionuclide") | 1 | U |  | DCID 4020 “PET Radionuclide”
2 |  | HAS ACQ CONTEXT | CODE | EV (F-61FDB, SRT, "Radiopharmaceutical agent") | 1 | U |  | DCID 4021 “PET Radiopharmaceutical”
3 |  | HAS ACQ CONTEXT | NUM | EV (R-42806, SRT, "Half-life of radiopharmaceutical") | 1 | U |  | UNITS = EV (s, UCUM, "s")
4 |  | HAS ACQ CONTEXT | DATETIME | EV (123003, DCM, "Radiopharmaceutical Start DateTime") | 1 | U |  |
5 |  | HAS ACQ CONTEXT | DATETIME | EV (123004, DCM, "Radiopharmaceutical Stop DateTime") | 1 | U |  |
6 |  | HAS ACQ CONTEXT | NUM | EV (123005, DCM, "Radiopharmaceutical Volume") | 1 | U |  | UNITS = DT (cm3, UCUM, "cm3")
7 |  | HAS ACQ CONTEXT | NUM | EV (123006, DCM, "Radionuclide Total Dose") | 1 | U |  | UNITS = DT (Bq, UCUM, "Bq")
8 |  | HAS ACQ CONTEXT | NUM | EV (123007, DCM, "Radiopharmaceutical Specific Activity") | 1 | U |  | UNITS = DT (Bq/mol, UCUM, "Bq/mol")
9 |  | HAS ACQ CONTEXT | CODE | EV (G-C340, SRT, "Route of Administration") | 1 | U |  | BCID 11 “Route of Administration”
10 |  | HAS ACQ CONTEXT | NUM | EV (123009, DCM, "Radionuclide Syringe Counts") | 1 | U |  | UNITS = DT ({counts}/s, UCUM "counts/s")
11 |  | HAS ACQ CONTEXT | NUM | EV (123010, DCM, "Radionuclide Residual Syringe Counts") | 1 | U |  | UNITS = DT ({counts}/s, UCUM "counts/s")
12 |  | HAS ACQ CONTEXT | NUM | EV (126203, DCM, "PET Radionuclide Incubation Time") | 1 | U |  | UNITS = EV (min, UCUM, "min")
13 |  | HAS ACQ CONTEXT | NUM | EV (14749-6, LN, "Glucose") | 1 | U |  | UNITS = EV (mmol/l, UCUM, "mmol/l")
14 |  | HAS ACQ CONTEXT | DATE | EV (127857, DCM, "Glucose Measurement Date") | 1 | MC | IF Row 13 Glucose is present and does not contain Observation DateTime (0040,A032). |
15 |  | HAS ACQ CONTEXT | TIME | EV (127858, DCM, "Glucose Measurement Time") | 1 | MC | IF Row 13 Glucose is present and does not contain Observation DateTime (0040,A032). |

Content Item Descriptions

Row 3 | Half-life of radiopharmaceutical | The units for half life are chosen to be seconds, to match the units used for Radionuclide Half Life (0018,1075). See Section C.8.9.2 “PET Isotope Module” in PS3.3 .
Row 14 | Glucose Measurement Date | In an earlier edition of the standard, an incorrect DCM code was used for this concept, which was already assigned as (109081, DCM, "Prospective gating").
Row 15 | Glucose Measurement Time | In an earlier edition of the standard, an incorrect DCM code was used for this concept, which was already assigned as (109082, DCM, "Retrospective gating").

### TID 2000 Basic Diagnostic Imaging Report

Basic report Template for general diagnostic imaging interpretation reports.

Can only be instantiated at the root node and cannot be included in other Templates.

Type:

  Non-Extensible

Order:

  Significant

Root:

  Yes

Table TID 2000. Basic Diagnostic Imaging Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | BCID 7000 “Diagnostic Imaging Report Document Titles” | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1-n | U |  |
3 | > | HAS CONCEPT MOD | CODE | EV (122142, DCM, "Acquisition Device Type") | 1-n | U |  | DCID 29 “Acquisition Modality”
4 | > | HAS CONCEPT MOD | CODE | EV (123014, DCM, "Target Region") | 1-n | U |  |
5 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
6 | > | HAS CONCEPT MOD | INCLUDE | DTID 1210 “Equivalent Meaning(s) of Concept Name” | 1-n | U |  |
7 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
8 | > | CONTAINS | CONTAINER | BCID 7001 “Diagnostic Imaging Report Headings” | 1-n | U |  |
9 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
10 | >> |  | INCLUDE | DTID 2002 “Report Narrative” | 1 | M |  |

No Content Items other than those defined in Observation Context TID 1001 “Observation Context” may be the target of a HAS OBS CONTEXT relationship when TID 2000 “Basic Diagnostic Imaging Report” is invoked.

Content Item Descriptions

Rows 2, 3, 4 | The content of rows 2, 3, and 4 shall not be inconsistent with the meaning of the report title of row 1. If the report title does not include the concepts of the procedure type, modality, or target site (e.g., the generic "Diagnostic Imaging Report"), these rows may provide post-coordination of those concepts. If the report title does include such concepts (e.g., "CT Head Report"), they may be encoded duplicatively to support report categorization and search.

### TID 2001 Basic Diagnostic Imaging Report Observations

Individual numeric or image observations that may be useful for inclusion as individual findings or as the source of inferences in a report.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 2001. Basic Diagnostic Imaging Report Observations

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | IMAGE | BCID 7003 “Diagnostic Imaging Report Purposes of Reference” | 1 | MC | XOR Rows 2, 3, 4, 5. |
2 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1 | MC | XOR Rows 1, 3, 4, 5. Shall not be present if the NUM value type is not supported by the IOD. |
3 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1 | MC | XOR Rows 1, 2, 4, 5. Shall not be present if the NUM value type is not supported by the IOD. |
4 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1 | MC | XOR Rows 1, 2, 3, 5. Shall not be present if the NUM value type is not supported by the IOD. |
5 |  |  | INCLUDE | DTID 1404 “Numeric Measurement” | 1 | MC | XOR Rows 1, 2, 3, 4. Shall not be present if the NUM value type is not supported by the IOD. |

### TID 2002 Report Narrative

The Report Narrative allows recording of text, code, and numeric observations. The order of Content Items in the Template is not significant; the order of Content Items in a SOP Instance may be significant to the narrative flow of the report.

Type:

  Non-Extensible

Order:

  Non-Significant

Root:

  No

Table TID 2002. Report Narrative

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CODE | BCID 7002 “Diagnostic Imaging Report Elements” | 1-n | U |  |
2 | > | INFERRED FROM | INCLUDE | DTID 2001 “Basic Diagnostic Imaging Report Observations” | 1-n | U |  |
3 |  | CONTAINS | TEXT | BCID 7002 “Diagnostic Imaging Report Elements” | 1-n | U |  |
4 | > | INFERRED FROM | INCLUDE | DTID 2001 “Basic Diagnostic Imaging Report Observations” | 1-n | U |  |
5 |  | CONTAINS | INCLUDE | DTID 2001 “Basic Diagnostic Imaging Report Observations” | 1-n | U |  |

### TID 2005 Transcribed Diagnostic Imaging Report

Basic report Template for general diagnostic imaging interpretation reports produced in a dictation/transcription workflow. SR documents encoded using this Template are intended to be transformable to HL7 Clinical Document Architecture format (see Section X.3 “Transcribed Diagnostic Imaging CDA Instance Content” in PS3.17 and Annexes in PS3.20).

This Template can be instantiated only at the root node, and cannot be included in other Templates.

Observation Context shall be inherited from outside the SR Content tree, and shall not be changed within the Content tree. To satisfy the requirement that Observer Context is inherited, either or both the Author Observer Sequence (0040,A078) or the Verifying Observer Sequence (0040,A073) from the SR Document Module must be present in the SOP Instance.

### Note

See Section C.17.5 “Observation Context Encoding” in PS3.3 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  Yes

Table TID 2005. Transcribed Diagnostic Imaging Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | BCID 7000 “Diagnostic Imaging Report Document Titles” | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1-n | U |  |
3 | > | HAS CONCEPT MOD | CODE | EV (122142, DCM, "Acquisition Device Type") | 1-n | U |  | DCID 29 “Acquisition Modality”
4 | > | HAS CONCEPT MOD | CODE | EV (123014, DCM, "Target Region") | 1-n | U |  |
5 | > | HAS CONCEPT MOD | CODE | EV (121049, DCM, "Language of Content Item and Descendants") | 1 | M |  | DCID 5000 “Languages”
6 | > | CONTAINS | CONTAINER | BCID 7001 “Diagnostic Imaging Report Headings” | 1-n | M |  |
7 | >> | CONTAINS | TEXT | BCID 7002 “Diagnostic Imaging Report Elements” | 1 | U |  |
8 | > | CONTAINS | CONTAINER | EV (55113-5, LN, "Key Images") | 1-n | U |  |
9 | >> | CONTAINS | TEXT | EV (113012, DCM, "Key Object Description") | 1 | U |  |
10 | >> | CONTAINS | IMAGE | Purpose of Reference is not used | 1-n | M |  |

Content Item Descriptions

Rows 2, 3, 4 | The content of rows 2, 3, and 4 shall not be inconsistent with the meaning of the report title of row 1. If the report title does not include the concepts of the procedure type, modality, or target site (e.g., the generic "Diagnostic Imaging Report"), these rows may provide post-coordination of those concepts. If the report title does include such concepts (e.g., "CT Head Report"), they may be encoded duplicatively to support report categorization and search.
Row 6 | CONTAINER Concept Name may be absent.
Row 10 | IMAGE Concept Name shall be absent

### TID 2006 Imaging Report With Conditional Radiation Exposure and Protection Information

This Template is used for general imaging reports for both radiation producing and non-radiation producing modalities.

For radiation producing modalities, radiation exposure and protection information is required, such as to support nationally-specific legal or standard requirements.

It contains mandatory sections, each of which may appear only once in objects instantiated from the Template, including the medical content of the report that comprises relevant medical history data, information on the current request (i.e., clinical question that is expected to be answered by the requested procedure), impressions on the current imaging procedure that has been performed, and radiation exposure and protection information.

This Template is a specialization of TID 2000 “Basic Diagnostic Imaging Report”, in that it uses the same structure of headings and content, but mandates the presence and order of specific headings, and extends the subordinate content with specific Content Items.

Type:

  Non-Extensible

Order:

  Non-Significant

Root:

  Yes

Table TID 2006. Imaging Report With Conditional Radiation Exposure and Protection Information

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | BCID 7000 “Diagnostic Imaging Report Document Titles” | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1-n | U |  |
3 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
4 | > | HAS CONCEPT MOD | INCLUDE | DTID 1210 “Equivalent Meaning(s) of Concept Name” | 1-n | U |  |
5 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
6 | > | CONTAINS | CONTAINER | EV (55111-9, LN, "Current Procedure Descriptions") | 1 | M |  |
7 | >> |  | INCLUDE | DTID 2007 “Imaging Procedure Description” | 1 | M |  |
8 | > | CONTAINS | CONTAINER | EV (55114-3, LN, "Prior Procedure Descriptions") | 1-n | MC | IF relevant prior procedures have been performed. |
9 | >> |  | INCLUDE | DTID 2007 “Imaging Procedure Description” | 1 | M |  |
10 | > | CONTAINS | CONTAINER | EV (11329-0, LN, "History") | 1 | M |  |
11 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
12 | >> |  | INCLUDE | DTID 2002 “Report Narrative” | 1 | M |  |
13 | > | CONTAINS | CONTAINER | EV (55115-0, LN, "Request") | 1 | M |  |
14 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
15 | >> |  | INCLUDE | DTID 2002 “Report Narrative” | 1 | M |  |
16 | > | CONTAINS | CONTAINER | EV (19005-8, LN, "Impressions") | 1 | M |  |
17 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
18 | >> |  | INCLUDE | DTID 2002 “Report Narrative” | 1 | M |  |
19 | > | CONTAINS | INCLUDE | DTID 2008 “Radiation Exposure and Protection Information” | 1 | MC | IF the current procedure exposes the patient to ionizing radiation. |
20 | > | CONTAINS | CONTAINER | BCID 7001 “Diagnostic Imaging Report Headings” | 1-n | U |  |
21 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
22 | >> |  | INCLUDE | DTID 2002 “Report Narrative” | 1 | M |  |

No Content Items other than those defined in TID 1001 “Observation Context” may be the target of a HAS OBS CONTEXT relationship when TID 2006 “Imaging Report With Conditional Radiation Exposure and Protection Information” is invoked.

Content Item Descriptions

Row 2 | Even though this information is related to the content of Row 6 in TID 2007 “Imaging Procedure Description”, it is present here for consistency with other report Templates.
Rows 5, 6 | Information on at least one of the following person observers is mandatory: 1) "Performing Physician" 2) "Performing Technologist". (For those person observers, requirement types as specified in TID 1003 “Person Observer Identifying Attributes” apply. That means that "Person Observer Name" is the only mandatory attribute).
Row 20 | Each heading (concept code from CID 7001 “Diagnostic Imaging Report Headings”) may appear only once, and may not repeat the headings (concept codes) used when instantiating any other rows of this Template.

### TID 2007 Imaging Procedure Description

Contains information related to the procedure.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 2007. Imaging Procedure Description

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
2 |  | CONTAINS | TEXT | EV (123014, DCM, "Target Region") | 1 | MC | XOR with Row 3 |
3 |  | CONTAINS | CODE | EV (123014, DCM, "Target Region") | 1 | MC | XOR with Row 2 | DCID 4028 “Craniofacial Anatomic Regions” , DCID 4030 “CT, MR and PET Anatomy Imaged” , DCID 4031 “Common Anatomic Regions”
4 | > | HAS CONCEPT MOD | CODE | EV (122142, DCM, "Acquisition Device Type") | 1-n | U |  | DCID 29 “Acquisition Modality”
5 |  | CONTAINS | TEXT | EV (121065, DCM, "Procedure Description") | 1 | M |  |
6 |  | CONTAINS | DATE | EV (111060, DCM, "Study Date") | 1 | M |  | Shall be equal to the Study Date (0020,0020) in the General Study Module in the images to which this report applies.
7 |  | CONTAINS | TIME | EV (111061, DCM, "Study Time") | 1 | U |  | If present, shall be equal to the Study Time (0020,0030) in the General Study Module in the images to which this report applies.
8 |  | CONTAINS | COMPOSITE | EV (113701, DCM, "X-Ray Radiation Dose Report") | 1-n | U |  |

### TID 2008 Radiation Exposure and Protection Information

Contains information related to the radiation exposure and protection of the patient, as is required by national legal requirements or standards.

Other information about the current procedure is described in TID 2006 “Imaging Report With Conditional Radiation Exposure and Protection Information” and not repeated here.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 2008. Radiation Exposure and Protection Information

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (73569-6, LN, "Radiation Exposure and Protection Information") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
3 | > | CONTAINS | CODE | EV (364320009, SCT, "Pregnancy observable") | 1 | MC | IF female patient of child-bearing age | DCID 6096 “Pregnancy Status”
4 | > | CONTAINS | TEXT | EV (18785-6, LN, "Indications for Procedure") | 1 | M |  |
5 | > | CONTAINS | PNAME | EV (113850, DCM, "Irradiation Authorizing ") | 1 | M |  |
6 | > | CONTAINS | TEXT | EV (113921, DCM, "Radiation Exposure") | 1 | MC | IFF ionizing radiation is applied in the context of the current procedure |
7 | > | CONTAINS | TEXT | EV (440252007, SCT, "Administration of radiopharmaceutical") | 1 | MC | IFF radioactive substance is administered in the context of the current procedure |

Content Item Descriptions

Row 5 | The clinician responsible for determining that the irradiating procedure was appropriate for the indications.
Row 6 | A textual, human-readable description of the radiation exposure is all that is required by this Template (such as is sufficient to comply with, for example, German law). Detailed specification of exposure is out of the scope of this Template. Such information may be given in a separate SR instances such as described in TID 10001 “Projection X-Ray Radiation Dose” or TID 10011 “CT Radiation Dose”, and referenced from TID 2007 “Imaging Procedure Description”.

### TID 2010 Key Object Selection

The Key Object Selection Template is intended for flagging one or more significant images, waveforms, or other composite SOP Instances. Key Object Selection contains:

- coded document title stating the reason for significance of the referenced objects in the Key Object Selection,

- optional free form text comment in an explicitly identified language, and

- optional identification of the observer (device or person) that created the Key Object Selection.

### Note

1. For instance, when this Template is used to identify images rejected for quality reasons, the device or person performing the quality assessment is identified in observation context items (invoked through TID 1002 “Observer Context”). The reason for rejection can be included both as a code used as a concept modifier for the document title, and as text description.

2. The order of object references may be significant, e.g., when the title concept is "For Conference".

3. Instances referenced in a Key Object Selection Document may be securely referenced by Digital Signature or MAC mechanisms within the SR Document General Module (see PS3.3).

The Template can only be instantiated at the root node and cannot be included in other Templates. The Template is not extensible; that is, no other Content Items may be added to this Template, or the Templates that are included, recursively.

Type:

  Non-Extensible

Order:

  Non-Significant

Root:

  Yes

Table TID 2010. Key Object Selection

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DCID 7010 “Key Object Selection Document Title” | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (113011, DCM, "Document Title Modifier") | 1-n | U |  |
3 | > | HAS CONCEPT MOD | CODE | EV (113011, DCM, "Document Title Modifier") | 1 | UC | IF Row 1 Concept Name = (113001, DCM, "Rejected for Quality Reasons") or (113010, DCM, "Quality Issue") | DCID 7011 “Rejected for Quality Reasons”
4 | > | HAS CONCEPT MOD | CODE | EV (113011, DCM, "Document Title Modifier") | 1 | MC | IF Row 1 Concept Name = (113013, DCM, "Best In Set") | DCID 7012 “Best in Set”
5 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
6 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
7 | > | CONTAINS | TEXT | EV (113012, DCM, "Key Object Description") | 1 | U |  |
8 | > | CONTAINS | IMAGE | Purpose of Reference shall not be present | 1-n | MC | At least one of Rows 8, 9 and 10 shall be present |
9 | > | CONTAINS | WAVEFORM | Purpose of Reference shall not be present | 1-n | MC | At least one of Rows 8, 9 and 10 shall be present |
10 | > | CONTAINS | COMPOSITE | Purpose of Reference shall not be present | 1-n | MC | At least one of Rows 8, 9 and 10 shall be present |

### TID 2020 Spectacle Prescription Report

The Spectacle Prescription Report is a structured report used to represent the prescription for a patient. Usually a prescription is for both eyes, but sometimes just one. The Spectacle Prescription Report defines a refractive correction relative to which visual acuity may be measured subjectively, and thus may be referenced by a Visual Acuity Measurements Storage SOP Instance.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 2020. Spectacle Prescription Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111671, DCM, "Spectacle Prescription Report") | 1 | M |  | Root node
2 | > | CONTAINS | CONTAINER | EV (111688, DCM, "Right Eye Rx") | 1 | UC | IF Right Spectacle Lens is prescribed |
3 | >> | CONTAINS | INCLUDE | DTID 2021 “Template for Spectacle Prescription Details” | 1 | M |  |
4 | > | CONTAINS | CONTAINER | EV (111689, DCM, "Left Eye Rx") | 1 | UC | IF Left Spectacle Lens is prescribed |
5 | >> | CONTAINS | INCLUDE | DTID 2021 “Template for Spectacle Prescription Details” | 1 | M |  |
6 | > | CONTAINS | NUM | EV (111679, DCM, "Distance Pupillary Distance") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
7 | > | CONTAINS | NUM | EV (111680, DCM, "Near Pupillary Distance") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
8 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

### TID 2021 Template for Spectacle Prescription Details

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 2021. Spectacle Prescription Details

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | NUM | EV (F-02FB4, SRT, "Sphere") | 1 | M |  | UNITS = EV ([diop], UCUM, "diopters")
2 |  | CONTAINS | NUM | EV (F-A2143, SRT, "Cylinder Power") | 1 | UC | IF Cylinder is prescribed | UNITS = EV ([diop], UCUM, "diopters")
3 |  | CONTAINS | NUM | EV (F-02FB7, SRT, "Axis") | 1 | MC | IF ROW 2 is Present | UNITS = EV (deg, UCUM, "degrees")
4 |  | CONTAINS | NUM | EV (111672, DCM, "Add Near") | 1 | UC | IF Add Near is prescribed | UNITS = EV ([diop], UCUM, "diopters")
5 |  | CONTAINS | NUM | EV (111673, DCM, "Add Intermediate") | 1 | UC | IF Add Intermediate is prescribed | UNITS = EV ([diop], UCUM, "diopters")
6 |  | CONTAINS | NUM | EV (111674, DCM, "Add Other") | 1 | UC | IF Add Other is prescribed | UNITS = EV ([diop], UCUM, "diopters")
7 |  | CONTAINS | NUM | EV (111675, DCM, "Horizontal Prism Power") | 1 | UC | IF Horizontal Prism is prescribed | UNITS = EV ([p'diop], UCUM, "prism diopters")
8 |  | CONTAINS | CODE | EV (111676, DCM, "Horizontal Prism Base") | 1 | MC | IF ROW 7 is present | DCID 4214 “Ophthalmic Horizontal Directions”
9 |  | CONTAINS | NUM | EV (111677, DCM, "Vertical Prism Power") | 1 | UC | IF Vertical Prism is prescribed | UNITS = EV ([p'diop], UCUM, "prism diopters")
10 |  | CONTAINS | CODE | EV (111678, DCM, "Vertical Prism Base") | 1 | MC | IF Row 9 is present | DCID 4215 “Ophthalmic Vertical Directions”

### TID 2100 Macular Grid Thickness and Volume Report

The Macular Grid Thickness and Volume Report is a structured report encoding the macular grid thickness and volume values derived from ophthalmic images, such as ophthalmic OPT images. This may encode measurements of either or both eyes.

The macular grid conveyed by this report is based upon the grid employed by the Early Treatment of Diabetic Retinopathy Study (ETDRS) to measure area and proximity of macular edema to the anatomic center (fovea) of the macula. See ETDRS Report Number 10.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 2100. Macular Grid Thickness and Volume Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111690, DCM, "Macular Grid Thickness and Volume Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 2101 “Macular Grid Thickness and Volume Measurement” | 1 | MC | IF Row 5 is absent. | $Laterality = EV (G-A100, SRT, "Right")
5 | > | CONTAINS | INCLUDE | DTID 2101 “Macular Grid Thickness and Volume Measurement” | 1 | MC | IF Row 4 is absent. | $Laterality = EV (G-A101, SRT, "Left")

### TID 2101 Macular Grid Thickness and Volume Measurement

This Template encodes the macular grid thickness and volume measurements for a single eye.

Table TID 2101. Parameters

Parameter Name | Parameter Usage
--- | ---
$Laterality | Which eye

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 2101. Macular Grid Thickness and Volume Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | EV (T-AA000, SRT, "Eye")
3 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | $Laterality
4 | > | CONTAINS | NUM | EV (57108-3, LN, "Macular Grid.Center Point Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
5 | > | CONTAINS | NUM | EV (57109-1, LN, "Macular Grid.Center Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
6 | > | CONTAINS | NUM | EV (57110-9, LN, "Macular Grid.Inner Superior Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
7 | > | CONTAINS | NUM | EV (57111-7, LN, "Macular Grid.Inner Nasal Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
8 | > | CONTAINS | NUM | EV (57112-5, LN, "Macular Grid.Inner Inferior Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
9 | > | CONTAINS | NUM | EV (57113-3, LN, "Macular Grid.Inner Temporal Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
10 | > | CONTAINS | NUM | EV (57114-1, LN, "Macular Grid.Outer Superior Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
11 | > | CONTAINS | NUM | EV (57115-8, LN, "Macular Grid.Outer Nasal Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
12 | > | CONTAINS | NUM | EV (57116-6, LN, "Macular Grid.Outer Inferior Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
13 | > | CONTAINS | NUM | EV (57117-4, LN, "Macular Grid.Outer Temporal Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
14 | > | CONTAINS | NUM | EV (57118-2, LN, "Macular Grid.Total Volume") | 1 | M |  | UNITS = EV (mm3, UCUM, "mm3")
15 | > | CONTAINS | NUM | EV (111691, DCM, "Number of Images Used for Macular Measurements") | 1 | M |  | UNITS = EV ({images}, UCUM, "images")
16 | > | CONTAINS | NUM | EV (111692, DCM, "Number of Samples Used per Image") | 1 | M |  | UNITS = EV ({samples}, UCUM, "samples")
17 | > | CONTAINS | NUM | EV (111693, DCM, "Analysis Quality Rating") | 1 | M |  | UNITS = EV ({0:100}, UCUM, "range:0:100") Value = 0 - 100
18 | >> | HAS OBS CONTEXT | INCLUDE | DTID 2102 “Quality Rating Identification” | 1 | M |  |
19 | > | CONTAINS | NUM | EV (111694, DCM, "Image Set Quality Rating") | 1 | M |  | UNITS = EV ({0:100}, UCUM, "range:0:100") Value = 0 - 100
20 | >> | HAS OBS CONTEXT | INCLUDE | DTID 2102 “Quality Rating Identification” | 1 | M |  |
21 | > | CONTAINS | NUM | EV (111029, DCM, "Image Quality Rating") | 1-n | U |  | UNITS = EV ({0:100}, UCUM, "range:0:100") Value = 0 - 100
22 | >> | INFERRED FROM | IMAGE | No purpose of reference | 1 | M |  |
23 | >> | HAS OBS CONTEXT | INCLUDE | DTID 2102 “Quality Rating Identification” | 1 | M |  |
24 | > | CONTAINS | CODE | EV (111696, DCM, "Visual Fixation Quality During Acquisition") | 1 | U |  | DCID 4220 “Visual Fixation Quality During Acquisition”
25 | >> | HAS CONCEPT MOD | CODE | EV (111697, DCM, "Visual Fixation Quality Problem") | 1-n | U |  | DCID 4221 “Visual Fixation Quality Problem”
26 | > | CONTAINS | CODE | EV (111698, DCM, "Ophthalmic Macular Grid Problem") | 1-n | U |  | DCID 4222 “Ophthalmic Macular Grid Problem”
27 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

### TID 2102 Quality Rating Identification

This Template specifies the algorithm (and parameters) used to create a quality rating for an image or image set.

It is expected that the identified algorithm will create a consistent quality rating when analyzing a given image. If the algorithm allows change to its parameters that would alter the quality rating created, the specific parameters used should be specified.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 2102. Quality Rating Identification

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (111001, DCM, "Algorithm Name") | 1 | M |  |
2 |  |  | TEXT | EV (111003, DCM, "Algorithm Version") | 1 | M |  |
3 |  |  | TEXT | EV (122405, DCM, "Algorithm Manufacturer") | 1 | M |  |
4 |  |  | TEXT | EV (111002, DCM, "Algorithm Parameters") | 1-n | U |  |

## Procedure Log IOD Templates

### TID 3001 Procedure Log

The Procedure Log Template is intended for the representation of reports or logs of time-stamped events occurring during an image-guided interventional or other procedure.

This Template does not require a particular ordering of the subsidiary Content Items.

### Note

1. The Procedure Log IOD (PS3.3) requires ordering by Observation DateTime; thus log entries of different types (i.e., specified by different Rows in the Template) may appear in any order.

2. While this Template is extensible, the Procedure Log IOD forbids Container Content Items subsidiary to the top level Container.

Type:

  Extensible

Order:

  Non-Significant

Root:

  Yes

Table TID 3001. Procedure Log

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DCID 3400 “Procedure Log Titles” | 1 | M |  | Root node
2 | > |  | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
3 | > |  | INCLUDE | DTID 3601 “Procedure Context” | 1 | M |  |
4 | > | HAS ACQ CONTEXT | TEXT | EV (121121, DCM, "Room identification") | 1 | U |  |
5 | > | HAS ACQ CONTEXT | TEXT | EV (121122, DCM, "Equipment identification") | 1-n | U |  |
6 | > | CONTAINS | TEXT | DCID 3401 “Types of Log Notes” | 1-n | U |  |
7 | >> |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
8 | > | CONTAINS | CODE | EV (121123, DCM, "Patient Status or Event") | 1-n | U |  | DCID 3402 “Patient Status and Events”
9 | >> |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
10 | > | CONTAINS | PNAME | DCID 3404 “Staff Actions” | 1-n | U |  |
11 | >> |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
12 | > | CONTAINS | TEXT | DCID 3427 “Equipment Events” | 1-n | U |  | Equipment identifier
13 | >> |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
14 | > | CONTAINS | INCLUDE | DTID 3100 “Procedure Action” | 1-n | U |  |
15 | > | CONTAINS | INCLUDE | DTID 3101 “Image Acquisition” | 1-n | U |  |
16 | > | CONTAINS | INCLUDE | DTID 3102 “Waveform Acquisition” | 1-n | U |  |
17 | > | CONTAINS | INCLUDE | DTID 3103 “Referenced Object” | 1-n | U |  |
18 | > | CONTAINS | INCLUDE | DTID 3104 “Consumables” | 1-n | U |  |
19 | > | CONTAINS | INCLUDE | DTID 3105 “Lesion Identification and Properties” | 1-n | U |  |
20 | > | CONTAINS | INCLUDE | DTID 3106 “Drugs/Contrast Administered” | 1-n | U |  |
21 | > | CONTAINS | INCLUDE | DTID 3107 “Device Used” | 1-n | U |  |
22 | > | CONTAINS | INCLUDE | DTID 3108 “Intervention” | 1-n | U |  |
23 | > | CONTAINS | CODE | EV (DD-60002, SRT, "Complication of Procedure") | 1-n | U |  | DCID 3413 “Adverse Outcomes”
24 | >> |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
25 | > | CONTAINS | INCLUDE | DTID 3109 “Measurements” | 1-n | U |  |
26 | > | CONTAINS | INCLUDE | DTID 3110 “Impressions or Findings” | 1-n | U |  |
27 | > | CONTAINS | INCLUDE | DTID 3111 “Percutaneous Entry” | 1-n | U |  |
28 | > | CONTAINS | INCLUDE | DTID 3112 “Specimen Obtained” | 1-n | U |  |
29 | > | CONTAINS | INCLUDE | DTID 3113 “Patient Support” | 1-n | U |  |
30 | > | CONTAINS | INCLUDE | DTID 3114 “Patient Assessment” | 1-n | U |  |
31 | > | CONTAINS | INCLUDE | DTID 3115 “ECG ST Assessment” | 1-n | U |  |

Content Item Descriptions

Row 2 | Includes TID 1002 “Observer Context”, which shall be used to record the identity of the person responsible for recording the log, as well as all other participants in the procedure, even though these personnel may not technically be "observers" of the Procedure Log. As participants in the procedure, they are potential sources for events and observations recorded in the Log. TID 1002 “Observer Context” allows the specification of the person's role in the organization (e.g., physician, nurse), as well as the role in the procedure (e.g., circulating, performing, etc.).
Row 5 | Shall be used to record the identity of the major equipment used in the procedure.
Row 6 | May be used to record any event not covered by a specific log entry Template.

### TID 3010 Log Entry Qualifiers

The Log Entry Qualifiers Template provides a common means for adding additional description to a procedure log Content Item. It allows identification of a source for the procedure log entry (other than the recording observer for the log as a whole), a free text comment, a link to a particular Procedure Action item, a link to a particular lesion, or the date/time of recording (if different than the time of the event occurrence recorded in the Observation DateTime of the parent Content Item).

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3010. Log Entry Qualifiers

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
2 |  | HAS PROPERTIES | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
3 |  | HAS OBS CONTEXT | TEXT | EV (121124, DCM, "Procedure Action ID") | 1-n | U |  |
4 |  | HAS OBS CONTEXT | TEXT | EV (121151, DCM, "Lesion Identifier") | 1-n | U |  | Up to 3 numeric characters
5 |  | HAS OBS CONTEXT | DATETIME | EV (121125, DCM, "DateTime of Recording of Log Entry") | 1 | U |  |
6 |  | INFERRED FROM | IMAGE |  | 1-n | U |  |
7 |  | INFERRED FROM | WAVEFORM |  | 1-n | U |  |
8 |  | INFERRED FROM | COMPOSITE |  | 1-n | U |  |
9 |  | HAS OBS CONTEXT | CODE | EV (121135, DCM, "Observation DateTime Qualifier") | 1 | U |  | DCID 3430 “DateTime Qualifiers”

Content Item Descriptions

Row 3 | Procedure Action ID allows linking recorded events to a particular action, step, or phase of a procedure. See description for TID 3100 “Procedure Action”.
Row 4 | Lesion Identifier is specified as a numeric text string, and allows linking recorded events to the diagnosis or therapy of particular lesion. See description for TID 3105 “Lesion Identification and Properties”.

### TID 3100 Procedure Action

The Procedure Action Template is intended for the recording of the beginning or end of procedure steps or action items in a procedure. The level of granularity of the recorded events is not specified, and may vary between institutions, or even be at multiple levels within a single procedure log. There is no requirement for the real-world procedure step or action item recorded with this Template to end before another one begins; there may be overlapping or simultaneous procedure steps or action items.

This log entry Template may be used to record the start or stop of timers.

Other recorded events in the procedure may be linked to a particular step or action item by Procedure Action ID (see TID 3010 “Log Entry Qualifiers”).

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3100. Procedure Action

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DCID 3421 “Procedure Action” | 1 | M |  | BCID 3405 “Procedure Action Values”
2 | > | HAS PROPERTIES | TEXT | EV (121124, DCM, "Procedure Action ID") | 1 | M |  |
3 | > | HAS PROPERTIES | PNAME | BCID 7453 “Performing Roles” | 1-n | U |  |
4 | > | HAS PROPERTIES | NUM | EV (121128, DCM, "Procedure Action Duration") | 1 | U |  |
5 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
6 | > | HAS PROPERTIES | UIDREF | EV (121126, DCM, "Performed Procedure Step SOP Instance UID") | 1 | MC | IFF a Performed Procedure Step SOP Class is used to provide status of the Procedure Step |
7 | > | HAS PROPERTIES | UIDREF | EV (121127, DCM, "Performed Procedure Step SOP Class UID") | 1 | MC | IFF a Performed Procedure Step SOP Class is used to provide status of the Procedure Step |

Content Item Descriptions

Row 2 | The value of the Procedure Action ID shall be uniquely associated with the step or action within the context of the Study, and may be used to associate various Procedure Log entries with the step or action.
Row 3 | May be used to record the identity of staff roles for the purpose of this Procedure Action, which may differ from their roles in the procedure as a whole.

### TID 3101 Image Acquisition

The Image Acquisition Template allows recording of the essential parameters of a digital image acquired during the procedure.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3101. Image Acquisition

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | IMAGE | EV (121138, DCM, "Image Acquired") | 1 | M |  |
2 | > | HAS ACQ CONTEXT | UIDREF | EV (112002, DCM, "Series Instance UID") | 1 | M |  |
3 | > | HAS ACQ CONTEXT | CODE | EV (121139, DCM, "Modality") | 1 | M |  | DCID 29 “Acquisition Modality” Derived from referenced image SOP Instance attribute (0008,0060)
4 | > | HAS PROPERTIES | NUM | EV (121140, DCM, "Number of Frames") | 1 | U |  |
5 | > | HAS PROPERTIES | TEXT | EV (121141, DCM, "Image Type") | 1 | U |  | From referenced image SOP Instance attribute (0008,0008)
6 | > | HAS ACQ CONTEXT | NUM | EV (112011, DCM, "Positioner Primary Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
7 | > | HAS ACQ CONTEXT | NUM | EV (112012, DCM, "Positioner Secondary Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
8 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3102 Waveform Acquisition

The Waveform Acquisition Template allows recording of the essential parameters of a digital waveform acquired during the procedure.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3102. Waveform Acquisition

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | WAVEFORM | EV (121143, DCM, "Waveform Acquired") | 1 | M |  |
2 | > | HAS ACQ CONTEXT | CODE | EV (121139, DCM, "Modality") | 1 | M |  | DCID 29 “Acquisition Modality” Derived from referenced waveform SOP Instance attribute (0008,0060)
3 | > | HAS ACQ CONTEXT | NUM | EV (121142, DCM, "Acquisition Duration") | 1 | U |  |
4 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3103 Referenced Object

The Referenced Object Template allows reference to measurement or report objects, such as prior medical reports, laboratory results, hemodynamic measurement reports, or quantitative analysis reports.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3103. Referenced Object

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | COMPOSITE | BCID 3407 “Purpose of Reference to Object” | 1 | M |  |
2 | > | HAS PROPERTIES | CODE | EV (121144, DCM, "Document Title") | 1 | MC | IFF Row 1 references an SR object | Root node concept of referenced SR object
3 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3104 Consumables

The Consumables Template allows recording of devices (e.g., catheters or stents), drugs, or contrast agents accessed in a procedure. This Content Item is directed towards inventory control and billing. The actual clinical use of the particular consumable is recorded using TID 3106 “Drugs/Contrast Administered” or TID 3107 “Device Used”.

This Template allows recording both consumable retrieval from, and return to, inventory or stock, and disposal of used material. The quantity involved in each recorded transaction may be specified.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3104. Consumables

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DCID 3408 “Actions With Consumables” | 1 | M |  | Vendor or local bar coded values
2 | > | HAS PROPERTIES | TEXT | DCID 3426 “Consumables Parameters” | 1-n | U |  |
3 | > | HAS PROPERTIES | NUM | EV (121146, DCM, "Quantity of Material") | 1 | U |  |
4 | > | HAS PROPERTIES | CODE | EV (121147, DCM, "Billing Code") | 1 | U |  | local billing codes
5 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3105 Lesion Identification and Properties

The Lesion Identification and Properties Template allows recording the identification of each lesion addressed in a procedure. The lesion identifier may be used to relate diagnostic or therapeutic actions with their target lesion (see Row 4 in TID 3010 “Log Entry Qualifiers”). This Content Item may include the initial visually estimated measurements of stenosis or TIMI flow; measured values from a quantitative measurement report may be referenced indirectly (through TID 3103 “Referenced Object”), or by quotation (TID 3109 “Measurements”). Subsequent (e.g., post-intervention) stenosis measurements may be encoded using TID 3109 “Measurements”, with the Lesion Identifier conveyed through its subsidiary TID 3010 “Log Entry Qualifiers” Template.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3105. Lesion Identification and Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (121151, DCM, "Lesion Identifier") | 1 | M |  | Up to 3 numeric characters
2 | > | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3604 “Arterial Lesion Locations”
3 | >> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
4 | > | HAS PROPERTIES | CODE | EV (121153, DCM, "Lesion Risk") | 1 | U |  | DCID 3418 “Lesion Risk”
5 | > | HAS PROPERTIES | NUM | EV (R-101BB, SRT, "Lumen Diameter Stenosis") | 1 | U |  | UNITS = EV (%, UCUM, "%")
6 | >> | HAS CONCEPT MOD | CODE | EV (G-72BB, SRT, "Catheterization Procedure Phase") | 1 | M |  | EV (G-7293, SRT, "Baseline Phase")
7 | >> | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | U |  | DCID 3745 “Calculation Method”
8 | > | HAS PROPERTIES | CODE | EV (122109, DCM, "Baseline TIMI Flow") | 1 | UC | IFF Row 2 specifies a coronary artery | DCID 3713 “TIMI Flow Characteristics”
9 | > | HAS PROPERTIES | CODE | EV (122131, DCM, "Degree of Thrombus") | 1 | U |  | DCID 3714 “Thrombus”
10 | > | HAS PROPERTIES | CODE | EV (F-01740, SRT, "Lesion Margin Characteristics") | 1 | U |  | DCID 3715 “Lesion Margin”
11 | > | HAS PROPERTIES | CODE | EV (122134, DCM, "Vessel Morphology") | 1-n | U |  | DCID 3712 “Vessel Descriptors”
12 | > | HAS PROPERTIES | CODE | EV (122132, DCM, "Severity of Calcification") | 1 | U |  | DCID 3716 “Severity”
13 | > | HAS PROPERTIES | IMAGE | DT (121080, DCM, "Best illustration of finding") | 1 | U |  |
14 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

Content Item Descriptions

Row 1 | Lesion Identifier is specified as a numeric text string in order to facilitate transcoding to DICOM Attribute (0018,3105) Lesion Number and to formats for outcomes registries, such as the ACC National Cardiovascular Data Registry™.

### TID 3106 Drugs/Contrast Administered

The Drugs/Contrast Administered Template allows the recording of the start or end of that type of event, together with its parameters. If start and end are represented by a single log entry (e.g., for an injection), the concept name "Drug/contrast administered" shall be used.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3106. Drugs/Contrast Administered

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DCID 3409 “Administration of Drugs/Contrast” | 1 | M |  | BCID 10 “Interventional Drug” or BCID 12 “Radiographic Contrast Agent”
2 | > | HAS PROPERTIES | TEXT | EV (121145, DCM, "Description of Material") | 1 | U |  |
3 | > | HAS PROPERTIES | CODE | EV (G-C340, SRT, "Route of administration") | 1 | U |  | BCID 11 “Route of Administration”
4 | > | HAS PROPERTIES | NUM | DCID 3410 “Numeric Parameters of Drugs/Contrast” | 1-n | U |  |
5 | > | HAS PROPERTIES | PNAME | EV (121152, DCM, "Person administering drug/contrast") | 1 | U |  |
6 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3107 Device Used

The Device Used Template allows recording of the use of interventional diagnostic and therapeutic devices.

The identification of one device used to deploy another device (e.g., a balloon catheter to deploy a stent) may be described with two entries, with one identified as a deployment device in the Concept Modifier of Row 6 of this Template, and linked by the same Procedure Action ID in the Log Entry Qualifiers of the included TID 3010 “Log Entry Qualifiers”.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3107. Device Used

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DCID 3422 “Device Use Actions” | 1 | M |  | BCID 3429 “Catheterization Devices”
2 | > | HAS PROPERTIES | CODE | EV (121150, DCM, "Device Code") | 1-n | U |  | Vendor or local bar coded values
3 | > | HAS PROPERTIES | TEXT | EV (121145, DCM, "Description of Material") | 1 | U |  |
4 | > | HAS PROPERTIES | NUM | DCID 3423 “Numeric Device Characteristics” | 1-n | U |  |
5 | > | HAS PROPERTIES | CODE | EV (G-C0E9, SRT, "Procedure site") | 1 | U |  | BCID 3630 “Cardiovascular Anatomic Locations”
6 | > | HAS CONCEPT MOD | CODE | EV (G-C0E8, SRT, "Has Intent") | 1 | U |  | DT (121155, DCM, "Deployment")
7 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3108 Intervention

The Intervention Template allows recording of interventions, including atherectomy, angioplasty, stent placement, brachytherapy, etc. The record may include reference to an image that documents the intervention.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3108. Intervention

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (122090, DCM, "Intervention Action") | 1 | M |  | DCID 3412 “Intervention Actions and Status”
2 | > | HAS PROPERTIES | CODE | EV (G-C0E9, SRT, "Procedure site") | 1 | M |  | DCID 3604 “Arterial Lesion Locations”
3 | >> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
4 | > | HAS PROPERTIES | TEXT | EV (121154, DCM, "Intervention attempt identifier") | 1 | M |  | Up to 3 numeric characters
5 | > | HAS PROPERTIES | CODE | EV (G-C50A, SRT, "Uses Equipment") | 1-n | U |  | BCID 3411 “Intracoronary Devices”
6 | >> | HAS CONCEPT MOD | CODE | EV (122111, DCM, "Primary Intervention Device") | 1 | MC | IF Device is Primary for this Lesion | DCID 230 “Yes-No”
7 | > | HAS PROPERTIES | NUM | DCID 3425 “Intervention Parameters” | 1-n | U |  |
8 | > | HAS PROPERTIES | IMAGE | BCID 7003 “Diagnostic Imaging Report Purposes of Reference” | 1 | U |  |
9 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

Content Item Descriptions

Row 4 | Intervention attempt Identifier is specified as a numeric text string, and shall be treated as the ordinal of the recorded attempted intervention within this procedure (i.e., "1" for the first attempted intervention, "2" for the second, etc.).

### TID 3109 Measurements

The Measurements Template allows recording of significant measurements, such as vital signs, laboratory results, hemodynamic measurements, or quantitative analysis measurements. These measurements are often quoted from another source, which would be documented in the included TID 3010 “Log Entry Qualifiers”.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3109. Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | No BCID | 1 | U |  |
2 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
3 | > | HAS PROPERTIES | INCLUDE | DTID 310 “Measurement Properties” | 1 | U |  |
4 |  |  | CODE | No BCID | 1 | U |  |
5 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3110 Impressions or Findings

The Impressions or Findings Template allows the recording of unconfirmed (provisional) impressions or findings noted during the procedure. It is not intended to convey the Cath Lab Clinical Report (the formal report from the performing physician), although it may be used (like any Procedure Log entry) for the subsequent construction of the Cath Lab Clinical Report.

A finding that is supported by a specific image frame may reference that image in the INFERRED FROM / IMAGE row of the included TID 3010 “Log Entry Qualifiers” Template.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3110. Impressions or Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121071, DCM, "Finding") | 1 | U |  | BCID 3728 “Cath Findings”
2 | > | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
3 | > | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | U |  |
4 | >> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
5 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
6 |  |  | TEXT | BCID 3419 “Findings Titles” | 1 | U |  |
7 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

Content Item Descriptions

Row 3 | Finding Site has no Baseline Context ID specified. Typically terms would be drawn from coronary segments, other arterial segments, myocardial segments, etc.

### TID 3111 Percutaneous Entry

The Percutaneous Entry Template allows recording of the opening or closing of invasive access ports.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3111. Percutaneous Entry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121156, DCM, "Percutaneous Entry Action") | 1 | M |  | DCID 3403 “Percutaneous Entry”
2 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
3 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3112 Specimen Obtained

The Specimen Obtained Template allows recording of obtaining a specimen, and the identifiers for that specimen. This is particularly designed for blood samples that will be analyzed for blood oxygen-related measurements. The analysis of the sample may be recorded in one or more log entries using TID 3109 “Measurements”, or in a separate Structured Report SOP Instance referenced by a log entry using TID 3103 “Referenced Object”.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3112. Specimen Obtained

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121123, DCM, "Patient Status or Event") | 1 | M |  | DCID 3515 “Specimen Collection”
2 | > | HAS ACQ CONTEXT | CODE | EV (R-00254, SRT, "Specimen Type") | 1 | UC | IFF specimen is blood sample | DCID 3520 “Blood Source Type”
3 | > | HAS ACQ CONTEXT | CODE | EV (G-C0E9, SRT, "Procedure site") | 1 | U |  | BCID 3630 “Cardiovascular Anatomic Locations”
4 | > | HAS PROPERTIES | INCLUDE | DTID 1009 “Subject Context, Specimen” | 1 | U |  |

### TID 3113 Patient Support

The Patient Support Template allows recording of the use of various support technologies, including oxygen, ventilation, pacing, etc.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3113. Patient Support

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DCID 3530 “Oxygen Administration Actions” | 1 | U |  | DCID 3531 “Oxygen Administration”
2 | > | HAS PROPERTIES | NUM | EV (121160, DCM, "Oxygen Administration Rate") | 1 | MC | IFF Row 1 Concept is (121161, DCM, "Begin Oxygen Administration") | UNITS = DT (l/min, UCUM, "l/min")
3 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
4 |  |  | CODE | DCID 3550 “Circulatory Support Actions” | 1 | U |  | DCID 3553 “Circulatory Support”
5 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
6 |  |  | CODE | DCID 3551 “Ventilation Actions” | 1 | U |  | DCID 3554 “Ventilation”
7 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
8 |  |  | CODE | DCID 3552 “Pacing Actions” | 1 | U |  | DCID 3555 “Pacing”
9 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3114 Patient Assessment

The Patient Assessment Template allows recording of the assessment of the patient's cardiovascular, neurological, and/or respiratory condition. A particular use of this Template is for "vital signs", which are a specific subset of mandatory patient assessment measurements.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3114. Patient Assessment

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121123, DCM, "Patient Status or Event") | 1 | M |  | DT (121165, DCM, "Patient Assessment Performed") DT (PA-00500, SRT, "Observation of Vital Signs")
2 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (F-008EC, SRT, "Systolic blood pressure") $Units = DCID 3500 “Pressure Units” $Method = BCID 3560 “Blood Pressure Methods”
3 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (F-008ED, SRT, "Diastolic blood pressure") $Units = DCID 3500 “Pressure Units”
4 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (8867-4, LN, "Heart rate") $Units = EV ({H.B.}/min, UCUM, "BPM")
5 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (8310-5, LN, "Body temperature") $Units = EV (Cel, UCUM, "C")
6 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = DCID 3526 “Blood Gas Saturation” $Units = EV (%, UCUM, "%")
7 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (F-21000, SRT, "Respiration rate") $Units = EV (/min, UCUM, "breaths/min")
8 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1-n | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (122195, DCM, "Pulse Strength") $Method = BCID 3442 “Peripheral Pulse Methods” $TargetSite = BCID 3440 “Peripheral Pulse Locations” $Units = DT ({0:4}, UCUM, "range 0:4")
9 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (F-009EA, SRT, "Pain Score") $Units = DT ({1:10}, UCUM, "range 1:10")
10 | > | HAS PROPERTIES | CODE | DT (8884-9, LN, "Cardiac Rhythm") | 1 | U |  | BCID 3415 “Cardiac Rhythms”
11 | > | HAS PROPERTIES | CODE | DT (9304-7, LN, "Respiration Rhythm") | 1 | U |  | BCID 3416 “Respiration Rhythms”
12 | > | HAS PROPERTIES | CODE | DT (F-043E6, SRT, "Respiration Assessment") | 1 | U |  | BCID 3448 “Airway Assessment”
13 | > | HAS PROPERTIES | CODE | DT (F-046D8, SRT, "Skin condition") | 1-n | U |  | BCID 3446 “Skin Condition”
14 | > | HAS PROPERTIES | CODE | DT (F-04317, SRT, "Patient mental state assessment") | 1 | U |  | No BCID
15 | > | HAS PROPERTIES | TEXT | BCID 3441 “Patient Assessments” | 1-n | U |  |

Content Item Descriptions

Row 8 | Pulse Strength allows the assessment of the patient's pulse at multiple locations using the Topographical concept modifier. It may also be used for a single pulse strength measurement from an unspecified location, as is typical of vital signs assessments.
Row 16 | Allows free text description of patient assessments that are not expressible by coded entries of Rows 10 to 14.

### TID 3115 ECG ST Assessment

The ECG ST Assessment Template allows recording of the assessment of changes in the patient ECG relative to baseline.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3115. ECG ST Assessment

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121123, DCM, "Patient Status or Event") | 1 | M |  | DT (R-41D8B, SRT, "ECG Analysis")
2 | > | HAS PROPERTIES | NUM | DT (122099, DCM, "ST change from baseline") | 1-n | M |  | UNITS = EV (uV, UCUM, "uV")
3 | >> | HAS CONCEPT MOD | CODE | DT (122148, DCM, "Lead ID") | 1 | M |  | BCID 3001 “ECG Leads”

## Quantitative Ventricular Analysis Report SR IOD Templates

The Templates that comprise the Quantitative Ventricular Analysis SR are interconnected as in Figure A-3:

Figure A-3. Quantitative Ventricular Analysis Report SR IOD Template Structure

### TID 3202 Ventricular Analysis

The Ventricular Analysis Template provides a CONTAINER with a structure for reporting the result of the ventricular analysis.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3202. Ventricular Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122292, DCM, "Quantitative Ventriculography Report") | 1 | M |  |
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
4 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | U |  |
5 | > | CONTAINS | CONTAINER | EV (122144, DCM, "Quantitative Analysis") | 1-n | M |  |
6 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
7 | >> | HAS OBS CONTEXT | TEXT | EV (111001, DCM, "Algorithm Name") | 1 | M |  |
8 | >> | HAS OBS CONTEXT | TEXT | EV (111003, DCM, "Algorithm Version") | 1 | M |  |
9 | >> | HAS OBS CONTEXT | TEXT | EV (122405, DCM, "Algorithm Manufacturer") | 1 | M |  |
10 | >> | CONTAINS | IMAGE | EV (121112, DCM, "Source of Measurement") | 1-n | M |  |
11 | >>> | HAS CONCEPT MOD | CODE | EV (G-A60B, SRT, "Cardiac Phase") | 1 | M |  | DCID 12233 “Cardiac Phase”
12 | >>> | HAS CONCEPT MOD | CODE | EV (111031, DCM, "Image View") | 1 | MC | If Biplane Analysis | DCID 3466 “Plane Identification”
13 | >> | HAS ACQ CONTEXT | INCLUDE | DTID 3205 “Calibration” | 1-2 | U | VM = 1: Single plane analysis, VM = 2: Biplane analysis | $CalibrationPlane = DCID 3466 “Plane Identification”
14 | >> | HAS ACQ CONTEXT | INCLUDE | DTID 3520 “Hemodynamic Clinical Context” | 1 | U |  |
15 | >> | CONTAINS | INCLUDE | DTID 3206 “VA Main Results” | 1 | M |  |
16 | >> | CONTAINS | INCLUDE | DTID 3207 “AA Main Results” | 1 | U |  |
17 | >> | CONTAINS | INCLUDE | DTID 3208 “Frame-to-Frame Results” | 1 | U |  |
18 | >> | CONTAINS | INCLUDE | DTID 3209 “Centerline Wall Motion” | 1-2 | U | VM = 1: Single plane analysis, VM = 2: Biplane analysis |
19 | >> | CONTAINS | INCLUDE | DTID 3210 “Radial Based Wall Motion” | 1-2 | U | VM = 1: Single plane analysis, VM = 2: Biplane analysis |
20 | >> | CONTAINS | INCLUDE | DTID 3211 “Landmark Based Wall Motion” | 1-2 | U | VM = 1: Single plane analysis, VM = 2: Biplane analysis |

Content Item Descriptions

Row 7 | Identifies the Ventricular Analysis program
Row 8 | Identifies the Ventricular Analysis program version
Row 9 | Identifies the Ventricular Analysis program manufacturer
Row 10 | Identifies the ES and ED images on which the analysis is based, for frame by frame analysis the analyzed image are specified in the frame by frame results (3208) Template

### TID 3205 Calibration

The Calibration Template consists of a CONTAINER, with a structure for reporting of the calibration of images used in the analysis.

Table TID 3205. Parameters

Parameter Name | Parameter Usage
--- | ---
$CalibrationPlane | XA Imaging plane

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3205. Calibration

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122505, DCM, "Calibration") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111031, DCM, "Image View") | 1 | U |  | $CalibrationPlane
3 | > | HAS OBS CONTEXT | TEXT | EV (111001, DCM, "Algorithm Name") | 1 | MC | IF different from Analysis program specified in the invoking Template |
4 | > | HAS OBS CONTEXT | TEXT | EV (111003, DCM, "Algorithm Version") | 1 | MC | IF different from Analysis program specified in the invoking Template |
5 | > | HAS OBS CONTEXT | TEXT | EV (122405, DCM, "Algorithm Manufacturer") | 1 | MC | IF different from Analysis program specified in the invoking Template |
6 | > | CONTAINS | CODE | EV (122422, DCM, "Calibration Method") | 1 | M |  | DCID 3452 “Calibration Methods”
7 | > | CONTAINS | CODE | EV (122421, DCM, "Calibration Object") | 1 | MC | If row 6 value specifies Calibration Object Used | DCID 3451 “Calibration Objects”
8 | > | CONTAINS | NUM | EV (122423, DCM, "Calibration Object Size") | 1 | MC | If row 6 value specifies Calibration Object Used | DCID 3510 “Catheter Size Units”
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (111026, DCM, "Horizontal Pixel Spacing") $Unit = DT (mm/{pixel}, UCUM, "mm/pixel") $ImagePurpose = EV (121112, DCM, "Source of Measurement")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (111066, DCM, "Vertical Pixel Spacing") $Unit = DT (mm/{pixel}, UCUM, "mm/pixel") $ImagePurpose = EV (121112, DCM, "Source of Measurement")
11 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 3 | Identifies the Calibration program
Row 4 | Identifies the Calibration program version
Row 5 | Identifies the Calibration program manufacturer
Row 7 | Besides a Sphere and a Catheter, a Distance can be identified as a Calibration Object. In this case a distance measurement of a known dimension of the object is used to calculate the pixel size.
Row 8 | The catheter size units is also used to specify the size of other calibration objects (e.g., sphere)
Row 9, 10 | Spacing in the patient body. Point to a single frame containing the image used for calibration if applicable, the actual measurements may be indicated by a SCOORD (see TID 320 “Image or Spatial Coordinates”, row 3)
Row 11 | Secondary Capture image with calibration position

### TID 3206 VA Main Results

The VA Main Results Template consists of a CONTAINER with a structure for reporting the main ventricular analysis measurements.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3206. VA Main Results

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3462 “Chamber Identification”
3 | > | CONTAINS | CODE | EV (122429, DCM, "Volume Method") | 1 | M |  | DCID 3453 “Cardiac Volume Methods”
4 | > | CONTAINS | NUM | EV (122435, DCM, "Regression Volume Exponent") | 1 | U |  | Unit = DT (1, UCUM, "no units")
5 | > | CONTAINS | NUM | EV (122431, DCM, "Regression Slope ED") | 1 | U |  | Unit = DT ({ratio}, UCUM, "ratio")
6 | > | CONTAINS | NUM | EV (122432, DCM, "Regression Offset ED") | 1 | U |  | Unit = DT (ml, UCUM, "ml")
7 | > | CONTAINS | NUM | EV (122433, DCM, "Regression Slope ES") | 1 | U |  | Unit = DT ({ratio}, UCUM, "ratio")
8 | > | CONTAINS | NUM | EV (122434, DCM, "Regression Offset ES") | 1 | U |  | Unit = DT (ml, UCUM, "ml")
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = DCID 3467 “Ejection Fraction” $Unit = DT (%, UCUM, "%")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3468 “ED Volume” $Unit = DT (ml, UCUM, "ml")
11 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3469 “ES Volume” $Unit = DT (ml, UCUM, "ml")
12 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $Unit = DT (ml, UCUM, "ml")
13 | > | CONTAINS | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  | Unit = DT ({H.B.}/min, UCUM, "BPM")
14 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3468 “ED Volume” $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (ml/m2, UCUM, "ml/m2")
15 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3468 “ED Volume” $ModType = EV (121425, DCM, "Index") $ModValue = EV (29463-7, LN, "Patient Weight") $Unit = DT (ml/kg, UCUM, "ml/kg")
16 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3469 “ES Volume” $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (ml/m2, UCUM, "ml/m2")
17 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3469 “ES Volume” $ModType = EV (121425, DCM, "Index") $ModValue = EV (29463-7, LN, "Patient Weight") $Unit = DT (ml/kg, UCUM, "ml/kg")
18 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (ml/m2, UCUM, "ml/m2")
19 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $ModType = EV (121425, DCM, "Index") $ModValue = EV (29463-7, LN, "Patient Weight") $Unit = DT (ml/kg, UCUM, "ml/kg")
20 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32100, SRT, "Cardiac Output") $Unit = DT (l/min, UCUM, "l/min")
21 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32110, SRT, "Cardiac Index") $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (l/min/m2, UCUM, "l/min/m2")
22 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122445, DCM, "Wall Thickness") $Unit = DT (mm, UCUM, "mm")
23 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122446, DCM, "Wall Volume") $Unit = DT (ml, UCUM, "ml")
24 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $Unit = DT (g, UCUM, "gram")
25 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (g/m2, UCUM, "gram/m2")
26 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $ModType = EV (121425, DCM, "Index") $ModValue = EV (29463-7, LN, "Patient Weight") $Unit = DT (g/kg, UCUM, "gram/kg")
27 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122448, DCM, "Wall Stress") $Unit = DT (dyn/cm2, UCUM, "dynes/cm2")
28 | > | CONTAINS | IMAGE | No purpose of reference | 1-n | U |  |

Content Item Descriptions

Row 28 | Secondary Capture image with ED and/or ES contours

### TID 3207 AA Main Results

The AA Main Results Template consists of a CONTAINER with a structure for reporting the main atrial analysis measurements.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3207. AA Main Results

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3462 “Chamber Identification”
3 | > | CONTAINS | CODE | EV (122429, DCM, "Volume Method") | 1 | M |  | DCID 3453 “Cardiac Volume Methods”
4 | > | CONTAINS | NUM | EV (122435, DCM, "Regression Volume Exponent") | 1 | U |  | Unit = DT (1, UCUM, "no units")
5 | > | CONTAINS | NUM | EV (122431, DCM, "Regression Slope ED") | 1 | U |  | Unit = DT ({ratio}, UCUM, "ratio")
6 | > | CONTAINS | NUM | EV (122432, DCM, "Regression Offset ED") | 1 | U |  | Unit = DT (ml, UCUM, "ml")
7 | > | CONTAINS | NUM | EV (122433, DCM, "Regression Slope ES") | 1 | U |  | Unit = DT ({ratio}, UCUM, "ratio")
8 | > | CONTAINS | NUM | EV (122434, DCM, "Regression Offset ES") | 1 | U |  | Unit = DT (ml, UCUM, "ml")
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3468 “ED Volume” $Unit = DT (ml, UCUM, "ml")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3469 “ES Volume” $Unit = DT (ml, UCUM, "ml")
11 | > | CONTAINS | IMAGE | No purpose of reference | 1-n | U |  |

Content Item Descriptions

Row 11 | Secondary Capture image with ED and/or ES contours

### TID 3208 Frame-to-Frame Results

The Frame-to-Frame Result Template consists of a CONTAINER providing measurements derived from the angiographic images on frame-by-frame basis.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3208. Frame-To-Frame Result

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122499, DCM, "Frame to Frame Analysis")
3 | > | CONTAINS | IMAGE | EV (121112, DCM, "Source of Measurement") | 1-2 | M | VM = 1: Single plane analysis, VM = 2: Biplane analysis |
4 | > | CONTAINS | CODE | EV (122429, DCM, "Volume Method") | 1 | M |  | DCID 3453 “Cardiac Volume Methods”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = DCID 3471 “Estimated Volumes” $TargetSite = DCID 3462 “Chamber Identification” $Unit = DT (ml, UCUM, "ml")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (122445, DCM, "Wall Thickness") $Unit = DT (mm, UCUM, "mm")
7 | > | CONTAINS | IMAGE | No purpose of reference | 1-n | U |  |

Content Item Descriptions

Row 3 | Identifies each frame analyzed, using the multi-valued Referenced Frame Number (0008,1160) attribute of the IMAGE Content Item.
Row 5, 6 | Includes one measurement for each frame referenced in Row 3.
Row 7 | Secondary Capture image with ventricular contours

### TID 3209 Centerline Wall Motion

The Centerline Wall Motion Template consists of a CONTAINER providing measurements of the centerline wall motion.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3209. Centerline Wall Motion

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122449, DCM, "Centerline Wall Motion Analysis")
3 | >> | HAS CONCEPT MOD | CODE | EV (122410, DCM, "Contour Realignment") | 1 | M |  | DCID 3458 “Contour Realignment”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 100 | M |  | $Measurement = EV (122450, DCM, "Normalized Chord Length") $Unit = DT (%, UCUM, "%")
5 | > | CONTAINS | NUM | EV (122411, DCM, "Threshold Value") | 1 | M |  | Values shall be 1, 2 or 3 UNITS = EV ({sd}, UCUM, "Standard Deviations")
6 | > | CONTAINS | CONTAINER | EV (122451, DCM, "Abnormal Region") | 1-6 | U |  |
7 | >> | CONTAINS | CODE | EV (F-32050, SRT, "Cardiac Wall Motion") | 1 | M |  | DCID 3703 “Wall Motion”
8 | >> | CONTAINS | CODE | EV (R-404F0, SRT, "Circumferential Extent") | 1 | U |  | DCID 3460 “Circumferential Extent”
9 | >> | CONTAINS | NUM | EV (122452, DCM, "First Chord of Abnormal Region") | 1 | M |  | Unit = DT (1, UCUM, "no unit")
10 | >> | CONTAINS | NUM | EV (122453, DCM, "Last Chord of Abnormal Region") | 1 | M |  | Unit = DT (1, UCUM, "no unit")
11 | > | CONTAINS | CONTAINER | EV (122417, DCM, "Regional Abnormal Wall Motion ") | 1-4 | U |  |
12 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3461 “Regional Extent”
13 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122459, DCM, "Territory Region Severity") $ModType = EV (F-32050, SRT, "Cardiac Wall Motion") $ModValue = DCID 3703 “Wall Motion” $Unit = DT ({sd}, UCUM, "Standard Deviations")
14 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122461, DCM, "Opposite Region Severity") $ModType = EV (F-32050, SRT, "Cardiac Wall Motion") $ModValue = DCID 3703 “Wall Motion” $Unit = DT ({sd}, UCUM, "Standard Deviations")
15 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 4 | Normalized lengths of the chords determined between ED and ES contour. The measurement Template allows the specification of the statistical properties of the normal population and of the chord measurement relative to the population.
Row 8 | If the Circumferential Extent is not specified no limitations to the boundaries for regions are assumed.
Row 11 | The Regional Abnormal Wall Motion container holds the information on the severity of the decreased or increased wall motion of the 4 predefined regions as described in [Sheehan, 1986].
Row 12 | The name of the region with an abnormal ventricular wall motion as described in [Sheehan, 1986].
Row 13 | The severity of the wall motion abnormality expressed in Standard Deviations above or below normal in the territory region as described in [Sheehan, 1986].
Row 14 | The severity of the wall motion abnormality expressed in Standard Deviations above or below normal in the opposite region as described in [Sheehan, 1986].
Row 15 | Secondary Capture image with centerline analysis result

### TID 3210 Radial Based Wall Motion

The Radial Based Wall Motion Template consists of a CONTAINER providing measurements of the radial based wall motion.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3210. Radial Based Wall Motion

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122493, DCM, "Radial Based Wall Motion Analysis")
3 | >> | HAS CONCEPT MOD | CODE | EV (122410, DCM, "Contour Realignment") | 1 | M |  | DCID 3458 “Contour Realignment”
4 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | M |  |
5 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3718 “Myocardial Wall Segments in Projection”
6 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122495, DCM, "Regional Contribution to Ejection Fraction") $Unit = DT (%, UCUM, "%")
7 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (122496, DCM, "Radial Shortening") $Unit = DT (%, UCUM, "%")
8 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 6 | The CREF values of the 6 regions determined for the radial based wall motion
Row 7 | The shortening of the measured radials within the region
Row 8 | Secondary Capture image with radial based analysis result

### TID 3211 Landmark Based Wall Motion

The Landmark Based Wall Motion Template consists of a CONTAINER providing measurements of the landmark based wall motion.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3211. Landmark Based Wall Motion

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122497, DCM, "Landmark Based Wall Motion Analysis")
3 | >> | HAS CONCEPT MOD | CODE | EV (122410, DCM, "Contour Realignment") | 1 | M |  | DCID 3458 “Contour Realignment”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = EV (122498, DCM, "Slice Contribution to Ejection Fraction") $Unit = DT (%, UCUM, "%")
5 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 5 | M |  |
6 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3718 “Myocardial Wall Segments in Projection”
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122495, DCM, "Regional Contribution to Ejection Fraction") $Unit = DT (%, UCUM, "%")
8 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 8 | Secondary Capture image with Landmark Based Analysis result

## Quantitative Arterial Analysis Report SR IOD Templates

The Templates that comprise the Quantitative Arterial Analysis SR are interconnected as in Figure A-4:

Figure A-4. Quantitative Arterial Analysis Report SR IOD Template Structure

### TID 3213 Quantitative Arterial Analysis

The Quantitative Arterial Analysis Template consists of a CONTAINER with a structure for reporting the result of the quantitative arterial analysis process.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3213. Quantitative Arterial Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122291, DCM, "Quantitative Arteriography Report") | 1 | M |  |
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
4 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | U |  |
5 | > | HAS OBS CONTEXT | TEXT | EV (111001, DCM, "Algorithm Name") | 1 | M |  |
6 | > | HAS OBS CONTEXT | TEXT | EV (111003, DCM, "Algorithm Version") | 1 | M |  |
7 | > | HAS OBS CONTEXT | TEXT | EV (122405, DCM, "Algorithm Manufacturer") | 1 | M |  |
8 | > | CONTAINS | INCLUDE | DTID 3214 “Analyzed Segment” | 1-n | M |  |

Content Item Descriptions

Row 5 | Identifies the Arterial Analysis program
Row 6 | Identifies the Arterial Analysis program version
Row 7 | Identifies the Arterial Analysis program manufacturer

### TID 3214 Analyzed Segment

The Analyzed Segment Template consists of a CONTAINER providing quantitative arterial analysis measurements derived from the angiographic images.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3214. Analyzed Segment

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3604 “Arterial Lesion Locations”
3 | > | CONTAINS | IMAGE | EV (121112, DCM, "Source of Measurement") | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 3205 “Calibration” | 1 | M |  |
5 | > | HAS ACQ CONTEXT | INCLUDE | DTID 3520 “Hemodynamic Clinical Context” | 1 | U |  |
6 | > | HAS ACQ CONTEXT | CODE | EV (G-72BB, SRT, "Catheterization Procedure Phase") | 1 | U |  | DCID 3651 “Hemodynamic Measurement Phase”
7 | > | CONTAINS | SCOORD | EV (122507, DCM, "Left Contour") | 1 | M |  | GRAPHIC TYPE = POLYLINE
8 | >> | R-SELECTED FROM | IMAGE |  | 1 | M |  | Must reference Row 3
9 | > | CONTAINS | SCOORD | EV (122508, DCM, "Right Contour") | 1 | M |  | GRAPHIC TYPE = POLYLINE
10 | >> | R-SELECTED FROM | IMAGE |  | 1 | M |  | Must reference Row 3
11 | > | CONTAINS | INCLUDE | DTID 3219 “Segment Values” | 1 | M |  |
12 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-404FB, SRT, "Minimum") $Unit = DT (mm, UCUM, "mm")
13 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (G-A437, SRT, "Maximum") $Unit = DT (mm, UCUM, "mm")
14 | > | CONTAINS | CONTAINER | EV (122509, DCM, "Diameter Graph") | 1 | U |  |
15 | >> | CONTAINS | NUM | EV (122511, DCM, "Graph Increment") | 1 | M |  | Value = 1 UNITS = DT ({pixels}, UCUM, "pixels")
16 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Unit = DT (mm, UCUM, "mm")
17 | > | CONTAINS | NUM | EV (122382, DCM, "Site of Luminal Minimum ") | 1 | U |  | UNITS = DT ({pixels}, UCUM, "pixels")
18 | > | CONTAINS | NUM | EV (122516, DCM, "Site of Luminal Maximum") | 1 | U |  | UNITS = DT ({pixels}, UCUM, "pixels")
19 | > | CONTAINS | INCLUDE | DTID 3215 “Angiographic Lesion Analysis” | 1-n | U |  |
20 | > | CONTAINS | INCLUDE | DTID 3217 “Sub-segmental Data” | 1-n | U |  |
21 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 1 | Observation DateTime (0040,A032) of container needs to be flagged with the time of the analysis
Row 6 | Numeric coordinates (x,y) identifying the contour points from proximal to distal of left contour. Left is relative to the direction of the blood flow.
Row 9 | Numeric coordinates (x,y) identifying the contour points from proximal to distal of right contour. Right is relative to the direction of the blood flow.
Row 12, 13 | Positions are relative to the midpoint between the first left and right contour points and measured along the midline between the left and right contour.
Row 14 | The X-axis represents the pixel points of the midline of the vessel from proximal to distal. The points on the midline are not necessarily equidistant.
Row 16 | For each point of the midline of the vessel a measurement value for the diameter is calculated.
Row 17, 18 | The positions in the graph are related to the points on the midline of the vessel.
Row 21 | Secondary Capture image with Arterial Analysis contour.

Definition of Left and Right defined by the direction of the blood flow as in Figure A-4b:

Figure A-4b. Direction of Blood Flow

### TID 3215 Angiographic Lesion Analysis

The Angiographic Lesion Analysis Template consists of a CONTAINER providing quantitative arterial analysis measurements derived for an obstruction in a total analyzed segment.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3215. Angiographic Lesion Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (F-03FCD, SRT, "Lesion Finding") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121151, DCM, "Lesion Identifier") | 1 | M |  |
3 | >> | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3604 “Arterial Lesion Locations”
4 | >>> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-404FB, SRT, "Minimum") $Unit = DT (mm, UCUM, "mm")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Method = DCID 3470 “Vessel Lumen Cross-sectional Area Calculation Methods” $Derivation = EV (R-404FB, SRT, "Minimum") $Unit = DT (mm2, UCUM, "mm2")
7 | > | CONTAINS | CODE | EV (122430, DCM, "Reference Method") | 1 | M |  | DCID 3465 “QA Reference Methods”
8 | > | CONTAINS | CONTAINER | EV (122438, DCM, "Reference Points") | 1 | U |  |
9 | >> | CONTAINS | NUM | EV (122337, DCM, "Relative Position") | 1-n | M |  | UNITS = DT (mm, UCUM, "mm")
10 | >>> | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Unit = DT (mm, UCUM, "mm")
11 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $TargetSite = (122382, DCM, "Site of Luminal Minimum") $Unit = DT (mm, UCUM, "mm")
12 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Derivation = EV (122404, DCM, "Reconstructed") $TargetSite = (122382, DCM, "Site of Luminal Minimum") $Unit = DT (mm2, UCUM, "mm2")
13 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-41D2D, SRT, "Calculated") $TargetSite = EV (122481, DCM, "Contour Start") $Unit = DT (mm, UCUM, "mm")
14 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-41D2D, SRT, "Calculated") $TargetSite = EV (122482, DCM, "Contour End") $Unit = DT (mm, UCUM, "mm")
15 | > | CONTAINS | INCLUDE | DTID 3218 “Position in Arterial Segment” | 1 | M |  |
16 | > | CONTAINS | CONTAINER | EV (122517, DCM, "Densitometric Luminal Cross-sectional Area Graph") | 1 | U |  |
17 | >> | CONTAINS | NUM | EV (122511, DCM, "Graph Increment") | 1 | M |  | Value = 1 UNITS = DT ({pixels}, UCUM, "pixels")
18 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Unit = (mm2, UCUM, "mm2")
19 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Derivation = EV (R-41D2D, SRT, "Calculated") $Method = EV (122474, DCM, "Densitometric method") $TargetSite = EV (122481, DCM, "Contour Start") $Unit = (mm2, UCUM, "mm2")
20 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Derivation = EV (R-41D2D, SRT, "Calculated") $Method = EV (122474, DCM, "Densitometric method") $TargetSite = EV (122482, DCM, "Contour End") $Unit = (mm2, UCUM, "mm2")
21 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (R-101BC, SRT, "Stenotic Lesion Length") $Unit = DT (mm, UCUM, "mm")
22 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (R-101BB, SRT, "Lumen Diameter Stenosis") $Unit = DT (%, UCUM, "%")
23 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (R-101BA, SRT, "Lumen Area Stenosis") $Method = DCID 3470 “Vessel Lumen Cross-sectional Area Calculation Methods” $Unit = DT (%, UCUM, "%")
24 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (122372, DCM, "Lumen Volume") $Method = DCID 3470 “Vessel Lumen Cross-sectional Area Calculation Methods” $Unit = DT (mm3, UCUM, "mm3")
25 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122542, DCM, "Plaque Area") $Unit = DT (mm2, UCUM, "mm2")
26 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122376, DCM, "Total Plaque Volume") $Unit = DT (mm3, UCUM, "mm3")
27 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122544, DCM, "Diameter Symmetry") $Unit = DT ({ratio}, UCUM, "ratio")
28 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122545, DCM, "Area Symmetry") $Unit = DT ({ratio}, UCUM, "ratio")
29 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122546, DCM, "Inflow Angle") $Unit = DT (deg, UCUM, "deg")
30 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122547, DCM, "Outflow Angle") $Unit = DT (deg, UCUM, "deg")
31 | > | CONTAINS | INCLUDE | DTID 3216 “Stenotic Flow Reserve” | 1 | U |  |
32 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 8 | Set of user defined reference position for method that requires local reference position.
Row 9 | Distance of local reference position from an arbitrary landmark.
Row 10 | Diameter at a local reference position.
Row 11 | The reference diameter for the arterial lesion calculated with the applicable reference method
Row 12 | The reference area for the arterial lesion calculated with the applicable reference method
Row 13 | The diameter measurement at the start of the reconstruction line in the diameter graph (TID 3214 “Analyzed Segment” Row 14)
Row 14 | The diameter measurement at the end of the reconstruction line in the diameter graph (TID 3214 “Analyzed Segment” Row 14)
Row 15 | The positions of the lesion, borders of the lesion, etc.
Row 16 | The graph with the calculated cross sectional area results based on the densitometric method
Row 18 | The cross sectional area measurements calculated based on the densitometric method
Row 19 | The cross sectional area measurement at the start of the reconstruction line in the area graph
Row 20 | The cross sectional area measurement at the end of the reconstruction line in the area graph
Row 21 | Measured along the midline of the left and right contour
Row 22 | The diameter stenosis is calculated as follows: (Reference Luminal Diameter - Minimum Luminal Diameter / Reference Luminal Diameter) * 100%
Row 23 | The circular and the densitometric area stenosis are calculated respectively as: (Reference Vessel Lumen Cross-Sectional Area - Minimum Luminal Circular Area / Reference Vessel Lumen Cross-Sectional Area) * 100%(Reference Vessel Lumen Cross-Sectional Area - Minimum Luminal Densitometric Area / Reference Vessel Lumen Cross-Sectional Area) * 100%
Row 24 | Estimated lumen volume between proximal border and distal border of lesion (TID 3218 “Position in Arterial Segment”, row 1 and 2)
Row 32 | Secondary Capture image with obstruction analysis contour

### TID 3216 Stenotic Flow Reserve

The Stenotic Flow Reserve Template consists of a CONTAINER providing quantitative arterial analysis measurements derived for an obstruction in a total analyzed segment.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3216. Stenotic Flow Reserve

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122548, DCM, "Stenotic Flow Reserve") $Unit = DT ({ratio}, UCUM, "ratio")
2 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122549, DCM, "Poiseuille Resistance") $Unit = DT (mm[Hg].s/cm, UCUM, "mmHG.s/cm")
3 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122550, DCM, "Turbulence Resistance") $Unit = DT (mm[Hg].s2/cm2, UCUM, "mmHG.s2/cm2")
4 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122555, DCM, "Estimated Normal Flow") $Unit = DT (ml/s, UCUM, "ml/s")
5 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122551, DCM, "Pressure Drop at SFR") $Unit = DT (mm[Hg], UCUM, "mmHg")
6 |  | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 6 | Secondary Capture image with SFR analysis contour

### TID 3217 Sub-segmental Data

The Sub-segmental Data Template consists of a CONTAINER providing quantitative arterial analysis measurements derived for a sub-segment in a total analyzed segment.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3217. Sub-Segmental Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3604 “Arterial Lesion Locations”
3 | >> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
4 | > | CONTAINS | CODE | EV (122554, DCM, "Segmentation Method") | 1 | M |  | DCID 3456 “Sub-segment Methods”
5 | > | CONTAINS | INCLUDE | DTID 3219 “Segment Values” | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3218 “Position in Arterial Segment” | 1 | M |  |
7 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 7 | Secondary Capture image with obstruction analysis contour

### TID 3218 Position in Arterial Segment

The Position in Arterial Segment Template consists of the position Content Items common for the Angiographic Lesion Analysis and Sub-Segmental Data.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3218. Position in Arterial Segment

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122528, DCM, "Position of Proximal Border") $Unit = DT (mm, UCUM, "mm")
2 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122529, DCM, "Position of Distal Border") $Unit = DT (mm, UCUM, "mm")
3 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122382, DCM, "Site of Luminal Minimum") $Unit = DT (mm, UCUM, "mm")
4 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122516, DCM, "Site of Luminal Maximum") $Unit = DT (mm, UCUM, "mm")
5 |  | CONTAINS | NUM | EV (122528, DCM, "Position of Proximal Border") | 1 | UC | IFF TID 3214 “Analyzed Segment” Row 14 is present | UNITS = DT ({pixels}, UCUM, "pixels")
6 |  | CONTAINS | NUM | EV (122529, DCM, "Position of Distal Border") | 1 | UC | IFF TID 3214 “Analyzed Segment” Row 14 is present | UNITS = DT ({pixels}, UCUM, "pixels")
7 |  | CONTAINS | NUM | EV (122382, DCM, "Site of Luminal Minimum") | 1 | UC | IFF TID 3214 “Analyzed Segment” Row 14 is present | UNITS = DT ({pixels}, UCUM, "pixels")
8 |  | CONTAINS | NUM | EV (122516, DCM, "Site of Luminal Maximum") | 1 | UC | IFF TID 3214 “Analyzed Segment” Row 14 is present | UNITS = DT ({pixels}, UCUM, "pixels")

Content Item Descriptions

Row 1, 2, 3, 4 | Positions are relative to the midpoint of the first left and right contour points and measured along the midline of the left and right contour
Row 5, 6, 7, 8 | The positions are relative to the measurement locations of the Diameter Graph of TID 3214 “Analyzed Segment” row 14.

### TID 3219 Segment Values

The Segment Values Template consists of Content Items providing quantitative arterial analysis measurements for a total analyzed segment or sub segment.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3219. Segment Values

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122510, DCM, "Length Luminal Segment") $Unit = DT (mm, UCUM, "mm")
2 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-404FB, SRT, "Minimum") $Unit = DT (mm, UCUM, "mm")
3 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (G-A437, SRT, "Maximum") $Unit = DT (mm, UCUM, "mm")
4 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-00317, SRT, "Mean") $Unit = DT (mm, UCUM, "mm")
5 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-10047, SRT, "Standard Deviation") $Unit = DT (mm, UCUM, "mm")

Content Item Descriptions

Row 1 | Measured along the midline of the left and right contour.

## IVUS Report Templates

The Templates that comprise the IVUS Report within the Evidence Report IOD are interconnected as shown in Figure A-5.

Figure A-5. IVUS Report Template Hierarchy

### TID 3250 IVUS Report

The IVUS Report Template is the root structure for the representation of IVUS measurements acquired during a catheterization procedure.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 3250. IVUS Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122325, DCM, "IVUS Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
4 | > |  | INCLUDE | DTID 3601 “Procedure Context” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | U |  |
6 | > | CONTAINS | CONTAINER | EV (111028, DCM, "Image Library") | 1 | U |  |
7 | >> | CONTAINS | IMAGE | No purpose of reference | 1-n | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3251 “IVUS Vessel” | 1-n | M |  |

### TID 3251 IVUS Vessel

The IVUS Vessel Template provides a structure for grouping one or more lesions analyzed and/or treated during a single phase of a catheterization procedure, according to vessel (or arterial location).

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3251. IVUS Vessel

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | U |  | DCID 3604 “Arterial Lesion Locations”
3 | >> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
4 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | UC | IFF anatomy has laterality | DCID 244 “Laterality”
5 | > | HAS ACQ CONTEXT | CODE | EV (G-72BB, SRT, "Catheterization Procedure Phase") | 1 | U |  | DCID 3480 “IVUS Procedure Phases”
6 | > | CONTAINS | CODE | EV (122134, DCM, "Vessel Morphology") | 1-n | U |  | CID 3712 “Vessel Descriptors”
7 | > | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | U |  |
8 | > | CONTAINS | CODE | EV (115, NCDR [2.0b], "Dissection in segment") | 1 | U |  | DCID 230 “Yes-No”
9 | > | CONTAINS | INCLUDE | DTID 3252 “IVUS Lesion” | 1-n | U |  |

### TID 3252 IVUS Lesion

The IVUS Lesion Template provides a structure for grouping measurements and observations made on a single lesion during an Intravascular Ultrasound Procedure.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3252. IVUS Lesion

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (F-03FCD, SRT, "Lesion Finding") | 1 | M |  |
2 | > | HAS OBS CONTEXT | TEXT | EV (121151, DCM, "Lesion Identifier") | 1 | M |  | Up to 3 numeric characters
3 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | U |  | DCID 3604 “Arterial Lesion Locations”
4 | >>> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
5 | > | HAS ACQ CONTEXT | INCLUDE | DTID 3107 “Device Used” | 1-n | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3253 “IVUS Measurements” | 1 | MC | One or both of rows 6 & 7 must be present |
7 | > | CONTAINS | INCLUDE | DTID 3254 “IVUS Qualitative Assessments” | 1 | MC | One or both of rows 6 & 7 must be present |
8 | > | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | U |  |

Content Item Descriptions

Row 2 | Lesion Identifier is specified as a numeric text string in order to facilitate trans-coding to DICOM Attribute (0018,3105) Lesion Number and to formats for outcomes registries, such as the ACC National Cardiovascular Data Registry™. ### Note Also see TID 3105 “Lesion Identification and Properties”.
Row 3 | Finding site may span multiple segments with the proximal and distal extent specified by separate items. These may not be totally contained with the segment specified at the Vessel level.

### TID 3253 IVUS Measurements

The IVUS measurements Template groups together simple distance, area and angle measurements, along with derived measurements that made during an IVUS procedure. Refer to the " ACC Clinical Expert Consensus Document on Standards for Acquisition, measurement and Reporting of Intravascular Ultrasound Studies (IVUS) "for more information.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3253. IVUS Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3481 “IVUS Distance Measurements” $Units = EV (mm, UCUM, "mm") $Derivation = DCID 3488 “Min/Max/Mean” $TargetSite = BCID 3486 “Vascular Measurement Sites”
2 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3482 “IVUS Area Measurements” $Units = EV (mm2, UCUM, "mm2") $Derivation = DCID 3488 “Min/Max/Mean” $TargetSite = BCID 3486 “Vascular Measurement Sites”
3 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3483 “IVUS Longitudinal Measurements” $Units = EV (mm, UCUM, "mm")
4 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (122355, DCM, "Arc of Calcium") $Units = EV (deg, UCUM, "degrees") $TargetSite = BCID 3486 “Vascular Measurement Sites”
5 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (R-101BA, SRT, "Lumen Area Stenosis") $Units = EV (%, UCUM, "%")
6 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122354, DCM, "Plaque Burden") $Units = EV (%, UCUM, "%") $TargetSite = BCID 3486 “Vascular Measurement Sites”
7 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3484 “IVUS Indices and Ratios” $Units = EV ({ratio}, UCUM, "ratio") $TargetSite = BCID 3486 “Vascular Measurement Sites”
8 |  |  | INCLUDE | DTID 3255 “IVUS Volume Measurement” | 1-n | U |  |
9 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122339, DCM, "Stent Volume Obstruction") $Units = EV (%, UCUM, "%")

### TID 3254 IVUS Qualitative Assessments

The IVUS Qualitative Assessments Template groups together the qualitative properties of a lesion that are observed during an IVUS procedure. Refer to the "ACC Clinical Expert Consensus Document on Standards for Acquisition, measurement and Reporting of Intravascular Ultrasound Studies (IVUS) " for more information.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3254. IVUS Qualitative Assessments

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (122133, DCM, "Lesion Morphology") | 1-n | U |  | DCID 3491 “IVUS Lesion Morphologies”
2 |  |  | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3494 “IVUS Non Morphological Findings”
3 | > |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
4 |  |  | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (D3-81310, SRT, "Arterial Dissection")
5 | > | HAS CONCEPT MOD | CODE | EV (122387, DCM, "Dissection Classification") | 1 | U |  | DCID 3492 “Vascular Dissection Classifications”
6 | > |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
7 |  |  | CODE | EV (122391, DCM, "Relative Stenosis Severity") | 1 | U |  | DCID 3493 “IVUS Relative Stenosis Severities”
8 |  |  | CODE | EV (108, NCDR [2.0b], "Previously Dilated Lesion ") | 1 | U |  | DCID 3750 “Previously Dilated Lesion”
9 |  |  | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (122393, DCM, "Restenotic Lesion")
10 | > |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
11 |  |  | CODE | EV (111009, DCM, "Calcification Type") | 1 | U |  | DCID 3489 “Calcium Distribution”

### TID 3255 IVUS Volume Measurement

The IVUS Volume Measurement Template contains information describing an IVUS Volumetric measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3255. IVUS Volume Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = DCID 3485 “IVUS Volume Measurements” $Units = EV (mm3, UCUM, "mm3") $TargetSite = BCID 3487 “Intravascular Volumetric Regions”
2 | > | HAS PROPERTIES | NUM | EV (122336, DCM, "Vascular Volume measurement length") | 1 | U |  | UNITS = DT (mm, UCUM, "mm")
3 | > | HAS PROPERTIES | NUM | EV (122337, DCM, "Relative position") | 1 | U |  | UNITS = DT (mm, UCUM, "mm")
4 | >> | HAS CONCEPT MOD | CODE | EV (122340, DCM, "Fiducial feature") | 1 | M |  | DCID 3496 “IVUS Fiducial Points”

## Stress Testing Report Templates

### TID 3300 Stress Testing Report

The Stress Testing Report Template is the root structure for the representation of measurements and findings of a stress testing procedure.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 3300. Stress Testing Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (18752-6, LN, "Stress Testing Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DCID 3200 “Stress Test Procedure”
3 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
5 | > | CONTAINS | CONTAINER | EV (18785-6, LN, "Indications for Procedure") | 1 | U |  |
6 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3201 “Indications for Stress Test”
7 | >> | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1 | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | M |  |
9 | > | CONTAINS | INCLUDE | DTID 3802 “Cardiovascular Patient History” | 1 | U |  |
10 | > | CONTAINS | INCLUDE | DTID 3301 “Stress Test Procedure Description” | 1 | M |  |
11 | > | CONTAINS | INCLUDE | DTID 3303 “Stress Test Phase Data” | 1-n | M |  |
12 | > | CONTAINS | INCLUDE | DTID 3311 “Stress Test Summary” | 1 | U |  |
13 | > | CONTAINS | INCLUDE | DTID 3318 “Comparison to Prior Stress Exam” | 1 | U |  |
14 | > | CONTAINS | INCLUDE | DTID 3320 “Conclusions and Recommendations” | 1 | U |  |

### TID 3301 Stress Test Procedure Description

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3301. Stress Test Procedure Description

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (55111-9, LN, "Current Procedure Descriptions") | 1 | M |  |
2 | > | CONTAINS | CODE | DT (109056, DCM, "Stress Protocol") | 1 | U |  | BCID 3261 “Stress Protocols”
3 | > | CONTAINS | TEXT | DT (109056, DCM, "Stress Protocol") | 1 | U |  |
4 | > | CONTAINS | CODE | DT (10:11345, MDC, "Lead System") | 1 | U |  | BCID 3263 “Electrode Placement Values”
5 | > | CONTAINS | CODE | DT (A-17200, SRT, "Exerciser Device") | 1 | U |  | BCID 3203 “Exerciser Device”
6 | > | CONTAINS | CODE | DT (G-C11C, SRT, "Pharmacological Stress Agent") | 1 | MC | IFF Pharmacological Stress used | BCID 3204 “Stress Agents”
7 | > | CONTAINS | CONTAINER | EV (122700, DCM, "Indications for Pharmacological Stress") | 1 | MC | IFF Pharmacological Stress used |
8 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3205 “Indications for Pharmacological Stress Test”
9 | > | CONTAINS | CODE | DT (P0-0099A, SRT, "Imaging procedure") | 1 | MC | IFF imaging used in procedure | DCID 3206 “Non-invasive Cardiac Imaging Procedures”
10 | > | CONTAINS | CODE | DT (125203, DCM, "Acquisition Protocol") | 1 | UC | IFF Nuclear imaging | DCID 3110 “Nuclear Cardiology Protocols”
11 | > | CONTAINS | CODE | DT (125203, DCM, "Acquisition Protocol") | 1 | UC | IFF PET imaging | DCID 3106 “PET Cardiology Protocols”
12 | > | CONTAINS | TEXT | DT (121141, DCM, "Image Type") | 1 | UC | IFF Nuclear or PET imaging | STATIC, DYNAMIC, or GATED. See note.
13 | > | CONTAINS | CODE | DT (RID11248, RADLEX, "Cardiac Gating") | 1 | U |  | DCID 3104 “Cardiac Synchronization Technique”
14 | > | CONTAINS | CODE | DT (125203, DCM, "Acquisition Protocol") | 1 | UC | IFF Contrast echocardiography | DT (P5-B3090, SRT, "Contrast echocardiography")
15 | > | CONTAINS | CODE | DT (113743, DCM, "Patient Orientation") | 1 | U |  | DCID 19 “Patient Orientation”
16 | >> | HAS CONCEPT MOD | CODE | EV (113744, DCM, "Patient Orientation Modifier") | 1 | U |  | DCID 20 “Patient Orientation Modifier”
17 | > | CONTAINS | TEXT | DT (121065, DCM, "Procedure Description") | 1 | U |  |
18 | > | CONTAINS | DATETIME | DT (122701, DCM, "Procedure Time Base") | 1 | U |  | See note.

Content Item Descriptions

Row 12 | Image Type may be copied from the NM Image SOP Instance attribute Image Type (0008,0008) value 3, or from the PET Image SOP Instance attribute Series Type (0054,1000).
Row 18 | The Procedure Time Base is the time from which elapsed times are measured. The Study Time (0008,0030) may include the patient prep period, while this Procedure Time Base is typically established when baseline data collection begins.

### TID 3303 Stress Test Phase Data

The Stress Test Phase Data Template provides a structure for measurements acquired during a single procedure phase.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3303. Stress Test Phase Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  | See note.
2 | > | HAS ACQ CONTEXT | CODE | EV (G-7292, SRT, "Procedure phase") | 1 | MC | XOR row 3 | BCID 3207 “Stress Test Procedure Phases”
3 | > | HAS ACQ CONTEXT | CODE | EV (G-7292, SRT, "Procedure phase") | 1 | MC | IFF Nuclear Imaging; XOR row 2 | DCID 3101 “Cardiac Procedural State Values”
4 | > | HAS ACQ CONTEXT | TEXT | EV (G-7292, SRT, "Procedure phase") | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 3301 “Stress Test Procedure Description” | 1 | MC | IFF protocol changed from initial specification |
6 | > | HAS ACQ CONTEXT | NUM | EV (109055, DCM, "Protocol Stage") | 1 | U |  | UNITS = DT ({stage}, UCUM, "stage")
7 | > | CONTAINS | INCLUDE | DTID 3304 “Stress Test Measurement Group” | 1-n | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3307 “NM/PET Perfusion Measurement Group” | 1 | MC | IFF Nuclear or PET Imaging |
9 | > | CONTAINS | INCLUDE | DTID 3309 “Stress Echo Measurement Group” | 1 | UC | IFF Echocardiography Imaging |

Content Item Descriptions

Row 1 | The Container shall have a specific Content Item Observation DateTime (0040,A032) attribute to indicate the time at which the phase began.

### TID 3304 Stress Test Measurement Group

Each instance of the Stress Test Measurement Group represents a group of data elements acquired at approximately the same instant, and conventionally rendered as row in a tabular presentation. It is typically generated during the Stress exam whenever a time interval elapses (for example, every minute of the phase), when a technician observes data worth capturing, or when measurements exceed a given range.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3304. Stress Test Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  | See note.
2 | > | CONTAINS | NUM | DT (F-031F9, SRT, "Time since start of exam") | 1 | M |  | UNITS = DT (min, UCUM, "min")
3 | > | CONTAINS | NUM | DT (122710, DCM, "Time since start of stage") | 1 | M |  | UNITS = DT (min, UCUM, "min")
4 | > | CONTAINS | NUM | DT (122702, DCM, "Treadmill speed") | 1 | U |  | UNITS = DCID 3212 “Treadmill Speed”
5 | > | CONTAINS | NUM | DT (122703, DCM, "Treadmill gradient") | 1 | U |  | UNITS = EV (%, UCUM, "%")
6 | > | CONTAINS | NUM | DT (122704, DCM, "Ergometer power") | 1 | U |  | UNITS = EV (W, UCUM, "Watts")
7 | > | CONTAINS | NUM | DT (122709, DCM, "Activity workload") | 1 | U |  | UNITS = DT ([MET], UCUM, "METS")
8 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DT (122706, DCM, "Rating of Perceived Exertion") $Method = BCID 3239 “Perceived Exertion Scales”
9 | > | CONTAINS | NUM | DT (122705, DCM, "Pharmacological Stress Agent Dose Rate") | 1 | MC | IFF Pharmacological Stress used | UNITS = EV (ug/kg/min, UCUM, "ug/kg/min")
10 | > | CONTAINS | INCLUDE | DTID 3106 “Drugs/Contrast Administered” | 1 | U |  | See note.
11 | > | CONTAINS | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  | UNITS = EV ({H.B.}/min, UCUM, "BPM")
12 | > | CONTAINS | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
13 | > | CONTAINS | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
14 | > | CONTAINS | NUM | DT (122707, DCM, "Number of Ectopic Beats") | 1 | U |  | UNITS = EV ({beats}, UCUM, "beats")
15 | >> | HAS PROPERTIES | NUM | DT (R-40861, SRT, "Period of collection") | 1 | M |  | UNITS = DT (min, UCUM, "min")
16 | >> | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | U |  | BCID 3234 “Ectopic Beat Morphology”
17 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DT (F-03204, SRT, "ST Elevation") $Units = DT (mV, UCUM, "mV") $TargetSite = DCID 3001 “ECG Leads”
18 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DT (F-38279, SRT, "ST Depression") $Units = DT (mV, UCUM, "mV") $TargetSite = DCID 3001 “ECG Leads”
19 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3228 “ECG Timing Measurements” $Units = DT (ms, UCUM, "ms") $TargetSite = DCID 3001 “ECG Leads” See note.
20 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3227 “QTc Measurements” $Units = DT (ms, UCUM, "ms") $TargetSite = DCID 3001 “ECG Leads” $Equation = DCID 3678 “QT Correction Algorithms” See note.
21 | >> | INFERRED FROM | NUM | DT (2:16000, MDC, "RR Interval for QTc") | 1 | U |  | UNITS = DT (ms, UCUM, "ms") See note.
22 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3229 “ECG Axis Measurements” $Units = DT (deg, UCUM, "deg") See note.
23 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3526 “Blood Gas Saturation” $Units = EV (%, UCUM, "%")
24 | > | CONTAINS | NUM | DT (122708, DCM, "Double Product") | 1 | U |  | UNITS = DT (mm[Hg].{H.B.}/min, UCUM, "mmHg.BPM")
25 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3220 “Stress Symptoms”
26 | > | CONTAINS | CODE | EV (F-00033, SRT, "ECG Finding") | 1-n | U |  | DCID 3230 “ECG Findings”
27 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

Content Item Descriptions

Row 1 | The Container shall have a specific Content Item Observation DateTime (0040,A032) attribute to indicate the time at which the measurements were made.
Row 10 | Included TID 3106 “Drugs/Contrast Administered” allows the recording of test medications other than the Pharmacological Stress Agent identified in TID 3301 “Stress Test Procedure Description”.
Rows 17, 18 | ECG ST elevation/depression is measured in units of mV, but is conventionally reported in units of mm, based on strip recordings with scaling of 100 uV/mm. The display application should render these measurements in units meaningful to the user.
Row 19 | Note that the MDC codes for "per lead" measurements specified in CID 3228 “ECG Timing Measurements” are base codes for post-coordination with lead identifiers conveyed in the Target Site modifier in TID 300 “Measurement”. MDC also defines pre-coordinated codes that include both the measurement and the lead, which may be used in this row.
Row 20 | Note that the MDC code for "QTc interval per lead" specified in CID 3227 “QTc Measurements” is a base code for post-coordination with lead identifiers conveyed in the Target Site modifier in TID 300 “Measurement”. MDC also defines pre-coordinated codes that include both the measurement and the lead, which may be used in this row. Note that TID 300 “Measurement” enables the encoding of a non-standard correction algorithm, either as a local code, or as a TEXT Method Citation (see TID 300 “Measurement” row 12).
Row 21 | R-R interval used for QT correction algorithm
Row 22 | Recommended range for ECG axis measurements is -90° to +270°

### TID 3307 NM/PET Perfusion Measurement Group

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3307. NM/PET Perfusion Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DCID 3108 “NM/PET Procedures”
3 | > | CONTAINS | CODE | EV (F-61FDB, SRT, "Radiopharmaceutical agent") | 1 | M |  | DCID 3111 “Nuclear Cardiology Radiopharmaceuticals”
4 | > | CONTAINS | NUM | EV (123006, DCM, "Radionuclide Total Dose") | 1 | M |  | DCID 3083 “Units of Radioactivity”
5 | > | CONTAINS | DATETIME | EV (123003, DCM, "Radiopharmaceutical Start DateTime") | 1 | M |  |
6 | > | CONTAINS | NUM | DT (122711, DCM, "Exercise duration after stress agent injection") | 1 | U |  | UNITS = DT (min, UCUM, "min")
7 | > | CONTAINS | DATETIME | EV (122712, DCM, "Imaging Start DateTime") | 1 | M |  |
8 | > | CONTAINS | CODE | EV (122713, DCM, "Attenuation correction") | 1 | U |  | BCID 3112 “Attenuation Correction”
9 | >> | HAS PROPERTIES | CODE | EV (111001, DCM, "Algorithm Name") | 1 | U |  | BCID 3117 “Attenuation Correction Methods”
10 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | BCID 3113 “Types of Perfusion Defects”
11 | >> | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | BCID 3717 “Myocardial Wall Segments”
12 | >> | HAS PROPERTIES | CODE | EV (112025, DCM, "Size Descriptor") | 1 | M |  | BCID 252 “S-M-L Size Descriptor”
13 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | M |  | BCID 3716 “Severity”
14 | > | CONTAINS | CODE | EV (F-02220, SRT, "Left Ventricular Function") | 1 | U |  | BCID 3119 “LV Function”
15 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | BCID 3716 “Severity”
16 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (10230-1, LN, "LV Ejection Fraction") $Units = EV (%, UCUM, "%") $Derivation = DT (R-41D2D, SRT, "Calculated")
17 | > | CONTAINS | CODE | EV (F-02236, SRT, "Left Ventricular Size") | 1 | U |  | BCID 3122 “Ventricular Enlargement”
18 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8821-1, LN, "Left Ventricular ED Volume") $Units = EV (ml, UCUM, "ml")
19 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8823-7, LN, "Left Ventricular ES Volume") $Units = EV (ml, UCUM, "ml")
20 | > | CONTAINS | INCLUDE | DTID 5204 “Wall Motion Analysis” | 1 | U |  | $Procedure = DCID 3108 “NM/PET Procedures”

### TID 3309 Stress Echo Measurement Group

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3309. Stress Echo Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (P5-B3000, SRT, "Echocardiography")
3 | > | CONTAINS | INCLUDE | DTID 5203 “Echo Measurement” | 1-n | U |  | $Measurement = DCID 12200 “Echocardiography Left Ventricle” $Method = CID 12227 “Echocardiography Measurement Method”
4 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | IF Row 3 measurement concept is in CID 12222 “Orifice Flow Properties” | EV (T-32600, SRT, "Left Ventricle")
5 | > | CONTAINS | INCLUDE | DTID 5203 “Echo Measurement” | 1-n | U |  | $Measurement = DCID 12211 “Echocardiography Aortic Valve” $Method = CID 12227 “Echocardiography Measurement Method”
6 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | IF Row 5 measurement concept is in CID 12222 “Orifice Flow Properties” | EV (T-35400, SRT, "Aortic Valve")
7 | > | CONTAINS | INCLUDE | DTID 5203 “Echo Measurement” | 1-n | U |  | $Measurement = DCID 12207 “Echocardiography Mitral Valve” $Method = CID 12227 “Echocardiography Measurement Method”
8 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | IF Row 7 measurement concept is in CID 12222 “Orifice Flow Properties” | EV (T-35300, SRT, "Mitral Valve")
9 | > | CONTAINS | INCLUDE | DTID 5203 “Echo Measurement” | 1-n | U |  | $Measurement = DCID 12208 “Echocardiography Tricuspid Valve” $Method = CID 12227 “Echocardiography Measurement Method”
10 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | IF Row 9 measurement concept is in CID 12222 “Orifice Flow Properties” | EV (T-35100, SRT, "Tricuspid Valve")
11 | > | CONTAINS | INCLUDE | DTID 5204 “Wall Motion Analysis” | 1 | U |  | $Procedure = DT (P5-B3121, SRT, "Echocardiography for Determining Ventricular Contraction")

Content Item Descriptions

Rows 3-10 | These invocations of TID 5203 “Echo Measurement” do not include an inherited Findings Site concept, for example as in the invocations of TID 5203 “Echo Measurement” from TID 5202 “Echo Section”. Echo measurements that do not have the associated Finding Site pre-coordinated in the measurement concept (i.e., the orifice flow measurements of CID 12222 “Orifice Flow Properties”), shall have the Finding Site explicitly post-coordinated with a Concept Modifier (Rows 4, 6, 8, and 10). This Template does not include the concept of an Image Library, for example as used in TID 5200 “Echocardiography Procedure Report”. Image Content Items in the Echo Measurement Template shall be included with by-value relationships, not with by-reference relationships.

### TID 3311 Stress Test Summary

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3311. Stress Test Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121111, DCM, "Summary") | 1 | U |  |
3 | > | CONTAINS | INCLUDE | DTID 3312 “Physiological Summary” | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 3313 “Stress ECG Summary” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 3317 “Stress Imaging Summary” | 1 | U |  |
6 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (D3-13025, SRT, "Exercise-induced angina") See note.
7 | >> |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | M |  |
8 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3220 “Stress Symptoms”
9 | > | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1 | U |  |
10 | > | CONTAINS | CODE | DT (G-0180, SRT, "Reason for stopping test") | 1 | U |  | DCID 3221 “Stress Test Termination Reasons”
11 | > | CONTAINS | NUM | DT (122715, DCM, "Pharmacological Stress Agent Dose") | 1 | U |  | DT (mg/kg, UCUM, "mg/kg")

Content Item Descriptions

Rows 6-7 | These rows allow an explicit finding of presence or absence of exercise-induced angina through the TID 1350 “Negation Modifier, Presence of Finding” Concept Modifier "Presence of property"

### TID 3312 Physiological Summary

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3312. Physiological Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DT (40443-4, LN, "Resting Heart Rate") | 1 | M |  | UNITS = DT ({H.B.}/min, UCUM, "BPM")
2 |  |  | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | M |  | UNITS = DCID 3500 “Pressure Units”
3 | > | HAS CONCEPT MOD | CODE | EV (109054, DCM, "Patient State") | 1 | M |  | DT (F-01604, SRT, "Resting State")
4 |  |  | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | M |  | UNITS = DCID 3500 “Pressure Units”
5 | > | HAS CONCEPT MOD | CODE | EV (109054, DCM, "Patient State") | 1 | M |  | DT (F-01604, SRT, "Resting State")
6 |  |  | NUM | DT (F-04F92, SRT, "Target HR") | 1 | M |  | UNITS = DT ({H.B.}/min, UCUM, "BPM")
7 |  |  | NUM | DT (F-04FA6, SRT, "Maximum HR Achieved") | 1 | M |  | UNITS = DT ({H.B.}/min, UCUM, "BPM")
8 |  |  | NUM | DT (F-04FA6, SRT, "Maximum HR Achieved") | 1 | M |  | UNITS = EV (%, UCUM, "%")
9 | > | HAS CONCEPT MOD | CODE | EV (121425, DCM, "Index") | 1 | M |  | DT (F-04F92, SRT, "Target HR")
10 |  |  | NUM | DT (122716, DCM, "Maximum Power Output Achieved") | 1 | U |  | UNITS = DT (W, UCUM, "Watts")
11 |  |  | NUM | DT (122717, DCM, "Peak activity workload") | 1 | U |  | UNITS = DT ([MET], UCUM, "METS")
12 |  |  | CODE | DT (F-04F9F, SRT, "HR Response") | 1 | U |  | DCID 3210 “Speed of Response”
13 |  |  | NUM | DT (F-00E11, SRT, "Maximum systolic blood pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
14 |  |  | NUM | DT (F-00E21, SRT, "Maximum diastolic blood pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
15 |  |  | CODE | DT (F-04F74, SRT, "BP Response") | 1 | U |  | DCID 3210 “Speed of Response”
16 |  |  | NUM | DT (122718, DCM, "Peak Double Product") | 1 | U |  | UNITS = DT (mm[Hg].{H.B.}/min, UCUM, "mmHg.BPM")
17 |  |  | NUM | DT (F-031F8, SRT, "Total Exercise duration") | 1 | U |  | UNITS = DT (min, UCUM, "min")
18 |  |  | NUM | DT (F-031F7, SRT, "Total test duration") | 1 | U |  | UNITS = DT (min, UCUM, "min")
19 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DT (F-04FCC, SRT, "Functional capacity") See note.
20 |  |  | TEXT | DT (F-04FCC, SRT, "Functional capacity") | 1 | U |  |
21 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (122760, DCM, "Stress test score") $Method = BCID 3238 “Stress Scoring Scales”
22 |  |  | NUM | DT (F-04FCA, SRT, "Heart rate recovery time") | 1 | U |  | UNITS = DT (s, UCUM, "s")
23 |  |  | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | BCID 3213 “Stress Hemodynamic Findings”
24 |  |  | CODE | EV (F-00F4E, SRT, "Cardiovascular event risk") | 1 | U |  | BCID 3118 “Level of Risk”

Content Item Descriptions

Row 22 | Numerical scoring of a patient's functional capacity shall include the range of the scoring system in the Units of Measurement (see Section 7.2.2), and may include a coded identifier for the scoring system in the Method concept modifier of TID 300 “Measurement”.

### TID 3313 Stress ECG Summary

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3313. Stress ECG Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (F-00033, SRT, "ECG Finding") | 1 | U |  | Device Generated Test Summary
2 | > | HAS OBS CONTEXT | CODE | EV (121005, DCM, "Observer Type") | 1 | M |  | EV (121007, DCM, "Device")
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1004 “Device Observer Identifying Attributes” | 1 | M |  |
4 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (F-03204, SRT, "ST Elevation") $Units = DT (mV, UCUM, "mV") $Derivation = EV (G-A437, SRT, "Maximum") $TargetSite = DCID 3001 “ECG Leads”
5 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (F-38279, SRT, "ST Depression") $Units = DT (mV, UCUM, "mV") $Derivation = EV (G-A437, SRT, "Maximum") $TargetSite = DCID 3001 “ECG Leads”
6 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (F-38287, SRT, "T wave alternans") $Units = DT (uV, UCUM, "uV") $Derivation = EV (G-A437, SRT, "Maximum") $TargetSite = DCID 3001 “ECG Leads”
7 |  |  | CODE | EV (F-38035, SRT, "ST Segment Finding") | 1 | U |  | BCID 3231 “ST Segment Findings”
8 | > | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | U |  | BCID 3232 “ST Segment Location”
9 | > | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | U |  | BCID 3233 “ST Segment Morphology”
10 |  |  | NUM | DT (122707, DCM, "Number of Ectopic Beats") | 1 | U |  | UNITS = EV ({beats}, UCUM, "beats")
11 | > | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | U |  | BCID 3234 “Ectopic Beat Morphology”
12 |  |  | CODE | DT (8884-9, LN, "Cardiac Rhythm") | 1-2 | U |  | BCID 3415 “Cardiac Rhythms” See note.
13 | > | HAS CONCEPT MOD | CODE | EV (109054, DCM, "Patient State") | 1 | M |  | DCID 3102 “Rest-Stress”
14 |  |  | CODE | EV (F-00033, SRT, "ECG Finding") | 1-n | U |  | BCID 3230 “ECG Findings”
15 | > | HAS CONCEPT MOD | CODE | EV (109054, DCM, "Patient State") | 1 | U |  | BCID 3262 “ECG Patient State Values”

Content Item Descriptions

Rows 4-14 | Each observation (measurement or finding) may have a specific Content Item Observation DateTime attribute to indicate the time in the procedure at which the observation was made (e.g., time of maximum heart rate, or time of occurrence of an arrhythmia).
Row 12-13 | This Concept and the associated Concept Modifier may be instantiated twice, once for resting state measurements, once for stress.

### TID 3317 Stress Imaging Summary

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3317. Stress Imaging Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DT (122739, DCM, "Overall study quality") | 1 | M |  | BCID 3114 “Study Quality”
2 |  |  | CODE | DT (113010, DCM, "Quality Issue") | 1 | U |  | BCID 3115 “Stress Imaging Quality Issues”
3 |  |  | CODE | EV (121071, DCM, "Finding") | 1 | U |  | BCID 3116 “NM Extracardiac Findings”
4 |  |  | INCLUDE | DTID 300 “Measurement” | 1-2 | U |  | $Measurement = DT (F-04F76, SRT, "Perfusion defect extent") $ModType = EV (109054, DCM, "Patient State") $ModValue = DCID 3102 “Rest-Stress” $Units = EV (%, UCUM, "%") See note.
5 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DT (F-04FCD, SRT, "Stress ischemia extent") $Units = EV (%, UCUM, "%")
6 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DT (122762, DCM, "Number of diseased vessel territories") $Units = EV ({territories}, UCUM, "territories")
7 |  |  | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (D3-13040, SRT, "Coronary artery disease")
8 | > | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | M |  | BCID 3016 “Major Coronary Arteries”
9 |  |  | CODE | EV (F-0238D, SRT, "Myocardial perfusion") | 1-2 | U |  | BCID 3120 “Perfusion Findings”
10 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | BCID 3463 “Ventricle Identification”
11 | > | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | U |  | BCID 3121 “Perfusion Morphology”
12 |  |  | CODE | EV (121071, DCM, "Finding") | 1 | U |  | DT (D4-31124, SRT, "Transient cavitary dilatation")
13 | > |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
14 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DT (F-04FB4, SRT, "Transient cavitary dilatation ratio") $Units = EV ({ratio}, UCUM, "ratio")
15 |  |  | INCLUDE | DTID 300 “Measurement” | 1-2 | U |  | $Measurement = EV (10230-1, LN, "LV Ejection Fraction") $ModType = EV (109054, DCM, "Patient State") $ModValue = DCID 3102 “Rest-Stress” $Units = EV (%, UCUM, "%") See note.

Content Item Descriptions

Row 4 | This row may be instantiated twice, once for resting state measurements, once for stress.
Row 15 | The LVEF code specified in this row is defined in LOINC with method "imaging". LVEF measurement by ultrasound may also be encoded elsewhere in the Content Tree (e.g., in TID 3309 “Stress Echo Measurement Group”) with LOINC code 18043-0, which has method "ultrasound". It is recommended that such findings from the per-phase measurements be summarized here with the generic "LVEF by Imaging" concept code.

### TID 3318 Comparison to Prior Stress Exam

This Template describes changes in findings from a prior stress exam. Comparison is to only one prior exam, even though the generic concept name for the Template uses the plural "exams".

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3318. Comparison to Prior Stress Exam

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111424, DCM, "Comparison to previous exams") | 1 | M |  |
2 | > | CONTAINS | CODE | DT (121058, DCM, "Procedure Reported") | 1 | U |  | DCID 3200 “Stress Test Procedure”
3 | >> | HAS PROPERTIES | DATETIME | EV (122146, DCM, "Procedure DateTime") | 1 | U |  |
4 | >> | HAS PROPERTIES | UIDREF | EV (121018, DCM, "Procedure Study Instance UID") | 1 | U |  |
5 | >> | HAS PROPERTIES | COMPOSITE | EV (122075, DCM, "Prior report for current patient") | 1-n | U |  |
6 | > | CONTAINS | CODE | DT (F-03D1D, SRT, "Exercise tolerance") | 1 | U |  | BCID 3236 “Tolerance Comparison Findings”
7 | > | CONTAINS | CODE | DT (F-0238D, SRT, "Myocardial Perfusion") | 1 | U |  | BCID 3235 “Perfusion Comparison Findings”
8 | >> | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1-n | U |  | BCID 3215 “Perfusion Finding Method”
9 | > | CONTAINS | CODE | DT (F-02225, SRT, "LV Wall motion") | 1-n | U |  | BCID 3237 “Wall Motion Comparison Findings”
10 | >> | HAS CONCEPT MOD | CODE | EV (109054, DCM, "Patient State") | 1 | M |  | DCID 3102 “Rest-Stress”
11 | > | CONTAINS | CODE | EV (111424, DCM, "Comparison to previous exams") | 1 | U |  | BCID 3217 “Comparison Finding”
12 | > | CONTAINS | NUM | DT (122768, DCM, "Difference in Ejection Fraction") | 1 | U |  | UNITS = EV (%, UCUM, "%")
13 | > | CONTAINS | NUM | DT (122769, DCM, "Difference in ED LV Volume") | 1 | U |  | UNITS = EV (ml, UCUM, "ml")
14 | > | CONTAINS | NUM | DT (122769, DCM, "Difference in ED LV Volume") | 1 | U |  | UNITS = EV (ml/m2, UCUM, "ml/m2")
15 | >> | HAS CONCEPT MOD | CODE | EV (121425, DCM, "Index") | 1 | M |  | DT (8277-6, LN, "BSA")

### TID 3320 Conclusions and Recommendations

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 3320. Conclusions and Recommendations

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121076, DCM, "Conclusions") | 1 | MC | IF Completion Flag (0040,A491) = COMPLETE |
2 | > | CONTAINS | TEXT | EV (121077, DCM, "Conclusion") | 1 | U |  |
3 | > | CONTAINS | CODE | EV (F-00033, SRT, "ECG Finding") | 1 | M |  | DCID 3208 “Summary Codes Exercise ECG”
4 | > | CONTAINS | CODE | EV (F-01969, SRT, "Imaging Finding") | 1 | M |  | DCID 3209 “Summary Codes Stress Imaging”
5 |  |  | CONTAINER | EV (121074, DCM, "Recommendations") | 1 | U |  |
6 | > | CONTAINS | TEXT | EV (121075, DCM, "Recommendation") | 1 | U |  |

## Hemodynamics Report Templates

The Templates that comprise the Hemodynamic Report are interconnected as shown in Figure A-6.

Figure A-6. Hemodynamic Report Template Hierarchy

Figure A-6 shows only the use of Templates specific to the Hemodynamic Report; it does not show common structural Templates such as TID 300 “Measurement”.

### Note

Figure A-6 shows the relationship of Templates; it does not show the structural hierarchy of Content Items in the IOD. See Figure L-1 “Hemodynamics Report Structure” in PS3.17 .

### TID 3500 Hemodynamics Report

The Hemodynamic Report Template is the root structure for the representation of measurements acquired during a procedure in a cardiac catheterization lab.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 3500. Hemodynamics Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122120, DCM, "Hemodynamics Report") | 1 | M |  | Root node
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
3 | > |  | INCLUDE | DTID 3601 “Procedure Context” | 1 | M |  |
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | M |  |
5 | > | HAS ACQ CONTEXT | INCLUDE | DTID 3603 “Procedure Environmental Characteristics” | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3501 “Hemodynamics Measurement Group” | 1-n | M |  |
7 | > | CONTAINS | INCLUDE | DTID 3570 “Summary, Hemodynamics” | 1 | U |  |

### TID 3501 Hemodynamics Measurement Group

The Hemodynamic Measurement Group Template provides a structure for measurements acquired during a single procedure phase in a cardiac catheterization lab.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3501. Hemodynamics Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS ACQ CONTEXT | CODE | EV (G-72BB, SRT, "Catheterization Procedure Phase") | 1 | M |  | DCID 3651 “Hemodynamic Measurement Phase”
3 | > | HAS ACQ CONTEXT | INCLUDE | DTID 3520 “Hemodynamic Clinical Context” | 1 | U |  |
4 | > | HAS ACQ CONTEXT | TEXT | EV (121124, DCM, "Procedure Action ID") | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 3510 “Vital Signs” | 1-n | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3504 “Arterial Pressure Measurement” | 1-n | U |  |
7 | > | CONTAINS | INCLUDE | DTID 3505 “Atrial Pressure Measurement” | 1-n | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3506 “Venous Pressure Measurement” | 1-n | U |  |
9 | > | CONTAINS | INCLUDE | DTID 3507 “Ventricular Pressure Measurement” | 1-n | U |  |
10 | > | CONTAINS | INCLUDE | DTID 3508 “Gradient Measurement” | 1-n | U |  |
11 | > | CONTAINS | INCLUDE | DTID 3509 “Blood Velocity Measurement” | 1-n | U |  |
12 | > | CONTAINS | INCLUDE | DTID 3515 “Cardiac Output Measurement by Indicator Dilution” | 1-n | U |  |
13 | > | CONTAINS | INCLUDE | DTID 3516 “Blood Lab Measurements” | 1-n | U |  |
14 | > | CONTAINS | INCLUDE | DTID 3560 “Derived Hemodynamic Measurements” | 1-n | U |  |
15 | > | CONTAINS | INCLUDE | DTID 3714 “ECG Lead Measurements” | 1-n | U |  |

Content Item Descriptions

Row 4 | Procedure Action ID allows linkage between the hemodynamic measurements recorded in this Template and a procedure step or phase recorded in the Procedure Log, e.g., using TID 3100 “Procedure Action”.

### TID 3504 Arterial Pressure Measurement

The Arterial Pressure Measurement Template consists of a CONTAINER, with an acquisition context of the measurement anatomic site, and containing systolic, diastolic, and mean measurements. This implies that the name of the measurement is effectively post-coordinated from the measurement name, the Hemodynamic Measurement Group container (procedure phase) name, and the acquisition context (finding site) value.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3504. Arterial Pressure Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (P2-36102, SRT, "Arterial pressure measurements") | 1 | M |  |
2 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | M |  | $LocationName = EV (G-C0E3, SRT, "Finding Site") $LocationValue = DCID 3606 “Arterial Source Locations”
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (8480-6, LN, "Intravascular arterial Systolic pressure") $Units = DCID 3500 “Pressure Units”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (8462-4, LN, "Intravascular arterial Diastolic pressure") $Units = DCID 3500 “Pressure Units”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (8478-0, LN, "Intravascular arterial mean pressure") $Units = DCID 3500 “Pressure Units”
6 | > | CONTAINS | INCLUDE | DTID 3550 “Pressure Waveform Measurements” | 1-n | U |  |

### TID 3505 Atrial Pressure Measurement

The Atrial Pressure Measurement Template consists of a CONTAINER, with an acquisition context of the measurement anatomic site, and containing a-wave, v-wave, and mean measurements. This implies that the name of the measurement is effectively post-coordinated from the measurement name, the Hemodynamic Measurement Group container (procedure phase) name, and the acquisition context (finding site) value.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3505. Atrial Pressure Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122121, DCM, "Atrial pressure measurements") | 1 | M |  |
2 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | M |  | $LocationName = EV (G-C0E3, SRT, "Finding Site") $LocationValue = DCID 3608 “Atrial Source Locations”
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (109016, DCM, "A-wave peak pressure") $Units = DCID 3500 “Pressure Units”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (109034, DCM, "V-wave peak pressure") $Units = DCID 3500 “Pressure Units”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (F-31150, SRT, "Mean blood pressure") $Units = DCID 3500 “Pressure Units”
6 | > | CONTAINS | INCLUDE | DTID 3550 “Pressure Waveform Measurements” | 1-n | U |  |

### TID 3506 Venous Pressure Measurement

The Venous Pressure Measurement Template consists of a CONTAINER, with an acquisition context of the measurement anatomic site, and containing a mean measurement. This implies that the name of the measurement is effectively post-coordinated from the measurement name, the Hemodynamic Measurement Group container (procedure phase) name, and the acquisition context (finding site) value.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3506. Venous Pressure Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (P2-36110, SRT, "Venous pressure measurements") | 1 | M |  |
2 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | M |  | $LocationName = EV (G-C0E3, SRT, "Finding Site") $LocationValue = DCID 3607 “Venous Source Locations”
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (F-31150, SRT, "Mean blood pressure") $Units = DCID 3500 “Pressure Units”
4 | > | CONTAINS | INCLUDE | DTID 3550 “Pressure Waveform Measurements” | 1-n | U |  |

### TID 3507 Ventricular Pressure Measurement

The Ventricular Pressure Measurement Template consists of a CONTAINER, with an acquisition context of the measurement anatomic site, and containing systolic and end-diastolic measurements. This implies that the name of the measurement is effectively post-coordinated from the measurement name, the Hemodynamic Measurement Group container (procedure phase) name, and the acquisition context (finding site) value.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3507. Ventricular Pressure Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122122, DCM, "Ventricular pressure measurements") | 1 | M |  |
2 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | M |  | $LocationName = EV (G-C0E3, SRT, "Finding Site") $LocationValue = DCID 3609 “Ventricular Source Locations”
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | IFF $LocationValue selected in row 2 is (T-32600, SRT, "Left Ventricle") or subsite thereof | $Measurement = EV (F-03E0D, SRT, "Left Ventricular Systolic blood pressure") $Units = DCID 3500 “Pressure Units”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | IFF $LocationValue selected in row 2 is (T-32600, SRT, "Left Ventricle") or subsite thereof | $Measurement = EV (F-03E0E, SRT, "Left Ventricular End Diastolic pressure") $Units = DCID 3500 “Pressure Units”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | IFF $LocationValue selected in row 2 is (T-32500, SRT, "Right Ventricle") or subsite thereof | $Measurement = EV (F-03DFE, SRT, "Right Ventricular Systolic blood pressure") $Units = DCID 3500 “Pressure Units”
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | IFF $LocationValue selected in row 2 is (T-32500, SRT, "Right Ventricle") or subsite thereof | $Measurement = EV (F-03E02, SRT, "Right Ventricular End Diastolic pressure") $Units = DCID 3500 “Pressure Units”
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | IFF $LocationValue selected in row 2 is (D4-31120, SRT, "Common Ventricle") | $Measurement = EV (122194, DCM, "Ventricular Systolic blood pressure") $Units = DCID 3500 “Pressure Units”
8 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | IFF $LocationValue selected in row 2 is (D4-31120, SRT, "Common Ventricle") | $Measurement = EV (122191, DCM, "Ventricular End Diastolic pressure") $Units = DCID 3500 “Pressure Units”
9 | > | CONTAINS | INCLUDE | DTID 3550 “Pressure Waveform Measurements” | 1-n | U |  |

### TID 3508 Gradient Measurement

The Gradient Measurement Template consists of a CONTAINER, with an acquisition context of the measurement anatomic site, and containing the gradient measurement. This implies that the name of the measurement is effectively post-coordinated from the measurement name, the Hemodynamic Measurement Group container (procedure phase) name, and the acquisition context (finding site) value.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3508. Gradient Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122123, DCM, "Gradient assessment") | 1 | M |  |
2 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | MC | XOR with Rows 3 & 4 IFF single location is appropriate | $LocationName = EV (G-C0E3, SRT, "Finding Site") $LocationValue = DCID 3610 “Gradient Source Locations”
3 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | MC | XOR with Row 2 IFF a dual location is appropriate | $LocationName = EV (121116, DCM, "Proximal Finding Site") $LocationValue = DCID 3630 “Cardiovascular Anatomic Locations”
4 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | MC | XOR with Row 2 IFF a dual location is appropriate | $LocationName = EV (121117, DCM, "Distal Finding Site") $LocationValue = DCID 3630 “Cardiovascular Anatomic Locations”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = EV (F-023F7, SRT, "Pressure Gradient") $Units = DCID 3500 “Pressure Units” $Derivation = DCID 3627 “Measurement Type”
6 | > | CONTAINS | INCLUDE | DTID 3550 “Pressure Waveform Measurements” | 1-n | U |  |

Content Item Descriptions

Row 5 | Is used to encode the gradient measurements. Uses TID 300 “Measurement” for the Content Item structure of the measurements. That Template allows an INFERRED FROM relationship to other numeric measurements used in the computation or derivation of the current measurement. In the case of a gradient measurement, the pressure measurements at the distal and proximal locations may thus be explicitly conveyed.

### TID 3509 Blood Velocity Measurement

The Blood Velocity Measurement Template consists of a CONTAINER, with an acquisition context of the measurement anatomic site, and containing primary (not derived) velocity measurements, e.g., from a Doppler catheter. Derived velocity measurements may be recorded using TID 3560 “Derived Hemodynamic Measurements”.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3509. Blood Velocity Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122124, DCM, "Blood velocity measurements") | 1 | M |  |
2 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | M |  | $LocationName = EV (G-C0E9, SRT, "Procedure site") $LocationValue = BCID 3606 “Arterial Source Locations”
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = DCID 3612 “Blood Velocity Measurements” $Units = EV (mm/s, UCUM, "mm/s")
4 | > | CONTAINS | INCLUDE | DTID 3550 “Pressure Waveform Measurements” | 1-n | U |  |

### TID 3510 Vital Signs

The Vital Signs Template consists of a CONTAINER containing the various vital signs measurements. These measurements may be acquired automatically from patient monitoring equipment, or may be entered based on manual measurements.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3510. Vital Signs

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (8716-3, LN, "Vital Signs") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-008EC, SRT, "Systolic blood pressure") $Units = DCID 3500 “Pressure Units” $Method = BCID 3560 “Blood Pressure Methods”
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-008ED, SRT, "Diastolic blood pressure") $Units = DCID 3500 “Pressure Units”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8867-4, LN, "Heart rate") $Units = EV ({H.B.}/min, UCUM, "BPM")
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8310-5, LN, "Body temperature") $Units = EV (Cel, UCUM, "C")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3526 “Blood Gas Saturation” $Units = EV (%, UCUM, "%")
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-21000, SRT, "Respiration rate") $Units = EV (/min, UCUM, "breaths/min")
8 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122195, DCM, "Pulse Strength") $Units = DT ({0:4}, UCUM, "range 0:4")
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-009EA, SRT, "Pain Score") $Units = DT ({1:10}, UCUM, "range 1:10")
10 | > | CONTAINS | CODE | DT (8884-9, LN, "Cardiac Rhythm") | 1 | U |  | BCID 3415 “Cardiac Rhythms”
11 | > | CONTAINS | CODE | DT (9304-7, LN, "Respiration Rhythm") | 1 | U |  | BCID 3416 “Respiration Rhythms”

### TID 3515 Cardiac Output Measurement by Indicator Dilution

The Cardiac Output Measurement by Indicator Dilution Template consists of a CONTAINER containing the measurement and significant parameters of the technical method. If the measurement is based on a DICOM Hemodynamic Waveform SOP Instance, that object may also be referenced.

### Note

Fick Cardiac Output is encoded in TID 3560 “Derived Hemodynamic Measurements”.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3515. Cardiac Output Measurement By Indicator Dilution

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (P2-34201, SRT, "Cardiac Output measurement") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (8737-9, LN, "Cardiac Output by Indicator Dilution") $Units = EV (l/min, UCUM, "l/min") $Method = DCID 3628 “Cardiac Output Methods” $WavePurpose = DT (121112, DCM, "Source of Measurement")
3 | > | HAS ACQ CONTEXT | NUM | EV (122319, DCM, "Catheter Size") | 1 | MC | IFF Row 2 indicates a thermal method | UNITS = DCID 3510 “Catheter Size Units”
4 | > | HAS ACQ CONTEXT | NUM | EV (122320, DCM, "Injectate Temperature") | 1 | MC | IFF Row 2 indicates a thermal method | UNITS = EV (Cel, UCUM, "C")
5 | > | HAS ACQ CONTEXT | NUM | EV (122321, DCM, "Injectate Volume") | 1 | M |  | UNITS = DT (ml, UCUM, "ml")
6 | > | HAS ACQ CONTEXT | NUM | EV (122322, DCM, "Calibration Factor") | 1 | M |  | UNITS = DT (1, UCUM, "no units")

### TID 3516 Blood Lab Measurements

The Blood Lab Measurements Template provides for the recording of measurements made on blood samples obtained during a catheterization procedure. The type and anatomic source of the blood is recorded as acquisition context. The results from the blood chemistry measurement system are quoted; the measurement names may be pre-coordinated with the type or source of the blood, or generic measurement names may be reported. In the latter case, the full measurement concept name may be effectively post-coordinated using the recorded acquisition context.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3516. Blood Lab Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122125, DCM, "Blood lab measurements") | 1 | M |  |
2 | > | HAS ACQ CONTEXT | CODE | EV (R-00254, SRT, "Specimen Type") | 1 | M |  | DCID 3520 “Blood Source Type”
3 | > | HAS ACQ CONTEXT | CODE | EV (G-C0E9, SRT, "Procedure site") | 1 | M |  | BCID 3630 “Cardiovascular Anatomic Locations”
4 | > |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (718-7, LN, "Hemoglobin") $Units = DT (g/dl, UCUM, "g/dl")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3524 “Blood Gas Pressures” $Units = DCID 3500 “Pressure Units”
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3525 “Blood Gas Content” $Units = DT (ml/dl, UCUM, "ml/dl")
8 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3528 “Blood pH” $Units = EV ([pH], UCUM, "pH")
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3526 “Blood Gas Saturation” $Units = EV (%, UCUM, "%")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3527 “Blood Base Excess” $Units = DT (meq/dl, UCUM, "meq/dl")
11 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122183, DCM, "Blood temperature") $Units = EV (Cel, UCUM, "C")

### TID 3520 Hemodynamic Clinical Context

The Clinical Context Template allows the recording of information about the patient's clinical state that may affect interpretation of the hemodynamic measurements. This Template includes several coded conditions (e.g., complications, drugs, physiological challenges, etc.), each of which may have a Concept Modifier of "Relative Time". This Modifier indicates the temporal relationship of the hemodynamic measurements to the conditions recorded in this Template. E.g., the Content Item structure:

[CONTAINER] "Findings"

>HAS ACQ CONTEXT "Cath Procedure Phase" "Post-intervention phase"

>HAS ACQ CONTEXT [CONTAINER] "Clinical Context"

>>CONTAINS [CODE] "Complication" "Arrhythmia"

>>>HAS CONCEPT MOD [CODE] "Relative Time" "After"

>CONTAINS [CONTAINER] "Arterial Measurements"…

conveys the semantics that this group of measurements for the post-intervention phase of a cath procedure was made after an occurrence of arrhythmia in the patient.

In the absence of a Relative Time modifier, the acquisition context is during the identified event or state.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3520. Hemodynamic Clinical Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122127, DCM, "Clinical Context") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (109054, DCM, "Patient State") | 1-n | U |  | BCID 3602 “Hemodynamic Patient State”
3 | > | CONTAINS | TEXT | EV (109054, DCM, "Patient State") | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (DD-60002, SRT, "Complication of Procedure") $ConditionValue = DCID 3413 “Adverse Outcomes”
5 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (122086, DCM, "Contrast administered") $ConditionValue = BCID 12 “Radiographic Contrast Agent”
6 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (122083, DCM, "Drug administered") $ConditionValue = BCID 10 “Interventional Drug”
7 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (109059, DCM, "Physiological challenges") $ConditionValue = BCID 3271 “Hemodynamic Physiological Challenges”
8 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (122138, DCM, "Circulatory Support") $ConditionValue = DCID 3553 “Circulatory Support”
9 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (P2-2290D, SRT, "Controlled ventilation") $ConditionValue = DCID 3554 “Ventilation”
10 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (P2-35000, SRT, "Cardiac Pacing") $ConditionValue = BCID 3555 “Pacing”
11 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (P0-00000, SRT, "Procedure") $ConditionValue = BCID 3405 “Procedure Action Values”

### TID 3521 Relative Time

The Relative Time Template is invoked by 3520 Hemodynamic Acquisition Context Template. It specifies an applicable clinical context condition and the time of the current measurements relative to that condition.

Table TID 3521. Parameters

Parameter Name | Parameter Usage
--- | ---
$ConditionName | Coded term for Concept Name of condition
$ConditionValue | Context Group for condition values

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 3521. Relative Time

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | $ConditionName | 1 | M |  | $ConditionValue
2 | > | HAS CONCEPT MOD | CODE | EV (G-D709, SRT, "Relative time") | 1 | U |  | BCID 3600 “Relative Times”

### TID 3530 Hemodynamic Acquisition Context

The Hemodynamic Acquisition Context Template is invoked by the various hemodynamic measurement Templates.

Table TID 3530. Parameters

Parameter Name | Parameter Usage
--- | ---
$LocationName | Coded term for Concept Name of measurement location
$LocationValue | Context Group for appropriate anatomic locations

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3530. Hemodynamic Acquisition Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | CODE | $LocationName | 1 | M |  | $LocationValue
2 | > | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | BCID 3019 “Cardiovascular Anatomic Location Modifiers”
3 |  | HAS ACQ CONTEXT | CODE | EV (G-C036, SRT, "Measurement Method ") | 1 | U |  | BCID 3241 “Hemodynamic Measurement Techniques”
4 |  | HAS ACQ CONTEXT | WAVEFORM | DT (121112, DCM, "Source of Measurement") | 1 | UC | XOR Row 5 |
5 |  | HAS ACQ CONTEXT | TCOORD | DT (121112, DCM, "Source of Measurement") | 1 | UC | XOR Row 4 |
6 | > | SELECTED FROM | WAVEFORM |  | 1 | M |  |

### TID 3550 Pressure Waveform Measurements

The Pressure Waveform Measurements Template is invoked by the various hemodynamic measurement Templates for recording general measurements made in conjunction with the specific required measurements for that anatomic location.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3550. Pressure Waveform Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3611 “Pressure Measurements” $Units = DCID 3500 “Pressure Units”
2 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3612 “Blood Velocity Measurements” $Units = EV (mm/s, UCUM, "mm/s")
3 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3613 “Hemodynamic Time Measurements” $Units = DT (ms, UCUM, "ms")
4 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32100, SRT, "Cardiac Output") $Units = EV (l/min, UCUM, "l/min")
5 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $Units = DT (ml, UCUM, "ml")
6 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (10230-1, LN, "LV Ejection Fraction") $Units = EV (%, UCUM, "%")
7 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8867-4, LN, "Heart rate") $Units = DT ({H.B.}/min, UCUM, "BPM")
8 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-21000, SRT, "Respiration rate") $Units = DT (/min, UCUM, "breaths/min")
9 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (109025, DCM, "Max dp/dt") $Units = DT (mm[Hg]/s, UCUM, "mmHg/s")
10 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (109026, DCM, "Max neg dp/dt") $Units = DT (mm[Hg]/s, UCUM, "mmHg/s")
11 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122190, DCM, "Max dp/dt/P") $Units = DT (/s, UCUM, "/s")
12 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122192, DCM, "Indicator appearance time") $Units = DT (s, UCUM, "s")
13 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122193, DCM, "Maximum pressure acceleration") $Units = DT (mm[Hg]/s2, UCUM, "mmHg/s/s")
14 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3617 “Valve Flows” $Units = DT (ml/min, UCUM, "ml/min")

### TID 3560 Derived Hemodynamic Measurements

The Derived Hemodynamic Measurements Template consists of a CONTAINER containing measurements derived from one or more other measurements. These measurements are associated with a particular procedure phase, but not necessarily with a particular anatomic location.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3560. Derived Hemodynamic Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122126, DCM, "Derived Hemodynamic Measurements") | 1 | U |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3614 “Valve Areas, Non-mitral” $Units = EV (cm2, UCUM, "cm2") $Equation = DT (122262, DCM, "Area = Flow / 44.5 * sqrt(Gradient[mmHg]) ")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-02320, SRT, "Mitral Valve Area") $Units = EV (cm2, UCUM, "cm2") $Equation = DT (122263, DCM, "MVA = Flow / 38.0 * sqrt(Gradient[mmHg]) ")
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3615 “Valve Areas” $ModType = EV (121425, DCM, "Index") $ModValue = EV (8277-6, LN, "Body Surface Area") $Units = DT (cm2/m2, UCUM, "cm2/m2")
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3616 “Hemodynamic Period Measurements” $Units = DT (s/min, UCUM, "s/min")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3617 “Valve Flows” $Units = DT (ml/s, UCUM, "ml/s")
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32110, SRT, "Cardiac Index") $Units = DT (l/min/m2, UCUM, "l/min/m2")
8 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3529 “Arterial / Venous Content” $Units = DT (ml/dl, UCUM, "ml/dl")
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3618 “Hemodynamic Flows” $Units = DT (l/min, UCUM, "l/min")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8736-1, LN, "FICK Cardiac Output") $Units = DT (l/min, UCUM, "l/min")
11 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8750-2, LN, "FICK Cardiac Index") $Units = DT (l/min/m2, UCUM, "l/min/m2")
12 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122229, DCM, "Arteriovenous difference") $Units = DT (ml/dl, UCUM, "ml/dl")
13 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = BCID 3620 “Hemodynamic Ratios” $Units = DT ({ratio}, UCUM, "ratio")
14 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122237, DCM, "Corrected Sinus Node Recovery Time") $Units = DT (ms, UCUM, "ms")
15 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8861-7, LN, "Left Ventricular Stroke Work") $Units = DT (gf.m, UCUM, "gf.m")
16 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8862-5, LN, "Right Ventricular Stroke Work") $Units = DT (gf.m, UCUM, "gf.m")
17 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8863-3, LN, "Left Ventricular Stroke Work Index") $Units = DT (gf.m/m2, UCUM, "gf.m/m2")
18 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8864-1, LN, "Right Ventricular Stroke Work Index") $Units = DT (gf.m/m2, UCUM, "gf.m/m2")
19 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122238, DCM, "Max volume normalized to 50mmHg pulse pressure") $Units = DT (ml, UCUM, "ml")
20 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122239, DCM, "Oxygen Consumption") $Units = DT (ml/min, UCUM, "ml/min") $Equation = BCID 3664 “Oxygen Consumption Equations and Tables”
21 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (19217-9, LN, "Oxygen partial pressure at 50% saturation (P50) ") $Units = DCID 3500 “Pressure Units” $Equation = BCID 3666 “P50 Equations”
22 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (11556-8, LN, "Blood Oxygen partial pressure") $Units = DCID 3500 “Pressure Units”
23 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3618 “Hemodynamic Flows” $ModType = EV (121425, DCM, "Index") $ModValue = EV (8277-6, LN, "Body Surface Area") $Units = DT (l/min/m2, UCUM, "l/min/m2")
24 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3619 “Hemodynamic Resistance Measurements” $Units = DCID 3502 “Hemodynamic Resistance Units”
25 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3619 “Hemodynamic Resistance Measurements” $ModType = EV (121425, DCM, "Index") $ModValue = EV (8277-6, LN, "Body Surface Area") $Units = DCID 3503 “Indexed Hemodynamic Resistance Units”
26 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122227, DCM, "Left to Right Flow") $Units = DT (l/min, UCUM, "l/min")
27 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122228, DCM, "Right to Left Flow") $Units = DT (l/min, UCUM, "l/min")
28 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $Units = DT (ml, UCUM, "ml")
29 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $ModType = EV (121425, DCM, "Index") $ModValue = EV (8277-6, LN, "Body Surface Area") $Units = DT (ml/m2, UCUM, "ml/m2")
30 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-042BA, SRT, "Total blood volume") $Units = DT (l, UCUM, "l")
31 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3667 “Framingham Scores” $Units = DT (%, UCUM, "%") $Equation = DCID 3668 “Framingham Tables”

### TID 3570 Summary, Hemodynamics

This Template allows the recording of any significant numeric measurements or findings.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3570. Summary, Hemodynamics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121111, DCM, "Summary") | 1 | U |  |
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  |
4 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3640 “Hypertension”
5 | >> |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
6 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
7 | > | CONTAINS | CODE | DCID 3641 “Hemodynamic Assessments” | 1-n | U |  | DCID 3642 “Degree Findings”
8 | >> | HAS PROPERTIES | CODE | EV (G-C0B2, SRT, "Condition") | 1 | U |  | EV (R-102B9, SRT, "Large v-wave")
9 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (R-102BA, SRT, "Diastolic pressure equalization")

### TID 3601 Procedure Context

The Procedure Context Template describes acquisition context for measurements made or events recorded in a procedure.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3601. Procedure Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | TEXT | EV (121065, DCM, "Procedure Description") | 1 | U |  | Defaults to Study Description (0008,1030) of the General Study Module
2 |  | HAS ACQ CONTEXT | CODE | EV (G-C0E8, SRT, "Has Intent") | 1 | U |  | BCID 3629 “Procedure Intent”
3 |  | HAS ACQ CONTEXT | CODE | EV (G-C09C, SRT, "Procedure Priority") | 1 | U |  | BCID 3414 “Procedure Urgency”
4 |  | HAS OBS CONTEXT | CODE | EV (121023, DCM, "Procedure Code") | 1-n | U |  | Defaults to Procedure Code Sequence (0008,1032) of General Study Module

### TID 3602 Cardiovascular Patient Characteristics

This Template describes the characteristics of the patient that are specific to the current clinical presentation (visit). Patient Characteristic concepts in this Template, which may replicate attributes in the Patient Study Module, are included here as possible targets of by-reference relationships from other Content Items in the SR tree.

### Note

Several of the concepts in this Template duplicate concepts in TID 1007 “Subject Context, Patient”. The difference in use is that this Template has those concepts as primary observations of the patient, while in TID 1007 “Subject Context, Patient” the concepts are used to set (or reset) the context for other observations.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3602. Cardiovascular Patient Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121118, DCM, "Patient Characteristics") | 1 | M |  |
2 | > | CONTAINS | NUM | EV (121033, DCM, "Subject Age") | 1 | M |  | UNITS = DCID 7456 “Units of Measure for Age”
3 | > | CONTAINS | CODE | EV (121032, DCM, "Subject Sex") | 1 | M |  | DCID 7455 “Sex”
4 | > | CONTAINS | NUM | EV (8302-2, LN, "Patient Height") | 1 | M |  | UNITS = EV (cm, UCUM, "cm")
5 | > | CONTAINS | NUM | EV (29463-7, LN, "Patient Weight") | 1 | M |  | UNITS = EV (kg, UCUM, "kg")
6 | > | CONTAINS | NUM | EV (122221, DCM, "Thorax diameter, sagittal") | 1 | U |  | UNITS = EV (cm, UCUM, "cm")
7 | > | CONTAINS | NUM | EV (8277-6, LN, "Body Surface Area") | 1 | MC | IF BSA used for indexed measurements in SOP Instance | UNITS = EV (m2, UCUM, "m2")
8 | >> | INFERRED FROM | CODE | EV (8278-4, LN, "Body Surface Area Formula") | 1 | U |  | BCID 3663 “Body Surface Area Equations”
9 | > | CONTAINS | NUM | EV (F-01860, SRT, "Body Mass Index") | 1 | U |  | UNITS = EV (kg/m2, UCUM, "kg/m2")
10 | >> | INFERRED FROM | CODE | EV (121420, DCM, "Equation") | 1 | U |  | DT (122265, DCM, "BMI = Wt/Ht^2")
11 | > | CONTAINS | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  | UNITS = EV ({H.B.}/min, UCUM, "BPM")
12 | > | CONTAINS | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
13 | > | CONTAINS | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
14 | > | CONTAINS | CODE | DT (8884-9, LN, "Cardiac Rhythm") | 1 | U |  | BCID 3415 “Cardiac Rhythms”
15 | > | CONTAINS | NUM | EV (F-03D8C, SRT, "Chest Circumference") | 1 | U |  | UNITS = EV (cm, UCUM, "cm")
16 | > | CONTAINS | TEXT | EV (F-009E4, SRT, "Breast size") | 1 | U |  | Bra size as text string
17 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1 | U |  | DCID 3202 “Chest Pain”
18 | > | CONTAINS | CODE | EV (F-04FCC, SRT, "Functional capacity") | 1 | U |  | DCID 3719 “Canadian Clinical Classification”
19 | > | CONTAINS | CODE | EV (F-04FCC, SRT, "Functional capacity") | 1 | U |  | DCID 3736 “NYHA Classification”
20 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  |
21 | > | CONTAINS | TEXT | EV (121110, DCM, "Patient Presentation") | 1 | U |  |

Content Item Descriptions

Rows 11-13 | Cardiac vital signs, for use when the SR SOP Instance does not record vital signs at multiple procedure phases or stages.
Row 16 | Breast size for interpretation of attenuation in nuclear medicine imaging

### TID 3603 Procedure Environmental Characteristics

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3603. Procedure Environmental Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122222, DCM, "Procedure Environmental Characteristics") | 1 | M |  |
2 | > | CONTAINS | NUM | EV (122223, DCM, "Room oxygen concentration") | 1 | U |  | UNITS = EV (%, UCUM, "%")
3 | > | CONTAINS | NUM | EV (122224, DCM, "Room temperature") | 1 | U |  | UNITS = EV (Cel, UCUM, "C")
4 | > | CONTAINS | NUM | EV (122225, DCM, "Room Barometric pressure") | 1 | U |  | UNITS = DT (mbar, UCUM, "millibar")

## ECG Report Templates

### TID 3700 ECG Report

The ECG Report Template is the root structure for the representation of analysis of an ECG waveform, potentially in comparison to a prior ECG waveform analysis. The analyzed waveform may or may not be stored as a DICOM SOP Instance.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 3700. ECG Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (28010-7, LN, "ECG Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | U |  | DCID 3670 “ECG Procedure Types”
3 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
5 | > | CONTAINS | CONTAINER | EV (18785-6, LN, "Indications for Procedure") | 1 | U |  |
6 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3671 “Reason for ECG Exam”
7 | >> | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1 | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3802 “Cardiovascular Patient History” | 1 | U |  |
9 | > | CONTAINS | INCLUDE | DTID 3704 “Patient Characteristics for ECG” | 1 | U |  |
10 | > | CONTAINS | INCLUDE | DTID 3702 “Prior ECG Exam” | 1 | U |  |
11 | > | CONTAINS | INCLUDE | DTID 3708 “ECG Waveform Information” | 1 | M |  |
12 | > | CONTAINS | CONTAINER | EV (122144, DCM, "Quantitative Analysis") | 1 | M |  |
13 | >> | CONTAINS | INCLUDE | DTID 3713 “ECG Global Measurements” | 1 | U |  |
14 | >> | CONTAINS | INCLUDE | DTID 3714 “ECG Lead Measurements” | 1-n | U |  | One instantiation per reported lead
15 | > | CONTAINS | INCLUDE | DTID 3717 “ECG Qualitative Analysis” | 1 | U |  |
16 | > | CONTAINS | INCLUDE | DTID 3719 “Summary, ECG” | 1 | U |  |

### TID 3701 Clinical Context, ECG (Retired)

This Template has been retired (see PS3.16-2009).

### TID 3702 Prior ECG Exam

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3702. Prior ECG Exam

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (55114-3, LN, "Prior Procedure Descriptions") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (122140, DCM, "Comparison with Prior Exam Done") | 1 | M |  | DCID 230 “Yes-No”
3 | > | CONTAINS | DATETIME | EV (122146, DCM, "Procedure DateTime") | 1 | U |  |
4 | > | CONTAINS | UIDREF | EV (121018, DCM, "Procedure Study Instance UID") | 1 | U |  |
5 | > | CONTAINS | COMPOSITE | EV (122075, DCM, "Prior report for current patient") | 1 | U |  |
6 | > | CONTAINS | WAVEFORM | EV (121112, DCM, "Source of Measurement") | 1 | U |  |

### TID 3704 Patient Characteristics for ECG

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3704. Patient Characteristics for ECG

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121118, DCM, "Patient Characteristics") | 1 | M |  |
2 | > | CONTAINS | NUM | EV (121033, DCM, "Subject Age") | 1 | M |  | UNITS = DCID 7456 “Units of Measure for Age”
3 | > | CONTAINS | CODE | EV (121032, DCM, "Subject Sex") | 1 | M |  | DCID 7455 “Sex”
4 | > | CONTAINS | NUM | EV (8302-2, LN, "Patient Height") | 1 | U |  | UNITS = EV (cm, UCUM, "cm")
5 | > | CONTAINS | NUM | EV (29463-7, LN, "Patient Weight") | 1 | U |  | UNITS = EV (kg, UCUM, "kg")
6 | > | CONTAINS | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
7 | > | CONTAINS | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
8 | > | CONTAINS | CODE | EV (109054, DCM, "Patient State") | 1 | U |  | DCID 3262 “ECG Patient State Values”
9 | > | CONTAINS | CODE | EV (R-00728, SRT, "Pacemaker in situ") | 1 | U |  | DCID 3672 “Pacemakers”
10 | > | CONTAINS | CODE | EV (R-0077C, SRT, "ICD in situ") | 1 | U |  | DCID 3692 “ICDs”

### TID 3708 ECG Waveform Information

The ECG Waveform Information Template provides reference to, and important parameters of, the analyzed waveform.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3708. ECG Waveform Information

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (55111-9, LN, "Current Procedure Descriptions") | 1 | M |  |
2 | > | CONTAINS | WAVEFORM | EV (121112, DCM, "Source of Measurement") | 1 | U |  |
3 | > | CONTAINS | CODE | EV (10:11345, MDC, "Lead System") | 1 | U |  | DCID 3263 “Electrode Placement Values”
4 | > | CONTAINS | TEXT | EV (122142, DCM, "Acquisition Device Type") | 1 | U |  |
5 | > | CONTAINS | TEXT | EV (121122, DCM, "Equipment Identification") | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 1003 “Person Observer Identifying Attributes” | 1 | U |  | Person performing the ECG acquisition
7 | > | CONTAINS | TEXT | EV (121121, DCM, "Room Identification") | 1 | U |  |
8 | > | CONTAINS | DATETIME | EV (122146, DCM, "Procedure DateTime") | 1 | M |  |
9 | > | CONTAINS | NUM | DCID 3690 “ECG Control Variables Numeric” | 1-n | U |  |
10 | > | CONTAINS | TEXT | DCID 3691 “ECG Control Variables Text” | 1-n | U |  |
11 | > | CONTAINS | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | U |  |

### TID 3713 ECG Global Measurements

The ECG Global Measurements Template provides a structure for measurements calculated across the ECG waveform as a whole (multiple leads).

As an Extensible Template, applications may include any ECG global measurements, such as angles of the electrical vector of various ECG waves. The recommended vocabulary for such concepts is ISO/IEEE 11073-10102.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3713. ECG Global Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122158, DCM, "ECG Global Measurements") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 3715 “ECG Measurement Source” | 1 | U |  |
3 | > | CONTAINS | NUM | DT (2:16020, MDC, "Atrial Heart Rate") | 1 | U |  | UNITS = EV ({H.B.}/min, UCUM, "BPM")
4 | > | CONTAINS | NUM | DT (2:16016, MDC, "Ventricular Heart Rate") | 1 | M |  | UNITS = EV ({H.B.}/min, UCUM, "BPM")
5 | > | CONTAINS | NUM | DT (2:16160, MDC, "QT interval global") | 1 | M |  | UNITS = EV (ms, UCUM, "ms")
6 | > | CONTAINS | NUM | DT (2:15876, MDC, "QTc interval global") | 1 | U |  | UNITS = EV (ms, UCUM, "ms")
7 | >> | HAS PROPERTIES | CODE | DT (111001, DCM, "Algorithm Name") | 1 | U |  | DCID 3678 “QT Correction Algorithms”
8 | > | CONTAINS | NUM | DT (2:15872, MDC, "PR interval global") | 1 | M |  | UNITS = EV (ms, UCUM, "ms")
9 | > | CONTAINS | NUM | DT (2:16156, MDC, "QRS duration global") | 1 | M |  | UNITS = EV (ms, UCUM, "ms")
10 | > | CONTAINS | NUM | DT (2:16168, MDC, "RR interval global") | 1 | M |  | UNITS = EV (ms, UCUM, "ms")
11 | > | CONTAINS | NUM | DCID 3689 “ECG Global Waveform Durations” | 1-n | U |  | UNITS = EV (ms, UCUM, "ms")
12 | > | CONTAINS | NUM | DCID 3229 “ECG Axis Measurements” | 1-n | U |  | UNITS = EV (deg, UCUM, "deg")
13 | > | CONTAINS | NUM | DT (2:16032, MDC, "Count of all beats") | 1 | U |  | UNITS = EV ({beats}, UCUM, "beats")
14 | > | CONTAINS | NUM | DT (122707, DCM, "Number of Ectopic Beats") | 1 | U |  | UNITS = EV ({beats}, UCUM, "beats")
15 | >> | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | U |  | BCID 3234 “Ectopic Beat Morphology”

### TID 3714 ECG Lead Measurements

The ECG Lead Measurements Template provides a structure for measurements calculated on individual ECG leads.

As an Extensible Template, applications may include any ECG per lead measurements, such as integrals over time of various ECG wave voltages. The recommended vocabulary for such concepts is ISO/IEEE 11073-10102.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3714. ECG Lead Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122159, DCM, "ECG Lead Measurements") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (122148, DCM, "Lead ID") | 1 | M |  | BCID 3001 “ECG Leads”
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 3715 “ECG Measurement Source” | 1 | U |  |
4 | > | CONTAINS | NUM | DCID 3687 “Electrophysiology Waveform Durations” | 1-n | U |  | UNITS = EV (ms, UCUM, "ms")
5 | > | CONTAINS | NUM | DCID 3688 “Electrophysiology Waveform Voltages” | 1-n | U |  | UNITS = EV (mV, UCUM, "mV")
6 | > | CONTAINS | CODE | EV (F-38035, SRT, "ST Segment Finding") | 1 | U |  | DCID 3233 “ST Segment Morphology”
7 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3680 “ECG Lead Noise Descriptions”

### TID 3715 ECG Measurement Source

The ECG Measurement Source Template provides a structure for identifying the particular cardiac cycle, or beat, in an analyzed ECG waveform used for the measurement group for which this Template provides Observation Context. The cardiac cycle is identified by beat number, and optionally by specific temporal coordinates within a DICOM ECG Waveform SOP Instance.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3715. ECG Measurement Source

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (122149, DCM, "Beat Number") | 1 | U |  | Up to three numeric characters
2 |  |  | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | DCID 3676 “Lead Measurement Technique”
3 |  |  | TCOORD | EV (121112, DCM, "Source of Measurement") | 1 | U |  |
4 | > | SELECTED FROM | WAVEFORM |  | 1 | U |  |

Content Item Descriptions

Row 1 | Beat Number is specified as a numeric text string, and shall be treated as the ordinal of the beat (cardiac cycle) within the waveform acquisition for this lead that was analyzed for the measurements in this container (i.e., "1" for the first beat, "2" for the second, etc.). If absent, the measurements may have been made by a technique across multiple cycles as specified in Row 2 Measurement Method.
Rows 3 and 4 | Source of measurement identify the specific channel and time period within a DICOM ECG Waveform SOP Instance that was analyzed for the measurements in this container.

### TID 3717 ECG Qualitative Analysis

The ECG Qualitative Analysis Template allows a free text qualitative interpretation of the analyzed ECG, as well as a structure for a coded analysis.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3717. Qualitative Analysis, ECG

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (122145, DCM, "Qualitative Analysis") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (F-00033, SRT, "ECG Finding") | 1 | MC | At least one of rows 2 and 3 shall be present |
3 | > | CONTAINS | CODE | EV (F-00033, SRT, "ECG Finding") | 1-n | MC | At least one of rows 2 and 3 shall be present | BCID 3230 “ECG Findings”
4 | >> | HAS CONCEPT MOD | TEXT | EV (121051, DCM, "Equivalent Meaning of Value") | 1 | U |  | Interpretive statement
5 | >> | INFERRED FROM | CODE | EV (F-00033, SRT, "ECG Finding") | 1-n | U |  | No BCID; may use implementation-specific codes
6 | >>> | HAS CONCEPT MOD | TEXT | EV (121051, DCM, "Equivalent Meaning of Value") | 1 | U |  | Interpretive statement

Content Item Descriptions

Row 3-4 | ECG Finding provides one or more coded interpretive statements using standard or implementation-specific codes. Each coded finding will include a Code Meaning (0008,0104) using the LO Value Representation (64 characters); longer human-readable text strings may be conveyed in the Row 4 Equivalent Meaning of Value Content Item..
Row 5-6 | Each primary ECG Finding of Row 3 may have multiple supporting coded findings in Row 5, with longer human-readable text strings if necessary in Row 6.

### TID 3718 ECG Interpretive Statement (Retired)

This Template is retired. See PS3.16-2009.

### TID 3719 Summary, ECG

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3719. Summary, ECG

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121111, DCM, "Summary") | 1 | U |  |
3 | > | CONTAINS | CODE | DT (18810-2, LN, "ECG overall finding") | 1 | U |  | DCID 3677 “Summary Codes ECG”

## Cath Lab Clinical Report Templates

The Templates that comprise the Cardiac Catheterization Report are interconnected as shown in Figure A-7.

Figure A-7. Cardiac Catheterization Report Template Hierarchy

### Note

Figure A-7 shows the relationship of Templates; it does not show the structural hierarchy of Content Items in the IOD.

### TID 3800 Cardiac Catheterization Report Root

The Cardiac Cath Report provides the overall clinical results of the catheterization procedure and interventions. In many cases, more detailed information is optionally available in other reports (Hemodynamic Measurements, Procedure Log, etc.). That information is collected and summarized here (and referenced when available).

When a Discharge Summary section (row 12) is included, this report Template covers the full set of information required for submission to the ACC NCDR™ (version 2.0) registry.

### Note

1. The information required for such a submission must sometimes be reformatted from a single concept in these Templates to two data elements for the registry, or vice versa.

2. This Template is expected to be used with the Basic Text SR or Enhanced SR IOD.3. This Cardiac Cath Report Template is a baseline Template within these SOP Classes that may be replaced; it is therefore in no sense binding for exchange of this type of report. It is solely an example of a possible encoding of the Cardiac Cath Report.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 3800. Cardiac Catheterization Report Root

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (18745-0, LN, "Cardiac Catheterization Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DCID 3739 “Cath Procedure Type”
3 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
4 | > |  | INCLUDE | DTID 3601 “Procedure Context” | 1 | M |  |
5 | > | CONTAINS | INCLUDE | DTID 3802 “Cardiovascular Patient History” | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3803 “Patient Presentation, Cath” | 1 | M |  |
7 | > | CONTAINS | INCLUDE | DTID 3806 “Cath Procedure” | 1 | M |  |
8 | > | CONTAINS | INCLUDE | DTID 3810 “Cardiac Catheterization Findings” | 1 | M |  |
9 | > | CONTAINS | INCLUDE | DTID 3807 “Percutaneous Coronary Intervention Procedure” | 1 | U |  |
10 | > | CONTAINS | INCLUDE | DTID 3809 “Other Interventional Procedures” | 1-n | U |  |
11 | > | CONTAINS | INCLUDE | DTID 3820 “Adverse Outcomes, Cath” | 1 | M |  |
12 | > | CONTAINS | INCLUDE | DTID 3824 “Summary, Cath” | 1 | M |  |
13 | > | CONTAINS | INCLUDE | DTID 3828 “Discharge Summary, Cath” | 1 | U |  |

### TID 3802 Cardiovascular Patient History

This Template contains information about a cardiovascular patient's past medical history that is relevant for the interpretation of the SR document of which it is part. This information is considered to have some degree of "persistence" across different episodes of care.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3802. Cardiovascular Patient History

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (11329-0, LN, "History") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (11329-0, LN, "History") | 1 | U |  |
3 | > | CONTAINS | CONTAINER | DT (11450-4, LN, "Problem List") | 1 | U |  |
4 | >> | CONTAINS | TEXT | DCID 3769 “Concern Types” | 1-n | U |  |
5 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1-n | U |  | $Problem = DCID 3756 “Cardiac Patient Risk Factors”
6 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1 | U |  | $Problem = EV (G-023F, SRT, "History of Diabetes mellitus") $Therapy = DCID 3722 “Diabetic Therapy”
7 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1 | U |  | $Problem = EV (G-0269, SRT, "History of Hypertension") $Therapy = DCID 3760 “Hypertension Therapy”
8 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1 | U |  | $Problem = EV (R-102B8, SRT, "History of Hypercholesterolemia") $Therapy = DCID 3761 “Antilipemic Agents”
9 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1 | U |  | $Problem = EV (D3-30000, SRT, "Arrhythmia") $Therapy = DCID 3762 “Antiarrhythmic Agents”
10 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1 | U |  | $Problem = EV (G-03AA, SRT, "History of Myocardial infarction") $ModType = DT (122170, DCM, "Type of Myocardial Infarction") $ModValue = DCID 3723 “MI Types” $Therapy = DCID 3764 “Myocardial Infarction Therapies”
11 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1 | U |  | $Problem = EV (G-0069, SRT, "History of Kidney disease") $Stage = DCID 3778 “Stages”
12 | > | CONTAINS | CONTAINER | DT (29762-2, LN, "Social History") | 1 | U |  |
13 | >> | CONTAINS | TEXT | EV (F-02455, SRT, "Social History") | 1 | U |  |
14 | >> | CONTAINS | TEXT | DCID 3774 “Social History” | 1-n | U |  |
15 | >> | CONTAINS | CODE | EV (F-93109, SRT, "Tobacco Smoking Behavior") | 1 | U |  | DCID 3724 “Smoking History”
16 | >> | CONTAINS | CODE | DT (F-931D4, SRT, "Drug misuse behavior") | 1 | U |  | DT (D9-30400, SRT, "Cocaine Abuse")
17 | > | CONTAINS | CONTAINER | DT (10167-5, LN, "Past Surgical History") | 1 | U |  |
18 | >> | CONTAINS | TEXT | EV (11329-0, LN, "History") | 1-n | U |  |
19 | >> | CONTAINS | INCLUDE | DTID 3830 “Procedure Properties” | 1-n | U |  | $ProcType = DT (P0-009C3, SRT, "Surgical Procedure") $Procedure = DCID 3721 “Cardiovascular Surgeries”
20 | > | CONTAINS | CONTAINER | DT (30954-2, LN, "Relevant Diagnostic Tests and/or Laboratory Data") | 1 | U |  |
21 | >> | CONTAINS | TEXT | EV (11329-0, LN, "History") | 1-n | U |  |
22 | >> | CONTAINS | INCLUDE | DTID 3830 “Procedure Properties” | 1-n | U |  | $ProcType = DT (P0-00002, SRT, "Diagnostic procedure") $Procedure = DCID 3757 “Cardiac Diagnostic Procedures”
23 | >> | CONTAINS | NUM | DT (2086-7, LN, "Cholesterol.in HDL") | 1 | U |  | UNITS = EV (mg/dl, UCUM, "mg/dl")
24 | >> | CONTAINS | NUM | DT (2089-1, LN, "Cholesterol.in LDL") | 1 | U |  | UNITS = EV (mg/dl, UCUM, "mg/dl")
25 | > | CONTAINS | CONTAINER | DT (10160-0, LN, "History of Medication Use") | 1 | U |  |
26 | >> | CONTAINS | TEXT | DT (111516, DCM, "Medication Type") | 1-n | U |  |
27 | >>> | HAS PROPERTIES | CODE | DT (33999-4, LN, "Status") | 1 | U |  | DCID 3773 “Use Status”
28 | >> | CONTAINS | CODE | DT (111516, DCM, "Medication Type") | 1-n | U |  |
29 | >>> | HAS PROPERTIES | NUM | DT (G-C0B7, SRT, "Dosage") | 1 | U |  |
30 | >>> | HAS PROPERTIES | CODE | DT (33999-4, LN, "Status") | 1 | U |  | DCID 3773 “Use Status”
31 | > | CONTAINS | CONTAINER | DT (10157-6, LN, "History of Family Member Diseases") | 1 | U |  |
32 | >> | CONTAINS | TEXT | EV (11329-0, LN, "History") | 1-n | U |  |
33 | >> | CONTAINS | CODE | EV (G-03E5, SRT, "Family history of clinical finding") | 1-n | U |  | DCID 3758 “Cardiovascular Family History”
34 | >>> | HAS PROPERTIES | CODE | EV (G-C32E, SRT, "Subject relationship") | 1 | M |  | DCID 7451 “Family Member”
35 | > | CONTAINS | CONTAINER | DT (46264-8, LN, "History of medical device use") | 1 | U |  |
36 | >> | CONTAINS | TEXT | EV (11329-0, LN, "History") | 1 | U |  |
37 | >> | CONTAINS | INCLUDE | DTID 3831 “Medical Device Use” | 1-n | U |  | $Device = DCID 3777 “Implanted Devices”

### TID 3803 Patient Presentation, Cath

This Template describes the aspects of the patient that are specific to this clinical presentation (admission).

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3803. Patient Presentation, Cath

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121110, DCM, "Patient Presentation") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (122128, DCM, "Patient Transferred From") | 1 | U |  |
3 | > | CONTAINS | DATETIME | EV (15, NCDR [2.0b], "Admission DateTime") | 1 | U |  |
4 | > | CONTAINS | CODE | EV (17, NCDR [2.0b], "Admission Status") | 1 | U |  | DCID 3729 “Admission Status”
5 | > | CONTAINS | CODE | EV (18, NCDR [2.0b], "Insurance Payor Type") | 1 | U |  | DCID 3730 “Insurance Payor”
6 | > | CONTAINS | CODE | EV (46, NCDR [2.0b], "Congestive Heart Failure Prior to Procedure") | 1 | U |  | DCID 230 “Yes-No”
7 | > | CONTAINS | CODE | EV (47, NCDR [2.0b], "NYHA Classification") | 1 | UC | IFF Row 6 Value code meaning is <yes> | DCID 3736 “NYHA Classification”
8 | > | CONTAINS | CODE | EV (48, NCDR [2.0b], "Noninvasive Testing - Ischemia") | 1 | U |  | DCID 3737 “Non-invasive Test - Ischemia”
9 | > | CONTAINS | CODE | EV (49, NCDR [2.0b], "Pre-Cath Angina Type") | 1 | U |  | DCID 3738 “Pre-Cath Angina Type”
10 | > | CONTAINS | CODE | EV (50, NCDR [2.0b], "Pre-Cath Canadian Classification") | 1 | U |  | DCID 3719 “Canadian Clinical Classification”
11 | > | CONTAINS | CODE | EV (51, NCDR [2.0b], "Acute Coronary Syndrome Time Period") | 1 | UC | IFF Row 9 Value code meaning is <ACS> | DCID 3735 “Acute Coronary Syndrome Time Period”
12 | > | CONTAINS | CONTAINER | EV (18785-6, LN, "Indications for Procedure") | 1 | U |  |
13 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3727 “Indications for Catheterization”
14 | > | CONTAINS | NUM | EV (10230-1, LN, "LV Ejection Fraction") | 1-n | U |  | UNITS = EV (%, UCUM, "%")
15 | >> | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement method") | 1 | U |  | DCID 3744 “EF Testing Method”
16 | >> |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
17 | > | CONTAINS | CODE | EV (121069, DCM, "Previous Finding") | 1-n | U |  | DCID 3700 “Cath Diagnosis”
18 | > | CONTAINS | TEXT | EV (121110, DCM, "Patient Presentation") | 1 | U |  |

### TID 3806 Cath Procedure

This Template describes the patient-related information about this specific clinical encounter (catheterization).

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3806. Cath Procedure

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (55111-9, LN, "Current Procedure Descriptions") | 1 | M |  |
2 | > | CONTAINS | DATETIME | EV (52, NCDR [2.0b], "Procedure DateTime") | 1 | M |  |
3 | > | CONTAINS | TEXT | EV (53, NCDR [2.0b], "Procedure Number in this admission") | 1 | U |  | Up to three numeric characters
4 | > | CONTAINS | TEXT | EV (121065, DCM, "Procedure Description") | 1 | U |  |
5 | > | CONTAINS | COMPOSITE | EV (121120, DCM, "Cath Lab Procedure Log") | 1-n | U |  |
6 | > | CONTAINS | NUM | EV (55, NCDR [2.0b], "Fluoroscopy Time") | 1 | U |  | UNITS = DT (min, UCUM, "min")
7 | > | CONTAINS | NUM | EV (122130, DCM, "Dose Area Product") | 1 | U |  | UNITS = DT (mGy.cm2, UCUM, "mGy.cm2")
8 | > | CONTAINS | PNAME | EV (76, NCDR [2.0b], "Catheterization Operator") | 1 | M |  |
9 | > | CONTAINS | PNAME | EV (121088, DCM, "Fellow") | 1-n | U |  |
10 | > | CONTAINS | PNAME | BCID 7453 “Performing Roles” | 1-n | U |  |
11 | > | CONTAINS | CODE | EV (122129, DCM, "PCI during this procedure") | 1 | U |  | DCID 230 “Yes-No”
12 | > | CONTAINS | CONTAINER | EV (F-04460, SRT, "Medication Given") | 1 | M |  |
13 | >> | CONTAINS | CODE | EV (57, NCDR [2.0b], "Thrombolytics") | 1 | U |  | DCID 3740 “Thrombolytic Administration”
14 | >> | CONTAINS | CODE | EV (58, NCDR [2.0b], "IIb/IIIa Blockade") | 1 | U |  | DCID 3741 “Medication Administration, Lab Visit”
15 | >> | CONTAINS | CODE | EV (59, NCDR [2.0b], "Heparin") | 1-n | U |  | DCID 3742 “Medication Administration, PCI”
16 | >> | CONTAINS | CODE | EV (60, NCDR [2.0b], "Aspirin") | 1 | U |  | DCID 3741 “Medication Administration, Lab Visit”
17 | >> | CONTAINS | CODE | EV (61, NCDR [2.0b], "Clopidogrel/Ticlopidine") | 1 | U |  | DCID 3743 “Clopidogrel/Ticlopidine Administration”
18 | >> | CONTAINS | TEXT | EV (122083, DCM, "Drug administered") | 1-n | U |  |
19 | > | CONTAINS | CODE | EV (122138, DCM, "Circulatory Support") | 1-n | U |  | DCID 3553 “Circulatory Support”
20 | > | CONTAINS | CODE | EV (74, NCDR [2.0b], "Percutaneous Entry Site") | 1 | M |  | DCID 3746 “Percutaneous Entry Site”
21 | > | CONTAINS | CODE | EV (75, NCDR [2.0b], "Closure Device") | 1 | U |  | DCID 3747 “Percutaneous Closure”

Content Item Descriptions

Row 3 | Procedure Number (this admission) is specified as a numeric text string, and shall be treated as the ordinal of this catheterization procedure within the admission (i.e., "1" for the first catheterization, "2" for the second, etc.).

### TID 3807 Percutaneous Coronary Intervention Procedure

This Template describes the various aspects of a coronary intervention.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3807. Percutaneous Coronary Intervention Procedure

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (55111-9, LN, "Current Procedure Descriptions") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (122061, DCM, "Percutaneous Coronary Intervention")
3 | > | CONTAINS | PNAME | EV (121114, DCM, "Performing Physician") | 1 | M |  |
4 | > | CONTAINS | PNAME | EV (121088, DCM, "Fellow") | 1-n | U |  |
5 | > | CONTAINS | PNAME | DCID 7452 “Organizational Roles” | 1-n | U |  |
6 | > | CONTAINS | CODE | EV (G-C09C, SRT, "Procedure Priority") | 1 | M |  | DCID 3414 “Procedure Urgency”
7 | > | CONTAINS | CONTAINER | EV (18785-6, LN, "Indications for Procedure") | 1 | U |  |
8 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3726 “Indications for Coronary Intervention”
9 | >> | CONTAINS | CODE | EV (122172, DCM, "Acute MI Present") | 1 | U |  | DCID 230 “Yes-No”
10 | >>> | HAS PROPERTIES | CODE | DT (122170, DCM, "Type of Myocardial Infarction") | 1 | U |  | DCID 3723 “MI Types”
11 | >>> | HAS PROPERTIES | DATETIME | EV (122173, DCM, "ST Elevation Onset DateTime") | 1 | U |  |
12 | >> | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1-n | U |  |
13 | > | CONTAINS | NUM | EV (122175, DCM, "Number of lesion interventions attempted") | 1 | M |  | UNITS = EV (1, UCUM, "units")
14 | > | CONTAINS | NUM | EV (122176, DCM, "Number of lesion interventions successful") | 1 | M |  | UNITS = EV (1, UCUM, "units")
15 | > | CONTAINS | CODE | EV (122177, DCM, "Procedure Result") | 1 | M |  | DCID 3749 “PCI Procedure Result”
16 | > | CONTAINS | TEXT | EV (122177, DCM, "Procedure Result") | 1 | U |  |
17 | > | CONTAINS | INCLUDE | DTID 3808 “Lesion Intervention Information” | 1-n | M |  |

Content Item Descriptions

Rows 8 and 12 | Allow the recording of findings as either codes or as text; the same finding shall not be recorded as both.
Rows 15 and 16 | Allow the recording of procedure results as either codes or as text, but not as both.

### TID 3808 Lesion Intervention Information

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3808. Lesion Intervention Information

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122178, DCM, "Lesion Intervention Information") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 3105 “Lesion Identification and Properties” | 1 | M |  |
3 | > | CONTAINS | CODE | EV (108, NDR [2.0b], "Previous Dilation") | 1 | U |  | DCID 3750 “Previously Dilated Lesion”
4 | > | CONTAINS | CODE | EV (103, NDR [2.0b], "Guidewire crossing lesion") | 1 | U |  | DCID 3752 “Guidewire Crossing”
5 | > | CONTAINS | CODE | EV (G-C50A, SRT, "Uses Equipment") | 1-n | M |  | DCID 3411 “Intracoronary Devices”
6 | >> | HAS CONCEPT MOD | CODE | EV (122111, DCM, "Primary Intervention Device") | 1 | MC | IF Device is Primary for this Lesion | DCID 230 “Yes-No”
7 | >> | HAS PROPERTIES | TEXT | EV (121145, DCM, "Description of Material") | 1 | U |  |
8 | >> | HAS PROPERTIES | NUM | DCID 3423 “Numeric Device Characteristics” | 1-n | U |  |
9 | >> | HAS PROPERTIES | NUM | DCID 3425 “Intervention Parameters” | 1-n | U |  |
10 | >> | HAS PROPERTIES | DATETIME | EV (122105, DCM, "DateTime of Intervention") | 1 | U |  |
11 | >> | HAS PROPERTIES | NUM | EV (122106, DCM, "Duration of Intervention") | 1 | U |  | UNITS = EV (s, UCUM, "s")
12 | > | CONTAINS | NUM | EV (R-101BB, SRT, "Lumen Diameter Stenosis") | 1 | M |  | UNITS = EV (%, UCUM, "%")
13 | >> | HAS CONCEPT MOD | CODE | EV (G-72BB, SRT, "Catheterization Procedure Phase") | 1 | M |  | EV (G-7298, SRT, "Post-intervention Phase")
14 | >> | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | U |  | DCID 3745 “Calculation Method”
15 | >> |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
16 | > | CONTAINS | CODE | EV (122110, DCM, "Post- Intervention TIMI Flow") | 1 | U |  | DCID 3713 “TIMI Flow Characteristics”
17 | > | CONTAINS | CODE | EV (115, NCDR [2.0b], "Dissection in segment observed") | 1 | U |  | DCID 230 “Yes-No”
18 | > | CONTAINS | CODE | EV (116, NCDR [2.0b], "Acute closure observed") | 1 | U |  | DCID 230 “Yes-No”
19 | > | CONTAINS | CODE | EV (117, NCDR [2.0b], "Acute closure re-opened") | 1 | UC | IFF Row 18 value is <yes> | DCID 230 “Yes-No”
20 | > | CONTAINS | CODE | EV (118, NCDR [2.0b], "Perforation occurred") | 1 | U |  | DCID 230 “Yes-No”
21 | > | CONTAINS | IMAGE | DT (121080, DCM, "Best illustration of finding") | 1 | U |  |
22 | > | CONTAINS | TEXT | DT (122177, DCM, "Procedure Result") | 1 | U |  |

### TID 3809 Other Interventional Procedures

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3809. Other Interventional Procedures

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (55111-9, LN, "Current Procedure Descriptions") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | BCID 3406 “Non-coronary Transcatheter Interventions”
3 | > | CONTAINS | TEXT | EV (121065, DCM, "Procedure Description") | 1 | U |  |
4 | > | CONTAINS | CODE | DT (121065, DCM, "Procedure Description") | 1 | U |  | No BCID
5 | > | CONTAINS | TEXT | DT (122177, DCM, "Procedure Result") | 1 | U |  |

Content Item Descriptions

Rows 3 and 4 | Allow the recording of procedure description as either code or as text; the same description shall not be recorded as both.

### TID 3810 Cardiac Catheterization Findings

The Cardiac Catheterization Findings Template provides the structure for the diagnostic findings of the cath procedure, organized into sub-sections based on type of sub-procedure. It also provides for top-level summary findings and diagnoses.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3810. Cardiac Catheterization Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 3812 “Hemodynamic Findings” | 1 | U |  |
3 | > | CONTAINS | INCLUDE | DTID 3817 “Coronary Arteriography Findings” | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 3814 “Left Ventriculography Findings” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 3815 “Right Ventriculography Findings” | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3818 “Other Cardiographic Findings” | 1-n | U |  |
7 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3700 “Cath Diagnosis”
8 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
9 | > | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1-n | U |  |

Content Item Descriptions

Rows 7 and 9 | Allow the recording of findings as either codes or as text; the same finding shall not be recorded as both.

### TID 3812 Hemodynamic Findings

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3812. Hemodynamic Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (PA-50030, SRT, "Hemodynamic measurements")
3 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3640 “Hypertension”
4 | >> |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
5 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
6 | > | CONTAINS | CODE | DCID 3641 “Hemodynamic Assessments” | 1-n | U |  | DCID 3642 “Degree Findings”
7 | >> | HAS PROPERTIES | CODE | EV (G-C0B2, SRT, "Condition") | 1 | U |  | EV (R-102B9, SRT, "Large v-wave")
8 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (R-102BA, SRT, "Diastolic pressure equalization")
9 | > | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | U |  | $Report = DT (122120, DCM, "Hemodynamics Report")

Content Item Descriptions

Row 4 | (Through TID 3819 “Common Findings”) may be used to encode any significant hemodynamic numeric measurements. For reference, see TID 3550 “Pressure Waveform Measurements” and TID 3560 “Derived Hemodynamic Measurements”.

### TID 3814 Left Ventriculography Findings

The information contained here about the left ventricle is relatively qualitative in nature. It is a high-level summary of the more detailed information that may be contained in an optional Quantitative Ventricular Analysis report. This Template addresses findings about any ventricular septal defect (Row 7), the myocardial wall (Row 11), and about the aortic root (Row 16).

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3814. Left Ventriculography Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (P5-30041, SRT, "Left Ventriculography")
3 | > | CONTAINS | CODE | EV (F-30117, SRT, "Left Ventricular Function - Finding") | 1 | M |  | DCID 242 “Normal-Abnormal”
4 | > |  | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (10230-1, LN, "LV Ejection Fraction") $Units = EV (%, UCUM, "%") $Method = DCID 3748 “Angiographic EF Testing Method” $Derivation = DCID 3745 “Calculation Method”
5 | > | CONTAINS | CODE | EV (F-0224E, SRT, "Left Ventricular Cavity Size") | 1 | U |  | DCID 3705 “Chamber Size”
6 | > | CONTAINS | CODE | EV (F-02225, SRT, "Left Ventricular Contractility") | 1 | U |  | DCID 3706 “Overall Contractility”
7 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (D4-31150, SRT, "Ventricular Septal Defect")
8 | >> | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | U |  | DCID 3707 “VSD Description”
9 | > | CONTAINS | INCLUDE | DTID 3816 “Ventricular Assessment” | 1 | U |  |
10 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | U |  |
11 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (T-D075D, SRT, "Myocardial Wall")
12 | >> | CONTAINS | CODE | EV (18179-2, LN, "Wall Segment") | 1-n | M |  | BCID 3717 “Myocardial Wall Segments”
13 | >>> | HAS PROPERTIES | CODE | EV (F-32050, SRT, "Cardiac Wall Motion") | 1 | M |  | DCID 3703 “Wall Motion”
14 | >>> | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | U |  | DCID 3704 “Myocardium Wall Morphology Findings”
15 | >>> | HAS PROPERTIES | NUM | DT (G-C1E3, SRT, "Score") | 1 | U |  |
16 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | U |  |
17 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (F-04403, SRT, "Aortic Root")
18 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | M |  | DCID 3709 “Aortic Root Description”

### TID 3815 Right Ventriculography Findings

The information contained here about right ventricle is relatively qualitative in nature. It is a high-level summary of the more detailed information that may be contained in an optional Quantitative Ventricular Analysis report.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3815. Right Ventriculography Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (P5-3003F, SRT, "Right Ventriculography")
3 | > |  | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (10231-9, LN, "RV Ejection Fraction")) $Units = EV (%, UCUM, "%") $Method = DCID 3748 “Angiographic EF Testing Method” $Derivation = DCID 3745 “Calculation Method”
4 | > | CONTAINS | CODE | EV (F-022A1, SRT, "Right Ventricular Cavity Size") | 1 | U |  | DCID 3705 “Chamber Size”
5 | > | CONTAINS | CODE | EV (F-0227A, SRT, "Right Ventricular Contractility") | 1 | U |  | DCID 3706 “Overall Contractility”
6 | > | CONTAINS | INCLUDE | DTID 3816 “Ventricular Assessment” | 1 | U |  |

### TID 3816 Ventricular Assessment

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3816. Ventricular Assessment

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | U |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3701 “Cardiac Valves and Tracts”
3 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | M |  | DCID 3711 “Valvular Abnormalities”
4 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
5 | >> | HAS PROPERTIES | NUM | DT (G-C1E3, SRT, "Score") | 1 | U |  |
6 |  | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | U |  | $Report = DT (122292, DCM, "Quantitative Ventriculography Report")

### TID 3817 Coronary Arteriography Findings

The information contained here about with regards to coronary artery lesions is relatively qualitative in nature. It is a high-level summary of the more detailed information that may be contained in an optional Quantitative Coronary Arteriography report. This Template addresses findings about the individual arteries (Row 4), and about individual lesions (Row 9).

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3817. Coronary Arteriography Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (P5-30100, SRT, "Coronary Arteriography")
3 | > | CONTAINS | CODE | EV (F-04404, SRT, "Coronary artery feature") | 1 | U |  | DCID 3710 “Coronary Dominance”
4 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | U |  |
5 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | BCID 3015 “Coronary Arteries”
6 | >>> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | BCID 3019 “Cardiovascular Anatomic Location Modifiers”
7 | >> | CONTAINS | CODE | EV (122134, DCM, "Vessel Morphology") | 1-n | U |  | DCID 3712 “Vessel Descriptors”
8 | >> | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | U |  |
9 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | U |  |
10 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (R-002EF, SRT, "Coronary artery lesion (culprit)")
11 | >> | CONTAINS | INCLUDE | DTID 3105 “Lesion Identification and Properties” | 1 | M |  |
12 | >> | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | U |  | $Report = DT (122291, DCM, "Quantitative Arteriography Report")

### TID 3818 Other Cardiographic Findings

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3818. Other Cardiographic Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | BCID 3428 “Imaging Procedures”
3 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | U |  | BCID 3630 “Cardiovascular Anatomic Locations”
4 | > | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | M |  |

### TID 3819 Common Findings

Table TID 3819. Parameters

Parameter Name | Parameter Usage
--- | ---
$Report | Title of composite object (evidence document) referenced

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3819. Common Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (121071, DCM, "Finding") | 1 | U |  |
2 |  |  | IMAGE | DT (121080, DCM, "Best illustration of finding") | 1 | U |  |
3 |  |  | NUM | No BCID | 1 | U |  |
4 | > |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
5 |  |  | COMPOSITE | DT (122073, DCM, "Current procedure evidence") | 1 | U |  |
6 | > | HAS PROPERTIES | CODE | EV (121144, DCM, "Document Title") | 1 | U |  | $Report

Content Item Descriptions

Row 3 | May be used to encode any significant image- or waveform-based numeric measurements, with post-coordination of the Concept Name using child Content Items (with relationship HAS CONCEPT MOD), as permitted by Section 6.2.4. The source of the measurement may be noted using the Quotation Template in Row 4.

### TID 3820 Adverse Outcomes, Cath

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3820. Adverse Outcomes, Cath

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121113, DCM, "Complications") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (DD-60002, SRT, "Complication of Procedure") | 1-n | U |  | DCID 3755 “Cath Complications”
3 | >> | HAS CONCEPT MOD | CODE | EV (G-D709, SRT, "Relative Time") | 1 | U |  | DCID 12102 “Temporal Periods Relating to Procedure or Therapy”
4 | > | CONTAINS | CODE | EV (DD-60002, SRT, "Complication of Procedure") | 1-n | U |  | DCID 3754 “Vascular Complications”
5 | >> | HAS CONCEPT MOD | CODE | EV (G-D709, SRT, "Relative Time") | 1 | U |  | DCID 12102 “Temporal Periods Relating to Procedure or Therapy”
6 | > | CONTAINS | TEXT | EV (DD-60002, SRT, "Complication of Procedure") | 1-n | U |  |
7 | > | CONTAINS | CODE | EV (122179, DCM, "Peri-procedural MI occurred") | 1 | U |  | DCID 230 “Yes-No”
8 | >> | INFERRED FROM | NUM | EV (122181, DCM, "CK-MB peak") | 1 | U |  | UNITS = EV ([iU], UCUM, "International unit")
9 | >>> | HAS PROPERTIES | NUM | EV (R-0038B, SRT, "Normal Range Upper Limit") | 1 | M |  | UNITS = EV ([iU], UCUM, "International unit")
10 | >> | INFERRED FROM | NUM | EV (122180, DCM, "CK-MB baseline") | 1 | M |  | UNITS = EV ([iU], UCUM, "International unit")
11 | > | CONTAINS | IMAGE | DT (121080, DCM, "Best illustration of finding") | 1-n | U |  |

Content Item Descriptions

Rows 2, 3 and 4 | Allow the recording of outcomes as either codes or as text; the same outcome shall not be recorded as both.

### TID 3824 Summary, Cath

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 3824. Summary, Cath

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3728 “Cath Findings”
3 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
4 | > | CONTAINS | TEXT | EV (121111, DCM, "Summary") | 1 | U |  |
5 | > | CONTAINS | TEXT | EV (121075, DCM, "Recommendation") | 1-n | U |  |

### TID 3828 Discharge Summary, Cath

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3828. Discharge Summary, Cath

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121115, DCM, "Discharge Summary") | 1 | M |  |
2 | > | CONTAINS | DATETIME | EV (122163, DCM, "Discharge DateTime") | 1 | U |  |
3 | > | CONTAINS | CODE | EV (122164, DCM, "Coronary Artery Bypass During This Admission") | 1 | U |  | DCID 230 “Yes-No”
4 | >> | HAS PROPERTIES | CODE | EV (G-C09C, SRT, "Procedure Priority") | 1 | U |  | BCID 3414 “Procedure Urgency”
5 | >> | HAS PROPERTIES | DATETIME | EV (122146, DCM, "Procedure DateTime") | 1 | U |  |
6 | > | CONTAINS | CODE | EV (122166, DCM, "Death During This Admission") | 1 | U |  | DCID 230 “Yes-No”
7 | >> | HAS PROPERTIES | DATETIME | EV (122165, DCM, "DateTime of Death") | 1 | U |  |
8 | >> | HAS PROPERTIES | CODE | EV (25, NCDR [2.0b], "Primary Cause of Death") | 1 | U |  | DCID 3733 “Primary Cause of Death”
9 | >> | HAS PROPERTIES | CODE | EV (122167, DCM, "Death During Catheterization") | 1 | U |  | DCID 230 “Yes-No”
10 | > | CONTAINS | TEXT | EV (121111, DCM, "Summary") | 1 | U |  |

### TID 3829 Problem Properties

Table TID 3829. Parameters

Parameter Name | Parameter Usage
--- | ---
$Problem | Coded Value or Context Group for problem
$ModType | Modifier Name for Concept Name of problem
$ModValue | Modifier Value for Concept Name of problem
$Therapy | Coded Value or Context Group for therapy received for problem
$Stage | Coded Value or Context Group for problem or disease stage

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3829. Problem Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121430, DCM, "Concern") | 1 | M |  |
2 | > | CONTAINS | DATETIME | DT (121431, DCM, "DateTime Concern Noted") | 1 | U |  |
3 | > | CONTAINS | DATETIME | DT (121432, DCM, "DateTime Concern Resolved") | 1 | U |  |
4 | > | CONTAINS | CODE | DCID 3769 “Concern Types” | 1 | M |  | $Problem
5 | >> | HAS CONCEPT MOD | CODE | $ModType | 1-n | U |  | $ModValue
6 | >> | HAS PROPERTIES | DATETIME | DT (111526, DCM, "DateTime Started") | 1 | U |  |
7 | >> | HAS PROPERTIES | DATETIME | DT (121433, DCM, "DateTime Problem Resolved") | 1 | U |  |
8 | >> | HAS PROPERTIES | CODE | DT (33999-4, LN, "Status") | 1 | U |  | DCID 3770 “Problem Status”
9 | >> | HAS PROPERTIES | CODE | DT (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
10 | >> | HAS PROPERTIES | CODE | DT (G-C16B, SRT, "Stage") | 1 | U |  | $Stage
11 | > | CONTAINS | CODE | DT (11323-3, LN, "Health status") | 1 | U |  | DCID 3772 “Health Status”
12 | > | CONTAINS | CODE | EV (P0-0000E, SRT, "Therapy") | 1-n | U |  | $Therapy
13 | >> | HAS PROPERTIES | CODE | DT (33999-4, LN, "Status") | 1 | U |  | DCID 3773 “Use Status”
14 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

### TID 3830 Procedure Properties

Table TID 3830. Parameters

Parameter Name | Parameter Usage
--- | ---
$ProcType | Coded Value for class of procedure
$Procedure | Coded Value or Context Group for procedure
$ModType | Modifier Name for Concept Name of procedure
$ModValue | Modifier Value for Concept Name of procedure

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3830. Procedure Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | $ProcType | 1 | M |  | $Procedure
2 | > | HAS CONCEPT MOD | CODE | $ModType | 1-n | U |  | $ModValue
3 | > | HAS PROPERTIES | DATETIME | DT (111526, DCM, "DateTime Started") | 1 | U |  |
4 | > | HAS PROPERTIES | COMPOSITE | EV (R-42B89, SRT, "Clinical Report") | 1-n | U |  |
5 | >> | HAS PROPERTIES | CODE | EV (121144, DCM, "Document Title") | 1 | U |  |
6 | > | HAS PROPERTIES | TEXT | EV (R-42B89, SRT, "Clinical Report") | 1-n | U |  | Description of report with URL or other reference for report
7 | > | HAS PROPERTIES | TEXT | DT (121434, DCM, "Service Delivery Location") | 1 | U |  |
8 | > | HAS PROPERTIES | PNAME | DT (121435, DCM, "Service Performer") | 1 | UC | IF Service Performer is a person |
9 | > | HAS PROPERTIES | TEXT | DT (121435, DCM, "Service Performer") | 1 | UC | IF Service Performer is an organization |
10 | > | HAS PROPERTIES | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

### TID 3831 Medical Device Use

Table TID 3831. Parameters

Parameter Name | Parameter Usage
--- | ---
$Device | Coded Value for type of device

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3831. Medical Device Use

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121436, DCM, "Medical Device Used") | 1 | M |  | $Device
2 | > | HAS PROPERTIES | DATETIME | DT (111526, DCM, "DateTime Started") | 1 | U |  |
3 | > | HAS PROPERTIES | DATETIME | DT (111527, DCM, "DateTime Ended") | 1 | U |  |
4 | > | HAS PROPERTIES | CODE | DT (33999-4, LN, "Status") | 1 | U |  | DCID 3773 “Use Status”
5 | > | HAS PROPERTIES | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

## CT/MR Cardiovascular Analysis Report Templates

### TID 3900 CT/MR Cardiovascular Analysis Report

Root Template of the Non-invasive Computed Tomography and Magnetic Resonance Cardiovascular Analysis Report.

This Template contains the top level structure and includes subordinate Templates for the various analyses.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 3900. CT/MR Cardiovascular Analysis Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122600, DCM, "Cardiovascular Analysis Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure Reported") | 1-n | M |  | DCID 3820 “Non-invasive Vascular Procedures”
3 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
5 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3802 “Cardiovascular Patient History” | 1 | U |  |
7 | > | CONTAINS | INCLUDE | DTID 3901 “Procedure Summary” | 1 | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3902 “Vascular Analysis” | 1 | U |  | $AnalysisPerformed = EV (122605, DCM, "Vascular Morphological Analysis")
9 | > | CONTAINS | INCLUDE | DTID 3902 “Vascular Analysis” | 1 | U |  | $AnalysisPerformed = EV (122606, DCM, "Vascular Functional Analysis")
10 | > | CONTAINS | INCLUDE | DTID 3920 “Ventricular Analysis” | 1 | U |  |
11 | > | CONTAINS | INCLUDE | DTID 3927 “Report Summary” | 1-n | U |  |

### TID 3901 Procedure Summary

Contains summaries related to the performed procedures.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3901. Procedure Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (55111-9, LN, "Current Procedure Descriptions") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121065, DCM, "Procedure Description") | 1-n | M |  |
3 | > | CONTAINS | CODE | DT (RID11248, RADLEX, "Cardiac Gating") | 1 | U |  | DCID 3104 “Cardiac Synchronization Technique”

### TID 3902 Vascular Analysis

Contains either morphological or functional vascular measurement results of an analysis

Table TID 3902. Parameters

Parameter Name | Parameter Usage
--- | ---
$AnalysisPerformed | Analysis Performed

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3902. Vascular Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | $AnalysisPerformed
3 | > | CONTAINS | INCLUDE | DTID 3905 “Calcium Scoring Results” | 1 | UC | IFF the value of row 2 equals EV (122605, DCM, "Vascular Morphological Analysis") |
4 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-D0767, SRT, "Blood Vessel of Head") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12105 “Intracranial Cerebral Vessels” $AnalysisPerformed = $AnalysisPerformed
5 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-D0767, SRT, "Blood Vessel of Head") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12105 “Intracranial Cerebral Vessels” $AnalysisPerformed = $AnalysisPerformed
6 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-D0767, SRT, "Blood Vessel of Head") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 12106 “Intracranial Cerebral Vessels (Unilateral)” $AnalysisPerformed = $AnalysisPerformed
7 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-45005, SRT, "Artery of Neck") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12104 “Extracranial Arteries” $AnalysisPerformed = $AnalysisPerformed
8 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-45005, SRT, "Artery of Neck") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12104 “Extracranial Arteries” $AnalysisPerformed = $AnalysisPerformed
9 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-47040, SRT, "Artery of Lower Extremity ") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12109 “Lower Extremity Arteries” $AnalysisPerformed = $AnalysisPerformed
10 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-47040, SRT, "Artery of Lower Extremity ") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12109 “Lower Extremity Arteries” $AnalysisPerformed = $AnalysisPerformed
11 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-49403, SRT, "Vein of Lower Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12110 “Lower Extremity Veins” $AnalysisPerformed = $AnalysisPerformed
12 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-49403, SRT, "Vein of Lower Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12110 “Lower Extremity Veins” $AnalysisPerformed = $AnalysisPerformed
13 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-47020, SRT, "Artery of Upper Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12107 “Upper Extremity Arteries” $AnalysisPerformed = $AnalysisPerformed
14 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-47020, SRT, "Artery of Upper Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12107 “Upper Extremity Arteries” $AnalysisPerformed = $AnalysisPerformed
15 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-49103, SRT, "Vein of Upper Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12108 “Upper Extremity Veins” $AnalysisPerformed = $AnalysisPerformed
16 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-49103, SRT, "Vein of Upper Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12108 “Upper Extremity Veins” $AnalysisPerformed = $AnalysisPerformed
17 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-71019, SRT, "Vascular Structure of Kidney") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12115 “Renal Vessels” $AnalysisPerformed = $AnalysisPerformed
18 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-71019, SRT, "Vascular Structure of Kidney") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12115 “Renal Vessels” $AnalysisPerformed = $AnalysisPerformed
19 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-46002, SRT, "Artery of Abdomen") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12111 “Abdominal Arteries (Lateral)” $AnalysisPerformed = $AnalysisPerformed
20 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-46002, SRT, "Artery of Abdomen") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12111 “Abdominal Arteries (Lateral)” $AnalysisPerformed = $AnalysisPerformed
21 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-46002, SRT, "Artery of Abdomen") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 12112 “Abdominal Arteries (Unilateral)” $AnalysisPerformed = $AnalysisPerformed
22 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-487A0, SRT, "Vein of Abdomen") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12113 “Abdominal Veins (Lateral)” $AnalysisPerformed = $AnalysisPerformed
23 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-487A0, SRT, "Vein of Abdomen") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12113 “Abdominal Veins (Lateral)” $AnalysisPerformed = $AnalysisPerformed
24 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-487A0, SRT, "Vein of Abdomen") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 12114 “Abdominal Veins (Unilateral)” $AnalysisPerformed = $AnalysisPerformed
25 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-44000, SRT, "Pulmonary Artery Structure") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 3829 “Pulmonary Arteries” $AnalysisPerformed = $AnalysisPerformed
26 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-43000, SRT, "Coronary Artery Structure") $Anatomy = DCID 3015 “Coronary Arteries” $AnalysisPerformed = $AnalysisPerformed
27 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-48400, SRT, "Cardiac Vein Structure") $Anatomy = DCID 3839 “Coronary Veins” $AnalysisPerformed = $AnalysisPerformed
28 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-48581, SRT, "Pulmonary Venous Structure") $Anatomy = DCID 3840 “Pulmonary Veins” $AnalysisPerformed = $AnalysisPerformed

### TID 3905 Calcium Scoring Results

Contains the calcium scoring results related to plaque findings, vessels or the whole body.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3905. Calcium Scoring Results

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122603, DCM, "Calcium Scoring Analysis")
3 | > | CONTAINS | NUM | EV (122657, DCM, "Agatston Score Threshold") | 1 | U |  | UNITS = DT ([hnsf'U], UCUM, "Hounsfield unit")
4 | > | CONTAINS | NUM | EV (122658, DCM, "Calcium Mass Threshold") | 1 | U |  | UNITS = DT (mg/cm3, UCUM, "mg/cm3")
5 | > | CONTAINS | NUM | EV (122659, DCM, "Calcium Scoring Calibration") | 1 | U |  | UNITS = DT (mg/[hnsf'U].cm3, UCUM, "mg/[hnsf'U].cm3")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (112058, DCM, "Calcium Score") $Method = EV (112055, DCM, "Agatston Scoring Method") $Units = DT (1, UCUM, "no units")
7 | > | CONTAINS | NUM | EV (122660, DCM, "Calcium Volume") | 1 | U |  | UNITS = EV (mm3, UCUM, "mm3")
8 | > | CONTAINS | NUM | EV (122661, DCM, "Calcium Mass") | 1 | U |  | UNITS = EV (mg, UCUM, "mg")
9 | > | CONTAINS | NUM | EV (F-02A3B, SRT, "Number of Lesions") | 1 | U |  | UNITS = DT ({lesions}, UCUM, "lesions")
10 | > | CONTAINS | INCLUDE | DTID 3909 “Best Illustration of Findings” | 1-n | U |  |

### TID 3906 Vascular Section Measurements

Sections of vascular measurements are section containers of an anatomical region consisting of measurement group containers containing the measurements.

Table TID 3906. Parameters

Parameter Name | Parameter Usage
--- | ---
$VascularSection | The concept name of the region or structure of which the anatomy is part
$SectionLaterality | The laterality (if any) of the anatomy in this section heading
$Anatomy | The concept name of the vascular anatomy
$AnalysisPerformed | The context of the measurements performed during the analysis

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3906. Vascular Section Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | $VascularSection | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IFF $SectionLaterality has a value | $SectionLaterality
3 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | M |  |
4 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | $Anatomy
5 | >> | CONTAINS | CODE | EV (122686, DCM, "Parent Vessel Finding") | 1-n | U |  | DCID 3810 “Vascular Morphology”
6 | >>> |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
7 | >> | CONTAINS | INCLUDE | DTID 3905 “Calcium Scoring Results” | 1 | UC | IF the value of $AnalysisPerformed equals (122605, DCM, "Vascular Morphological Analysis") |
8 | >> | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | U |  |
9 | >>> | HAS CONCEPT MOD | CODE | EV (125101, DCM, "Vessel Branch") | 1-n | UC | IF concept value of row 4 is not equal to (T-43000, SRT, "Coronary Artery Structure") | DCID 12117 “Vessel Branch Modifiers”
10 | >>> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical Modifier") | 1 | UC | IF concept value of row 4 is not equal to (T-43000, SRT, "Coronary Artery Structure") | DCID 12116 “Vessel Segment Modifiers”
11 | >>> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical Modifier") | 1 | UC | IF concept value of row 4 equals (T-43000, SRT, "Coronary Artery Structure") | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
12 | >>> | CONTAINS | INCLUDE | DTID 3907 “Vessel Measurements” | 1 | U |  |
13 | >>> | CONTAINS | INCLUDE | DTID 3908 “Vascular Lesion” | 1-n | UC | IF the value of$AnalysisPerformed equals (122605, DCM, "Vascular Morphological Analysis") |
14 | >>> | CONTAINS | INCLUDE | DTID 3910 “Flow Quantification” | 1 | UC | IF the value of$AnalysisPerformed equals (122606, DCM, "Vascular Functional Analysis") |

Content Item Descriptions

Row 3 | This Findings container allows an application to group related vessels or branches
Row 5 | The characteristics associated with the parent vessel shall also be reported in the findings Container for the parent vessel. Negative findings (characteristics not present) need not be reported in the parent vessel Container.

### TID 3907 Vessel Measurements

Contains measurements made on vessel level.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3907. Vessel Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | NUM | EV (R-101BB, SRT, "Lumen Diameter Stenosis") | 1 | U |  | UNITS = DT (%, UCUM, "%")
2 |  | CONTAINS | NUM | EV (R-101BA, SRT, "Lumen Area Stenosis") | 1 | U |  | UNITS = DT (%, UCUM, "%")
3 |  | CONTAINS | NUM | EV (121206, DCM, "Distance") | 1-n | U |  | UNITS = DT (mm, UCUM, "mm")
4 | > | HAS CONCEPT MOD | CODE | EV (122340, DCM, "Fiducial Feature") | 2 | M |  |
5 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
6 | > |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1 | U |  |
7 |  | CONTAINS | NUM | EV (G-0364, SRT, "Vessel Lumen Diameter") | 1-n | U |  | UNITS = DT (mm, UCUM, "mm")
8 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | U |  | DCID 3488 “Min/Max/Mean”
9 | > | HAS PROPERTIES | NUM | EV (122337, DCM, "Relative Position") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
10 | >> | HAS CONCEPT MOD | CODE | EV (122340, DCM, "Fiducial Feature") | 1 | M |  | DCID 3837 “Fiducial Feature”

Content Item Descriptions

Rows 3-5 | The distance between two identified fiducial features
Rows 7-10 | Measurement of vessel diameter made at a position relative to a fiducial feature
Row 9 | A positive value indicates a distance in the direction of flow within the vessel

### TID 3908 Vascular Lesion

Specifies properties and the features of a vascular lesion detected during the analysis. In addition it is possible to reference or include growing of lesions over time by adding references to previous reports or by adding previous examination results.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3908. Vascular Lesion

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (F-03FCD, SRT, "Lesion Finding") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121151, DCM, "Lesion Identifier") | 1 | M |  |
3 | > | CONTAINS | INCLUDE | DTID 3909 “Best Illustration of Findings” | 1-n | U |  |
4 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1-n | U |  |
5 | > | CONTAINS | NUM | EV (122337, DCM, "Relative Position") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
6 | >> | HAS CONCEPT MOD | CODE | EV (122340, DCM, "Fiducial Feature") | 1 | M |  | DCID 3837 “Fiducial Feature”
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (G-0364, SRT, "Vessel Lumen Diameter") $Derivation = DCID 3838 “Diameter Derivation” $FindingSite = DCID 3486 “Vascular Measurement Sites” $Units = DT (mm, UCUM, "mm")
8 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Derivation = DCID 3838 “Diameter Derivation” $FindingSite = DCID 3486 “Vascular Measurement Sites” $Units = DT (mm2, UCUM, "mm2")
9 | > | CONTAINS | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | M |  | DCID 3810 “Vascular Morphology”
10 | >> |  | INCLUDE | DTID 3909 “Best Illustration of Findings” | 1-n | U |  |
11 | >> | HAS PROPERTIES | TEXT | EV (121106, DCM, "Comment") | 1-n | U |  |
12 | >> |  | INCLUDE | DTID 3911 “Plaque Properties” | 1 | MC | IFF value of row 9 equals (M-01470, SRT, "Plaque") |
13 | >> |  | INCLUDE | DTID 3912 “Stenosis Properties” | 1 | MC | IFF value of row 9 equals (M-3400A, SRT, "Stenosis") |
14 | >> |  | INCLUDE | DTID 3913 “Aneurysm Properties” | 1 | MC | IFF value of row 9 equals (M-32200, SRT, "Aneurysm") |
15 | >> |  | INCLUDE | DTID 3914 “Arterial Dissection Properties” | 1 | MC | IFF value of row 9 equals (D3-81310, SRT, "Arterial Dissection") |
16 | >> | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | MC | IFF value of row 9 equals (M-520F8, SRT, "Vascular Sclerosis") | DCID 3817 “Vascular Sclerosis Types”
17 | >> |  | INCLUDE | DTID 3915 “Vascular Occlusion Properties” | 1 | MC | IFF value of row 9 equals EV (M-34000, SRT, "Occlusion") |
18 | >> |  | INCLUDE | DTID 3916 “Stent Properties” | 1 | MC | IFF value of row 9 equals (A-25500, SRT, "Stent") |

Content Item Descriptions

Row 5 | A positive value indicates a distance in the direction of flow within the vessel For example: An aneurysm with relative position -4 mm from the renal arteries would begin superior to the renal arteries.
Row 7, 8 | These rows are associated with the position of the most significant effect of the lesion, i.e., maximum diameter of aneurysm or the minimum diameter of stenosis

### TID 3909 Best Illustration of Findings

Specification of images, waveforms, spatial and temporal coordinates used to illustrate findings.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3909. Best Illustration of Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | IMAGE | EV (121080, DCM, "Best illustration of finding") | 1 | U |  |
2 |  | CONTAINS | WAVEFORM | EV (121080, DCM, "Best illustration of finding") | 1 | U |  |
3 |  | CONTAINS | SCOORD | EV (121080, DCM, "Best illustration of finding") | 1 | U |  |
4 | > | SELECTED FROM | IMAGE | no purpose of reference | 1 | M |  |
5 |  | CONTAINS | TCOORD | EV (121080, DCM, "Best illustration of finding") | 1 | U |  |
6 | > | SELECTED FROM | SCOORD | no purpose of reference | 1 | MC | XOR row 8, 9 |
7 | >> | SELECTED FROM | IMAGE | no purpose of reference | 1 | M |  | must be a multi-frame image
8 | > | SELECTED FROM | WAVEFORM | no purpose of reference | 1 | MC | XOR row 6, 9 |
9 | > | SELECTED FROM | IMAGE | no purpose of reference | 1 | MC | XOR row 6, 8 | must be a multi-frame image

### TID 3910 Flow Quantification

Contains the flow quantification measurement results

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3910. Flow Quantification

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122604, DCM, "Flow Quantification")
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 3929 “Cardiovascular Analysis Observation Context” | 1 | U |  |
4 | > | HAS OBS CONTEXT | DATETIME | EV (G-D321, SRT, "Start DateTime") | 1 | M |  |
5 | > | HAS OBS CONTEXT | DATETIME | EV (G-D320, SRT, "Stop DateTime") | 1 | M |  |
6 | > | CONTAINS | INCLUDE | DTID 3990 “Two Dimensional Measurement Graph” | 1 | U |  | $MeasurementGraph = EV (122667, DCM, "Blood velocity vs. time of cardiac cycle") $X-Concept = EV (122666, DCM, "Time relative to R-wave peak") $Y-Concept = EV (F-0319E, SRT, "Arterial Velocity") $X-AxisUnits = DT (ms, UCUM, "ms") $Y-AxisUnits = DT (cm/s, UCUM, "cm/s")
7 | > | CONTAINS | NUM | EV (122642, DCM, "Velocity Encoding Minimum Value") | 1 | U |  | UNITS = DT (cm/s, UCUM, "cm/s")
8 | > | CONTAINS | NUM | EV (122643, DCM, "Velocity Encoding Maximum Value") | 1 | U |  | UNITS = DT (cm/s, UCUM, "cm/s")
9 | > | CONTAINS | CONTAINER | EV (125007, DCM, "Measurement Group") | 1-n | M |  |
10 | >> | HAS CONCEPT MOD | TEXT | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | XOR row 11 |
11 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | XOR row 10 |
12 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122207, DCM, "Blood velocity, peak") $Units = DT (cm/s, UCUM, "cm/s")
13 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122205, DCM, "Blood velocity, mean") $Units = DT (cm/s, UCUM, "cm/s")
14 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-39200, SRT, "Blood Flow") $Derivation = EV (R-00317, SRT, "Mean") $Units = DT (ml/s, UCUM, "ml/s")
15 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-2 | U |  | $Measurement = EV (F-39200, SRT, "Blood Flow") $ModType = EV (G-C048, SRT, "Direction of flow") $ModValue = DCID 12221 “Flow Direction” $Units = DT (ml/s, UCUM, "ml/s")
16 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122645, DCM, "Net Forward Volume") $Units = DT (ml, UCUM, "ml")
17 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122645, DCM, "Net Forward Volume") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8277-6, LN, "BSA") $Units = DT (ml/m2, UCUM, "ml/m2")
18 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Derivation = DCID 3488 “Min/Max/Mean” $Units = DT (mm2, UCUM, "mm2")

### TID 3911 Plaque Properties

Properties of a plaque finding

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3911. Plaque Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | NUM | EV (122376, DCM, "Total Plaque Volume") | 1 | U |  | UNITS = DT (mm3, UCUM, "mm3")
2 |  | HAS PROPERTIES | CODE | EV (G-A428, SRT, "Margin") | 1 | U |  | DCID 3715 “Lesion Margin”
3 |  | HAS PROPERTIES | CODE | EV (M-01000, SRT, "Morphological Abnormal Structure") | 1-n | M |  | DCID 3802 “Plaque Structures”
4 |  | HAS PROPERTIES | INCLUDE | DTID 3905 “Calcium Scoring Results” | 1 | U |  |
5 |  | HAS PROPERTIES | CODE | EV (121071, DCM, "Finding") | 1 | U |  | DT (R-102DA, SRT, "Contrast Media Seen in Plaque")

### TID 3912 Stenosis Properties

Properties of a stenosis finding

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3912. Stenosis Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | CODE | EV (G-C036, SRT, "Measurement method") | 1 | M |  | DCID 3804 “Stenosis Measurement Methods”
2 |  | HAS PROPERTIES | CODE | EV (G-D775, SRT, "Type of Stenosis") | 1 | U |  | DCID 3805 “Stenosis Types”
3 |  | HAS PROPERTIES | CODE | EV (G-C002, SRT, "Associated with") | 1 | U |  | DCID 3815 “Source of Vascular Finding”
4 |  | HAS PROPERTIES | CODE | EV (G-C2FE, SRT, "Shape") | 1 | U |  | DCID 3806 “Stenosis Shape”
5 |  | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (G-0364, SRT, "Vessel Lumen Diameter") $Derivation = DCID 3488 “Min/Max/Mean” $Units = DT (mm, UCUM, "mm")
6 |  | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Derivation = DCID 3488 “Min/Max/Mean” $Units = DT (mm2, UCUM, "mm2")
7 |  | HAS PROPERTIES | NUM | EV (R-101BC, SRT, "Stenotic Lesion Length") | 1 | U |  | UNITS = DT (mm, UCUM, "mm")
8 |  | HAS PROPERTIES | CODE | EV (R-101BC, SRT, "Stenotic Lesion Length") | 1 | U |  | DCID 3831 “Stenosis Length”
9 |  | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (R-101BA, SRT, "Lumen Area Stenosis") $Derivation = DCID 3488 “Min/Max/Mean” $Units = DT (%, UCUM, "%")
10 |  | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (R-101BB, SRT, "Lumen Diameter stenosis") $Derivation = DCID 3488 “Min/Max/Mean” $Units = DT (%, UCUM, "%")
11 |  | HAS PROPERTIES | CODE | EV (R-101BA, SRT, "Lumen Area Stenosis") | 1-n | U |  | DCID 3832 “Stenosis Grade”
12 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | M |  | DCID 3488 “Min/Max/Mean”
13 |  | HAS PROPERTIES | CODE | EV (R-101BB, SRT, "Lumen Diameter Stenosis") | 1-n | U |  | DCID 3832 “Stenosis Grade”
14 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | M |  | DCID 3488 “Min/Max/Mean”

### TID 3913 Aneurysm Properties

Properties of an aneurysm finding

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3913. Aneurysm Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | M |  | DCID 3808 “Aneurysm Types”
2 |  | HAS PROPERTIES | CODE | EV (G-C002, SRT, "Associated with") | 1 | U |  | DCID 3815 “Source of Vascular Finding”
3 |  | HAS PROPERTIES | INCLUDE | DTID 3917 “Aneurysm Measurements” | 1 | U |  |
4 |  | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | UC | IFF value of row 1 equals (M-32240, SRT, "Mixed Aneurysm") | DCID 3808 “Aneurysm Types”
5 | > | HAS PROPERTIES | INCLUDE | DTID 3917 “Aneurysm Measurements” | 1 | U |  |

### TID 3914 Arterial Dissection Properties

Properties of a arterial dissection finding

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3914. Arterial Dissection Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | CODE | EV (122387, DCM, "Dissection Classification") | 1 | M |  | DCID 3492 “Vascular Dissection Classifications”
2 |  | HAS PROPERTIES | CODE | EV (G-C150, SRT, "Etiology") | 1 | U |  | DCID 3809 “Associated Conditions”
3 |  | HAS PROPERTIES | NUM | EV (G-D7FE, SRT, "Length") | 1 | U |  | UNITS = DT (mm, UCUM, "mm")
4 |  | HAS PROPERTIES | CODE | EV (R-102DD, SRT, "Anatomic structure potentially involved in evolution of disease") | 1-n | U |  | DCID 3827 “Vessel Segments”

### TID 3915 Vascular Occlusion Properties

Properties of vascular occlusion finding

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3915. Vascular Occlusion Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | CODE | EV (G-D775, SRT, "Type of Stenosis") | 1 | M |  | DCID 3805 “Stenosis Types”
2 |  | HAS PROPERTIES | CODE | EV (G-C002, SRT, "Associated with") | 1 | U |  | DCID 3815 “Source of Vascular Finding”
3 |  | HAS PROPERTIES | CODE | EV (G-C2FE, SRT, "Shape") | 1 | U |  | DCID 3806 “Stenosis Shape”
4 |  | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (R-101BC, SRT, "Stenotic Lesion Length") $Method = DCID 3804 “Stenosis Measurement Methods” $Units = DT (mm, UCUM, "mm")

### TID 3916 Stent Properties

Properties of a stent finding

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3916. Stent Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | CODE | EV (122685, DCM, "Stent Composition") | 1-n | M |  | DCID 3814 “Stent Composition”
2 |  | HAS PROPERTIES | NUM | EV (R-101AD, SRT, "Vascular Stent Diameter") | 1 | U |  | UNITS = DT (mm, UCUM, "mm")
3 |  | HAS PROPERTIES | NUM | EV (R-101B0, SRT, "Vascular Stent Length") | 1 | U |  | UNITS = DT (mm, UCUM, "mm")
4 |  | HAS PROPERTIES | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3813 “Stent Findings”
5 | > |  | INCLUDE | DTID 3912 “Stenosis Properties” | 1 | MC | IFF value of row 4 equals (M-3400A, SRT, "Stenosis") |

### TID 3917 Aneurysm Measurements

Measurements of aneurysms. TID 300 “Measurement” invoked from this Template allows the measurement to reference an image used as the source of the measurement.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3917. Aneurysm Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-D7FE, SRT, "Length") $ModType = EV (G-C093, SRT, "Extent") $ModValue = DT (G-A143, SRT, "Longitudinal") $Units = DT (mm, UCUM, "mm")
2 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-D705, SRT, "Volume") $Method = DCID 3807 “Volume Measurement Methods” $Units = DT (mm3, UCUM, "mm3")
3 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (R-102DB, SRT, "Vessel Lumen Cross-Sectional Area Increase") $Units = DT (%, UCUM, "%")
4 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (R-102DB, SRT, "Vessel Lumen Cross-Sectional Area Increase") $Units = DT (mm2, UCUM, "mm2")
5 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (R-102DC, SRT, "Vessel Lumen Cross-Sectional Diameter Increase") $Units = DT (%, UCUM, "%")
6 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (R-102DC, SRT, "Vessel Lumen Cross-Sectional Diameter Increase") $Units = DT (mm, UCUM, "mm")

### TID 3920 Ventricular Analysis

Contains the ventricular functional measurement results.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3920. Ventricular Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122601, DCM, "Ventricular Analysis")
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 3929 “Cardiovascular Analysis Observation Context” | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 3921 “Ventricular Measurements” | 1-n | U |  | $Ventricle = EV (T-32600, SRT, "Left Ventricle")
5 | > | CONTAINS | INCLUDE | DTID 3921 “Ventricular Measurements” | 1-n | U |  | $Ventricle = EV (T-32500, SRT, "Right Ventricle")
6 | > | CONTAINS | INCLUDE | DTID 3925 “Ventricular Thickening Analysis” | 1-n | U |  |
7 | > | CONTAINS | INCLUDE | DTID 3926 “Myocardial Perfusion Analysis” | 1-n | U |  |

### TID 3921 Ventricular Measurements

Ventricular measurement results related to the volume of a ventricle.

Table TID 3921. Parameters

Parameter Name | Parameter Usage
--- | ---
$Ventricle | Describes if either the left or the right ventricle was examined

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3921. Ventricular Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | $Ventricle
3 | > | CONTAINS | INCLUDE | DTID 3922 “Absolute Values of Ventricular Measurements” | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 3923 “BSA-Normalized Ventricular Measurements” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 3924 “Heart Rate-Normalized Ventricular Measurements” | 1 | U |  |

### TID 3922 Absolute Values of Ventricular Measurements

Ventricular measurement results related to the absolute volume of a ventricle.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3922. Absolute Values of Ventricular Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (122608, DCM, "Absolute Values Of Ventricular Measurements") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3833 “Cardiac Ejection Fraction” $ModType = DT (122670, DCM, "Papillary Muscle Included/Excluded") $ModValue = DCID 3821 “Papillary Muscle Included/Excluded” $Method = DCID 3807 “Volume Measurement Methods” $Units = DT (%, UCUM, "%")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3835 “Cardiac Volume Measurements” $ModType = DT (122670, DCM, "Papillary Muscle Included/Excluded") $ModValue = DCID 3821 “Papillary Muscle Included/Excluded” $Units = DT (ml, UCUM, "ml")
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32100, SRT, "Cardiac Output") $ModType = DT (122670, DCM, "Papillary Muscle Included/Excluded") $ModValue = DCID 3821 “Papillary Muscle Included/Excluded” $Units = DT (l/min, UCUM, "l/min")
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $ModType = DT (122670, DCM, "Papillary Muscle Included/Excluded") $ModValue = DCID 3821 “Papillary Muscle Included/Excluded” $Units = DT (g, UCUM, "g")
6 | >> | HAS CONCEPT MOD | CODE | EV (R-4089A, SRT, "Cardiac Cycle Point") | 1 | U |  | DCID 12233 “Cardiac Phase”
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122616, DCM, "Peak Ejection Rate") $Units = DT (ml/s, UCUM, "ml/s")
8 | > | CONTAINS | NUM | EV (122617, DCM, "Peak Ejection Time") | 1 | U |  | UNITS = EV (s, UCUM, "s")
9 | >> | HAS CONCEPT MOD | CODE | EV (122611, DCM, "Reference Point") | 1 | M |  | EV (R-FAB5C, SRT, "End-Diastolic")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122618, DCM, "Peak Filling Rate") $Units = DT (ml/s, UCUM, "ml/s")
11 | > | CONTAINS | NUM | EV (122619, DCM, "Peak Filling Time") | 1 | U |  | UNITS = DT (s, UCUM, "s")
12 | >> | HAS CONCEPT MOD | CODE | EV (122611, DCM, "Reference Point") | 1 | M |  | DT (109070, DCM, "End-Systolic")

### TID 3923 BSA-Normalized Ventricular Measurements

Ventricular measurement results normalized based on the Body Surface Area

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3923. BSA-Normalized Ventricular Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (122609, DCM, "Normalized values of ventricular measurements") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121425, DCM, "Index") | 1 | M |  | DT (8277-6, LN, "Body Surface Area")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (8277-6, LN, "Body Surface Area") $Unit = DT (m2, UCUM, "m2")
4 | >> | INFERRED FROM | CODE | EV (8278-4, LN, "Body Surface Area Formula") | 1 | U |  | BCID 3663 “Body Surface Area Equations”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3835 “Cardiac Volume Measurements” $ModType = EV (121425, DCM, "Index") $ModValue = DT (8277-6, LN, "Body Surface Area") $Units = DT (ml/m2, UCUM, "ml/m2")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32110, SRT, "Cardiac Index") $Units = DT (ml/min/m2, UCUM, "(ml/min) /m2")
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-2 | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8277-6, LN, "Body Surface Area") $Units = DT (g/m2, UCUM, "g/m2")
8 | >> | HAS CONCEPT MOD | CODE | DT (122670, DCM, "Papillary Muscle Included/Excluded") | 1 | U |  | DCID 3821 “Papillary Muscle Included/Excluded”
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122618, DCM, "Peak Filling Rate") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8277-6, LN, "Body Surface Area") $Units = DT (ml/s/m2, UCUM, "(ml/s) /m2")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32070, SRT, "Peak Cardiac Ejection Fraction") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8277-6, LN, "Body Surface Area") $Units = DT (%/m2, UCUM, "%/m2")

### TID 3924 Heart Rate-Normalized Ventricular Measurements

Ventricular measurement results normalized based on the Heart Rate

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3924. Heart Rate-Normalized Ventricular Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (122609, DCM, "Normalized values of ventricular measurements") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121425, DCM, "Index") | 1 | M |  | DT (8867-4, LN, "Heart Rate")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3835 “Cardiac Volume Measurements” $ModType = EV (121425, DCM, "Index") $ModValue = DT (8867-4, LN, "Heart Rate") $Units = DT (ml/{H.B.}/min, UCUM, "ml/BPM")
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32100, SRT, "Cardiac Output") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8867-4, LN, "Heart Rate") $Units = DT (ml/min/{H.B.}/min, UCUM, "(ml/min) /BPM")
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122618, DCM, "Peak Filling Rate") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8867-4, LN, "Heart Rate") $Units = DT (ml/s/{H.B.}/min, UCUM, "(ml/s) /BPM")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32070, SRT, "Peak Cardiac Ejection Fraction") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8867-4, LN, "Heart Rate") $Units = DT (%/{H.B.}/min, UCUM, "%/BPM")

### TID 3925 Ventricular Thickening Analysis

Data of a ventricular wall thickening analysis

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3925. Thickening Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122607, DCM, "Thickening Analysis")
3 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | M |  |
4 | >> | CONTAINS | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | MC | XOR row 5 | DCID 3717 “Myocardial Wall Segments”
5 | >> | CONTAINS | TEXT | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | XOR row 4 |
6 | >> | CONTAINS | INCLUDE | DTID 3909 “Best Illustration of Findings” | 1-n | U |  |
7 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122445, DCM, "Wall Thickness") $ModType = EV (R-4089A, SRT, "Cardiac Cycle Point") $ModValue = DT (R-FAB5C, SRT, "End-Diastolic") $Units = DT (mm, UCUM, "mm")
8 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122445, DCM, "Wall Thickness") $ModType = EV (R-4089A, SRT, "Cardiac Cycle Point") $ModValue = DT (109070, DCM, "End-Systolic") $Units = DT (mm, UCUM, "mm")
9 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122624, DCM, "Wall Thickness Ratio end-systolic to end-diastolic") $Units = DT (%, UCUM, "%")
10 | >> | CONTAINS | CODE | EV (F-32050, SRT, "Cardiac Wall Motion") | 1 | U |  | DCID 3703 “Wall Motion”
11 | >> | CONTAINS | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | U |  | DCID 3704 “Myocardium Wall Morphology Findings”

### TID 3926 Myocardial Perfusion Analysis

Myocardial perfusion analysis results.

Perfusion measurements may be performed either for one or more ventricular segments (row 4) or for substructures inside ventricular segments (row 14)

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3926. Myocardial Perfusion Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122602, DCM, "Myocardial Perfusion Analysis")
3 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | M |  |
4 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | MC | XOR row 6 | DCID 3717 “Myocardial Wall Segments”
5 | >>> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical Modifier") | 1 | U |  | DCID 3843 “Myocardial Subsegment”
6 | >> | HAS CONCEPT MOD | TEXT | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | XOR row 4 |
7 | >> | HAS ACQ CONTEXT | CODE | EV (109054, DCM, "Patient State") | 1 | U |  | DCID 3101 “Cardiac Procedural State Values”
8 | >> | HAS ACQ CONTEXT | INCLUDE | DTID 3106 “Drugs/Contrast Administered” | 1-n | U |  |
9 | >> | CONTAINS | TEXT | EV (122627, DCM, "Curve Fit Method") | 1 | U |  |
10 | >> | CONTAINS | INCLUDE | DTID 3909 “Best Illustration of Findings” | 1-n | U |  |
11 | >> | CONTAINS | TEXT | EV (122628, DCM, "Baseline Result Correction") | 1 | U |  |
12 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3836 “Time-based Perfusion Measurements” $Units = EV (s, UCUM, "s")
13 | >> | CONTAINS | NUM | EV (122640, DCM, "Image Interval") | 1 | U |  | UNITS = EV (ms, UCUM, "ms")
14 | >> | CONTAINS | NUM | EV (122635, DCM, "MR Perfusion Peak") | 1 | U |  | UNITS = DT (1, UCUM, "No units")
15 | >> | CONTAINS | NUM | EV (122636, DCM, "MR Perfusion Slope") | 1 | U |  | UNITS = DT (1, UCUM, "No units")
16 | >> | CONTAINS | NUM | EV (122637, DCM, "MR Perfusion Time Integral") | 1 | U |  | UNITS = DT (1, UCUM, "No units")
17 | >> | CONTAINS | CONTAINER | EV (125007, DCM, "Measurement Group") | 1-n | U |  |
18 | >>> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3836 “Time-based Perfusion Measurements” $Units = EV (s, UCUM, "s")
19 | >>> | CONTAINS | NUM | EV (122635, DCM, "MR Perfusion Peak") | 1 | U |  | UNITS = DT (1, UCUM, "No units")
20 | >>> | CONTAINS | NUM | EV (122636, DCM, "MR Perfusion Slope") | 1 | U |  | UNITS = DT (1, UCUM, "No units")
21 | >>> | CONTAINS | NUM | EV (122637, DCM, "MR Perfusion Time Integral") | 1 | U |  | UNITS = DT (1, UCUM, "No units")
22 | >> | CONTAINS | CODE | EV (122664, DCM, "Late Contrast Enhancement") | 1 | U |  | DCID 230 “Yes-No”
23 | >>> | HAS ACQ CONTEXT | NUM | EV (122665, DCM, "Time after start of injection of contrast bolus") | 1 | M |  | UNITS = DT (s, UCUM, "s")
24 | >>> | HAS ACQ CONTEXT | NUM | EV (122668, DCM, "Time interval since detection of contrast bolus") | 1 | U |  | UNITS = DT (s, UCUM, "s")

Content Item Descriptions

Row 12 | Image Interval is appropriate only for equally time-spaced images

### TID 3927 Report Summary

Contains summary elements based on the findings of the report

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 3927. Report Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | BCID 7001 “Diagnostic Imaging Report Headings” | 1 | M |  |
2 | > | CONTAINS | CODE | BCID 7002 “Diagnostic Imaging Report Elements” | 1-n | U |  |
3 | >> |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1-n | U |  |
4 | >> |  | INCLUDE | DTID 321 “Waveform or Temporal Coordinates” | 1-n | U |  |
5 | > | CONTAINS | TEXT | BCID 7002 “Diagnostic Imaging Report Elements” | 1-n | U |  |
6 | >> |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1-n | U |  |
7 | >> |  | INCLUDE | DTID 321 “Waveform or Temporal Coordinates” | 1-n | U |  |

### TID 3929 Cardiovascular Analysis Observation Context

Defines the observation context for cardiovascular Functional Analysis

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3929. Cardiovascular Analysis Observation Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS OBS CONTEXT | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  | UNITS = DT ({H.B.}/min, UCUM, "BPM")
2 |  | HAS OBS CONTEXT | CODE | EV (8884-9, LN, "Cardiac Rhythm") | 1 | U |  | DCID 3826 “Heart Rhythm”
3 |  | HAS OBS CONTEXT | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | U |  | UNITS = DT (mm[Hg], UCUM, "mmHg")
4 |  | HAS OBS CONTEXT | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | U |  | UNITS = DT (mm[Hg], UCUM, "mmHg")
5 |  | HAS OBS CONTEXT | CODE | EV (F-043E6, SRT, "Respiration Observable") | 1 | U |  | DCID 3823 “Respiratory Status”
6 |  | HAS ACQ CONTEXT | INCLUDE | DTID 3106 “Drugs/Contrast Administered” | 1-n | U |  |

### TID 3990 Two Dimensional Measurement Graph

Generic Template representing arbitrary two-dimensional graphs.

Table TID 3990. Parameters

Parameter Name | Parameter Usage
--- | ---
$MeasurementGraph | Describes what the graph is about
$X-Concept | Concept of the X-Axis of the graph
$Y-Concept | Concept of the Y-Axis of the graph
$X-AxisUnit | Unit of the x-axis data elements
$Y-AxisUnit | Unit of the y-axis data elements

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 3990. Two Dimensional Measurement Graph

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | $MeasurementGraph | 1 | M |  |
2 | > | CONTAINS | CODE | EV (122698, DCM, "X-Concept") | 1 | M |  | $X-Concept
3 | > | CONTAINS | CODE | EV (122699, DCM, "Y-Concept") | 1 | M |  | $Y-Concept
4 | > | CONTAINS | CONTAINER | no concept name | 1-n | MC | IF Row 7, 8, or 9 not present |
5 | >> | CONTAINS | NUM | $X-Concept | 1 | M |  | UNITS = $X-AxisUnit
6 | >> | CONTAINS | NUM | $Y-Concept | 1 | M |  | UNITS = $Y-AxisUnit
7 | > | CONTAINS | IMAGE | $MeasurementGraph | 1 | U |  |
8 | > | CONTAINS | WAVEFORM | $MeasurementGraph | 1 | U |  |
9 | > | CONTAINS | COMPOSITE | $MeasurementGraph | 1 | U |  |

Content Item Descriptions

Rows 5-6 | The X-Concept values shall be monotonically increasing.
Row 7 | Secondary Capture Image containing a bitmap representation of the graph
Row 8 | Waveform containing a representation of the graph
Row 9 | Composite Object containing a rendered representation of the graph

## Mammography CAD SR IOD Templates

The Templates that comprise the Mammography CAD SR IOD are interconnected as in Figure A-8:

Figure A-8. Mammography CAD SR IOD Template Structure

In Figure A-8, '···' indicates possible recursive application of subordinate Templates.

### TID 4000 Mammography CAD Document Root

This Template forms the top of a content tree that allows a mammography CAD device to describe the results of detection and analysis of Mammographic evidence. This Template, together with its subordinate Templates, describes both the results for presentation to radiologists and partial product results for consumption by mammography CAD devices in subsequent mammography CAD reports.

This Template defines a Container that contains an Image Library, the mammography CAD results, and summaries of the detection and analysis algorithms performed. The Image Library contains the Image SOP Class and Instance UIDs, and selected attributes for each image referenced in either the algorithm summaries or mammography CAD results.

The Summary of Detections and Summary of Analyses sub-trees gather lists of algorithms attempted, grouped by success/failure status. Algorithms not attempted are not mentioned in these sub-trees. This information forms the basis for understanding why a mammography CAD report may produce no (or fewer than anticipated) results. Mammography CAD results are constructed bottom-up, starting from Single Image Findings (see TID 4006 “Mammography CAD Single Image Finding”), associated as Composite Features (see TID 4004 “Mammography CAD Composite Feature”), and from which Individual and Overall Impressions are formed.

See Figure E.1-1 “Top Levels of Mammography CAD SR Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  Yes

Table TID 4000. Mammography CAD Document Root

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111036, DCM, "Mammography CAD Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | CONTAINS | CONTAINER | EV (111028, DCM, "Image Library") | 1 | M |  |
4 | >> | CONTAINS | INCLUDE | DTID 4020 “CAD Image Library Entry” | 1-n | M |  | $ImageLaterality = DCID 6022 “Side” $ImageView = DCID 4014 “View for Mammography” $ImageViewMod = DCID 4015 “View Modifier for Mammography”
5 | > | CONTAINS | INCLUDE | DTID 4001 “Mammography CAD Overall Impression/Recommendation” | 1 | M |  |
6 | > | CONTAINS | CODE | EV (111064, DCM, "Summary of Detections") | 1 | M |  | DCID 6042 “Status of Results”
7 | >> | INFERRED FROM | INCLUDE | DTID 4015 “CAD Detections Performed” | 1 | MC | Shall be present unless the value of (111064, DCM, "Summary of Detections") is (111225, DCM, "Not Attempted") | $DetectionCode = DCID 6014 “Mammography Single Image Finding”
8 | > | CONTAINS | CODE | EV (111065, DCM, "Summary of Analyses") | 1 | M |  | DCID 6042 “Status of Results”
9 | >> | INFERRED FROM | INCLUDE | DTID 4016 “CAD Analyses Performed” | 1 | MC | Shall be present unless the value of (111065, DCM, "Summary of Analyses") is (111225, DCM, "Not Attempted") | $AnalysisCode = DCID 6043 “Types of Mammography CAD Analysis”

Content Item Descriptions

Image Library | The "Image Library" section of the Content Tree (TID 4000 “Mammography CAD Document Root”, row 3) shall include all Image SOP Instances from the Current Requested Procedure Evidence Sequence (0040,A375) attribute of the SR Document General module. If a portion of another instance of a Mammography CAD SR IOD is duplicated in the "Overall Impression/ Recommendation" section of the Content Tree, the "Image Library" shall also include all Image Library Entries referenced from the duplicated portions of the Mammography CAD SR.
Detections Performed | The "Detections Performed" and "Analyses Performed" sections of the Content Tree (TID 4000 “Mammography CAD Document Root”, rows 6 and 8) together shall reference all Image SOP Instances included in the Current Requested Procedure Evidence Sequence (0040,A375) attribute of the SR Document General module.
Analyses Performed

### TID 4001 Mammography CAD Overall Impression/Recommendation

This Template forms the top of the mammography CAD results sub-tree. The contents of this Template describe the overall impression the mammography CAD device had for the mammographic evidence presented and any recommendations that the mammography CAD device made. The details of the overall impression and recommendation are expressed in this instance of the Mammography CAD Impression/Recommendation Body (see TID 4002 “Mammography CAD Impression/Recommendation Body”). The data from which the details are inferred, are expressed in the Mammography CAD Individual Impression/Recommendations (see TID 4003 “Mammography CAD Individual Impression/Recommendation”), of which there may be several.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4001. Mammography CAD Overall Impression/Recommendation

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111017, DCM, "CAD Processing and Findings Summary") | 1 | M |  | DCID 6047 “CAD Processing and Findings Summary”
2 | > | HAS PROPERTIES | INCLUDE | DTID 4002 “Mammography CAD Impression/Recommendation Body” | 1 | U |  |
3 | > | INFERRED FROM | INCLUDE | DTID 4003 “Mammography CAD Individual Impression/Recommendation” | 1-n | MC | Shall be present if 1 or more (111059, DCM, "Single Image Finding") or (111015, DCM, "Composite Feature") Content Items are reported. |

Content Item Descriptions

CAD Processing and Findings Summary | This code value is used to express if and why the Overall Impression/Recommendation sub-tree is empty. The Summary of Detections and Summary of Analyses sub-trees of the Document Root node contain detail about which (if any) algorithms succeeded or failed. If the code value indicates that there were no findings, then the code value can be used to determine whether mammography CAD processing occurred successfully, without parsing the Summary of Detections and Summary of Analyses sub-trees.
Row 3 | There are no constraints regarding the 1-n multiplicity of the inclusion of TID 4003 “Mammography CAD Individual Impression/Recommendation” or its underlying structure, other than the TID 4001 “Mammography CAD Overall Impression/Recommendation” and TID 4003 “Mammography CAD Individual Impression/Recommendation” requirements. Individual Impression/Recommendation containers may be organized, for example per image, per finding or composite feature, or some combination thereof.

### TID 4002 Mammography CAD Impression/Recommendation Body

The details of an impression and recommendation are expressed in this Template. It is applied to both Mammography CAD Overall Impression/Recommendation (TID 4001 “Mammography CAD Overall Impression/Recommendation”) and Mammography CAD Individual Impression/Recommendation (TID 4003 “Mammography CAD Individual Impression/Recommendation”).

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4002. Mammography CAD Impression/Recommendation Body

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111005, DCM, "Assessment Category") | 1-n | MC | At least one of rows 1, 3, 5, 6, 8, 9 shall be present. | DCID 6026 “Mammography Assessment”
2 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 6022 “Side”
3 |  |  | CODE | EV (111023, DCM, "Differential Diagnosis/ Impression") | 1-n | MC | At least one of rows 1, 3, 5, 6, 8, 9 shall be present. | DCID 6002 “Change Since Last Mammogram or Prior Surgery”
4 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 6022 “Side”
5 |  |  | TEXT | EV (111033, DCM, "Impression Description") | 1 | MC | At least one of rows 1, 3, 5, 6, 8, 9 shall be present. |
6 |  |  | CODE | EV (111053, DCM, "Recommended Follow-up") | 1-n | MC | At least one of rows 1, 3, 5, 6, 8, 9 shall be present. | DCID 6028 “Mammography Recommended Follow-up”
7 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 6022 “Side”
8 |  |  | NUM | EV (111055, DCM, "Recommended Follow-up Interval") | 1 | MC | At least one of rows 1, 3, 5, 6, 8, 9 shall be present. May be present only if (111054, DCM, "Recommended Follow-up Date") is not present. | UNITS = DCID 6046 “Units of Follow-up Interval” Values = Integer ≥ 0, where 0 = immediate follow-up
9 |  |  | DATE | EV (111054, DCM, "Recommended Follow-up Date") | 1 | MC | At least one of rows 1, 3, 5, 6, 8, 9 shall be present. May be present only if (111055, DCM, "Recommended Follow-up Interval") is not present. | Shall be later than date of exam
10 |  |  | NUM | EV (111013, DCM, "Certainty of impression") | 1 | UC | May be present only if (111005, DCM, "Assessment Category"), (111023, DCM, "Differential Diagnosis/Impression") or (111033, DCM, "Impression Description") is present. | UNITS = EV (%, UCUM, "Percent") Values = 0 - 100
11 |  |  | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1-n | M |  |
12 |  |  | NUM | DCID 6142 “Calculated Value” | 1-n | U |  |
13 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 6022 “Side”
14 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | M |  | DCID 6140 “Calculation Methods”
15 | > | INFERRED FROM | TEXT | EV (112034, DCM, "Calculation Description") | 1 | U |  |

Content Item Descriptions

Certainty of Impression | The certainty that the device populating the Mammography CAD SR report places on this impression, where 0 equals no certainty and 100 equals certainty.
Impression Description | Free-form text describing the overall or an individual impression

### TID 4003 Mammography CAD Individual Impression/Recommendation

This Template collects an individual impression the mammography CAD device had for a lesion, non-lesion object, or correlation of related objects. The details of the impression and recommendation are expressed in the Mammography CAD Impression/Recommendation Body (see TID 4002 “Mammography CAD Impression/Recommendation Body”). The data from which the details are inferred are expressed in the Composite Features (see TID 4004 “Mammography CAD Composite Feature”) and/or Single Image Findings (see TID 4006 “Mammography CAD Single Image Finding”) of which there may be several.

The sub-tree headed by this Template is illustrated in Figure E.1-3 “Example of Individual Impression/Recommendation Levels of Mammography CAD SR Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4003. Mammography CAD Individual Impression/Recommendation

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111034, DCM, "Individual Impression/ Recommendation") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
3 | > | CONTAINS | INCLUDE | DTID 4002 “Mammography CAD Impression/Recommendation Body” | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 4004 “Mammography CAD Composite Feature” | 1-n | MC | At least one of rows 4, 5 shall be present. |
5 | > | CONTAINS | INCLUDE | DTID 4006 “Mammography CAD Single Image Finding” | 1-n | MC | At least one of rows 4, 5 shall be present. |

Content Item Descriptions

Rendering Intent | This Content Item constrains the SCP receiving the Mammography CAD SR IOD in its use of the contents of this Template and its Target Content Items. Mammography CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent mammography CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .

### TID 4004 Mammography CAD Composite Feature

This Template collects a composite feature for a lesion, non-lesion object, or correlation of related objects. The details of the composition are expressed in the Mammography CAD Composite Feature Body (see TID 4005 “Mammography CAD Composite Feature Body”). The data from which the details are inferred, are expressed in the Composite Features (see TID 4004 “Mammography CAD Composite Feature”) and/or Single Image Findings (see TID 4006 “Mammography CAD Single Image Finding”), of which there may be several.

A Composite Feature shall be INFERRED FROM any combination of two or more Composite Features or Single Image Findings or mixture thereof.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4004. Mammography CAD Composite Feature

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111015, DCM, "Composite Feature") | 1 | M |  | DCID 6016 “Mammography Composite Feature”
2 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
4 | > | HAS PROPERTIES | INCLUDE | DTID 4005 “Mammography CAD Composite Feature Body” | 1 | M |  |
5 | > | INFERRED FROM | INCLUDE | DTID 4004 “Mammography CAD Composite Feature” | 1-n | MC | At least two items shall be present: two of row 5, two of row 6, or one of each. |
6 | > | INFERRED FROM | INCLUDE | DTID 4006 “Mammography CAD Single Image Finding” | 1-n | MC | At least two items shall be present: two of row 5, two of row 6, or one of each. |
7 | > | HAS OBS CONTEXT | INCLUDE | DTID 4022 “CAD Observation Context” | 1 | MC | Shall be present only if this feature is incorporated from a different report than its parent. |

Content Item Descriptions

Rendering Intent | This Content Item constrains the SCP receiving the Mammography CAD SR IOD in its use of the contents of this Template and its Target Content Items. Mammography CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent mammography CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .

### TID 4005 Mammography CAD Composite Feature Body

The details of a composite feature are expressed in this Template. It is applied to Mammography CAD Composite Feature (TID 4004 “Mammography CAD Composite Feature”).

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4005. Mammography CAD Composite Feature Body

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111016, DCM, "Composite type") | 1 | M |  | DCID 6035 “Composite Feature Relations” The value shall be (111155, DCM, "Target content items are related contra-laterally") if the parent Content Item has code value (F-01792, SRT, "Focal asymmetric breast tissue") or (F-01793, SRT, "Asymmetric breast tissue").
2 |  |  | CODE | EV (111057, DCM, "Scope of Feature") | 1 | M |  | DCID 6036 “Scope of Feature”
3 |  |  | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
4 |  |  | NUM | EV (111011, DCM, "Certainty of Feature") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
5 |  |  | NUM | EV (111047, DCM, "Probability of cancer") | 1 | UC | May be present only if value of parent is not (111102, DCM, "Non-lesion") | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
6 |  |  | CODE | EV (111042, DCM, "Pathology") | 1-n | U |  | BCID 6030 “Mammography Pathology Codes”
7 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1400 “Linear Measurement” shall be used.
8 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1401 “Area Measurement” shall be used.
9 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1402 “Volume Measurement” shall be used.
10 |  |  | INCLUDE | DTID 4021 “Mammography CAD Geometry” | 1-n | U |  |
11 |  |  | NUM | DCID 6037 “Mammography Quantitative Temporal Difference Type” | 1-n | UC | May be present only if the value of (111016, DCM, "Composite type") is (111153, DCM, "Target content items are related temporally") | UNITS = DCID 7460 “Units of Linear Measurement” UNITS = DCID 7461 “Units of Area Measurement” UNITS = DCID 7462 “Units of Volume Measurement” UNITS = DT (1, UCUM, "no units")
12 | > | R-INFERRED FROM | NUM |  | 2 | U |  | The referenced numeric values shall have the same Concept Name. Their UNITS shall be the same as row 11
13 |  |  | CODE | EV (111049, DCM, "Qualitative Difference") | 1-n | UC | May be present only if the value of (111016, DCM, "Composite type") is (111153, DCM, "Target content items are related temporally") | DCID 6038 “Mammography Qualitative Temporal Difference Type”
14 | > | HAS PROPERTIES | TEXT | EV (111021, DCM, "Description of Change") | 1 | U |  |
15 | > | R-INFERRED FROM | CODE |  | 2 | M |  | The referenced code values shall have the same Concept Name and be from the same context group.
16 |  |  | CODE | EV (111048, DCM, "Quadrant location") | 1 | U |  | DCID 6020 “Quadrant Location”
17 |  |  | CODE | EV (111014, DCM, "Clockface or region") | 1 | U |  | DCID 6018 “Clockface Location or Region”
18 |  |  | CODE | EV (111020, DCM, "Depth") | 1 | U |  | DCID 6024 “Depth”
19 |  |  | CODE | EV (111035, DCM, "Lesion Density") | 1 | UC | May be present only if value of parent is (F-01791, SRT, "Mammographic breast mass") or (F-01796, SRT, "Mammography breast density") | DCID 6008 “Density Modifier”
20 |  |  | CODE | EV (M-020F9, SRT, "Shape") | 1 | UC | May be present only if value of parent is (F-01791, SRT, "Mammographic breast mass") or (F-01796, SRT, "Mammography breast density") | DCID 6004 “Mammography Characteristics of Shape”
21 |  |  | CODE | EV (111037, DCM, "Margins") | 1-n | UC | May be present only if value of parent is (F-01791, SRT, "Mammographic breast mass") or (F-01796, SRT, "Mammography breast density") | DCID 6006 “Mammography Characteristics of Margin”
22 |  |  | CODE | EV (111009, DCM, "Calcification Type") | 1-n | UC | May be present only if value of parent is (F-01775, SRT, "Calcification Cluster") or (F-01776, SRT, "Individual Calcification") | DCID 6010 “Mammography Calcification Types”
23 |  |  | CODE | EV (111008, DCM, "Calcification Distribution") | 1 | UC | May be present only if value of parent is (F-01775, SRT, "Calcification Cluster") | DCID 6012 “Calcification Distribution Modifier”
24 |  |  | NUM | EV (111038, DCM, "Number of calcifications") | 1 | UC | May be present only if value of parent is (F-01775, SRT, "Calcification Cluster") | UNITS = EV (1, UCUM, "no units") Value = Integer 1 - n
25 |  |  | NUM | DCID 6142 “Calculated Value” | 1-n | U |  |
26 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | M |  | DCID 6140 “Calculation Methods”
27 | > | INFERRED FROM | TEXT | EV (112034, DCM, "Calculation Description") | 1 | U |  |

Content Item Descriptions

Certainty of Feature | The likelihood that the feature analyzed, and classified by the CODE specified in the Composite Feature parent Template, is in fact that type of feature.
Volume Measurement | If dimensions for a volume are to be stated in terms of length, width, and depth, then one shall use 3 instances of TID 1400 “Linear Measurement”.
Row 11 | Values ≤ 0 are allowed. The two referenced numeric values are Target Content Items of the first generation Composite Feature or Single Image Finding children of this composite feature. Given the equation, A - B, the value representing A shall be referenced first.
Qualitative Difference | The two referenced code values are Target Content Items of the first generation Composite Feature or Single Image Finding children of this composite feature.

### TID 4006 Mammography CAD Single Image Finding

This Template describes a single image finding for a lesion or other object. The details of the finding are expressed in this Template and/or more specific Templates. The details from which a single image Calcification Cluster is inferred may be expressed in a number of Single Image Findings (see TID 4006 “Mammography CAD Single Image Finding”) of type Individual Calcification.

A Single Image Finding of type Breast Composition may be INFERRED FROM by-reference to a Single Image Finding of type Breast Geometry.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4006. Mammography CAD Single Image Finding

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111059, DCM, "Single Image Finding") | 1 | M |  | DCID 6014 “Mammography Single Image Finding”
2 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
3 | >> | HAS PROPERTIES | NUM | EV (111071, DCM, "CAD Operating Point") | 1 | UC | IFF value of row 2 is (111151, DCM, "Presentation Optional") and row 9 of TID 4017 “CAD Detection Performed” is present | UNITS = DT ({1:n}, UCUM, "range: 1:n"), where n is the maximum specified in Row 9 of TID 4017 “CAD Detection Performed”. Value is restricted to being an integer
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
5 | > | HAS PROPERTIES | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
6 | > | HAS PROPERTIES | NUM | EV (111012, DCM, "Certainty of Finding") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
7 | > | HAS PROPERTIES | NUM | EV (111047, DCM, "Probability of cancer") | 1 | UC | May be present unless value of parent is (F-01710, SRT, "Breast composition"), (111100, DCM, "Breast geometry"), (T-04100, SRT, "Nipple"), (111099, DCM, "Selected region"), (111101, DCM, "Image quality") or (111102, DCM, "Non-lesion") | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
8 | > | HAS PROPERTIES | INCLUDE | DTID 4021 “Mammography CAD Geometry” | 1 | MC | Shall be present unless value of parent is (F-01710, SRT, "Breast composition"), (111100, DCM, "Breast geometry") or (111101, DCM, "Image quality") |
9 | > | HAS PROPERTIES | INCLUDE | DTID 4007 “Mammography CAD Breast Composition” | 1 | MC | Shall be present only if value of parent is (F-01710, SRT, "Breast composition") |
10 | > | R-INFERRED FROM | CODE |  | 1-n | UC | May be present only if value of parent is (F-01710, SRT, "Breast composition") | Shall reference a (111059, DCM, "Single Image Finding") of value: EV (111100, DCM, "Breast geometry")
11 | > | HAS PROPERTIES | INCLUDE | DTID 4008 “Mammography CAD Breast Geometry” | 1 | MC | Shall be present only if value of parent is (111100, DCM, "Breast geometry") |
12 | > | HAS PROPERTIES | INCLUDE | DTID 4009 “Mammography CAD Individual Calcification” | 1 | UC | May be present only if value of parent is (F-01776, SRT, "Individual Calcification") |
13 | > | HAS PROPERTIES | INCLUDE | DTID 4010 “Mammography CAD Calcification Cluster” | 1 | UC | May be present only if value of parent is (F-01775, SRT, "Calcification Cluster") |
14 | > | HAS PROPERTIES | INCLUDE | DTID 4011 “Mammography CAD Density” | 1 | UC | May be present only if value of parent is (F-01796, SRT, "Mammography breast density") |
15 | > | HAS PROPERTIES | CODE | EV (111297, DCM, "Nipple Characteristic") | 1 | UC | May be present only if value of parent is (T-04100, SRT, "Nipple") | DCID 6039 “Nipple Characteristic”
16 | > | HAS PROPERTIES | INCLUDE | DTID 4012 “Mammography CAD Non-lesion” | 1 | MC | Shall be present only if value of parent is (111102, DCM, "Non-lesion") |
17 | > | HAS PROPERTIES | INCLUDE | DTID 4013 “Mammography CAD Selected Region” | 1 | MC | Shall be present only if value of parent is (111099, DCM, "Selected Region") |
18 | > | R-INFERRED FROM | IMAGE |  | 1 | MC | IF value of parent is (111101, DCM, "Image quality") and IFF row 19 is not present | Shall reference an IMAGE Content Item in the (111028, DCM, "Image Library")
19 | > | HAS PROPERTIES | SCOORD | EV (111030, DCM, "Image Region") | 1-n | MC | IF value of parent is (111101, DCM, "Image quality") and IFF row 18 is not present |
20 | >> | R-SELECTED FROM | IMAGE |  | 1 | M |  | All the (111030, DCM, "Image Region") Content Items in a single invocation of this Template shall reference the same IMAGE Content Item in the (111028, DCM, "Image Library")
21 | > | HAS PROPERTIES | INCLUDE | DTID 4014 “CAD Image Quality” | 1-n | MC | Shall be present only if value of parent is (111101, DCM, "Image quality") | $QualityFinding = DCID 6041 “Mammography Image Quality Finding” $QualityStandard = DCID 6045 “Mammography Types of Quality Control Standard”
22 | > | HAS PROPERTIES | NUM | DCID 6142 “Calculated Value” | 1-n | U |  |
23 | >> | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | M |  | DCID 6140 “Calculation Methods”
24 | >> | INFERRED FROM | TEXT | EV (112034, DCM, "Calculation Description") | 1 | U |  |
25 | > | INFERRED FROM | INCLUDE | DTID 4006 “Mammography CAD Single Image Finding” | 1-n | UC | May be present only if value of parent is (F-01775, SRT, "Calcification Cluster") | EV (F-01776, SRT, "Individual Calcification")
26 | > | HAS OBS CONTEXT | INCLUDE | DTID 4022 “CAD Observation Context” | 1 | MC | Shall be present only if this finding is incorporated from a different report than its parent. |

Content Item Descriptions

Rendering Intent | This Content Item constrains the SCP receiving the Mammography CAD SR IOD in its use of the contents of this Template and its Target Content Items. Mammography CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent mammography CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .
CAD Operating Point | Additional information to use when Rendering Intent is "Presentation Optional". A CAD Operating Point of zero is not sent, and is encoded as a Rendering Intent of "Presentation Required". See Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 and Section E.4 “CAD Operating Point” in PS3.17 .
Single Image Finding | A Single Image Finding (whose parent is a Single Image Finding of type Calcification Cluster) allows one level of nesting for the definition of individual calcifications within the cluster. To use this Template recursively, this Single Image Finding code value shall be "Individual Calcification".
Certainty of Finding | The likelihood that the finding detected, and classified by the CODE specified in the Single Image Finding parent Template, is in fact that type of finding.

### TID 4007 Mammography CAD Breast Composition

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4007. Mammography CAD Breast Composition

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (F-01710, SRT, "Breast composition") | 1 | MC | At least one of row 1 or 2 shall be present | DCID 6000 “Overall Breast Composition”
2 |  |  | NUM | EV (111046, DCM, "Percent Fibroglandular Tissue") | 1 | MC | At least one of row 1 or 2 shall be present | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100

Content Item Descriptions

Percent Fibroglandular Tissue | Percent of breast area that is mammographically dense, excluding pectoralis muscle.

### TID 4008 Mammography CAD Breast Geometry

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4008. Mammography CAD Breast Geometry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | SCOORD | EV (111007, DCM, "Breast Outline Including Pectoral Muscle Tissue") | 1 | M |  | GRAPHIC TYPE = {POLYLINE}
2 | > | R-SELECTED FROM | IMAGE |  | 1 | M |  | Shall reference an IMAGE Content Item in the (111028, DCM, "Image Library")
3 |  |  | SCOORD | EV (111045, DCM, "Pectoral Muscle Outline") | 1 | U |  | GRAPHIC TYPE = {POLYLINE}
4 | > | R-SELECTED FROM | IMAGE |  | 1 | M |  | Shall reference the same node as row 2

Content Item Descriptions

Figure A-8a. Example of Breast Outline Including Pectoral Muscle Tissue | Figure A-8b. Example of Pectoral Muscle Outline

### TID 4009 Mammography CAD Individual Calcification

This Template provides the detail specific to an individual calcification.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4009. Mammography CAD Individual Calcification

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111009, DCM, "Calcification Type") | 1-n | MC | At least one of rows 1, 2, 3 shall be present | DCID 6010 “Mammography Calcification Types”
2 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | MC | At least one of rows 1, 2, 3 shall be present | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1400 “Linear Measurement” shall be used.
3 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | MC | At least one of rows 1, 2, 3 shall be present | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1401 “Area Measurement” shall be used.
4 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1402 “Volume Measurement” shall be used.

### TID 4010 Mammography CAD Calcification Cluster

This Template provides the detail specific to a calcification cluster.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4010. Mammography CAD Calcification Cluster

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111009, DCM, "Calcification Type") | 1-n | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | DCID 6010 “Mammography Calcification Types”
2 |  |  | CODE | EV (111008, DCM, "Calcification Distribution") | 1 | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | DCID 6012 “Calcification Distribution Modifier”
3 |  |  | NUM | EV (111038, DCM, "Number of calcifications") | 1 | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | UNITS = EV (1, UCUM, "no units") Value = Integer > = 1
4 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1400 “Linear Measurement” shall be used.
5 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1401 “Area Measurement” shall be used.
6 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1402 “Volume Measurement” shall be used.

### TID 4011 Mammography CAD Density

This Template provides the detail specific to a density.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4011. Mammography CAD Density

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111035, DCM, "Lesion Density") | 1 | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | DCID 6008 “Density Modifier”
2 |  |  | CODE | EV (M-020F9, SRT, "Shape") | 1 | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | DCID 6004 “Mammography Characteristics of Shape”
3 |  |  | CODE | EV (111037, DCM, "Margins") | 1-n | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | DCID 6006 “Mammography Characteristics of Margin”
4 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1400 “Linear Measurement” shall be used.
5 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1401 “Area Measurement” shall be used.
6 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1402 “Volume Measurement” shall be used.

### TID 4012 Mammography CAD Non-lesion

This Template provides the detail specific to a finding other than a lesion (see CID 6040 “Non-lesion Object Type”).

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4012. Mammography CAD Non-lesion

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111039, DCM, "Object type") | 1 | M |  | DCID 6040 “Non-lesion Object Type”
2 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1400 “Linear Measurement” shall be used.
3 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1401 “Area Measurement” shall be used.
4 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1402 “Volume Measurement” shall be used.

### TID 4013 Mammography CAD Selected Region

This Template provides the detail specific to a selected region. A selected region is any mammography CAD derived arbitrary region of the image, whether within the breast outline or not. This can be use to delineate regions such as the intramammary fold.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4013. Mammography CAD Selected Region

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (111058, DCM, "Selected Region Description") | 1 | M |  |
2 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1400 “Linear Measurement” shall be used.
3 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1401 “Area Measurement” shall be used.
4 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID 1402 “Volume Measurement” shall be used.

### TID 4014 CAD Image Quality

This Template provides the detail specific to image quality. It allows the encoding of descriptors of image quality (e.g., CID 6041 “Mammography Image Quality Finding”) for a given image or region of an image. For instance, images with partial motion blur can be identified with the region noted.

Table TID 4014. Parameters

Parameter Name | Parameter Usage
--- | ---
$QualityFinding | Coded term or Context Group for Quality Finding
$QualityStandard | Coded term or Context Group for Quality Control Standard

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4014. CAD Image Quality

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111052, DCM, "Quality Finding") | 1 | M |  | $QualityFinding
2 | > | HAS PROPERTIES | CODE | EV (111050, DCM, "Quality Assessment") | 1 | U |  | DCID 6044 “Types of Image Quality Assessment”
3 | > | HAS PROPERTIES | CODE | EV (111051, DCM, "Quality Control Standard") | 1 | UC | Shall be present if row 2 is present. | $QualityStandard
4 | > | HAS PROPERTIES | NUM | EV (111029, DCM, "Image Quality Rating") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100

Content Item Descriptions

Image Quality Rating | A numeric value in the range 0 to 100, inclusive, where 0 is worst quality and 100 is best quality.

### TID 4015 CAD Detections Performed

This Template gathers two lists of detection algorithms attempted, grouped by success/failure status. Algorithms not attempted are not mentioned in this sub-tree of the Document Root (e.g., TID 4000 “Mammography CAD Document Root”). This information forms the basis for understanding why a CAD report may produce no (or fewer than anticipated) detection results.

The sub-tree formed by this Template is illustrated in Figure E.1-2 “Summary of Detections and Analyses Levels of Mammography CAD SR Content Tree” in PS3.17 .

Table TID 4015. Parameters

Parameter Name | Parameter Usage
--- | ---
$DetectionCode | Coded term or Context Group for Detection Performed

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4015. CAD Detections Performed

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111063, DCM, "Successful Detections") | 1 | MC | Shall be present only if value of parent is (111222, DCM, "Succeeded") or (111223, DCM, "Partially Succeeded") |
2 | > | CONTAINS | INCLUDE | DTID 4017 “CAD Detection Performed” | 1-n | M |  | $DetectionCode = $DetectionCode
3 |  |  | CONTAINER | EV (111025, DCM, "Failed Detections") | 1 | MC | Shall be present only if value of parent is (111224, DCM, "Failed") or (111223, DCM, "Partially Succeeded") |
4 | > | CONTAINS | INCLUDE | DTID 4017 “CAD Detection Performed” | 1-n | M |  | $DetectionCode = $DetectionCode

### TID 4016 CAD Analyses Performed

This Template gathers two lists of analysis algorithms attempted, grouped by success/failure status. Algorithms not attempted are not mentioned in this sub-tree of the Document Root (e.g., TID 4000 “Mammography CAD Document Root”). This information forms the basis for understanding why a CAD report may produce no (or fewer than anticipated) analysis results.

The sub-tree formed by this Template is illustrated in Figure E.1-2 “Summary of Detections and Analyses Levels of Mammography CAD SR Content Tree” in PS3.17 .

Table TID 4016. Parameters

Parameter Name | Parameter Usage
--- | ---
$AnalysisCode | Coded term or Context Group for Analysis Performed

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4016. CAD Analyses Performed

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111062, DCM, "Successful Analyses") | 1 | MC | Shall be present only if value of parent is (111222, DCM, "Succeeded") or (111223, DCM, "Partially Succeeded") |
2 | > | CONTAINS | INCLUDE | DTID 4018 “CAD Analysis Performed” | 1-n | M |  | $AnalysisCode = $AnalysisCode
3 |  |  | CONTAINER | EV (111024, DCM, "Failed Analyses") | 1 | MC | Shall be present only if value of parent is (111224, DCM, "Failed") or (111223, DCM, "Partially Succeeded") |
4 | > | CONTAINS | INCLUDE | DTID 4018 “CAD Analysis Performed” | 1-n | M |  | $AnalysisCode = $AnalysisCode

### TID 4017 CAD Detection Performed

This Template fully identifies a detection algorithm and the images and/or image regions on which it operated (see TID 4015 “CAD Detections Performed”).

Table TID 4017. Parameters

Parameter Name | Parameter Usage
--- | ---
$DetectionCode | Coded term or Context Group for Detection Performed

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4017. CAD Detection Performed

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111022, DCM, "Detection Performed") | 1 | M |  | $DetectionCode
2 | > | HAS PROPERTIES | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
3 | > | HAS PROPERTIES | IMAGE |  | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present |
4 | > | R-HAS PROPERTIES | IMAGE |  | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present | Shall reference IMAGE Content Item(s) in the (111028, DCM, "Image Library")
5 | > | HAS PROPERTIES | UIDREF | EV (112002, DCM, "Series Instance UID") | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present |
6 | > | HAS PROPERTIES | SCOORD | EV (111030, DCM, "Image Region") | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present |
7 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR row 8 |
8 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR row 7 | Shall reference an IMAGE Content Item in the (111028, DCM, "Image Library")
9 | > |  | INCLUDE | DTID 4023 “CAD Operating Points” | 1 | U |  |

Content Item Descriptions

CAD Algorithm Identification | If more than one detection algorithm has the same "Detection Performed" code value (e.g., CID 6014 “Mammography Single Image Finding”) then the "CAD Algorithm Identification" shall unambiguously distinguish between algorithms.
Rows 5, 6 | Mammography CAD SR: When this Template is invoked for the Mammography CAD SR, the Image Library is mandatory, thus only row 4 and/or row 6 shall be present. Chest CAD SR: When this Template is invoked for the Chest CAD SR, the Image Library is optional, thus any combination of rows 3, 4, 5 and 6 may be present. Colon CAD SR: When this Template is invoked for the Colon CAD SR, the Image Library does not exist, thus rows 3, 5, and/or 6 may be present and row 4 shall not be present.
Rows 5, 6 | Mammography CAD SR: When this Template is invoked for the Mammography CAD SR, the Image Library is mandatory, thus only row 8 shall be present. Chest CAD SR: When this Template is invoked for the Chest CAD SR, the Image Library is optional, thus row 7 or 8 may be present. Colon CAD SR: When this Template is invoked for the Colon CAD SR, the Image Library does not exist, thus only row 7 may be present.

### TID 4018 CAD Analysis Performed

This Template fully identifies an analysis algorithm and the images and/or image regions on which it operated (see TID 4016 “CAD Analyses Performed”).

Table TID 4018. Parameters

Parameter Name | Parameter Usage
--- | ---
$AnalysisCode | Coded term or Context Group for Analysis Performed

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4018. CAD Analysis Performed

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | $AnalysisCode
2 | > | HAS PROPERTIES | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
3 | > | HAS PROPERTIES | IMAGE |  | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present |
4 | > | R-HAS PROPERTIES | IMAGE |  | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present | Shall reference IMAGE Content Item(s) in the (111028, DCM, "Image Library")
5 | > | HAS PROPERTIES | UIDREF | EV (112002, DCM, "Series Instance UID") | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present |
6 | > | HAS PROPERTIES | SCOORD | EV (111030, DCM, "Image Region") | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present |
7 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 8 |
8 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 7 | Shall reference an IMAGE Content Item in the (111028, DCM, "Image Library")
9 | > |  | INCLUDE | DTID 4023 “CAD Operating Points” | 1 | U |  |

Content Item Descriptions

CAD Algorithm Identification | If more than one analysis algorithm has the same "Analysis Performed" code value (e.g., CID 6043 “Types of Mammography CAD Analysis”) then the "CAD Algorithm Identification" shall unambiguously distinguish between algorithms.
Rows 5, 6 | Mammography CAD SR: When this Template is invoked for the Mammography CAD SR, the Image Library is mandatory, and a total of at least two instances of row 4 or row 6 shall be present. Chest CAD SR: When this Template is invoked for the Chest CAD SR, the Image Library is optional, thus any combination of rows 3, 4, 5 and 6 may be present. Colon CAD SR: When this Template is invoked for the Colon CAD SR, the Image Library does not exist, thus rows 3, 5 and/or 6 may be present and row 4 shall not be present.
Rows 5, 6 | Mammography CAD SR: When this Template is invoked for the Mammography CAD SR, the Image Library is mandatory, thus only row 8 shall be present. Chest CAD SR: When this Template is invoked for the Chest CAD SR, the Image Library is optional, thus row 7 or 8 may be present. Colon CAD SR: When this Template is invoked for the Colon CAD SR, the Image Library does not exist, thus only row 7 may be present.

### TID 4019 CAD Algorithm Identification

This Template details the algorithm unambiguously. Re-state the software identification from the General Equipment Module of the SR IOD if all algorithms are unambiguously defined by that module.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4019. CAD Algorithm Identification

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (111001, DCM, "Algorithm Name") | 1 | M |  |
2 |  |  | TEXT | EV (111003, DCM, "Algorithm Version") | 1 | M |  |
3 |  |  | TEXT | EV (111002, DCM, "Algorithm Parameters") | 1-n | U |  |

### TID 4020 CAD Image Library Entry

Each instance of the Image Library Entry Template contains the Image SOP Class and Instance UIDs, and selected attributes for an image that facilitate spatial analysis without having to retrieve the entire set of referenced images. If values for the attributes are not present in the Image SOP Instance, then as many of the attributes as possible should be derived.

Table TID 4020. Parameters

Parameter Name | Parameter Usage
--- | ---
$ImageLaterality | Coded term or Context Group for Image Laterality
$ImageView | Coded term or Context Group for Image View
$ImageViewMod | Coded term or Context Group for Image View Modifier

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 4020. CAD Image Library Entry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | IMAGE |  | 1 | M |  |
2 | > | HAS ACQ CONTEXT | CODE | EV (111027, DCM, "Image Laterality") | 1 | MC | Shall be present if (0020,0062) is in the Image Instance | $ImageLaterality
3 | > | HAS ACQ CONTEXT | CODE | EV (111031, DCM, "Image View") | 1 | MC | Shall be present if (0054,0220) is in the Image Instance | $ImageView
4 | >> | HAS CONCEPT MOD | CODE | EV (111032, DCM, "Image View Modifier") | 1-n | MC | Shall be present if (0054,0222) is in the Image Instance | $ImageViewMod
5 | > | HAS ACQ CONTEXT | TEXT | EV (111044, DCM, "Patient Orientation Row") | 1 | MC | Shall be present if (0020,0020) is in the Image Instance |
6 | > | HAS ACQ CONTEXT | TEXT | EV (111043, DCM, "Patient Orientation Column") | 1 | MC | Shall be present if (0020,0020) is in the Image Instance |
7 | > | HAS ACQ CONTEXT | DATE | EV (111060, DCM, "Study Date") | 1 | MC | Shall be present if (0008,0020) is in the Image Instance |
8 | > | HAS ACQ CONTEXT | TIME | EV (111061, DCM, "Study Time") | 1 | MC | Shall be present if (0008,0030) is in the Image Instance |
9 | > | HAS ACQ CONTEXT | DATE | EV (111018, DCM, "Content Date") | 1 | MC | Shall be present if (0008,0023) is in the Image Instance |
10 | > | HAS ACQ CONTEXT | TIME | EV (111019, DCM, "Content Time") | 1 | MC | Shall be present if (0008,0033) is in the Image Instance |
11 | > | HAS ACQ CONTEXT | NUM | EV (111026, DCM, "Horizontal Pixel Spacing") | 1 | MC | Shall be present if (0018,1164) or (0028,0030) is in the Image Instance | UNITS = EV (um, UCUM, "micrometer") UNITS = EV (mm, UCUM, "millimeter")
12 | > | HAS ACQ CONTEXT | NUM | EV (111066, DCM, "Vertical Pixel Spacing") | 1 | MC | Shall be present if (0018,1164) or (0028,0030) is in the Image Instance | UNITS = EV (um, UCUM, "micrometer") UNITS = EV (mm, UCUM, "millimeter")
13 | > | HAS ACQ CONTEXT | NUM | EV (112011, DCM, "Positioner Primary Angle") | 1 | UC | May be present if (0018,1510) is in the Image Instance | UNITS = EV (deg, UCUM, "deg")
14 | > | HAS ACQ CONTEXT | NUM | EV (112012, DCM, "Positioner Secondary Angle") | 1 | UC | May be present if (0018,1511) is in the Image Instance | UNITS = EV (deg, UCUM, "deg")
15 | > | HAS ACQ CONTEXT | NUM | EV (112226, DCM, "Spacing between slices") | 1 | UC | May be computed from the Image Position (Patient) (0020,0032) projected onto the normal to the Image Orientation (Patient) (0020,0037) if present; may or may not be the same as the Spacing Between Slices (0018,0088) if present. | UNITS = EV (mm, UCUM, "millimeter")
16 | > | HAS ACQ CONTEXT | NUM | EV (112225, DCM, "Slice Thickness") | 1 | UC | May be present if Slice Thickness (0018,0050) is in the Image Instance. | UNITS = EV (mm, UCUM, "millimeter")
17 | > | HAS ACQ CONTEXT | UIDREF | EV (112227, DCM, "Frame of Reference UID") | 1 | UC | May be present if Frame of Reference UID (0020,0052) is in the Image Instance. |
18 | > | HAS ACQ CONTEXT | NUM | EV (110901, DCM, "Image Position (Patient) X") | 1 | UC | May be present if Image Position (Patient) (0020,0032) is in the Image Instance, and is the first value of Image Position (Patient) (0020,0032) for the referenced image or frame. | UNITS = EV (mm, UCUM, "millimeter")
19 | > | HAS ACQ CONTEXT | NUM | EV (110902, DCM, "Image Position (Patient) Y") | 1 | MC | Shall be present if Row 18 is present, and is the second value of Image Position (Patient) (0020,0032) in the Image Instance for the referenced image or frame. | UNITS = EV (mm, UCUM, "millimeter")
20 | > | HAS ACQ CONTEXT | NUM | EV (110903, DCM, "Image Position (Patient) Z") | 1 | MC | Shall be present if Row 18 is present, and is the second value of Image Position (Patient) (0020,0032) in the Image Instance for the referenced image or frame. | UNITS = EV (mm, UCUM, "millimeter")
21 | > | HAS ACQ CONTEXT | NUM | EV (110904, DCM, "Image Orientation (Patient) Row X") | 1 | UC | May be present if Image Position (Patient) (0020,0037) is in the Image Instance, and is the first value of Image Orientation (Patient) (0020,0037) for the referenced image or frame. | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
22 | > | HAS ACQ CONTEXT | NUM | EV (110905, DCM, "Image Orientation (Patient) Row Y") | 1 | MC | Shall be present if Row 21 is present, and is the second value of Image Orientation (Patient) (0020,0037) in the Image Instance for the referenced image or frame. | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
23 | > | HAS ACQ CONTEXT | NUM | EV (110906, DCM, "Image Orientation (Patient) Row Z") | 1 | MC | Shall be present if Row 21 is present, and is the third value of Image Orientation (Patient) (0020,0037) in the Image Instance for the referenced image or frame. | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
24 | > | HAS ACQ CONTEXT | NUM | EV (110907, DCM, "Image Orientation (Patient) Column X") | 1 | MC | Shall be present if Row 21 is present, and is the fourth value of Image Orientation (Patient) (0020,0037) in the Image Instance for the referenced image or frame. | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
25 | > | HAS ACQ CONTEXT | NUM | EV (110908, DCM, "Image Orientation (Patient) Column Y") | 1 | MC | Shall be present if Row 21 is present, and is the fifth value of Image Orientation (Patient) (0020,0037) in the Image Instance for the referenced image or frame. | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
26 | > | HAS ACQ CONTEXT | NUM | EV (110909, DCM, "Image Orientation (Patient) Column Z") | 1 | MC | Shall be present if Row 21 is present, and is the sixth value of Image Orientation (Patient) (0020,0037) in the Image Instance for the referenced image or frame. | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
27 | > | HAS ACQ CONTEXT | NUM | EV (110910, DCM, "Pixel Data Rows") | 1 | UC | May be present if Rows (0028,0010) is in the Image Instance. | UNITS = EV ({pixels}, UCUM, "pixels")
28 | > | HAS ACQ CONTEXT | NUM | EV (110911, DCM, "Pixel Data Columns") | 1 | MC | Shall be present if Row 27 is present, and is the value of Columns (0028,0011) in the Image Instance. | UNITS = EV ({pixels}, UCUM, "pixels")

Content Item Descriptions

Patient Orientation Row | First (row) and second (column) components of Patient Orientation (0020,0020) in the Image IOD. See Section C.7.6.1.1.1 in PS3.3 .
Patient Orientation Column
Horizontal Imager Pixel Spacing | The second component of Imager Pixel Spacing (0018,1164) in the Image IOD. See Section C.8.11.4 in PS3.3 .
Vertical Imager Pixel Spacing | The first component of Imager Pixel Spacing (0018,1164) in the Image IOD. See Section C.8.11.4 in PS3.3 .

### TID 4021 Mammography CAD Geometry

All geometry Template invocations require specification of the location of the center of the object. Outline is optional.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4021. Mammography CAD Geometry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | SCOORD | EV (111010, DCM, "Center") | 1 | M |  | GRAPHIC TYPE = {POINT}
2 | > | R-SELECTED FROM | IMAGE |  | 1 | M |  | Shall reference an IMAGE Content Item in the (111028, DCM, "Image Library")
3 |  |  | SCOORD | EV (111041, DCM, "Outline") | 1 | U |  |
4 | > | R-SELECTED FROM | IMAGE |  | 1 | M |  | Shall reference the same Content Item as row 2
5 |  |  | SCOORD | DCID 6166 “CAD Geometry Secondary Graphical Representation” | 1-n | U |  |
6 | > | R-SELECTED FROM | IMAGE |  | 1 | M |  | Shall reference the same Content Item as row 2

### TID 4022 CAD Observation Context

This Template is invoked when a Content Item, which may be the "root" of a sub-tree, is paraphrased from a prior SR document.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4022. CAD Observation Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | COMPOSITE | EV (111040, DCM, "Original Source") | 1 | MC | Shall be present if the original source is a DICOM object. |
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 |  |  | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |

### TID 4023 CAD Operating Points

This Template describes CAD operating points. The description is deliberately left flexible and optional to allow implementation at differing levels of complexity.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4023. CAD Operating Points

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | NUM | EV (111072, DCM, "Maximum CAD Operating Point") | 1 | M |  | UNITS = DT ([arb'U], UCUM, "arbitrary unit") Value is restricted to being an integer
2 |  | HAS PROPERTIES | NUM | EV (111092, DCM, "Recommended CAD Operating Point") | 1 | U |  | UNITS = DT ({0:n}, UCUM, "range: 0:n"), where n is the value specified in row 1 Value is restricted to being an integer
3 |  | HAS PROPERTIES | CONTAINER | EV (111093, DCM, "CAD Operating Point Table") | 1 | U |  |
4 | > | CONTAINS | CODE | EV (122698, DCM, "X-Concept") | 1 | M |  | DCID 6048 “CAD Operating Point Axis Label”
5 | > | CONTAINS | CODE | EV (122699, DCM, "Y-Concept") | 1 | M |  | DCID 6048 “CAD Operating Point Axis Label”
6 | > | CONTAINS | NUM | EV (111071, DCM, "CAD Operating Point") | 1-n | M | Number of instances of this row shall equal value of row 1, plus 1. | UNITS = DT ({0:n}, UCUM, "range: 0:n"), where n is the value of Row 1. Value is restricted to being an integer that is unique within the invocation of this Template.
7 | >> | HAS PROPERTIES | TEXT | EV (111081, DCM, "CAD Operating Point Description") | 1 | U |  |
8 | >> | HAS PROPERTIES | NUM | The value of Row 4 | 1 | U |  |
9 | >> | HAS PROPERTIES | NUM | The value of Row 5 | 1 | U |  |

Content Item Descriptions

Maximum CAD Operating Point | The maximum possible value of CAD Operating Point for this type of Detection Performed. No CAD Operating Point value recorded in the CAD Processing and Findings Summary sub-tree of the report for this type of Detection Performed shall exceed this value. The report may or may not contain Rendering Intent = "Presentation Optional" detections that are assigned the maximum value.
Recommended CAD Operating Point | A number indicating which of the CAD operating points is recommended by the creator of a CAD SR instance as the first operating point to be used when rendering the CAD SR instance contents. Subsequent changes to the displayed operating point are implementation dependent.

## Chest CAD SR IOD Templates

The Templates that comprise the Chest CAD SR IOD are interconnected as in Figure A-9.

Figure A-9. Chest CAD SR IOD Template Structure

In Figure A-9, '···' indicates possible recursive application of subordinate Templates.

### TID 4100 Chest CAD Document Root

This Template forms the top of a content tree that allows a chest CAD device to describe the results of detection and analysis of chest evidence. This Template, together with its subordinate Templates, describes both the results for presentation to radiologists and partial product results for consumption by chest CAD devices in subsequent chest CAD reports.

This Template defines a Container that contains an Image Library, the CAD results, and summaries of the detection and analysis algorithms performed. The Image Library contains the Image SOP Class and Instance UIDs, and selected attributes for each image referenced in either the algorithm summaries or chest CAD results.

The atomic CAD results of Single Image Findings and Composite Features are described in the Chest CAD Findings Summary sub-tree.

The Summary of Detections and Summary of Analyses sub-trees gather lists of algorithms attempted, grouped by success/failure status. Algorithms not attempted are not mentioned in these sub-trees. This information forms the basis for understanding why a chest CAD report may produce no (or fewer than anticipated) results. Chest CAD results are constructed bottom-up, starting from Single Image Findings (see TID 4104 “Chest CAD Single Image Finding”), associated as Composite Features (see TID 4102 “Chest CAD Composite Feature”).

See Figure F.1-1 “Top Levels of Chest CAD SR Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  Yes

Table TID 4100. Chest CAD Document Root

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (112000, DCM, "Chest CAD Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | CONTAINS | CONTAINER | EV (111028, DCM, "Image Library") | 1 | U |  |
4 | >> | CONTAINS | INCLUDE | DTID 4020 “CAD Image Library Entry” | 1-n | M |  | $ImageLaterality = DCID 244 “Laterality” $ImageView = DCID 4010 “DX View” $ImageViewMod = DCID 4011 “DX View Modifier”
5 | > | CONTAINS | INCLUDE | DTID 4101 “Chest CAD Findings Summary” | 1 | M |  |
6 | > | CONTAINS | CODE | EV (111064, DCM, "Summary of Detections") | 1 | M |  | DCID 6042 “Status of Results”
7 | >> | INFERRED FROM | INCLUDE | DTID 4015 “CAD Detections Performed” | 1 | MC | Shall be present unless the value of row 6 is (111225, DCM, "Not Attempted") | $DetectionCode = DCID 6101 “Chest Finding or Feature”, DCID 6102 “Chest Finding or Feature Modifier”
8 | > | CONTAINS | CODE | EV (111065, DCM, "Summary of Analyses") | 1 | M |  | DCID 6042 “Status of Results”
9 | >> | INFERRED FROM | INCLUDE | DTID 4016 “CAD Analyses Performed” | 1 | MC | Shall be present unless the value of row 8 is (111225, DCM, "Not Attempted") | $AnalysisCode = DCID 6137 “Types of CAD Analysis”

Content Item Descriptions

Image Library | The "Image Library" section of the Content Tree (TID 4100 “Chest CAD Document Root”, row 3) may include all Image SOP Instances from the Current Requested Procedure Evidence Sequence (0040,A375) attribute of the SR Document General module. If a portion of another instance of a Chest CAD SR IOD is duplicated in the "Chest CAD Findings Summary" section of the Content Tree, the "Image Library" may also include all Image Library Entries referenced from the duplicated portions of the Chest CAD SR. The Image Library is intended to be used in cases where the acquisition context Content Items differ from image to image, such as different views and/or laterality in projection X-Ray.
Detections Performed | The "Detections Performed" and "Analyses Performed" sections of the Content Tree (TID 4100 “Chest CAD Document Root”, rows 7 and 9) together shall reference all Image SOP Instances included in the Current Requested Procedure Evidence Sequence (0040,A375) attribute of the SR Document General module.
Analyses Performed

### TID 4101 Chest CAD Findings Summary

The contents of this Template describe the findings and aggregate features that the chest CAD device detected for the chest evidence presented. This Template forms the chest CAD results sub-tree of the Chest CAD Document Root (TID 4100 “Chest CAD Document Root”). The data from which the details are inferred are expressed in the Composite Features (see TID 4102 “Chest CAD Composite Feature”) and/or Single Image Findings (see TID 4104 “Chest CAD Single Image Finding”), of which there may be several.

The sub-tree headed by this Template is illustrated in Figure F.1-2 “Example of CAD Processing and Findings Summary Sub-Tree of Chest CAD SR Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4101. Chest CAD Findings Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111017, DCM, "CAD Processing and Findings Summary") | 1 | M |  | DCID 6047 “CAD Processing and Findings Summary”
2 | > | INFERRED FROM | INCLUDE | DTID 4102 “Chest CAD Composite Feature” | 1-n | U |  |
3 | > | INFERRED FROM | INCLUDE | DTID 4104 “Chest CAD Single Image Finding” | 1-n | U |  |
4 | > | HAS PROPERTIES | INCLUDE | DTID 4106 “Response Evaluation” | 1-n | U |  |

Content Item Descriptions

CAD Processing and Findings Summary | This code value is used to express if and why the Chest CAD Findings Summary sub-tree is empty. The Summary of Detections and Summary of Analyses sub-trees of the Document Root node contain detail about which (if any) algorithms succeeded or failed. If the code value indicates that there were no findings, then the code value can be used to determine whether chest CAD processing occurred successfully, without parsing the Summary of Detections and Summary of Analyses sub-trees.

### TID 4102 Chest CAD Composite Feature

This Template collects a composite feature for a lesion, anatomy, non-lesion object, or correlation of related objects (see TID 4101 “Chest CAD Findings Summary”). The details of the composition are expressed in the Chest CAD Composite Feature Body (see TID 4103 “Chest CAD Composite Feature Body”). The data from which the details are inferred, are expressed in the Composite Features (see TID 4102 “Chest CAD Composite Feature”) and/or Single Image Findings (see TID 4104 “Chest CAD Single Image Finding”), of which there may be several.

A Composite Feature shall be INFERRED FROM any combination of two or more Composite Features or Single Image Findings or mixture thereof.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4102. Chest CAD Composite Feature

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111015, DCM, "Composite Feature") | 1 | M |  | DCID 6101 “Chest Finding or Feature”
2 | > | HAS CONCEPT MOD | CODE | EV (112023, DCM, "Composite Feature Modifier") | 1 | U |  | DCID 6102 “Chest Finding or Feature Modifier”
3 | > | HAS CONCEPT MOD | TEXT | EV (112050, DCM, "Anatomic Identifier") | 1 | U |  |
4 | > | HAS CONCEPT MOD | CODE | EV (112003, DCM, "Associated Chest Component") | 1 | MC | Shall be present IFF value of row 1 is (112005, DCM, "Radiographic anatomy") | DCID 6100 “Chest Component Categories”
5 | > | HAS CONCEPT MOD | CODE | EV (112037, DCM, "Non-lesion Modifier") | 1 | UC | May be present IFF value of row 1 is (111102, DCM, "Non-lesion") | DCID 6139 “Non-lesion Modifiers”
6 | > | HAS CONCEPT MOD | CODE | EV (112038, DCM, "Osseous Modifier") | 1 | UC | May be present IFF value of row 2 is from DCID 6114 “Osseous Anatomy Finding or Feature” | DCID 6115 “Osseous Anatomy Modifiers”
7 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
8 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
9 | > | HAS OBS CONTEXT | CODE | EV (112016, DCM, "Baseline Category") | 1 | U |  | DCID 6145 “Baseline Category”
10 | > | HAS OBS CONTEXT | INCLUDE | DTID 4022 “CAD Observation Context” | 1 | MC | Shall be present IFF this feature is duplicated from a different report than its parent. |
11 | > | HAS OBS CONTEXT | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
12 | > | HAS PROPERTIES | INCLUDE | DTID 4103 “Chest CAD Composite Feature Body” | 1 | M |  |
13 | > | INFERRED FROM | INCLUDE | DTID 4102 “Chest CAD Composite Feature” | 1-n | MC | At least two items shall be present: two of row 13, two of row 14, or one of each. |
14 | > | INFERRED FROM | INCLUDE | DTID 4104 “Chest CAD Single Image Finding” | 1-n | MC | At least two items shall be present: two of row 13, two of row 14, or one of each. |

Content Item Descriptions

Anatomic Identifier | An identifier of an anatomic feature when a multiplicity of features of that type may be present, such as "Rib 1", "Rib 2" or thoracic vertebrae "T1" or "T2".
Rendering Intent | This Content Item constrains the SCP receiving the Chest CAD SR IOD in its use of the contents of this Template and its Target Content Items. Chest CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent chest CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .

### TID 4103 Chest CAD Composite Feature Body

The details of a composite feature are expressed in this Template. It is applied to Chest CAD Composite Feature (TID 4102 “Chest CAD Composite Feature”).

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4103. Chest CAD Composite Feature Body

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111016, DCM, "Composite type") | 1 | M |  | DCID 6035 “Composite Feature Relations”
2 |  |  | CODE | EV (111057, DCM, "Scope of Feature") | 1 | M |  | DCID 6036 “Scope of Feature”
3 |  |  | NUM | EV (111011, DCM, "Certainty of feature") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
4 |  |  | INCLUDE | DTID 4107 “Chest CAD Geometry” | 1 | U |  |
5 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  |
6 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  |
7 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  |
8 |  |  | INCLUDE | DTID 4105 “Chest CAD Descriptors” | 1 | U |  |
9 |  |  | NUM | DCID 6133 “Chest Quantitative Temporal Difference Type” | 1-n | UC | May be present IFF the value of row 1 is (111153, DCM, "Target content items are related temporally") |
10 | > | R-INFERRED FROM | NUM |  | 2 | U |  | The referenced numeric values shall have the same Concept Name. Their UNITS shall be the same as row 9
11 |  |  | CODE | EV (111049, DCM, "Qualitative Difference") | 1-n | UC | May be present only if the value of row 1 is (111153, DCM, "Target content items are related temporally") | DCID 6134 “Chest Qualitative Temporal Difference Type”
12 | > | HAS PROPERTIES | TEXT | EV (111021, DCM, "Description of Change") | 1 | U |  |
13 | > | R-INFERRED FROM | CODE |  | 2 | M |  | The referenced Content Items shall have the same Concept Name and their code values shall be from the same context group.

Content Item Descriptions

Certainty of Feature | The certainty of the CAD device that the feature analyzed and classified by the CODE, as specified in the Composite Feature parent Template, is in fact that type of feature.
Volume Measurement | If dimensions for a volume are to be stated in terms of length, width, and depth, then one shall use 3 instances of TID 1400 “Linear Measurement”.
Row 9 | Values ≤ 0 are allowed. The two referenced numeric values are Target Content Items of the first generation Composite Feature or Single Image Finding children of this composite feature. Given the equation, A - B, the value representing A shall be referenced first.
Qualitative Difference | The two referenced code values are Target Content Items of the first generation Composite Feature or Single Image Finding children of this composite feature.

### TID 4104 Chest CAD Single Image Finding

This Template describes a single image finding for a lesion or other object. The details of the finding are expressed in this Template and/or more specific Templates.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4104. Chest CAD Single Image Finding

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111059, DCM, "Single Image Finding") | 1 | M |  | DCID 6101 “Chest Finding or Feature”
2 | > | HAS CONCEPT MOD | CODE | EV (112024, DCM, "Single Image Finding Modifier") | 1 | U |  | DCID 6102 “Chest Finding or Feature Modifier”
3 | > | HAS CONCEPT MOD | TEXT | EV (112050, DCM, "Anatomic Identifier") | 1 | U |  |
4 | > | HAS CONCEPT MOD | CODE | EV (112003, DCM, "Associated Chest Component") | 1 | MC | Shall be present IFF value of row 1 is (112005, DCM, "Radiographic anatomy") | DCID 6100 “Chest Component Categories”
5 | > | HAS CONCEPT MOD | CODE | EV (112037, DCM, "Non-lesion Modifier") | 1 | UC | May be present IFF value of row 1 is (111102, DCM, "Non-lesion") | DCID 6139 “Non-lesion Modifiers”
6 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
7 | >> | HAS PROPERTIES | NUM | EV (111071, DCM, "CAD Operating Point") | 1 | UC | IFF value of row 6 is (111151, DCM, "Presentation Optional") and row 1 of TID 4023 “CAD Operating Points” is present for the finding identified in row 1 | UNITS = DT ({1:n}, UCUM, "range: 1:n"), where n is the maximum specified in Row 1 of TID 4023 “CAD Operating Points” for the finding identified in row 1. Value is restricted to being an integer
8 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
9 | > | HAS OBS CONTEXT | CODE | EV (112016, DCM, "Baseline Category") | 1 | U |  | DCID 6145 “Baseline Category”
10 | > | HAS OBS CONTEXT | INCLUDE | DTID 4022 “CAD Observation Context” | 1 | MC | Shall be present IFF this finding is duplicated from a different report than its parent. |
11 | > | HAS OBS CONTEXT | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
12 | > | HAS PROPERTIES | NUM | EV (111012, DCM, "Certainty of Finding") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
13 | > | HAS PROPERTIES | TEXT | EV (111058, DCM, "Selected Region Description") | 1 | MC | Shall be present IFF value of row 1 is (111099, DCM, "Selected region") |
14 | > | HAS PROPERTIES | INCLUDE | DTID 4107 “Chest CAD Geometry” | 1 | MC | Shall be present unless value of row 1 is (111101, DCM, "Image quality") |
15 | > | HAS PROPERTIES | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  |
16 | > | HAS PROPERTIES | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  |
17 | > | HAS PROPERTIES | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  |
18 | > | HAS PROPERTIES | INCLUDE | DTID 4105 “Chest CAD Descriptors” | 1 | U |  |
19 | > | INFERRED FROM | IMAGE |  | 1 | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") and rows 20 and 21 are not present |
20 | > | R-INFERRED FROM | IMAGE |  | 1 | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") and rows 19 and 21 are not present | Shall reference an IMAGE Content Item in the (111028, DCM, "Image Library")
21 | > | INFERRED FROM | SCOORD | EV (111030, DCM, "Image Region") | 1-n | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") and rows 19 and 20 are not present |
22 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR row 23 | All the row 21 Content Items in a single invocation of this Template shall reference the same IMAGE
23 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR row 22 | All the row 21 Content Items in a single invocation of this Template shall reference the same IMAGE Content Item in the (111028, DCM, "Image Library")
24 | > | HAS PROPERTIES | INCLUDE | DTID 4014 “CAD Image Quality” | 1 | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") | $QualityFinding = DCID 6135 “Image Quality Finding” $QualityStandard = DCID 6136 “Chest Types of Quality Control Standard”

Content Item Descriptions

Anatomic Identifier | An identifier of an anatomic feature when a multiplicity of features of that type may be present, such as "Rib 1", "Rib 2" or thoracic vertebrae "T1" or "T2".
Rendering Intent | This Content Item constrains the SCP receiving the Chest CAD SR IOD in its use of the contents of this Template and its Target Content Items. Chest CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent chest CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .
CAD Operating Point | Additional information to use when Rendering Intent is "Presentation Optional". A CAD Operating Point of zero is not sent, and is encoded as a Rendering Intent of "Presentation Required". See Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 and Section E.4 “CAD Operating Point” in PS3.17 .
Certainty of Finding | The certainty of the CAD device that the finding detected and classified by the Single Image Finding CODE specified is in fact that type of finding.

### TID 4105 Chest CAD Descriptors

This Template provides qualitative detail for a Single Image Finding or Composite Feature. It is applied to Chest CAD Composite Feature (TID 4102 “Chest CAD Composite Feature”) and Chest CAD Single Image Finding (TID 4104 “Chest CAD Single Image Finding”).

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4105. Chest CAD Descriptors

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (112025, DCM, "Size Descriptor") | 1 | U |  | DCID 6118 “Size Descriptor”
2 |  |  | CODE | EV (112026, DCM, "Width Descriptor") | 1 | U |  | DCID 6107 “Width Descriptor”
3 |  |  | CODE | EV (112015, DCM, "Border shape") | 1 | U |  | DCID 6119 “Chest Border Shape”
4 |  |  | CODE | EV (112007, DCM, "Border definition") | 1 | U |  | DCID 6120 “Chest Border Definition”
5 |  |  | CODE | EV (112014, DCM, "Orientation Descriptor") | 1 | U |  | DCID 6121 “Chest Orientation Descriptor”
6 |  |  | CODE | EV (112009, DCM, "Type of Content") | 1-n | U |  | DCID 6122 “Chest Content Descriptor”
7 |  |  | CODE | EV (112027, DCM, "Opacity Descriptor") | 1 | U |  | DCID 6123 “Chest Opacity Descriptor”
8 |  |  | CODE | EV (112013, DCM, "Location in Chest") | 1 | U |  | DCID 6124 “Location in Chest”
9 |  |  | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
10 |  |  | CODE | EV (112006, DCM, "Distribution Descriptor") | 1-n | U |  | DCID 6128 “Chest Distribution Descriptor”
11 |  |  | CODE | EV (112028, DCM, "Abnormal Distribution of Anatomic Structure") | 1 | U |  | DCID 6108 “Chest Anatomic Structure Abnormal Distribution”
12 |  |  | CODE | EV (112008, DCM, "Site involvement") | 1-n | U |  | DCID 6129 “Chest Site Involvement”
13 |  |  | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 6130 “Severity Descriptor”
14 |  |  | CODE | EV (112010, DCM, "Texture Descriptor") | 1 | U |  | DCID 6131 “Chest Texture Descriptor”
15 |  |  | CODE | EV (112030, DCM, "Calcification Descriptor") | 1 | U |  | DCID 6132 “Chest Calcification Descriptor”
16 |  |  | NUM | DCID 6142 “Calculated Value” | 1-n | U |  |
17 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | M |  | DCID 6140 “Calculation Methods”
18 | > | INFERRED FROM | NUM | EV (112032, DCM, "Threshold Attenuation Coefficient") | 1 | U |  | UNITS = EV ([hnsf'U], UCUM, "Hounsfield unit")
19 | > | INFERRED FROM | TEXT | EV (112034, DCM, "Calculation Description") | 1 | U |  |
20 |  |  | NUM | DCID 6141 “Attenuation Coefficient Measurements” | 1-n | U |  | UNITS = EV ([hnsf'U], UCUM, "Hounsfield unit")

### TID 4106 Response Evaluation

This Template provides a means to report response evaluation to cancer treatment, based on a method such as RECIST or WHO.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4106. Response Evaluation

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (112020, DCM, "Response Evaluation") | 1 | M |  |
2 | > | HAS OBS CONTEXT | CODE | EV (112021, DCM, "Response Evaluation Method") | 1 | M |  | DT (112022, DCM, "RECIST") or DT (112029, DCM, "WHO")
3 | > | CONTAINS | CODE | EV (112048, DCM, "Current Response") | 1 | U |  | DCID 6143 “Lesion Response”
4 | > | CONTAINS | CODE | EV (112049, DCM, "Best Overall Response") | 1 | U |  | DCID 6143 “Lesion Response”
5 | > | CONTAINS | NUM | EV (112051, DCM, "Measurement of Response") | 1 | U |  | UNITS not specified

### TID 4107 Chest CAD Geometry

All geometry Template invocations require specification of either the location of the center of the object, the outline, or both. Geometry is a property of single image findings (see TID 4104 “Chest CAD Single Image Finding”) and composite features (see TID 4103 “Chest CAD Composite Feature Body”).

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4107. Chest CAD Geometry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | SCOORD | EV (111010, DCM, "Center") | 1 | MC | At least one of rows 1, 4 shall be present. | GRAPHIC TYPE = {POINT}
2 | > | SELECTED FROM | IMAGE |  | 1 | MC | XOR row 3 |
3 | > | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR row 2 | Shall reference an IMAGE Content Item in the (111028, DCM, "Image Library")
4 |  |  | SCOORD | EV (111041, DCM, "Outline") | 1 | MC | At least one of rows 1, 4 shall be present. |
5 | > | SELECTED FROM | IMAGE |  | 1 | MC | XOR row 6 | Shall reference the same Content Item as row 2
6 | > | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR row 5 | Shall reference the same Content Item as row 3

### TID 4108 Tracking Identifier

This Template provides a means to identify an object for longitudinal tracking, potentially across multiple Structured Reports, over time.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4108. Tracking Identifier

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (112039, DCM, "Tracking Identifier") | 1 | MC | At least one of row 1 or 2 shall be present. | A string of characters with case being non-significant. Leading and trailing spaces and control characters are forbidden.
2 |  |  | UIDREF | EV (112040, DCM, "Tracking Unique Identifier") | 1 | MC | At least one of row 1 or 2 shall be present. |

Content Item Descriptions

Tracking Identifier | A human readable identifier for longitudinal tracking, e.g., "Watchlist Nodule 1".
Tracking Unique Identifier | This is distinct from the Observation UID (0040,A171) that may be present in the data set for each Content Item, which identifies only a specific observation, not an object tracked over time, and each tracked object may have many observations.

## Colon CAD SR IOD Templates

The Templates that comprise the Colon CAD SR IOD are interconnected as in Figure A-9b. In Figure A-9b, '···' indicates possible recursive application of subordinate Templates.

Figure A-9b. Colon CAD SR IOD Template Structure

### TID 4120 Colon CAD Document Root

This Template forms the top of a content tree that allows a colon CAD device to describe the results of detection and analysis of colon evidence. This Template, together with its subordinate Templates, describes both the results for presentation to radiologists and partial product results for consumption by colon CAD devices in subsequent colon CAD reports.

This Template defines a Container that contains the CAD results and summaries of the detection and analysis algorithms performed.

The atomic CAD results of Single Image Findings and Composite Features are described in the Colon CAD Findings Summary sub-tree.

The Summary of Detections and Summary of Analyses sub-trees gather lists of algorithms attempted, grouped by success/failure status. Algorithms not attempted are not mentioned in these sub-trees. This information forms the basis for understanding why a colon CAD report may produce no (or fewer than anticipated) results. Colon CAD results are constructed bottom-up, starting from Single Image Findings (see TID 4127 “Colon CAD Single Image Finding”), associated as Composite Features (see TID 4125 “Colon CAD Composite Feature”).

See Figure SS.1-1 “Top Levels of Colon CAD SR Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  Yes

Table TID 4120. Colon CAD Document Root

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (112220, DCM, "Colon CAD Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | CONTAINS | INCLUDE | DTID 4122 “CAD Common Image Properties Entry” | 1-n | M |  |
4 | > | CONTAINS | INCLUDE | DTID 4121 “Colon CAD Findings Summary” | 1 | M |  |
5 | > | CONTAINS | CODE | EV (111064, DCM, "Summary of Detections") | 1 | M |  | DCID 6042 “Status of Results”
6 | >> | INFERRED FROM | INCLUDE | DTID 4015 “CAD Detections Performed” | 1 | MC | Shall be present unless the value of row 5 is (111225, DCM, "Not Attempted") | $DetectionCode = DCID 6201 “Colon Finding or Feature”
7 | > | CONTAINS | CODE | EV (111065, DCM, "Summary of Analyses") | 1 | M |  | DCID 6042 “Status of Results”
8 | >> | INFERRED FROM | INCLUDE | DTID 4016 “CAD Analyses Performed” | 1 | MC | Shall be present unless the value of row 7 is (111225, DCM, "Not Attempted") | $AnalysisCode = DCID 6137 “Types of CAD Analysis”

Content Item Descriptions

Detections Performed | The "Detections Performed" and "Analyses Performed" sections of the Content Tree (TID 4120 “Colon CAD Document Root”, rows 6 and 8) together shall reference all Image SOP Instances included in the Current Requested Procedure Evidence Sequence (0040,A375) attribute of the SR Document General module.
Analyses Performed

### TID 4121 Colon CAD Findings Summary

The contents of this Template describe the findings and aggregate features that the colon CAD device detected for the colon evidence presented. This Template forms the colon CAD results sub-tree of the Colon CAD Document Root (TID 4120 “Colon CAD Document Root”). The data from which the details are inferred are expressed in the Composite Features (see TID 4125 “Colon CAD Composite Feature”) and/or Single Image Findings (see TID 4127 “Colon CAD Single Image Finding”), of which there may be several.

The sub-tree headed by this Template is illustrated in Figure F.1-2 “Example of CAD Processing and Findings Summary Sub-Tree of Chest CAD SR Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4121. Colon CAD Findings Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111017, DCM, "CAD Processing and Findings Summary") | 1 | M |  | DCID 6047 “CAD Processing and Findings Summary”
2 | > | HAS PROPERTIES | CODE | EV (112222, DCM, "Colon Overall Assessment") | 1 | U |  | DCID 6200 “Colon Overall Assessment”
3 | > | INFERRED FROM | INCLUDE | DTID 4125 “Colon CAD Composite Feature” | 1-n | U |  |
4 | > | INFERRED FROM | INCLUDE | DTID 4127 “Colon CAD Single Image Finding” | 1-n | U |  |

Content Item Descriptions

CAD Processing and Findings Summary | This code value is used to express if and why the Colon CAD Findings Summary sub-tree is empty. The Summary of Detections and Summary of Analyses sub-trees of the Document Root node contain detail about which (if any) algorithms succeeded or failed. If the code value indicates that there were no findings, then the code value can be used to determine whether colon CAD processing occurred successfully, without parsing the Summary of Detections and Summary of Analyses sub-trees.

### TID 4122 CAD Common Image Properties Entry

Each instance of the CAD Common Image Properties Entry Template contains selected attributes for a set of parallel contiguous equally spaced slices (with identical properties) from which CAD findings are derived.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 4122. CAD Common Image Properties Entry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (112224, DCM, "Image Set Properties") | 1 | M |  |
2 | > | CONTAINS | UIDREF | EV (112227, DCM, "Frame of Reference UID") | 1 | M |  |
3 | > | CONTAINS | UIDREF | EV (110180, DCM, "Study Instance UID") | 1 | M |  |
4 | > | CONTAINS | DATE | EV (111060, DCM, "Study Date") | 1 | M |  | Shall be taken from Study Date (0008,0020) in the Image Instances.
5 | > | CONTAINS | TIME | EV (111061, DCM, "Study Time") | 1 | M |  | Shall be taken from Study Time (0008,0030) in the Image Instances.
6 | > | CONTAINS | CODE | EV (121139, DCM, "Modality") | 1 | M |  | Shall be taken from Modality (0008,0060) in the Image Instances.
7 | > | CONTAINS | NUM | EV (111026, DCM, "Horizontal Pixel Spacing") | 1 | M |  | Shall be taken from value 2 of Pixel Spacing (0028,0030) in the Image Instances. UNITS = EV (mm/{pixel}, UCUM, "millimeters per pixel")
8 | > | CONTAINS | NUM | EV (111066, DCM, "Vertical Pixel Spacing") | 1 | M |  | Shall be taken from value 1 of Pixel Spacing (0028,0030) in the Image Instances. UNITS = EV (mm/{pixel}, UCUM, "millimeters per pixel")
9 | > | CONTAINS | NUM | EV (112225, DCM, "Slice Thickness") | 1 | M |  | Shall be taken from Slice Thickness (0018,0050) in the Image Instances. UNITS = EV (mm, UCUM, "millimeter")
10 | > | CONTAINS | NUM | EV (112226, DCM, "Spacing between slices") | 1 | M |  | Shall be computed from the Image Position (Patient) (0020,0032) projected onto the normal to the Image Orientation (Patient) (0020,0037); may or may not be the same as the Spacing Between Slices (0018,0088) if present. UNITS = EV (mm, UCUM, "millimeter")
11 | > | CONTAINS | CODE | EV (112228, DCM, "Recumbent Patient Position with respect to gravity") | 1 | MC | Required if Patient Position (0018,5100) is present in the image instances and has a value. | Shall be derived from Patient Position (0018,5100) in the Image Instances. DCID 6206 “Recumbent Patient Orientation for Colon”

### TID 4125 Colon CAD Composite Feature

This Template collects a composite feature for a lesion, non-lesion object, or correlation of related objects (see TID 4121 “Colon CAD Findings Summary”). The details of the composition are expressed in the Colon CAD Composite Feature Body (see TID 4126 “Colon CAD Composite Feature Body”). The data from which the details are inferred, are expressed in the Composite Features (see TID 4125 “Colon CAD Composite Feature”) and/or Single Image Findings (see TID 4127 “Colon CAD Single Image Finding”), of which there may be several.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4125. Colon CAD Composite Feature

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111015, DCM, "Composite Feature") | 1 | M |  | DCID 6201 “Colon Finding or Feature”
2 | > | HAS CONCEPT MOD | CODE | EV (112023, DCM, "Composite Feature Modifier") | 1 | U |  | DCID 6202 “Colon Finding or Feature Modifier”
3 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
4 | >> | HAS PROPERTIES | NUM | EV (111071, DCM, "CAD Operating Point") | 1 | UC | IFF value of row 3 is (111151, DCM, "Presentation Optional") and row 1 of TID 4023 “CAD Operating Points” is present for the feature identified in row 1. | UNITS = DT ({1:n}, UCUM, "range: 1:n"), where n is the maximum specified in Row 1 of TID 4023 “CAD Operating Points” for the feature identified in row 1. Value is restricted to being an integer.
5 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
6 | > | HAS OBS CONTEXT | INCLUDE | DTID 4022 “CAD Observation Context” | 1 | MC | Shall be present IFF this feature is duplicated from a different report than its parent. |
7 | > | HAS OBS CONTEXT | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
8 | > | HAS PROPERTIES | INCLUDE | DTID 4126 “Colon CAD Composite Feature Body” | 1 | M |  |
9 | > | INFERRED FROM | INCLUDE | DTID 4125 “Colon CAD Composite Feature” | 1-n | U |  |
10 | > | INFERRED FROM | INCLUDE | DTID 4127 “Colon CAD Single Image Finding” | 1-n | U |  |

Content Item Descriptions

Rendering Intent | This Content Item constrains the SCP receiving the Colon CAD SR IOD in its use of the contents of this Template and its Target Content Items. Colon CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent colon CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .

### TID 4126 Colon CAD Composite Feature Body

The details of a composite feature are expressed in this Template. It is applied to Colon CAD Composite Feature (TID 4125 “Colon CAD Composite Feature”).

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4126. Colon CAD Composite Feature Body

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111016, DCM, "Composite type") | 1 | M |  | DCID 6035 “Composite Feature Relations”
2 |  |  | CODE | EV (111057, DCM, "Scope of Feature") | 1 | M |  | DCID 6036 “Scope of Feature”
3 |  |  | NUM | EV (111011, DCM, "Certainty of feature") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
4 |  |  | INCLUDE | DTID 4129 “Colon CAD Geometry” | 1 | U |  |
5 |  |  | INCLUDE | DTID 4128 “Colon CAD Descriptors” | 1 | U |  |
6 |  |  | NUM | DCID 6207 “Colon Quantitative Temporal Difference Type” | 1-n | UC | May be present IFF the value of row 1 is (111153, DCM, "Target content items are related temporally") |
7 | > | R-INFERRED FROM | NUM |  | 2 | U |  | The referenced numeric values shall have the same Concept Name. Their UNITS shall be the same as row 6
8 |  |  | CODE | EV (111049, DCM, "Qualitative Difference") | 1-n | UC | May be present only if the value of row 1 is (111153, DCM, "Target content items are related temporally") | DCID 6134 “Chest Qualitative Temporal Difference Type”
9 | > | HAS PROPERTIES | TEXT | EV (111021, DCM, "Description of Change") | 1 | U |  |
10 | > | R-INFERRED FROM | CODE |  | 2 | M |  | The referenced Content Items shall have the same Concept Name and their code values shall be from the same context group.

Content Item Descriptions

Certainty of Feature | The CAD device's certainty that the feature analyzed and classified by the CODE, as specified in the Composite Feature parent Template is, in fact, that type of feature.
Row 6 | Values ≤ 0 are allowed. The two referenced numeric values are Target Content Items of the first generation Composite Feature or Single Image Finding children of this composite feature. Given the equation, A - B, the value representing A shall be referenced first.
Qualitative Difference | The two referenced code values are Target Content Items of the first generation Composite Feature or Single Image Finding children of this composite feature.

### TID 4127 Colon CAD Single Image Finding

This Template describes a single image finding for a lesion or other object. The details of the finding are expressed in this Template and/or more specific Templates.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4127. Colon CAD Single Image Finding

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111059, DCM, "Single Image Finding") | 1 | M |  | DCID 6201 “Colon Finding or Feature”
2 | > | HAS CONCEPT MOD | CODE | EV (112024, DCM, "Single Image Finding Modifier") | 1 | U |  | DCID 6202 “Colon Finding or Feature Modifier”
3 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
4 | >> | HAS PROPERTIES | NUM | EV (111071, DCM, "CAD Operating Point") | 1 | UC | IFF value of row 3 is (111151, DCM, "Presentation Optional") and row 1 of TID 4023 “CAD Operating Points” is present for the finding identified in row 1 | UNITS = DT ({1:n}, UCUM, "range: 1:n"), where n is the maximum specified in Row 1 of TID 4023 “CAD Operating Points” for the finding identified in row 1. Value is restricted to being an integer.
5 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
6 | > | HAS OBS CONTEXT | INCLUDE | DTID 4022 “CAD Observation Context” | 1 | MC | Shall be present IFF this finding is duplicated from a different report than its parent. |
7 | > | HAS OBS CONTEXT | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
8 | > | HAS PROPERTIES | NUM | EV (111012, DCM, "Certainty of Finding") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
9 | > | HAS PROPERTIES | TEXT | EV (111058, DCM, "Selected Region Description") | 1 | MC | Shall be present IFF value of row 1 is (111099, DCM, "Selected region") |
10 | > | HAS PROPERTIES | INCLUDE | DTID 4129 “Colon CAD Geometry” | 1 | MC | Shall be present unless value of row 1 is (111101, DCM, "Image quality") |
11 | > | HAS PROPERTIES | INCLUDE | DTID 4128 “Colon CAD Descriptors” | 1 | U |  |
12 | > | INFERRED FROM | IMAGE |  | 1 | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") and row 13 is not present |
13 | > | INFERRED FROM | SCOORD | EV (111030, DCM, "Image Region") | 1-n | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") and row 12 is not present |
14 | >> | SELECTED FROM | IMAGE |  | 1 | M |  | All the row 13 Content Items in a single invocation of this Template shall reference the same IMAGE
15 | > | HAS PROPERTIES | INCLUDE | DTID 4014 “CAD Image Quality” | 1 | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") | $QualityFinding = DCID 6135 “Image Quality Finding” $QualityStandard = DCID 6208 “Colon Types of Quality Control Standard”

Content Item Descriptions

Rendering Intent | This Content Item constrains the SCP receiving the Colon CAD SR IOD in its use of the contents of this Template and its Target Content Items. Colon CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent colon CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .
CAD Operating Point | Additional information to use when Rendering Intent is "Presentational Optional". A CAD Operating Point of zero is not sent, and is encoded as a Rendering Intent of "Presentation Required". See Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 and Section E.4 “CAD Operating Point” in PS3.17 .
Certainty of Finding | The certainty of the CAD device that the finding detected and classified by the Single Image Finding CODE specified is in fact that type of finding.

### TID 4128 Colon CAD Descriptors

This Template provides qualitative detail for a Single Image Finding or Composite Feature. It is applied to Colon CAD Composite Feature (TID 4125 “Colon CAD Composite Feature”) and Colon CAD Single Image Finding (TID 4127 “Colon CAD Single Image Finding”).

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4128. Colon CAD Descriptors

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | U |  | DCID 6209 “Colon Morphology Descriptor”
2 |  |  | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | U |  | DCID 6210 “Location in Intestinal Tract”
3 |  |  | CODE | EV (111014, DCM, "Clockface or region") | 1 | U |  | DCID 6205 “Clockface Location for Colon”
4 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 6212 “Calculated Value for Colon Findings” $Derivation = DCID 6140 “Calculation Methods” $DerivationParameter = EV (112032, DCM, "Threshold Attenuation Coefficient") $DerivationParameterUnits = EV ([hnsf'U], UCUM, "Hounsfield unit")
5 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  |
6 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  |
7 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  |
8 |  |  | INCLUDE | DTID 1406 “Three Dimensional Linear Measurement” | 1-n | U |  |
9 |  |  | NUM | DCID 6141 “Attenuation Coefficient Measurements” | 1-n | U |  | UNITS = EV ([hnsf'U], UCUM, "Hounsfield unit")
10 | > | HAS PROPERTIES | CODE | EV (112009, DCM, "Type of Content") | 1 | U |  | DCID 6211 “Colon CAD Material Description”

Content Item Descriptions

Row 3 | 12 o'clock position is the anterior direction of the patient regardless of the positioning with respect to gravity; clockwise is from the point of view of an observer located closer to the anus than the finding being observed.

### TID 4129 Colon CAD Geometry

All geometry Template invocations require specification of either the location of the center of the object, the outline, or both. Geometry is a property of single image findings (see TID 4127 “Colon CAD Single Image Finding”) and composite features (see TID 4125 “Colon CAD Composite Feature”).

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4129. Colon CAD Geometry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | SCOORD | EV (111010, DCM, "Center") | 1 | MC | At least one of rows 1, 3, 4, 6 or 10 shall be present. | GRAPHIC TYPE = {POINT}
2 | > | SELECTED FROM | IMAGE |  | 1 | M |  |
3 |  |  | SCOORD3D | EV (111010, DCM, "Center") | 1 | MC | At least one of rows 1, 3, 4, 6 or 10 shall be present. | GRAPHIC TYPE = {POINT}
4 |  |  | SCOORD | EV (111041, DCM, "Outline") | 1 | MC | At least one of rows 1, 3, 4, 6 or 10 shall be present. |
5 | > | SELECTED FROM | IMAGE |  | 1 | M |  |
6 |  |  | SCOORD3D | EV (111041, DCM, "Outline") | 1 | MC | At least one of rows 1, 3, 4, 6 or 10 shall be present. |
7 |  |  | SCOORD | DCID 6166 “CAD Geometry Secondary Graphical Representation” | 1-n | U |  |
8 | > | SELECTED FROM | IMAGE |  | 1 | M |  |
9 |  |  | SCOORD3D | DCID 6166 “CAD Geometry Secondary Graphical Representation” | 1-n | U |  |
10 |  |  | IMAGE | EV (112229, DCM, "Identifying Segment") | 1 | MC | At least one of rows 1, 3, 4, 6 or 10 shall be present. | Referenced image shall be a Segmentation and the Content Item shall include Referenced Segment Number (0062,000B)

## Breast Imaging Report Templates

The Templates that comprise the Breast Imaging Report are interconnected as in Figure A-10.

Figure A-10. Breast Imaging Report Template Structure

### TID 4200 Breast Imaging Report

This Template forms the content tree that allows a Breast Imaging Report device to describe the results of a radiologist's diagnostic interpretation of Breast Imaging (e.g., X-Ray mammography or breast ultrasound) evidence. This Template, together with its subordinate Templates, describes the results for presentation to clinicians, or for consumption by Breast Imaging Report devices for subsequent Breast Imaging Reports.

This Template shall be instantiated at the Root node only.

See Figure Q.1-1 “Top Level of Breast Imaging Report Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  Yes

Table TID 4200. Breast Imaging Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111400, DCM, "Breast Imaging Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | CONTAINS | INCLUDE | DTID 4202 “Breast Imaging Report Narrative” | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 4208 “Breast Imaging Report Supplementary Data” | 1 | U |  |

### TID 4201 Breast Imaging Procedure Reported

A procedure that is reported in a Breast Imaging Report is expressed in this Template. The results of more than one procedure may be included in a single report instance (see TID 4208 “Breast Imaging Report Supplementary Data”).

See Figure Q.1-2 “Breast Imaging Procedure Reported Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4201. Breast Imaging Procedure Reported

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DCID 6050 “Breast Procedure Reported”
2 | > | HAS CONCEPT MOD | CODE | EV (111464, DCM, "Procedure Modifier") | 1-n | U |  | DCID 6058 “Procedure Modifiers for Breast”
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | DCID 6022 “Side”
4 | > | HAS PROPERTIES | CODE | EV (111401, DCM, "Reason for procedure") | 1 | U |  | DCID 6051 “Breast Procedure Reason”
5 | >> | HAS CONCEPT MOD | CODE | EV (G-D709, SRT, "Relative time") | 1 | U |  | DCID 12102 “Temporal Periods Relating to Procedure or Therapy”
6 | >> | HAS CONCEPT MOD | CODE | EV (111402, DCM, "Clinical Finding") | 1-n | UC | IFF row 4 value is "Clinical Finding" | DCID 6055 “Breast Clinical Finding or Indicated Problem” Breast Clinical Finding or Indicated Problem
7 | >>> | HAS PROPERTIES | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 6022 “Side”
8 | > | HAS PROPERTIES | DATE | EV (111060, DCM, "Study Date") | 1 | U |  |

Content Item Descriptions

Row 5 "Relative time" | This Content Item indicates whether the value of "Reason for procedure" (row 4) is modified with "pre-" or "follow-up".

### TID 4202 Breast Imaging Report Narrative

This Template contains the narrative text sub-tree of the content tree of a Breast Imaging Report. The narrative summary may be subdivided into sections with section headings.

See Figure Q.1-3 “Breast Imaging Report Narrative Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4202. Breast Imaging Report Narrative

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111412, DCM, "Narrative Summary") | 1 | M |  |
2 | > | CONTAINS | CONTAINER | BCID 6052 “Breast Imaging Report Section Title” | 1-n | M |  |
3 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
4 | >> | CONTAINS | TEXT | BCID 6053 “Breast Imaging Report Elements” | 1 | M |  |
5 | >>> | INFERRED FROM | INCLUDE | DTID 350 “References to Supporting Evidence” | 1 | U |  |

### TID 4203 Breast Imaging Assessment

This Template provides the content of a Breast Imaging Assessment, for an overall assessment section for the entire report (see TID 4208 “Breast Imaging Report Supplementary Data”) or an assessment of a particular finding (see TID 4206 “Breast Imaging Report Finding Section”). This Template defines a code-based assessment of the interpretation results.

See Figure Q.1-5 “Breast Imaging Assessment Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4203. Breast Imaging Assessment

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111005, DCM, "Assessment Category") | 1 | M |  | DCID 6026 “Mammography Assessment”
2 |  |  | CODE | EV (111053, DCM, "Recommended Follow-up") | 1-n | U |  | BCID 6028 “Mammography Recommended Follow-up”
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 6022 “Side”
4 | > | HAS PROPERTIES | NUM | EV (111055, DCM, "Recommended Follow-up Interval") | 1 | U |  | UNITS = DCID 6046 “Units of Follow-up Interval” Values = Integer ≥ 0, where 0 = immediate follow-up
5 | > | HAS PROPERTIES | DATE | EV (111054, DCM, "Recommended Follow-up Date") | 1 | U |  |
6 | > | HAS PROPERTIES | INCLUDE | DTID 4207 “Breast Imaging Pathology Results” | 1-n | U |  |

### TID 4204 Breast Imaging Report Intervention Section

This Template defines a supplementary data section for an Intervention of the breast, for the Breast Imaging Report. It is included from TID 4208 “Breast Imaging Report Supplementary Data”.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 4204. Breast Imaging Report Intervention Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111463, DCM, "Supplementary Data for Intervention") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 4201 “Breast Imaging Procedure Reported” | 1 | M |  |
3 | > | CONTAINS | CODE | EV (A-00110, SRT, "Instrument") | 1 | U |  |
4 | >> | HAS PROPERTIES | TEXT | EV (111465, DCM, "Needle Gauge") | 1 | UC | XOR row 5 |
5 | >> | HAS PROPERTIES | CODE | EV (111465, DCM, "Needle Gauge") | 1 | UC | XOR row 4 |
6 | >> | HAS PROPERTIES | NUM | EV (111467, DCM, "Needle Length") | 1 | U |  | UNITS = EV (cm, UCUM, "centimeter")
7 | > | CONTAINS | NUM | EV (111436, DCM, "Number of passes") | 1 | U |  | UNITS = EV ({passes}, UCUM, "passes")
8 | > | CONTAINS | NUM | EV (111437, DCM, "Number of specimens") | 1 | U |  | UNITS = EV ({specimens}, UCUM, "specimens")
9 | > | CONTAINS | CODE | EV (111431, DCM, "Instrument Approach") | 1-n | U |  | DCID 6065 “Instrument Approach”
10 | > | CONTAINS | CODE | EV (111438, DCM, "Needle in target") | 1 | U |  | DCID 230 “Yes-No”
11 | > | CONTAINS | NUM | EV (111439, DCM, "Number of needles around target") | 1 | U |  | UNITS = EV ({needles}, UCUM, "needles")
12 | > | CONTAINS | CODE | EV (F-04460, SRT, "Medication given") | 1-n | U |  |
13 | > | CONTAINS | CODE | EV (111440, DCM, "Incision made") | 1 | U |  | DCID 230 “Yes-No”
14 | > | CONTAINS | CODE | EV (111123, DCM, "Marker placement") | 1 | U |  | DCID 230 “Yes-No”
15 | > | CONTAINS | CODE | EV (111442, DCM, "Confirmation of target") | 1 | U |  | DCID 6066 “Target Confirmation”
16 | > | CONTAINS | CODE | EV (DD-60002, SRT, "Complication of procedure") | 1-n | U |  | DCID 6062 “Interventional Procedure Complications”
17 | >> | HAS PROPERTIES | CODE | EV (111466, DCM, "Severity of Complication") | 1 | U |  | DCID 251 “Severity of Complication”
18 | > | CONTAINS | CONTAINER | EV (121027, DCM, "Specimen") | 1-n | U |  |
19 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1009 “Subject Context, Specimen” | 1 | U |  |
20 | >> | CONTAINS | CODE | EV (F-00E6D, SRT, "Color of fluid") | 1 | U |  | DCID 6067 “Fluid Color”
21 | >> | CONTAINS | CODE | EV (111456, DCM, "Action on fluid") | 1 | U |  | DT (111457, DCM, "Sent for analysis") DT (111458, DCM, "Discarded")
22 | >> | CONTAINS | CODE | EV (111455, DCM, "Occult blood test result") | 1 | U |  | DCID 250 “Positive-Negative”
23 | >> | CONTAINS | INCLUDE | DTID 4207 “Breast Imaging Pathology Results” | 1-n | U |  |
24 | > | CONTAINS | INCLUDE | DTID 350 “References to Supporting Evidence” | 1 | U |  |

### TID 4205 Breast Composition Section

This Template defines a Breast Composition section for the supplementary data sub-tree of the Breast Imaging Report. It is included from TID 4208 “Breast Imaging Report Supplementary Data”.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4205. Breast Composition Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (F-01710, SRT, "Breast composition") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 4201 “Breast Imaging Procedure Reported” | 1-n | U |  |
3 | > | CONTAINS | CODE | EV (F-01710, SRT, "Breast composition") | 1-n | MC | At least one of row 3, 5 shall be present | DCID 6000 “Overall Breast Composition”
4 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | DCID 6022 “Side”
5 | > | CONTAINS | NUM | EV (111046, DCM, "Percent Fibroglandular Tissue") | 1-n | MC | At least one of row 3, 5 shall be present | UNITS = EV (%, UCUM, "Percent")
6 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | DCID 6022 “Side”
7 | > | CONTAINS | CODE | EV (111350, DCM, "Breast background echo texture") | 1-2 | U |  | DCID 6151 “Background Echotexture”
8 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | DCID 6022 “Side”

### TID 4206 Breast Imaging Report Finding Section

This Template defines a supplementary data section for the Findings of the Breast Imaging Report. It is included from TID 4208 “Breast Imaging Report Supplementary Data”.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 4206. Breast Imaging Report Finding Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
3 | > | CONTAINS | INCLUDE | DTID 4201 “Breast Imaging Procedure Reported” | 1 | M |  |
4 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | M |  | DCID 6054 “Breast Imaging Findings”
5 | >> | HAS CONCEPT MOD | CODE | EV (111405, DCM, "Implant type") | 1-n | UC | May be present if value of row 4 is (A-04010, SRT, "Implant") | DCID 6059 “Breast Implant Types”
6 | >> | HAS PROPERTIES | INCLUDE | DTID 4203 “Breast Imaging Assessment” | 1 | U |  |
7 | >> | HAS PROPERTIES | CODE | EV (111014, DCM, "Clockface or region") | 1 | U |  | DCID 6018 “Clockface Location or Region”
8 | >> | HAS PROPERTIES | CODE | EV (111048, DCM, "Quadrant location") | 1 | U |  | DCID 6020 “Quadrant Location”
9 | >> | HAS PROPERTIES | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  |
10 | >> | HAS PROPERTIES | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  |
11 | >> | HAS PROPERTIES | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  |
12 | >> | HAS PROPERTIES | CODE | EV (111020, DCM, "Depth") | 1 | U |  | DCID 6024 “Depth”
13 | >> | HAS PROPERTIES | CODE | EV (111035, DCM, "Lesion Density") | 1 | U |  | DCID 6008 “Density Modifier”
14 | >> | HAS PROPERTIES | CODE | EV (M-020F9, SRT, "Shape") | 1-n | U |  | DCID 6004 “Mammography Characteristics of Shape”
15 | >> | HAS PROPERTIES | CODE | EV (111037, DCM, "Margins") | 1-n | U |  | DCID 6006 “Mammography Characteristics of Margin”
16 | >> | HAS PROPERTIES | CODE | EV (111009, DCM, "Calcification Type") | 1-n | U |  | DCID 6010 “Mammography Calcification Types”
17 | >> | HAS PROPERTIES | CODE | EV (111008, DCM, "Calcification Distribution") | 1 | U |  | DCID 6012 “Calcification Distribution Modifier”
18 | >> | HAS PROPERTIES | NUM | EV (111038, DCM, "Number of calcifications") | 1 | U |  | UNITS = EV ({calcifications}, UCUM, "calcifications") Value = Integer 1 - n
19 | >> | HAS PROPERTIES | CODE | EV (111407, DCM, "Implant finding") | 1-n | U |  | DCID 6072 “Breast Implant Findings”
20 | >> | HAS PROPERTIES | CODE | EV (G-C189, SRT, "Associated Finding") | 1-n | U |  | DCID 6056 “Associated Findings for Breast”
21 | >> | HAS PROPERTIES | NUM | EV (111406, DCM, "Number of similar findings") | 1 | U |  | UNITS = EV ({findings}, UCUM, "findings") Value = Integer 2 - n
22 | >> | HAS PROPERTIES | CODE | EV (F-01720, SRT, "Change since last mammogram") | 1-n | U |  | DCID 6002 “Change Since Last Mammogram or Prior Surgery”
23 | >> | HAS PROPERTIES | CODE | EV (111354, DCM, "Orientation") | 1 | U |  | DCID 6152 “Orientation”
24 | >> | HAS PROPERTIES | CODE | EV (111357, DCM, "Lesion boundary") | 1 | U |  | DCID 6153 “Lesion Boundary”
25 | >> | HAS PROPERTIES | CODE | EV (111360, DCM, "Echo pattern") | 1 | U |  | DCID 6154 “Echo Pattern”
26 | >> | HAS PROPERTIES | CODE | EV (111366, DCM, "Posterior acoustic features") | 1 | U |  | DCID 6155 “Posterior Acoustic Features”
27 | >> | HAS PROPERTIES | CODE | EV (111371, DCM, "Identifiable effect on surrounding tissues") | 1 | U |  | DCID 6015 “Single Image Finding from BI-RADS®”
28 | >> | HAS PROPERTIES | CODE | EV (111372, DCM, "Vascularity") | 1 | U |  | DCID 6157 “Vascularity”
29 | >> | HAS PROPERTIES | CODE | EV (111380, DCM, "Correlation to Other Findings") | 1-n | U |  | DCID 6158 “Correlation to Other Findings”
30 | >> | INFERRED FROM | INCLUDE | DTID 350 “References to Supporting Evidence” | 1 | U |  |

### TID 4207 Breast Imaging Pathology Results

This Template defines the pathology results for a procedure. It may be applied to a Breast Imaging Assessment (see TID 4203 “Breast Imaging Assessment”), or a Breast Imaging Intervention (see TID 4204 “Breast Imaging Report Intervention Section”).

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 4207. Breast Imaging Pathology Results

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111468, DCM, "Pathology Results") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 4201 “Breast Imaging Procedure Reported” | 1 | U |  |
3 | > | CONTAINS | DATETIME | EV (111469, DCM, "Sampling DateTime") | 1 | M |  |
4 | > | CONTAINS | CODE | EV (122177, DCM, "Procedure Result") | 1 | M |  | DCID 6063 “Interventional Procedure Results”
5 | > | CONTAINS | CODE | EV (111042, DCM, "Pathology") | 1-n | U |  | BCID 6030 “Mammography Pathology Codes”
6 | >> | HAS PROPERTIES | CODE | EV (111388, DCM, "Malignancy Type") | 1 | U |  | DCID 6159 “Malignancy Type”
7 | >> | HAS PROPERTIES | NUM | DCID 6165 “Breast Linear Measurements” | 1-n | U |  | UNITS = EV (mm, UCUM, "millimeter")
8 | >> | HAS PROPERTIES | CODE | EV (F-02900, SRT, "Histological grade finding") | 1 | U |  | BCID 6069 “Nottingham Combined Histologic Grade” BCID 6070 “Bloom-Richardson Histologic Grade”
9 | >>> | HAS CONCEPT MOD | CODE | EV (R-00258, SRT, "Histologic grade") | 1 | U |  | BCID 6071 “Histologic Grading Method”
10 | >> | HAS PROPERTIES | CODE | EV (R-00274, SRT, "Tumor margin status") | 1 | U |  | DT (111470, DCM, "Uninvolved"), DT (111471, DCM, "Involved")
11 | >> | HAS PROPERTIES | CODE | EV (111472, DCM, "Nipple involved") | 1 | U |  | DCID 230 “Yes-No”
12 | >> | HAS PROPERTIES | NUM | EV (111473, DCM, "Number of nodes removed") | 1 | U |  | UNITS = EV ({nodes}, UCUM, "nodes")
13 | >> | HAS PROPERTIES | NUM | EV (111474, DCM, "Number of nodes positive") | 1 | MC | Shall be present IFF value of row 12 is > 0 | UNITS = EV ({nodes}, UCUM, "nodes")
14 | >> | HAS PROPERTIES | CODE | EV (R-00465, SRT, "pT category finding") | 1 | U |  | DCID 6160 “Breast Primary Tumor Assessment From AJCC”
15 | >> | HAS PROPERTIES | CODE | EV (R-00463, SRT, "Node stage finding") | 1 | U |  | DCID 6161 “Clinical Regional Lymph Node Assessment for Breast”
16 | >> | HAS PROPERTIES | CODE | EV (R-00461, SRT, "Metastasis stage finding") | 1 | U |  | DCID 6162 “Assessment of Metastasis for Breast”
17 | >> | HAS PROPERTIES | CODE | EV (R-00443, SRT, "Tumor stage finding") | 1 | U |  | BCID 6068 “Tumor Stages From AJCC”
18 | >> | HAS PROPERTIES | CODE | EV (111475, DCM, "Estrogen receptor") | 1 | U |  | DCID 250 “Positive-Negative”
19 | >> | HAS PROPERTIES | CODE | EV (111476, DCM, "Progesterone receptor") | 1 | U |  | DCID 250 “Positive-Negative”
20 | >> | HAS PROPERTIES | NUM | EV (111477, DCM, "S Phase") | 1 | U |  | UNITS = EV (%, UCUM, "percent")

### TID 4208 Breast Imaging Report Supplementary Data

This Template forms a supplementary data sub-tree of the content tree of a Breast Imaging Report. Each subsection provides a specific type of supporting evidence to the narrative text sub-tree, for example, as coded and numeric data.

See Figure Q.1-4 “Breast Imaging Report Supplementary Data Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 4208. Breast Imaging Report Supplementary Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111414, DCM, "Supplementary Data") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 4201 “Breast Imaging Procedure Reported” | 1-n | M |  |
3 | > | CONTAINS | CODE | EV (111403, DCM, "Baseline screening mammogram") | 1 | U |  | DCID 230 “Yes-No”
4 | > | CONTAINS | CODE | EV (111404, DCM, "First mammogram ever") | 1 | U |  | DCID 230 “Yes-No”
5 | > | CONTAINS | INCLUDE | DTID 4205 “Breast Composition Section” | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 4206 “Breast Imaging Report Finding Section” | 1-n | U |  |
7 | > | CONTAINS | INCLUDE | DTID 4204 “Breast Imaging Report Intervention Section” | 1-n | U |  |
8 | > | CONTAINS | CONTAINER | EV (111413, DCM, "Overall Assessment") | 1 | U |  |
9 | >> | CONTAINS | INCLUDE | DTID 4203 “Breast Imaging Assessment” | 1 | M |  |

## OB-GYN Report Templates

### TID 5000 OB-GYN Ultrasound Procedure Report

This is the Template for the root of the content tree for the OB-GYN ultrasound procedure report.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 5000. OB-GYN Ultrasound Procedure Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125000, DCM, "OB-GYN Ultrasound Procedure Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 5001 “OB-GYN Patient Characteristics” | 1 | U |  |
5 | > | CONTAINS | CONTAINER | DT (111028, DCM, "Image Library") | 1 | U |  |
6 | >> | CONTAINS | IMAGE | No purpose of reference | 1-n | M |  |
7 | > | CONTAINS | INCLUDE | DTID 5002 “OB-GYN Procedure Summary Section” | 1 | U |  |
8 | > | CONTAINS | INCLUDE | DTID 5004 “Fetal Biometry Ratio Section” | 1-n | U |  |
9 | > | CONTAINS | INCLUDE | DTID 5005 “Fetal Biometry Section” | 1-n | U |  |
10 | > | CONTAINS | INCLUDE | DTID 5006 “Fetal Long Bones Section” | 1-n | U |  |
11 | > | CONTAINS | INCLUDE | DTID 5007 “Fetal Cranium Section” | 1-n | U |  |
12 | > | CONTAINS | INCLUDE | DTID 5009 “Fetal Biophysical Profile Section” | 1-n | U |  |
13 | > | CONTAINS | INCLUDE | DTID 5011 “Early Gestation Section” | 1-n | U |  |
14 | > | CONTAINS | INCLUDE | DTID 5010 “Amniotic Sac Section” | 1 | U |  |
15 | > | CONTAINS | INCLUDE | DTID 5015 “Pelvis and Uterus Section” | 1 | U |  |
16 | > | CONTAINS | INCLUDE | DTID 5012 “Ovaries Section” | 1 | U |  |
17 | > | CONTAINS | INCLUDE | DTID 5013 “Follicles Section” | 1 | U |  | $Laterality = EV (G-A101, SRT, "Left") $Number = EV (11879-4, LN, "Number of follicles in left ovary")
18 | > | CONTAINS | INCLUDE | DTID 5013 “Follicles Section” | 1 | U |  | $Laterality = EV (G-A100, SRT, "Right") $Number = EV (11880-2, LN, "Number of follicles in right ovary")
19 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | U |  |
20 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | EV (T-F6800, SRT, "Embryonic Vascular Structure")
21 | >> | CONTAINS | INCLUDE | DTID 5025 “OB-GYN Fetal Vascular Ultrasound Measurement Group” | 1 | M |  | $AnatomyGroup = DCID 12141 “Fetal Vasculature Anatomical Location”
22 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | U |  |
23 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | EV (T-D6007, SRT, "Pelvic Vascular Structure")
24 | >> | CONTAINS | INCLUDE | DTID 5026 “OB-GYN Pelvic Vascular Ultrasound Measurement Group” | 1 | M |  | $AnatomyGroup = DCID 12140 “Pelvic Vasculature Anatomical Location”

### TID 5001 OB-GYN Patient Characteristics

Patient Characteristic concepts in this Template, which may replicate attributes in the Patient Study Module, are included here as possible targets of by-reference relationships from other Content Items in the SR tree.

### Note

Several of the concepts in this Template duplicate concepts in TID 1007 “Subject Context, Patient”. The difference in use is that this Template has those concepts as primary observations of the patient, while in TID 1007 “Subject Context, Patient” the concepts are used to set (or reset) the context for other observations.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5001. OB-GYN Patient Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121118, DCM, "Patient Characteristics") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
3 | > | CONTAINS | NUM | EV (8302-2, LN, "Patient Height") | 1 | U |  |
4 | > | CONTAINS | NUM | EV (29463-7, LN, "Patient Weight") | 1 | U |  |
5 | > | CONTAINS | NUM | EV (11996-6, LN, "Gravida") | 1 | U |  |
6 | > | CONTAINS | NUM | EV (11977-6, LN, "Para") | 1 | U |  |
7 | > | CONTAINS | NUM | EV (11612-9, LN, "Aborta") | 1 | U |  |
8 | > | CONTAINS | NUM | EV (33065-4, LN, "Ectopic Pregnancies") | 1 | U |  |

### TID 5002 OB-GYN Procedure Summary Section

Observations of the procedure of immediate clinical interest.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5002. OB-GYN Procedure Summary Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | DATE | DCID 12003 “OB-GYN Dates” | 1-n | U |  |
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = BCID 12018 “OB-GYN Summary”
4 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1-n | U |  |
5 | >> |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1-n | U |  |
6 | > | CONTAINS | INCLUDE | BTID 5003 “OB-GYN Fetus Summary” | 1-n | UC | No more than 1 inclusion per fetus |

### TID 5003 OB-GYN Fetus Summary

The Fetus Summary Template is a container for summary data of a fetus.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5003. OB-GYN Procedure Fetus Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125008, DCM, "Fetus Summary") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this Template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1-n | U |  |
4 | >> |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 12019 “OB-GYN Fetus Summary” $Equation = DCID 12012 “OB Equations and Tables”

### TID 5004 Fetal Biometry Ratio Section

The Fetal Biometry Section Ratio Template is a container for common biometric ratios.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5004. Fetal Biometry Ratio Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125001, DCM, "Fetal Biometry Ratios") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this Template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | NUM | DCID 12004 “Fetal Biometry Ratios” | 1-n | M |  |
4 | >> | R-INFERRED FROM | NUM |  | 2 | U |  |
5 | >> | HAS PROPERTIES | INCLUDE | DTID 312 “Normal Range Properties” | 1 | U |  |

Content Item Descriptions

Row 3 | Numeric ratio related to fetal growth
Row 4 | Reference to the numerator and denominator of the ratio.

### TID 5005 Fetal Biometry Section

The Fetal Biometry Section Template is a container for common biometric groups.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5005. Fetal Biometry Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125002, DCM, "Fetal Biometry") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this Template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | INCLUDE | DTID 5008 “Fetal Biometry Group” | 1-n | M |  | $BiometryType = MemberOf {DCID 12005 “Fetal Biometry Measurements” $TargetSite = DCID 12020 “Fetal Biometry Anatomic Sites”

Content Item Descriptions

Row 3 | The group of measurements. Only one group per biometry type.

### TID 5006 Fetal Long Bones Section

The Long Bones Template is a container for biometric data of long bones.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5006. Fetal Long Bones Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125003, DCM, "Fetal Long Bones") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this Template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | INCLUDE | DTID 5008 “Fetal Biometry Group” | 1-n | M |  | $BiometryType = MemberOf {DCID 12006 “Fetal Long Bones Biometry Measurements” $TargetSite = DCID 12021 “Fetal Long Bone Anatomic Sites”

Content Item Descriptions

Row 3 | The group of measurements. Only one group per biometry type.

### TID 5007 Fetal Cranium Section

The Fetal Cranium Template is a container for groups of biometric data of the fetal cranium.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5007. Fetal Cranium Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125004, DCM, "Fetal Cranium") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this Template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | INCLUDE | DTID 5008 “Fetal Biometry Group” | 1-n | M |  | $BiometryType = MemberOf {DCID 12007 “Fetal Cranium” $TargetSite = DCID 12022 “Fetal Cranium Anatomic Sites”

Content Item Descriptions

Row 3 | The group of measurements. Only one group per biometry type.

### TID 5008 Fetal Biometry Group

The Biometry Group Template is container for a biometric value and its associated growth metrics.

Table TID 5008. Parameters

Parameter Name | Parameter Usage
--- | ---
$BiometryType | The concept name of the biometry measurement
$TargetSite | Value for Anatomic Location of the biometry measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5008. Fetal Biometry Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125005, DCM, "Biometry Group") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | MC | At least one of row 2 and 3 shall be present | $Measurement = $BiometryType $TargetSite = $TargetSite $Derivation = DCID 3627 “Measurement Type”
3 | > | CONTAINS | NUM | EV (18185-9, LN, "Gestational Age") | 1 | MC | At least one of row 2 and 3 shall be present | UNITS = EV (d, UCUM, "days")
4 | >> | INFERRED FROM | CODE | DCID 228 “Equation or Table” | 1 | U |  | DCID 12013 “Gestational Age Equations and Tables”
5 | >> | R-INFERRED FROM | NUM |  | 1-n | U |  |
6 | >> | HAS PROPERTIES | NUM | DCID 226 “Population Statistical Descriptors” | 1-n | U |  |
7 | > | CONTAINS | NUM | DCID 12017 “Growth Distribution Rank” | 1 | U |  |
8 | >> | INFERRED FROM | CODE | DCID 228 “Equation or Table” | 1 | U |  | DCID 12015 “Fetal Growth Equations and Tables”

Content Item Descriptions

Row 1 | Container to segregate biometry data by measurement type
Row 2 | The discrete measurements of the biometry type including derived measurements such as mean. One of the measurements may be flagged as selected for derived measurements. The anatomic location may be precoordinated in the measurement type, but may also be explicitly conveyed in the $TargetSite parameter, which then also allows laterality to be encoded within TID 300 “Measurement”.
Row 3 | The estimated gestational age derived from an equation or table based on the explicitly referenced R-INFERRED FROM Content Item, selected measurement or mean, in that order of preference.
Row 4 | The reference that defines the equation or table of GA derivation
Row 6 | The uncertainty/confidence limits of the gestational age
Row 7 | Expresses the rank of the selected or mean measurement of row 2 relative to the distribution specified in row 8.
Row 8 | This row specifies the CODE reference used to compute the percentile or Z-score.

### TID 5009 Fetal Biophysical Profile Section

This Template encodes scoring observations for fetal well-being evaluation as described by Manning, Antepartum Fetal Evaluation: Development of a Fetal Biophysical Profile Score, Am. J Obsetet Gynecol, 1980;136:787.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5009. Fetal Biophysical Profile Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125006, DCM, "Biophysical Profile") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this Template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | NUM | EV (11631-9, LN, "Gross Body Movement") | 1 | MC | At least one of row 3-7 shall be present | UNITS = DT ({0:2}, UCUM, "range 0:2")
4 | > | CONTAINS | NUM | EV (11632-7, LN, "Fetal Breathing") | 1 | MC | At least one of row 3-7 shall be present | UNITS = DT ({0:2}, UCUM, "range 0:2")
5 | > | CONTAINS | NUM | EV (11635-0, LN, "Fetal Tone") | 1 | MC | At least one of row 3-7 shall be present | UNITS = DT ({0:2}, UCUM, "range 0:2")
6 | > | CONTAINS | NUM | EV (11635-5, LN, "Fetal Heart Reactivity") | 1 | MC | At least one of row 3-7 shall be present | UNITS = DT ({0:2}, UCUM, "range 0:2")
7 | > | CONTAINS | NUM | EV (11630-1, LN, "Amniotic Fluid Volume") | 1 | MC | At least one of row 3-7 shall be present | UNITS = DT ({0:2}, UCUM, "range 0:2")
8 | > | CONTAINS | NUM | DT (11634-3, LN, "Biophysical Profile Sum Score") | 1 | U |  |

Content Item Descriptions

Row 3-7 | The numeric profile score of range 0-2
Row 8 | The sum of rows 3-7. The range is from 0 to the maximum possible score according the items scored in rows 3-7.

### TID 5010 Amniotic Sac Section

This Template specifies a container for amniotic sac quadrant diameters and a derived index.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5010. Amniotic Sac Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (T-F1300, SRT, "Amniotic Sac")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = DT (11627-7, LN, "Amniotic Fluid Index")
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 4 | U |  | $Measurement = DCID 12008 “OB-GYN Amniotic Sac”

Content Item Descriptions

Row 3 | The sum of the 4 quadrant diameters
Row 4 | The four amniotic sac quadrant diameters

### TID 5011 Early Gestation Section

The Early Gestation Section Template is a container for common, first trimester biometric groups.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5011. Early Gestation Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125009, DCM, "Early Gestation") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this Template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | INCLUDE | DTID 5008 “Fetal Biometry Group” | 1-n | M |  | $BiometryType = Member of {DCID 12009 “Early Gestation Biometry Measurements”}

### TID 5012 Ovaries Section

This Template contains metrics of ovary size.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5012. Ovaries Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (T-87000, SRT, "Ovary")
3 | > | CONTAINS | INCLUDE | DTID 5016 “LWH Volume Group” | 1 | U |  | $GroupName = EV (T-87000, SRT, "Ovary") $Width = EV (11829-9, LN, "Left Ovary Width") $Length = EV (11840-6, LN, "Left Ovary Length") $Height = EV (11857-0, LN, "Left Ovary Height") $Volume = EV (12164-0, LN, "Left Ovary Volume")
4 | > | CONTAINS | INCLUDE | DTID 5016 “LWH Volume Group” | 1 | U |  | $GroupName = EV (T-87000, SRT, "Ovary") $Width = EV (11830-7, LN, "Right Ovary Width") $Length = EV (11841-4, LN, "Right Ovary Length") $Height = EV (11858-8, LN, "Right Ovary Height") $Volume = EV (12165-7, LN, "Right Ovary Volume")

### TID 5013 Follicles Section

This Template contains follicle metrics for left or right ovarian follicles.

Table TID 5013. Parameters

Parameter Name | Parameter Usage
--- | ---
$Laterality | Ovary laterality
$Number | The number of follicles

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5013. Follicles Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (T-87600, SRT, "Ovarian Follicle")
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | $Laterality
4 | > | CONTAINS | NUM | $Number | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 5014 “Follicle Measurement Group” | 1-n | U |  |

### TID 5014 Follicle Measurement Group

This Template contains metrics for one ovarian follicle.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5014. Follicle Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125007, DCM, "Measurement Group") | 1 | M |  |
2 | > | HAS OBS CONTEXT | TEXT | EV (125010, DCM, "Identifier") | 1 | U |  | Unique among all groups of same laterality
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-D705, SRT, "Volume")
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (11793-7, LN, "Follicle Diameter") $Derivation = DCID 3627 “Measurement Type”

### TID 5015 Pelvis and Uterus Section

This Template contains general measurements in the pelvis and uterus.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5015. Pelvis and Uterus Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125011, DCM, "Pelvis and Uterus") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 5016 “LWH Volume Group” | 1 | U |  | $GroupName = EV (T-83000, SRT, "Uterus") $Width = EV (11865-3, LN, "Uterus Width") $Length = EV (11842-2, LN, "Uterus Length") $Height = EV (11859-6, LN, "Uterus Height") $Volume = EV (33192-6, LN, "Uterus Volume")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 12011 “Ultrasound Pelvis and Uterus” $TargetSite = DCID 12023 “Pelvis and Uterus Anatomic Sites” $Derivation = DCID 3627 “Measurement Type”

### TID 5016 LWH Volume Group

This Template is a container for a group of measurements that assess the size of an anatomical structure using a volume derived from perpendicular diameters.

Table TID 5016. Parameters

Parameter Name | Parameter Usage
--- | ---
$GroupName | The name of the volume group that is an anatomical structure
$Volume | Concept name of volume measurement
$Length | Concept name of length measurement
$Width | Concept name of width measurement
$Height | Concept name of height measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5016. LWH Volume Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | $GroupName | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | At least one of row 2, 3, 4, 5 shall be present | $Measurement = $Volume $TargetSite = $GroupName
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | MC | At least one of row 2, 3, 4, 5 shall be present | $Measurement = $Length $TargetSite = $GroupName $Derivation = DCID 3627 “Measurement Type”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | MC | At least one of row 2, 3, 4, 5 shall be present | $Measurement = $Width $TargetSite = $GroupName $Derivation = DCID 3627 “Measurement Type”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | MC | At least one of row 2, 3, 4, 5 shall be present | $Measurement = $Height $TargetSite = $GroupName $Derivation = DCID 3627 “Measurement Type”

### TID 5025 OB-GYN Fetal Vascular Ultrasound Measurement Group

This Template is an anatomy specific container of OB-GYN fetal vascular measurements.

Table TID 5025. Parameters

Parameter Name | Parameter Usage
--- | ---
$AnatomyGroup | The concept name of the vascular anatomy

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5025. OB-GYN Fetal Vascular Ultrasound Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | $AnatomyGroup | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this Template is invoked more than once to describe more than one fetus |
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IFF anatomy has laterality | DCID 244 “Laterality”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $MeasType = DCID 12119 “Vascular Ultrasound Property” $Derivation = DCID 3627 “Measurement Type”

Content Item Descriptions

Anatomy Group | Specifies the anatomical context of the observations in the group.

### TID 5026 OB-GYN Pelvic Vascular Ultrasound Measurement Group

This Template is an anatomy specific container of OB-GYN pelvic vascular measurements.

Table TID 5026. Parameters

Parameter Name | Parameter Usage
--- | ---
$AnatomyGroup | The concept name of the vascular anatomy

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5026. OB-GYN Pelvic Vascular Ultrasound Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | $AnatomyGroup | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IFF anatomy has laterality | DCID 244 “Laterality”
3 | > | HAS CONCEPT MOD | TEXT | EV (112050, DCM, "Anatomic Identifier") | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $MeasType = DCID 12119 “Vascular Ultrasound Property” $Derivation = DCID 3627 “Measurement Type”
5 | >> | HAS CONCEPT MOD | CODE | EV (125105, DCM, "Measurement Orientation") | 1 | U |  | DCID 12118 “Measurement Orientation”
6 | >> | HAS PROPERTIES | NUM | EV (125106, DCM, "Doppler Angle") | 1 | U |  | UNIT = EV (deg, UCUM, "deg")
7 | >> | HAS PROPERTIES | NUM | EV (125107, DCM, "Sample Volume Depth") | 1 | U |  | UNIT = EV (cm, UCUM, "cm")

Content Item Descriptions

Row 1 | Specifies the anatomical context of the observations in the group.
Row 3 | Differentiates between multiple structures such as the two umbilical arteries.

## Vascular Ultrasound Report Templates

### TID 5100 Vascular Ultrasound Report

This is the Template for the root the content tree for the vascular ultrasound procedure report.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5100. Vascular Ultrasound Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125100, DCM, "Vascular Ultrasound Procedure Report") | 1 | M |  |
2 | > | HAS OBS CONTEXT | CODE | EV (R-40FB8, SRT, "Temporal periods Relating to Procedure") | 1 | U |  | DCID 12102 “Temporal Periods Relating to Procedure or Therapy”
3 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
5 | > | CONTAINS | INCLUDE | DTID 5101 “Vascular Patient Characteristics” | 1 | U |  |
6 | > | CONTAINS | CONTAINER | EV (111028, DCM, "Image Library") | 1 | U |  |
7 | >> | CONTAINS | IMAGE | No purpose of reference | 1-n | M |  |
8 | > | CONTAINS | INCLUDE | DTID 5102 “Vascular Procedure Summary Section” | 1 | U |  |
9 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-D0767, SRT, "Blood Vessel of Head") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12105 “Intracranial Cerebral Vessels”
10 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-D0767, SRT, "Blood Vessel of Head") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12105 “Intracranial Cerebral Vessels”
11 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-D0767, SRT, "Blood Vessel of Head") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 12106 “Intracranial Cerebral Vessels (Unilateral)”
12 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-45005, SRT, "Artery of neck") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12104 “Extracranial Arteries” $AnatomyRatio = DCID 12123 “Carotid Ratios”
13 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-45005, SRT, "Artery of neck") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12104 “Extracranial Arteries” $AnatomyRatio = DCID 12123 “Carotid Ratios”
14 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-47040, SRT, "Artery of Lower Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12109 “Lower Extremity Arteries”
15 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-47040, SRT, "Artery of Lower Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12109 “Lower Extremity Arteries”
16 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-49403, SRT, "Vein of Lower Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12110 “Lower Extremity Veins”
17 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-49403, SRT, "Vein of Lower Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12110 “Lower Extremity Veins”
18 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-47020, SRT, "Artery Of Upper Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12107 “Upper Extremity Arteries”
19 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-47020, SRT, "Artery Of Upper Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12107 “Upper Extremity Arteries”
20 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-49103, SRT, "Vein Of Upper Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12108 “Upper Extremity Veins”
21 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-49103, SRT, "Vein Of Upper Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12108 “Upper Extremity Veins”
22 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-71019, SRT, "Vascular Structure Of Kidney") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12115 “Renal Vessels” $AnatomyRatio = DCID 12124 “Renal Ratios”
23 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-71019, SRT, "Vascular Structure Of Kidney") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12115 “Renal Vessels” $AnatomyRatio = DCID 12124 “Renal Ratios”
24 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-46002, SRT, "Artery of Abdomen") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12111 “Abdominal Arteries (Lateral)”
25 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-46002, SRT, "Artery of Abdomen") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12111 “Abdominal Arteries (Lateral)”
26 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-46002, SRT, "Artery of Abdomen") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 12112 “Abdominal Arteries (Unilateral)”
27 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-487A0, SRT, "Vein of Abdomen") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12113 “Abdominal Veins (Lateral)”
28 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-487A0, SRT, "Vein of Abdomen") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12113 “Abdominal Veins (Lateral)”
29 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-487A0, SRT, "Vein of Abdomen") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 12114 “Abdominal Veins (Unilateral)”
30 | > | CONTAINS | INCLUDE | DTID 5105 “Ultrasound Graft Section” | 1 | U |  |

### TID 5101 Vascular Patient Characteristics

Patient Characteristic concepts in this Template, which may replicate attributes in the Patient Study Module, are included here as possible targets of by-reference relationships from other Content Items in the SR tree.

### Note

Several of the concepts in this Template duplicate concepts in TID 1007 “Subject Context, Patient”. The difference in use is that this Template has those concepts as primary observations of the patient, while in TID 1007 “Subject Context, Patient” the concepts are used to set (or reset) the context for other observations.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5101. Vascular Patient Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121118, DCM, "Patient Characteristics") | 1 | M |  |
2 | > | CONTAINS | NUM | EV (121033, DCM, "Subject Age") | 1 | U |  | UNITS = DCID 7456 “Units of Measure for Age”
3 | > | CONTAINS | CODE | EV (121032, DCM, "Subject Sex") | 1 | U |  | DCID 7455 “Sex”
4 | > | CONTAINS | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  |
5 | > | CONTAINS | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | U |  |
6 | > | CONTAINS | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | U |  |

### TID 5102 Vascular Procedure Summary Section

Comments and observations of the procedure of immediate clinical interest.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5102. Vascular Procedure Summary Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | TEXT | DCID 12101 “Vascular Summary” | 1-n | M |  |

### TID 5103 Vascular Ultrasound Section

Sections of a vascular ultrasound report are section containers of an anatomic region consisting of measurement group containers that contain the measurements.

Table TID 5103. Parameters

Parameter Name | Parameter Usage
--- | ---
$SectionScope | The concept name of the section heading modifier
$SectionLaterality | The laterality (if any) of the anatomy in this section heading
$Anatomy | The concept name of the vascular anatomy
$AnatomyRatio | The concept name of anatomy-coordinated ratio concepts

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5103. Vascular Ultrasound Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | $SectionScope
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | $SectionLaterality
4 | > | CONTAINS | INCLUDE | DTID 5104 “Vascular Ultrasound Measurement Group” | 1-n | M |  | $AnatomyGroup = $Anatomy
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = $AnatomyRatio

### TID 5104 Vascular Ultrasound Measurement Group

This Template is an anatomy specific container of measurements.

Table TID 5104. Parameters

Parameter Name | Parameter Usage
--- | ---
$AnatomyGroup | The concept name of the vascular anatomy

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5104. Vascular Ultrasound Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | $AnatomyGroup | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical Modifier") | 1 | U |  | DCID 12116 “Vessel Segment Modifiers”
3 | > | HAS CONCEPT MOD | CODE | EV (125101, DCM, "Vessel Branch") | 1-n | U |  | DCID 12117 “Vessel Branch Modifiers”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = DCID 12119 “Vascular Ultrasound Property” $Derivation = DCID 3627 “Measurement Type”
5 | >> | HAS CONCEPT MOD | CODE | EV (R-4089A, SRT, "Cardiac Cycle Point") | 1 | U |  | DCID 12233 “Cardiac Phase”
6 | >> | HAS CONCEPT MOD | CODE | EV (R-41FFC, SRT, "Temporal period related to eating") | 1 | U |  | DT (G-A491, SRT, "Post-prandial")
7 | >> | HAS CONCEPT MOD | CODE | EV (125105, DCM, "Measurement Orientation") | 1 | U |  | DCID 12118 “Measurement Orientation”
8 | >> | HAS PROPERTIES | NUM | EV (125106, DCM, "Doppler Angle") | 1 | U |  | UNIT = EV (deg, UCUM, "degrees")
9 | >> | HAS PROPERTIES | NUM | EV (125107, DCM, "Sample Volume Depth") | 1 | U |  | UNIT = EV (cm, UCUM, "cm")

Content Item Descriptions

Row 1 | Specifies the anatomic context of the observations in the group.
Row 2 | Details the anatomical location, e.g., proximal, middle, or distal
Row 3 | The particular vessel branch, such as the inferior, medial or lateral
Row 5 | Cardiac phase (systolic, diastolic), especially for aorta measurements
Row 6 | Eating phase, especially for mesenteric and celiac measurements

### TID 5105 Ultrasound Graft Section

This Template is a container of measurements on a vascular graft.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5105. Ultrasound Graft Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (T-D000F, SRT, "Vascular Graft")
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
4 | > | HAS CONCEPT MOD | CODE | DT (G-D871, SRT, "Proximal anastomosis") | 1 | M |  | BCID 12103 “Vascular Ultrasound Anatomic Location”
5 | > | HAS CONCEPT MOD | CODE | DT (G-D872, SRT, "Distal Anastomosis") | 1 | M |  | BCID 12103 “Vascular Ultrasound Anatomic Location”
6 | > | HAS CONCEPT MOD | CODE | DT (125102, DCM, "Graft Type") | 1 | U |  | No BCID
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = DCID 12119 “Vascular Ultrasound Property”

Content Item Descriptions

Proximal anastomosis | The proximal location of the graft
Distal anastomosis | The distal location of the graft
Graft type | The type of graft, e.g., "in situ", "prosthetic", "autogenous"

## Echocardiography Procedure Report Templates

### TID 5200 Echocardiography Procedure Report

This Template forms the top of a content tree that allows an ultrasound device to describe the results of an adult echocardiography imaging procedure. It is instantiated at the root node. It can also be included in other Templates that need to incorporate echocardiography findings into another report as quoted evidence.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 5200. Echocardiography Procedure Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125200, DCM, "Adult Echocardiography Procedure Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
4 | > | CONTAINS | CONTAINER | DT (55111-9, LN, "Current Procedure Descriptions") | 1 | U |  |
5 | >> | CONTAINS | CODE | DT (125203, DCM, "Acquisition Protocol") | 1-n | M |  | BCID 12001 “Ultrasound Protocol Types”
6 | > | CONTAINS | INCLUDE | DTID 5201 “Echocardiography Patient Characteristics” | 1 | U |  |
7 | > | CONTAINS | CONTAINER | EV (111028, DCM, "Image Library") | 1 | U |  |
8 | >> | CONTAINS | IMAGE | No purpose of reference | 1-n | M |  |
9 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-32600, SRT, "Left Ventricle") $MeasType = DCID 12200 “Echocardiography Left Ventricle”
10 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-32500, SRT, "Right Ventricle") $MeasType = DCID 12204 “Echocardiography Right Ventricle”
11 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-32300, SRT, "Left Atrium") $MeasType = DCID 12205 “Echocardiography Left Atrium”
12 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-32200, SRT, "Right Atrium") $MeasType = DCID 12206 “Echocardiography Right Atrium”
13 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-35400, SRT, "Aortic Valve") $MeasType = DCID 12211 “Echocardiography Aortic Valve”
14 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-35300, SRT, "Mitral Valve") $MeasType = DCID 12207 “Echocardiography Mitral Valve”
15 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-35200, SRT, "Pulmonic Valve") $MeasType = DCID 12209 “Echocardiography Pulmonic Valve”
16 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-35100, SRT, "Tricuspid Valve") $MeasType = DCID 12208 “Echocardiography Tricuspid Valve”
17 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-42000, SRT, "Aorta") $MeasType = DCID 12212 “Echocardiography Aorta”
18 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-44000, SRT, "Pulmonary artery") $MeasType = DCID 12210 “Echocardiography Pulmonary Artery”
19 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-48600, SRT, "Vena Cava") $MeasType = DCID 12215 “Echocardiography Vena Cavae”
20 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-48581, SRT, "Pulmonary Venous Structure") $MeasType = DCID 12214 “Echocardiography Pulmonary Veins”
21 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-39050, SRT, "Pericardial cavity") $MeasType = DCID 12250 “Cardiac Ultrasound Common Linear Measurements”
22 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (P5-30031, SRT, "Cardiac Shunt Study") $MeasType = DCID 12217 “Echocardiography Cardiac Shunt”
23 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (D4-30000, SRT, "Congenital Anomaly of Cardiovascular System") $MeasType = DCID 12218 “Echocardiography Congenital”
24 | > | CONTAINS | INCLUDE | DTID 5204 “Wall Motion Analysis” | 1-n | U |  | $Procedure = DT (P5-B3121, SRT, "Echocardiography for Determining Ventricular Contraction")

Content Item Descriptions

Row 24 | The wall motion findings of stress stage. There may be multiple Template instances to report wall motion findings of multiple stages.

### TID 5201 Echocardiography Patient Characteristics

Patient Characteristic concepts in this Template, which may replicate attributes in the Patient Study Module, are included here as possible targets of by-reference relationships from other Content Items in the SR tree.

### Note

Several of the concepts in this Template duplicate concepts in TID 1007 “Subject Context, Patient”. The difference in use is that this Template has those concepts as primary observations of the patient, while in TID 1007 “Subject Context, Patient” the concepts are used to set (or reset) the context for other observations.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5201. Echocardiography Patient Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121118, DCM, "Patient Characteristics") | 1 | M |  |
2 | > | CONTAINS | NUM | EV (121033, DCM, "Subject Age") | 1 | U |  | UNITS = DCID 7456 “Units of Measure for Age”
3 | > | CONTAINS | CODE | EV (121032, DCM, "Subject Sex") | 1 | U |  | DCID 7455 “Sex”
4 | > | CONTAINS | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  |
5 | > | CONTAINS | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | U |  |
6 | > | CONTAINS | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | U |  |
7 | > | CONTAINS | NUM | EV (8277-6, LN, "Body Surface Area") | 1 | M |  |
8 | >> | INFERRED FROM | CODE | EV (8278-4, LN, "Body Surface Area Formula") | 1 | U |  | BCID 3663 “Body Surface Area Equations”

### TID 5202 Echo Section

This is a generic section heading Template for any of the anatomical headings. Measurements within a section heading appear as groups (by image mode, acquisition protocol, and/or protocol stage).

Table TID 5202. Parameters

Parameter Name | Parameter Usage
--- | ---
$SectionSubject | The subject modifier of the section heading container
$MeasType | The concept name of the measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5202. Echo Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | $SectionSubject
3 | > | CONTAINS | CONTAINER | DT (125007, DCM, "Measurement Group") | 1-n | M |  |
4 | >> | HAS CONCEPT MOD | CODE | EV (G-0373, SRT, "Image Mode") | 1 | U |  | BCID 12224 “Ultrasound Image Modes”
5 | >> | HAS CONCEPT MOD | CODE | DT (125203, DCM, "Acquisition Protocol") | 1 | U |  |
6 | >> | HAS CONCEPT MOD | TEXT | DT (125203, DCM, "Acquisition Protocol") | 1 | U |  |
7 | >> | HAS ACQ CONTEXT | CODE | EV (18139-6, LN, "Stage") | 1 | U |  | BCID 12002 “Ultrasound Protocol Stage Types”
8 | >> | CONTAINS | INCLUDE | DTID 5203 “Echo Measurement” | 1-n | M |  | $Measurement = $MeasType $Method = CID 12227 “Echocardiography Measurement Method”

Content Item Descriptions

Rows 4, 5 | Type of measurement group. May be grouped by image mode, or acquisition protocol, or some other user or manufacturer designated classification
Row 7 | For measurements acquired in a staged protocol, all measurements in a measurement group are acquired at the identified stage.

### TID 5203 Echo Measurement

Table TID 5203. Parameters

Parameter Name | Parameter Usage
--- | ---
$Measurement | Coded term or Context Group for Concept Name of measurement
$Method | Value for Measurement Method

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5203. Echo Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = $Measurement $Method = $Method $TargetSite = BCID 12236 “Echo Anatomic Sites” $TargetSiteMod = BCID 12237 “Echocardiography Anatomic Site Modifiers”
2 | > | HAS CONCEPT MOD | CODE | EV (G-C048, SRT, "Flow Direction") | 1 | U |  | BCID 12221 “Flow Direction”
3 | > | HAS CONCEPT MOD | CODE | EV (R-40899, SRT, "Respiratory Cycle Point") | 1 | U |  | DCID 12234 “Respiration State”
4 | > | HAS CONCEPT MOD | CODE | EV (R-4089A, SRT, "Cardiac Cycle Point") | 1 | U |  | DCID 12233 “Cardiac Phase”
5 | > | HAS ACQ CONTEXT | CODE | EV (G-0373, SRT, "Image Mode") | 1 | U |  | DCID 12224 “Ultrasound Image Modes”
6 | > | HAS ACQ CONTEXT | CODE | EV (111031, DCM, "Image View") | 1 | U |  | BCID 12226 “Echocardiography Image View”
7 | > | HAS ACQ CONTEXT | CODE | EV (18139-6, LN, "Stage") | 1 | U |  | BCID 12002 “Ultrasound Protocol Stage Types”

Content Item Descriptions

Row 1 | TID 300 specifies an "Equivalent Meaning of Concept Name" that allows the creating application to specify the preferred composed concept name representing the measurement and the associated post-coordination Concept Modifiers (e.g., the ASE terminology described in Section N.3 “Illustrative Mapping to ASE Concepts” in PS3.17 ).

### TID 5204 Wall Motion Analysis

The Wall Motion Analysis Template is used to document wall motion scoring for any imaging modality.

Table TID 5204. Parameters

Parameter Name | Parameter Usage
--- | ---
$Procedure | The imaging procedure used for wall motion analysis.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5204. Wall Motion Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | $Procedure
3 | > | HAS ACQ CONTEXT | CODE | EV (18139-6, LN, "Stage") | 1 | U |  | BCID 3207 “Stress Test Procedure Phases”
4 | > | CONTAINS | IMAGE | EV (125201, DCM, "Illustration of Finding") | 1 | U |  |
5 | > | CONTAINS | TEXT | EV (18118-0, LN, "LV Wall Motion Segmental Findings") | 1 | U |  |
6 | > | CONTAINS | NUM | DT (125202, DCM, "LV Wall Motion Score Index") | 1 | U |  |
7 | >> | HAS CONCEPT MOD | CODE | EV (G-E048, SRT, "Assessment Scale") | 1 | M |  | BCID 12238 “Wall Motion Scoring Schemes”
8 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | U |  |
9 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (T-D075D, SRT, "Myocardial Wall")
10 | >> | CONTAINS | CODE | EV (18179-2, LN, "Wall Segment") | 1-n | M |  | BCID 3717 “Myocardial Wall Segments”
11 | >>> | HAS PROPERTIES | CODE | EV (F-32050, SRT, "Cardiac Wall Motion") | 1 | MC | IF row 12 is absent | DCID 3703 “Wall Motion”
12 | >>> | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | MC | IF row 11 is absent | DCID 3704 “Myocardium Wall Morphology Findings”
13 | >>> | HAS PROPERTIES | NUM | DT (G-C1E3, SRT, "Score") | 1 | U |  |
14 | >>> | HAS PROPERTIES | NUM | EV (122624, DCM, "Wall Thickness Ratio end-systolic to end-diastolic") | 1 | U |  | UNITS = DT (%, UCUM, "%")

Content Item Descriptions

Row 3 | The stage of the protocol at which these findings were scored. This row may be absent if this is a generic, non-staged scoring.
Row 4 | Image that graphically depicts the segments and their scores.
Row 5 | Text narration accompanying this stage.
Row 6 | The composite score computed from the average of the scored segments
Row 7 | The type of scoring scheme used to score this exam.
Row 8 | A container of all of the individual segment findings for this stage. The container shall be present if the observer makes an assessment, including the assessment of Not Visualized. It shall not be present if no evaluation was made.
Rows 11, 12 | Scar/thinning (in Row 12) may accompany akinesis and dyskinesis (in Row 11).
Row 13 | A numeric designation for the score. Score ranges vary, typically 0-4 or 0-5. Numeric scores may depend on wall motion findings as well as morphology findings. See Table 5204-1 for conventional numeric assignment schemes. The UCUM annotation code enables specifying the numeric range, ({L:N}, UCUM, "scale L:N"), where L and N are the lower and upper ends of the range.

A description of the scoring schemes described in Table 5204-1 is available in Recommendations for Quantitation of the Left Ventricle by Two-Dimensional Echocardiography, Journal of the American Society of Echocardiography, Vol 2, No 5 358-367, Oct 1989.

Table 5204-1. Numeric Score Assignment for Segmental Findings

Conventional Numeric Assignment | Wall Motion Finding or Morphology Finding
--- | ---
4 Point | 5 Point | 5 Point with Graded Hypokinesis
-1 | (F-32050, SRT, "Cardiac Wall Motion") = (R-0030D, SRT, "Hyperkinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (R-0030D, SRT, "Hyperkinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (R-0030D, SRT, "Hyperkinesis")
0 | (F-32050, SRT, "Cardiac Wall Motion") = (122288, DCM, "Not Visualized") | (F-32050, SRT, "Cardiac Wall Motion") = (122288, DCM, "Not Visualized") | (F-32050, SRT, "Cardiac Wall Motion") = (122288, DCM, "Not Visualized")
1 | (F-32050, SRT, "Cardiac Wall Motion") = (R-00344, SRT, "Normal Wall Motion") | (F-32050, SRT, "Cardiac Wall Motion") = (R-00344, SRT, "Normal Wall Motion") | (F-32050, SRT, "Cardiac Wall Motion") = (R-00344, SRT, "Normal Wall Motion")
1.5 |  |  | (F-32050, SRT, "Cardiac Wall Motion") = (R-00327, SRT, "Mild Hypokinesis")
2 | (F-32050, SRT, "Cardiac Wall Motion") = (F-32056, SRT, "Hypokinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (F-32056, SRT, "Hypokinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (R-0032F, SRT, "Moderate Hypokinesis")
2.5 |  |  | (F-32050, SRT, "Cardiac Wall Motion") = (R-00370, SRT, "Severe Hypokinesis")
3 | (F-32050, SRT, "Cardiac Wall Motion") = (F-30004, SRT, "Akinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (F-30004, SRT, "Akinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (F-30004, SRT, "Akinesis")
4 | (F-32050, SRT, "Cardiac Wall Motion") = (F-32052, SRT, "Dyskinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (F-32052, SRT, "Dyskinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (F-32052, SRT, "Dyskinesis")
5 |  | (G-C504, SRT, "Associated Morphology") = (D3-10510, SRT, "Ventricular Aneurysm") | (G-C504, SRT, "Associated Morphology") = (D3-10510, SRT, "Ventricular Aneurysm")

### TID 5220 Pediatric, Fetal and Congenital Cardiac Ultrasound Reports

This Template forms the top of a content tree that allows an ultrasound application to describe the results of a Cardiac Ultrasound imaging procedure. It is instantiated at the root node.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 5220. Pediatric, Fetal and Congenital Cardiac Ultrasound Reports

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DCID 12245 “Cardiac Ultrasound Report Titles” | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
4 | > | CONTAINS | CONTAINER | EV (18785-6, LN, "Indications for Procedure") | 1 | U |  |
5 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 12246 “Cardiac Ultrasound Indication for Study”
6 | >> | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1 | U |  |
7 | > | CONTAINS | INCLUDE | DTID 3802 “Cardiovascular Patient History” | 1 | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | U |  |
9 | > | CONTAINS | INCLUDE | DTID 5225 “Cardiac Ultrasound Fetal Characteristics” | 1-n | U |  | No more than one inclusion per fetus
10 | > | CONTAINS | INCLUDE | DTID 5226 “Cardiac Ultrasound Summary Section” | 1 | U |  |
11 | > | CONTAINS | INCLUDE | DTID 5227 “Cardiac Ultrasound Fetal Summary Section” | 1-n | U |  | No more than one inclusion per fetus
12 | > | CONTAINS | CONTAINER | EV (111028, DCM, "Image Library") | 1 | U |  |
13 | >> | CONTAINS | IMAGE | No purpose of reference | 1-n | M |  |
14 | > | CONTAINS | INCLUDE | DTID 5221 “Cardiac Ultrasound Pediatric Echo Measurement Section” | 1 | U |  |
15 | > | CONTAINS | INCLUDE | DTID 5228 “Cardiac Ultrasound Fetal Measurement Section” | 1-n | UC | For Fetal Report only. | No more than one inclusion per fetus

Content Item Descriptions

Row 3 | For Fetal Report, this row establishes the subject context of the mother.
Row 7 | For Fetal Report, this row will be the patient history of the mother.
Row 8 | For Fetal Report, this row will be the Patient Characteristics for the mother.
Row 10 | For Fetal Report, this row will be the Summary Section for the mother.

### TID 5221 Cardiac Ultrasound Pediatric Echo Measurement Section

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5221. Cardiac Ultrasound Pediatric Echo Measurement Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12282 “Cardiac Ultrasound Venous Return Systemic Finding Sites” $MeasType = DCID 12264 “Cardiac Ultrasound Venous Return Systemic Measurements”
2 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12283 “Cardiac Ultrasound Venous Return Pulmonary Finding Sites” $MeasType = DCID 12263 “Cardiac Ultrasound Venous Return Pulmonary Measurements”
3 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12284 “Cardiac Ultrasound Atria and Atrial Septum Finding Sites” $MeasType = DCID 12265 “Cardiac Ultrasound Atria and Atrial Septum Measurements”
4 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12285 “Cardiac Ultrasound Atrioventricular Valves Finding Sites” $MeasType = DCID 12268 “Cardiac Ultrasound Atrioventricular Valves Measurements”
5 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12286 “Cardiac Ultrasound Interventricular Septum Finding Sites” $MeasType = DCID 12269 “Cardiac Ultrasound Interventricular Septum Measurements”
6 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12287 “Cardiac Ultrasound Ventricles Finding Sites” $MeasType = DCID 12259 “Cardiac Ultrasound Ventricles Measurements”
8 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12288 “Cardiac Ultrasound Outflow Tracts Finding Sites” $MeasType = DCID 12271 “Cardiac Ultrasound Outflow Tracts Measurements”
9 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12289 “Cardiac Ultrasound Semilunar Valves, Annulus and Sinuses Finding Sites” $MeasType = DCID 12272 “Cardiac Ultrasound Semilunar Valves, Annulate and Sinuses Measurements”
10 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12290 “Cardiac Ultrasound Pulmonary Arteries Finding Sites” $MeasType = DCID 12260 “Cardiac Ultrasound Pulmonary Artery”
11 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12291 “Cardiac Ultrasound Aorta Finding Sites” $MeasType = DCID 12274 “Cardiac Ultrasound Aorta Measurements”
12 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12292 “Cardiac Ultrasound Coronary Arteries Finding Sites” $MeasType = DCID 12275 “Cardiac Ultrasound Coronary Arteries Measurements”
13 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12293 “Cardiac Ultrasound Aortopulmonary Connections Finding Sites” $MeasType = DCID 12276 “Cardiac Ultrasound Aorto Pulmonary Connections Measurements”
14 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12294 “Cardiac Ultrasound Pericardium and Pleura Finding Sites” $MeasType = DCID 12277 “Cardiac Ultrasound Pericardium and Pleura Measurements”

### TID 5222 Pediatric, Fetal and Congenital Cardiac Ultrasound Section

This is a generic section heading Template for any of the anatomical headings. Measurements within a section heading appear as groups (by image mode or acquisition protocol).

Table TID 5222. Parameters

Parameter Name | Parameter Usage
--- | ---
$SectionSubject | The subject modifier of the section heading container
$MeasType | The concept name of the measurement

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5222. Pediatric, Fetal and Congenital Cardiac Ultrasound Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | $SectionSubject
3 | >> | HAS CONCEPT MOD | CODE | EV (P1-32006, SRT, "Heart valve replacement - prosthesis") | 1 | U |  | DCID 230 “Yes-No”
4 | > | CONTAINS | CONTAINER | DT (125007, DCM, "Measurement Group") | 1-n | M |  |
5 | >> | HAS CONCEPT MOD | CODE | EV (G-0373, SRT, "Image Mode") | 1 | U |  | BCID 12224 “Ultrasound Image Modes”
6 | >> | HAS CONCEPT MOD | TEXT | DT (125203, DCM, "Acquisition Protocol") | 1 | U |  |
7 | >> | CONTAINS | INCLUDE | DTID 5223 “Pediatric, Fetal and Congenital Cardiac Ultrasound Measurement” | 1-n | M |  | $Measurement = $MeasType $Method = CID 12227 “Echocardiography Measurement Method”

### TID 5223 Pediatric, Fetal and Congenital Cardiac Ultrasound Measurement

This Template provides for the post-coordination of a measurement with a variety of concept modifiers and acquisition context observations. When invoked from TID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section”, the measurement concept is implicitly post-coordinated with the concept modifiers of the Measurement Group (TID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” Rows 5 and 6), and with the Finding Site of the report section (TID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” Row 2). The finding site may be further specified within this Template by the Target Site and Target Site Modifiers (CID 12280 “Cardiac Ultrasound Target Sites” and CID 12281 “Cardiac Ultrasound Target Site Modifiers”).

The implicit finding site inherited from TID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” can be made explicit by using the same finding site concept in the Target Site (the measurement concept modifier), rather than a term from CID 12280 “Cardiac Ultrasound Target Sites”. This explicit post-coordination allows the use of one of the modifiers of CID 12281 “Cardiac Ultrasound Target Site Modifiers” to that finding site, as the Target Site Modifier requires an explicit Target Site in the measurement structure (TID 300 “Measurement” Rows 5 and 7). In fact, any child concept of the finding site in the SNOMED hierarchy may be used as the measurement Target Site.

The finding or target site may be identified by a concept from the SNOMED "clinical finding" or "morphological anomaly" hierarchies (e.g., D4-31220 "Atrial Septal Defect", or M-36700 "Effusion"), rather than the "anatomical structure" hierarchy. In this case, the meaning is inferred as "the anatomic location of the clinical finding or morphological anomaly, within the constraints of other implicit or explicit post-coordinated finding site concepts."

### Note

Thus when TID 5221 “Cardiac Ultrasound Pediatric Echo Measurement Section” Row 14 invokes TID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” with the section finding site concept (T-39000, SRT, "Pericardium"), and TID 5223 “Pediatric, Fetal and Congenital Cardiac Ultrasound Measurement” Row 1 applies the target site (M-36700, SRT, "Effusion"), the effective finding site is "pericardial effusion".

Table TID 5223. Parameters

Parameter Name | Parameter Usage
--- | ---
$Measurement | Coded term or Context Group for Concept Name of measurement
$Method | Value for Measurement Method

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5223. Pediatric, Fetal and Congenital Cardiac Ultrasound Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = $Measurement $Method = $Method $TargetSite = BCID 12280 “Cardiac Ultrasound Target Sites” $TargetSiteMod = BCID 12281 “Cardiac Ultrasound Target Site Modifiers” $Derivation = DCID 3838 “Diameter Derivation”
2 | > | HAS CONCEPT MOD | CODE | EV (121425, DCM, "Index") | 1 | U |  | DCID 3455 “Index Methods”
3 | > | HAS CONCEPT MOD | CODE | EV (G-C048, SRT, "Flow Direction") | 1 | U |  | BCID 12221 “Flow Direction”
4 | > | HAS CONCEPT MOD | CODE | EV (R-40899, SRT, "Respiratory Cycle Point") | 1 | U |  | DCID 12234 “Respiration State”
5 | > | HAS CONCEPT MOD | CODE | EV (R-4089A, SRT, "Cardiac Cycle Point") | 1 | U |  | DCID 12233 “Cardiac Phase”
6 | > | HAS ACQ CONTEXT | CODE | EV (G-0373, SRT, "Image Mode") | 1 | U |  | DCID 12224 “Ultrasound Image Modes”
7 | > | HAS ACQ CONTEXT | CODE | EV (111031, DCM, "Image View") | 1 | U |  | BCID 12226 “Echocardiography Image View”

Content Item Descriptions

Row 1 | For an index type of measurement, the concept name of this row 1 will still be the original measurement concept name; it is row 2 that gives the indication that row 1 is actually an index type of measurement. When this happens, the measurement value of row 1 should be a value after being indexed and the measurement unit of row 1 should be an index type of unit. For example, to insert a "Stroke Volume Index" measurement to this SR object, the concept name of row 1 will be "Stroke Volume", its numerical value will be the calculation result of "Stroke Volume /BSA" and its units are "ml/cm2".
Row 2 | When this row is available, the row 1 is an index calculation of the object.

### TID 5225 Cardiac Ultrasound Fetal Characteristics

Contains a list of Fetus Specific characteristics.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5225. Cardiac Ultrasound Fetal Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125015, DCM, "Fetus Characteristics") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this Template is invoked more than once to describe more than one fetus. |
3 | > | CONTAINS | NUM | EV (18185-9, LN, "Gestational Age") | 1 | U |  | UNITS = DCID 7456 “Units of Measure for Age”
4 | > | CONTAINS | DATE | EV (11778-8, LN, "EDD") | 1 | U |  |
5 | > | CONTAINS | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  |

### TID 5226 Cardiac Ultrasound Summary Section

Comments and observations of the procedure of immediate clinical interest.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5226. Cardiac Ultrasound Summary Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | MC | IF row 3 does not exist | BCID 12248 “Cardiac Ultrasound Summary Codes”
3 | > | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1-n | MC | IF row 2 does not exist |
4 | > | CONTAINS | CODE | EV (P0-009C3, SRT, "Surgical Procedure") | 1-n | U |  | BCID 12247 “Pediatric, Fetal and Congenital Cardiac Surgical Interventions”

### TID 5227 Cardiac Ultrasound Fetal Summary Section

Comments and observations of the procedure of immediate clinical interest.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5227. Cardiac Ultrasound Fetal Summary Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125008, DCM, "Fetus Summary") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this Template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | MC | IF row 4 does not exist | BCID 12249 “Cardiac Ultrasound Fetal Summary Codes”
4 | > | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1-n | MC | IF row 3 does not exist |
5 | > | CONTAINS | CODE | EV (P0-009C3, SRT, "Surgical Procedure") | 1-n | U |  | BCID 12247 “Pediatric, Fetal and Congenital Cardiac Surgical Interventions”

### TID 5228 Cardiac Ultrasound Fetal Measurement Section

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5228. Cardiac Ultrasound Fetal Measurement Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125016, DCM, "Fetal Measurements") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this Template is invoked more than once to describe more than one fetus. |
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 12279 “Cardiac Ultrasound Fetal General Measurements”
4 | > | CONTAINS | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = EV (T-F6845, SRT, "Ductus arteriosus") $MeasType = DCID 12218 “Echocardiography Congenital”
5 | > | CONTAINS | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = EV (T-F680F, SRT, "Ductus venosus") $MeasType = DCID 12218 “Echocardiography Congenital”
6 | > | CONTAINS | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = EV (T-F1810, SRT, "Umbilical artery") $MeasType = DCID 12218 “Echocardiography Congenital”
7 | > | CONTAINS | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = EV (T-D03B4, SRT, "Umbilical vein") $MeasType = DCID 12218 “Echocardiography Congenital”
8 | > | CONTAINS | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = EV (T-45600, SRT, "Middle cerebral artery") $MeasType = DCID 12218 “Echocardiography Congenital”

## Simplified Adult Echocardiography Templates

The templates that comprise the Simplified Adult Echocardiography Report are interconnected as in Figure A-10b.

Figure A-10b. Simplified Adult Echocardiography Template Structure

### TID 5300 Simplified Echo Procedure Report

This template forms the top of a content tree that allows an ultrasound device to describe the results of an adult echocardiography imaging procedure.

The template is instantiated at the root node. It can also be included in other templates that need to incorporate echocardiography findings into another report as quoted evidence.

This template does not include an Image Library. Image Content Items in the Echo Measurement templates (for example to indicate Source of Measurement) shall be included with by-value relationships, not with by-reference relationships.

Measurements in this template (except for the Wall Motion Analysis) are collected into one of three containers, each with a specific sub-template and constraints appropriate to the purpose of the container.

- Pre-coordinated Measurements

  - Are fully standardized measurements (many taken from the ASE practice guidelines).

  - Each has a single pre-coordinated standard code that fully captures the semantics of the measurement.

  - The only modifiers permitted are to indicate coordinates where the measurement was taken, provide a brief display label, and indicate which of a set of repeated measurements is the preferred value. Other modifiers are not permitted.

- Post-coordinated Measurements

  - Are non-standardized measurements that are performed with enough regularity to merit the control and configuration to capture the full semantics of the measurement. For example these measurements may include those configured on the cart by the vendor or user site. Some of these may be variants of the Pre-coordinated Measurements.

  - A set of mandatory and conditional modifiers with controlled vocabularies capture the essential semantics in a uniform way.

  - A single pre-coordinated code is also provided so that when the same type of measurement is encountered in the future, it is not necessary to parse and evaluate the full constellation of modifer values. Since this measurement has not been fully standardized, the pre-coordinated code may use a private coding scheme (e.g., from the vendor or user site).

- Adhoc Measurements

  - Are non-standardized measurements that do not merit the effort to track or configure all the details necessary to populate the set of modifiers required for a post-coordinated measurement.

  - The measurement code describes the elementary property measured.

  - Modifiers provide a brief display label and indicate coordinates where the measurement was taken. Other modifiers are not permitted.

For an example of this encoding and a discussion of the benefits and use cases, see Annex CCCC Populating The Simplified Echo Procedure Report Template (Informative) in PS3.17.

Type:

  Non-Extensible

Order:

  Significant

Root:

  Yes

Table TID 5300. Simplified Echo Procedure Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125200, DCM, "Adult Echocardiography Procedure Report") | 1 | M |  |
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
4 | > | CONTAINS | CONTAINER | DT (55111-9, LN, "Current Procedure Descriptions") | 1 | U |  |
5 | >> | CONTAINS | CODE | DT (125203, DCM, "Acquisition Protocol") | 1-n | M |  | BCID 12001 “Ultrasound Protocol Types”
6 | > | CONTAINS | CONTAINER | EV (18785-6, LN, "Indications for Procedure") | 1 | U |  |
7 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 12246 “Cardiac Ultrasound Indication for Study”
8 | >> | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1 | U |  |
9 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | U |  |
10 | > | CONTAINS | CONTAINER | EV (125301, DCM, "Pre-coordinated Measurements") | 1 | M |  |
11 | >> | CONTAINS | INCLUDE | DTID 5301 “Pre-coordinated Echo Measurement” | 1-n | M |  | $Measurement = DCID 12300 “Core Echo Measurements” $Preferred = DCID 12301 “Measurement Selection Reasons”
12 | > | CONTAINS | CONTAINER | EV (125302, DCM, "Post-coordinated Measurements") | 1 | M |  |
13 | >> | CONTAINS | INCLUDE | DTID 5302 “Post-coordinated Echo Measurement” | 1-n | U |  | $Preferred = DCID 12301 “Measurement Selection Reasons”
14 | > | CONTAINS | CONTAINER | EV (125303, DCM, "Adhoc Measurements") | 1 | M |  |
15 | >> | CONTAINS | INCLUDE | DTID 5303 “Adhoc Measurement” | 1-n | U |  | $Property =DCID 12304 “Echo Measured Properties”
16 | > | CONTAINS | INCLUDE | DTID 5204 “Wall Motion Analysis” | 1-n | U |  | $Procedure = DT (P5-B3121, SRT, "Echocardiography for Determining Ventricular Contraction")
17 | > | CONTAINS | CONTAINER | EV (125310, DCM, "Staged Measurements") | 1 | U |  |
18 | >> | HAS ACQ CONTEXT | CODE | EV (18139-6, LN, "Stage") | 1 | M |  | BCID 3207 “Stress Test Procedure Phases”
19 | >> | CONTAINS | CONTAINER | EV (125301, DCM, "Pre-coordinated Measurements") | 1 | M |  |
20 | >>> | CONTAINS | INCLUDE | DTID 5301 “Pre-coordinated Echo Measurement” | 1-n | U |  | $Measurement = DCID 12300 “Core Echo Measurements” $Preferred = DCID 12301 “Measurement Selection Reasons”
21 | >> | CONTAINS | CONTAINER | EV (125302, DCM, "Post-coordinated Measurements") | 1 | M |  |
22 | >>> | CONTAINS | INCLUDE | DTID 5302 “Post-coordinated Echo Measurement” | 1-n | U |  | $Preferred = DCID 12301 “Measurement Selection Reasons”
23 | >> | CONTAINS | CONTAINER | EV (125303, DCM, "Adhoc Measurements") | 1 | M |  |
24 | >>> | CONTAINS | INCLUDE | DTID 5303 “Adhoc Measurement” | 1-n | U |  | $Property =DCID 12304 “Echo Measured Properties”

Content Item Descriptions

Row 8 | A text string containing one or more sentences describing one or more indications, possibly with additional comments from the physician or tech.
Row 11 | These are measurements from a standardized list of pre-coordinated codes. See CID 12300 “Core Echo Measurements”. Measurements which do not correspond to the full semantics of one of the pre-coordinated codes in CID 12300 can likely be encoded in Row 13 instead. Multiple instances of the same measurement code may be present in the container. Each instance represents a different sample or derivation. This template makes no requirement that any or all samples be sent. For example, a mean value of all the samples of a given measurement could be sent without sending all or any of the samples from which the mean was calculated. Device configuration and/or operator interactions determine what measurements are sent.
Row 13 | These are measurements that can be encoded using a standardized structure of post-coordinated codes. Measurements which correspond to the full semantics of one of the pre-coordinated codes in CID 12300 “Core Echo Measurements” should be encoded in Row 11 instead. $Measurement shall be provided, but is not constrained to a CID. Multiple instances of the same measurement code may be present in the container. Each instance represents a different sample or derivation. This template makes no requirement that any or all samples be sent. For example, a mean value of all the samples of a given measurement could be sent without sending all or any of the samples from which the mean was calculated. Device configuration and/or operator interactions determine what measurements are sent.
Row 15 | These are adhoc measurements encoded with minimal semantics. Row 13 can be used to encode measurements with more complete semantics. $Units shall be provided, but is not constrained to a CID. Device configuration and/or operator interactions determine what measurements are sent.
Rows 17-24 | When present, these rows contain measurements and associate them with a specific stage of a staged procedure.

### TID 5301 Pre-coordinated Echo Measurement

This template codes numeric echo measurements where most of the details about the nature of the measurement have been pre-coordinated in the measurement code. In contrast, see TID 5302 “Post-coordinated Echo Measurement”.

The pre-coordinated measurement code is provided when this Template is included from a parent Template.

Table TID 5301. Parameters

Parameter Name | Parameter Usage
--- | ---
$Measurement | Coded term or Context Group for Concept Name of measurement
$Preferred | Flag the preferred value by indicating the reason it was selected as preferred.

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 5301. Pre-coordinated Echo Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | $Measurement | 1 | M |  |
2 | > | HAS PROPERTIES | CODE | EV (121404, DCM, "Selection Status") | 1 | MC | IFF this measurement has been selected as the single preferred value for the measured concept. | $Preferred = MemberOf {DCID 12301 “Measurement Selection Reasons”}
3 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | MC | IFF this measurement is not a sample. | EV R-00317, SRT, "Mean")
4 | > |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1-n | U |  | $Purpose = EV (121112, DCM, "Source of measurement")
5 | > |  | INCLUDE | DTID 321 “Waveform or Temporal Coordinates” | 1-n | U |  | $Purpose = EV (121112, DCM, "Source of measurement")
6 | > | HAS PROPERTIES | TEXT | EV (125309, DCM, "Short Label") | 1 | U |  |

Content Item Descriptions

Row 2 | The reason that this value was selected as the preferred value for the measured concept. The parent template may allow TID 5301 “Pre-coordinated Echo Measurement” to be included multiple times with the same Measurement Concept Name, for example to allow multiple samples of the measurement. A given Measurement Concept Name might appear only once in the instance, in which case this this row may or may not be present. A given Measurement Concept Name may appear multiple times, however this row shall not be present for more than one value of the given Measurement Concept Name. E.g. multiple measurements of (11706-9, LN, "Aortic Valve Peak Systolic Flow") may be present, but only one may be selected as preferred.
Row 3 | The method used to derive this measurement value from multiple samples of the Measurement Concept Name. If Row 3 is not present, then this measurement value is simply a single sample of the Measurement Concept Name. ### Note A measurement value that is a mean value of other measurements and was also selected as the preferred value because it is the mean will have both Row 2 and Row 3 present.
Row 6 | This may be used to label the measurement value when space is limited on the screen or report page. E.g. a Short Label of "LVIDD" might be provided for a measurement of the left ventricle internal diameter at end diastole. ### Note Short Labels are not standardized and may omit details of the measurement, thus it is not recommended to use them for purposes such as matching.

### TID 5302 Post-coordinated Echo Measurement

This template codes numeric echo measurements where most of the details about the nature of the measurement have been post-coordinated in modifiers and acquisition context. In contrast, see TID 5301 “Pre-coordinated Echo Measurement” .

This template is intended to be used for User-defined and Vendor-defined Echo Measurements.

Several modifier rows are conditional and are omitted when the modifier concept is not significant for the measurement encoded in the item. When these modifiers are included by the sender, it indicates that the modifier concept is significant and receivers will generally treat the measurements differently than similar measurements sent that omit that modifier.

### Note

The codes in the CIDs referenced below were sufficient to accurately encode all the best practice echo measurements recommended by the ASE. If, however, a new code is needed to record a specific User-defined or Vendor-defined measurement, most of the CIDs are extensible. It is not unreasonable to expect that measurements might be made at other Finding Sites than those listed in CID 12305 “Basic Echo Anatomic Sites” , or using Measurement Methods beyond those listed in CID 12227 “Echocardiography Measurement Method” .

The concept modifiers in the template below were sufficient to accurately encode all the best practice echo measurements recommended by the ASE. Although TID 5302 “Post-coordinated Echo Measurement” is extensible and adding new modifiers is not prohibited, the meaning and significance of such new modifiers will generally not be understood by receiving systems, delaying or preventing import of such measurements. Further, adding modifiers that replicate the meaning of an existing modifier is prohibited.

If such measurements cannot be encoded with the following structure, an implementation may choose to code the measurement in TID 5303 “Adhoc Measurement” , or to use TID 5200 “Echocardiography Procedure Report” instead of TID 5300 “Simplified Echo Procedure Report” .

Table TID 5302. Parameters

Parameter Name | Parameter Usage
--- | ---
$Measurement | Coded term or Context Group for Concept Name of measurement
$Preferred | Flag the preferred value by indicating the reason it was selected as preferred.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 5302. Post-coordinated Echo Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | $Measurement | 1 | M |  |
2 | > | HAS PROPERTIES | CODE | EV (121050, DCM, "Equivalent Meaning of Concept Name") | 1-n | U |  |
3 | > | HAS PROPERTIES | CODE | EV (121404, DCM, "Selection Status") | 1 | MC | IFF this measurement has been selected as the single preferred value for the measured concept. | $Preferred =MemberOf {DCID 12301 “Measurement Selection Reasons”}
4 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | MC | IFF this measurement is not a sample. | EV R-00317, SRT, "Mean")
5 | > |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1-n | U |  | $Purpose =EV (121112, DCM, "Source of measurement")
6 | > |  | INCLUDE | DTID 321 “Waveform or Temporal Coordinates” | 1-n | U |  | $Purpose = EV (121112, DCM, "Source of measurement")
7 | > | HAS CONCEPT MOD | CODE | EV (125306, DCM, "Measurement Type") | 1 | M |  | DCID 12303 “Echo Measurement Types”
8 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 12305 “Basic Echo Anatomic Sites”
9 | > | HAS CONCEPT MOD | CODE | EV (125305, DCM, "Finding Observation Type") | 1 | M |  | DCID 12302 “Echo Finding Observation Types”
10 | > | HAS CONCEPT MOD | CODE | EV (125307, DCM, "Measured Property") | 1 | M |  | DCID 12304 “Echo Measured Properties”
11 | > | HAS CONCEPT MOD | CODE | EV (G-C048, SRT, "Flow Direction") | 1 | MC | IFF Row 9 is (PA-50030, SRT, "Hemodynamic Measurements") and the Flow Direction is significant for this measurement. | DCID 12306 “Echo Flow Directions”
12 | > | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | MC | IFF the Measurement Method is significant for this measurement. | DCID 12227 “Echocardiography Measurement Method”
13 | > | HAS ACQ CONTEXT | CODE | EV (G-0373, SRT, "Image Mode") | 1 | MC | IFF the Image Mode is significant for this measurement. | DCID 12224 “Ultrasound Image Modes”
14 | > | HAS ACQ CONTEXT | CODE | EV (111031, DCM, "Image View") | 1 | MC | IFF the Image View is significant for this measurement. | DCID 12226 “Echocardiography Image View”
15 | > | HAS CONCEPT MOD | CODE | EV (R-4089A, SRT, "Cardiac Cycle Point") | 1 | MC | IFF the Cardiac Cycle Point is significant for this measurement. | DCID 12307 “Cardiac Phases and Time Points”
16 | > | HAS CONCEPT MOD | CODE | EV (R-40899, SRT, "Respiratory Cycle Point") | 1 | MC | IFF the Respiratory Cycle Point is significant for this measurement. | DCID 12234 “Respiration State”
17 | > | HAS CONCEPT MOD | CODE | EV (125308, DCM, "Measurement Divisor") | 1 | MC | IFF the value of Row 7 is (125313, DCM, "Indexed") or (G-D750, SRT, "Ratio") or (125314, DCM, "Fractional Change") |
18 | > | HAS PROPERTIES | TEXT | EV (125309, DCM, "Short Label") | 1 | U |  |

Content Item Descriptions

Row 1 | A fully pre-coordinated code that incorporates all the semantics of Rows 7-17 for this measurement. The code is intended to allow parsers to recognize post-coordinated measurements that have been previously encountered, thus facilitating incorporation of the measurement into databases, report templates, registries, etc. Typically these codes will be from a vendor or site specific coding scheme, e.g., 99ACME. Sending the same code consistently in different reports will depend on the recording system maintaining a stable list of these pre-coordinated codes. Such a list might be configured or internally generated and managed. This shall be populated by the recording system. If the recording system does not have a method to ensure that all occurrences of the same post-coordinated measurement use the same code, it shall use the code (125304, DCM, "Untrackable Measurement"). ### Note Two measurements with the same pre-coordinated code have, by definition, the same semantics (except for "Untrackable Measurements") Two measurements with the same constellation of modifier values have the same semantics but may have different pre-coordinated codes because they come from carts of different vendors who don't share the same code table come from carts of the same vendor, but the carts don't share the same code table come from the same cart, but it's code table has been modified come from the same cart, but it does not maintain a code table Two measurements with the same constellation of modifier values and different pre-coordinated codes have the same semantics and the receiver is entitled to treat them as the same (with respect to the scope of those modifiers) Recommended units for various Measured Properties (Row 10) can be found in the Units column of CID 12304 “Echo Measured Properties” . When the Measurement Type (Row 7) is (125313, DCM, "Indexed") , (G-D750, SRT, "Ratio") or (125314, DCM, "Fractional Change") , the Units for Row 1 corresponds to the fully calculated $Measurement, incorporating both the numerator (Row 10) and the denominator (Row 17). E.g. a measure of Left Ventricular Outflow Tract Diameter / BSA would have units of (cm/m2, UCUM, "cm/m2") in Row 1, (125313, DCM, "Indexed") in Row 7, (M-02550, SRT, "Diameter") in Row 10, and (8277-6, LN, "Body Surface Area") in Row 17.
Row 2 | One or more additional fully pre-coordinated codes which are semantically equivalent to the code in Row 1. This may be used to communicate known mappings, such as to national registry codes or other vendors' codes.
Row 3 | The reason that this value was selected as the preferred value for the measured concept. The parent template may allow TID 5301 “Pre-coordinated Echo Measurement” to be included multiple times with the same Measurement Concept Name, for example to allow multiple samples of the measurement. A given Measurement Concept Name might appear only once in the instance, in which case this this row may or may not be present. A given Measurement Concept Name may appear multiple times, however this row shall not be present for more than one value of a given measured concept. E.g. multiple measurements of (11706-9, LN, "Aortic Valve Peak Systolic Flow") may be present, but only one may be selected as preferred.
Row 4 | The method used to derive this measurement value from multiple samples of the Measurement Concept Name. If Row 3 is not present, then this measurement value is simply a single sample of the Measurement Concept Name. ### Note A measurement value that is a mean value of other samples and was also selected as the preferred value because it is the mean will have both Row 2 and Row 3 present. This row is not used to record whether the measurement value is a direct measurement vs a measurement calculated from an equation. Such information is recorded in Row 7.
Row 8 | The finding site reflects the anatomical location where the measurement is taken. CID 12305 “Basic Echo Anatomic Sites” contains the codes which proved to be sufficient for mapping the full set of ASE standard measurements. It is recommended to use these locations unless a more detailed location is truly necessary.
Row 9 | The finding observation type indicates the type of observation made at the finding site to produce the measurement. In many cases, for example Aortic Root Diameter, the structure of the finding site is being observed. In other cases, for example Mitral Valve Regurgitant Flow Peak Velocity, the finding site is the mitral valve, the hemodynamic flow (not the valve structure) is being observed, the measured property is the peak velocity, and the flow direction is retrograde.
Row 17 | The pre-coordinated code for the measurement that has been used as the denominator of this measurement. Only applies to measurements of type Indexed, Ratio or Fractional Change. The measurement referenced as the Measurement Divisor shall be present in the instance in which it is used. When Row 17 is present, any values in Rows 5-6, 8-16 shall reflect the numerator of the measurement rather than the Index, Ratio or Fractional Change as a whole. The rest of the rows, including the pre-coordinated measurement value, the pre-coordinated measurement code, the units and the short label, reflect the Index, Ratio or Fractional Change as a whole. E.g. in the case of an Indexed measurement, the value recorded in Row 1 has already been divided by the Index referenced in Row 17, and the Units in Row 1 match the indexed value, not the numerator Property described in Row 10. For a measurement of type Indexed, the numerator is divided by the Measurement Divisor. For a measurement of type Ratio, the numerator is divided by the Measurement Divisor and is unitless. For a measurement of type Fractional Change, the numerator is first subtracted from the Measurement Divisor and the result divided by the Measurement Divisor (i.e., (Divisor - Numerator) / Divisor ).
Row 18 | This may be used to label the measurement value when space is limited on the screen or report page. E.g. a Short Label of "LVIDD" might be provided for a measurement of the left ventricle internal diameter at end diastole. ### Note Short Labels are not standardized and may omit details of the measurement, thus it is not recommended to use them for purposes such as matching.

### TID 5303 Adhoc Measurement

This Template codes numeric echo measurements where most of the details about the nature of the measurement are not communicated. The measurement is identified in terms of the property measured, such as Length, Diameter, Area, Velocity etc. and some measurement context may established by reference to spatial coordinates on an image or a waveform. A displayable label is included but there is no managed code identifying the measurement.

The template is intended to be used to include adhoc, one-time measurements whose need is determined during imaging exam or reviewing session.

Measurements that are taken in an adhoc fashion but are selected from the set of pre-coordinated or post-coordinated measurements that are configured on the Ultrasound System should be coded using TID 5301 “Pre-coordinated Echo Measurement” or TID 5302 “Post-coordinated Echo Measurement”.

Table TID 5303. Parameters

Parameter Name | Parameter Usage
--- | ---
$Property | Property being measured

Type:

  Non-Extensible

Order:

  Significant

Root:

  No

Table TID 5303. Adhoc Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | $Property | 1 | M |  |
2 | > |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1-n | U |  | $Purpose = EV (121112, DCM, "Source of measurement")
3 | > |  | INCLUDE | DTID 321 “Waveform or Temporal Coordinates” | 1-n | U |  | $Purpose = EV (121112, DCM, "Source of measurement")
4 | > | HAS PROPERTIES | TEXT | EV (125309, DCM, "Short Label") | 1 | M |  |

Content Item Descriptions

Row 4 | This may be used to label the measurement value when space is limited on the screen or report page. E.g. a Short Label of "LVIDD" might be provided for a measurement of the left ventricle internal diameter at end diastole. ### Note Short Labels are not standardized and may omit details of the measurement, thus it is not recommended to use them for purposes such as matching.

## Implantation Plan SR Document Templates

The Templates that comprise the Implantation Plan SR Document IOD are interconnected as in Figure A-11.

Figure A-11. Implantation Plan SR Document IOD Template Structure

### TID 7000 Implantation Plan

This Template contains all the necessary information to position an Implant Assembly and its Components in a patient. Therefore, all the Components that comprise an Implant Assembly are listed. If the Implant Assembly consists of more than one Component, the relation between the Components will be described as well. It is also possible to describe the registration between the Components and the patient and between the Components themselves.

To reference the Components within this document the Implantation Plan Component ID is used.

The Component Connection links two Implantation Plan Components in a commutative way. This means that for each link between A and B only one Component Connection has to be defined and not two for A-B and B-A.

The terminology used is defined by illustration using the example in Figure A-12.

Figure A-12. Implant Assembly and Components Terminology

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 7000. Implantation Plan

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (112345, DCM, "Implantation Plan") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 351 “Previous Reports” | 1 | MC | IFF previous Implantation Plan Documents exist | Shall only reference other Implantation Plan Documents
5 | > | CONTAINS | INCLUDE | DTID 7001 “Related Implantation Reports” | 1 | MC | IFF related Implantation Plan Documents exist that are not referenced by row 4 | Shall only reference other Implantation Plan Documents
6 | > | CONTAINS | CONTAINER | EV (112360, DCM, "Implant Component List") | 1 | M |  |
7 | >> | CONTAINS | COMPOSITE | EV (112366, DCM, "Implant Assembly Template") | 1 | U |  | References an Implant Assembly Template SOP Instance
8 | >> | CONTAINS | CONTAINER | EV (112346, DCM, "Selected Implant Component") | 1-n | M |  |
9 | >>> | CONTAINS | TEXT | EV (112347, DCM, "Component ID") | 1 | M |  |
10 | >>> | CONTAINS | CODE | EV (112370, DCM, "Component Type") | 1 | MC | IFF Row 8 contains more than one item. | DCID 7306 “Human Hip Implant Planning Landmarks”
11 | >>> | CONTAINS | COMPOSITE | No purpose of reference | 1 | M |  | References an Implant Template Storage SOP Instance
12 | >>> | CONTAINS | UIDREF | EV (112227, DCM, "Frame Of Reference UID") | 1 | M |  |
13 | >>> | CONTAINS | COMPOSITE | EV (112371, DCM, "Manufacturer Implant Template") | 1 | M |  | References an Implant Template Storage SOP Instance
14 | > | CONTAINS | CONTAINER | EV (112355, DCM, "Assembly") | 1-n | U |  |
15 | >> | CONTAINS | CONTAINER | EV (112350, DCM, "Component Connection") | 1-n | M |  |
16 | >>> | CONTAINS | CONTAINER | EV (112374, DCM, "Connected Implantation Plan Component") | 2 | M |  |
17 | >>>> | CONTAINS | TEXT | EV (112347, DCM, "Component ID") | 1 | M |  | Defined in the Implant Component List CONTAINER
18 | >>>> | CONTAINS | TEXT | EV (112351, DCM, "Mating Feature Set ID") | 1 | M |  | Only one Component Connection per Mating Feature Set is allowed
19 | >>>> | CONTAINS | TEXT | EV (112352, DCM, "Mating Feature ID") | 1 | M |  |
20 | >>>> | CONTAINS | CONTAINER | EV (112362, DCM, "Degrees of Freedom Specification") | 1-n | U |  |
21 | >>>>> | CONTAINS | TEXT | EV (112363, DCM, "Degree of Freedom ID") | 1 | M |  |
22 | >>>>> | CONTAINS | NUM | EV (112376, DCM, "Degree of Freedom Exact Translational Value") | 1 | MC | IFF Row 23, 24, 25, 26 and 27 are absent | UNITS = EV (mm, UCUM, "mm")
23 | >>>>> | CONTAINS | NUM | EV (112377, DCM, "Degree of Freedom Minimum Translational Value") | 1 | MC | IFF row 22, 25, 26, and 27 are absent | UNITS = EV (mm, UCUM, "mm")
24 | >>>>> | CONTAINS | NUM | EV (112378, DCM, "Degree of Freedom Maximum Translational Value") | 1 | MC | IFF row 22, 25, 26, and 27 are absent | UNITS = EV (mm, UCUM, "mm")
25 | >>>>> | CONTAINS | NUM | EV (112379, DCM, "Degree of Freedom Exact Rotational Value") | 1 | MC | IFF row 22, 23, 24, 26 and 27 are absent | UNITS = EV (deg, UCUM, "degree")
26 | >>>>> | CONTAINS | NUM | EV (112380, DCM, "Degree of Freedom Minimum Rotational Value") | 1 | MC | IFF row 22, 23, 24 and 25 are absent | UNITS = EV (deg, UCUM, "degree")
27 | >>>>> | CONTAINS | NUM | EV (112381, DCM, "Degree of Freedom Maximum Rotational Value") | 1 | MC | IFF row 22, 23, 24 and 25 are absent | UNITS = EV (deg, UCUM, "degree")
28 | > | CONTAINS | CONTAINER | EV (112358, DCM, "Information used for planning") | 1 | U |  |
29 | >> | CONTAINS | CODE | EV (112375, DCM, "Planning Method") | 1 | U |  | BCID 7320 “Planning Methods”
30 | >> | CONTAINS | IMAGE | EV (112354, DCM, "Patient Image") | 1-n | U |  |
31 | >>> | HAS PROPERTIES | NUM | EV (111026, DCM, "Horizontal Pixel Spacing") | 1 | M |  | UNITS = EV (mm/{pixel}, UCUM, "mm/pixel")
32 | >>> | HAS PROPERTIES | NUM | EV (111066, DCM, "Vertical Pixel Spacing") | 1 | M |  | UNITS = EV (mm/{pixel}, UCUM, "mm/pixel")
33 | >> | CONTAINS | COMPOSITE | EV (112361, DCM, "Patient Data Used During Planning") | 1-n | U |  | References SOP Instances except Images
34 | >>> | HAS PROPERTIES | UIDREF | EV (112356, DCM, "User Selected Fiducial") | 1-n | MC | IFF row 33 references a Fiducial SOP Instance |
35 | >>>> | HAS CONCEPT MOD | TEXT | EV (112369, DCM, "Fiducial Intent") | 1 | U |  |
36 | > | CONTAINS | CONTAINER | EV (112367, DCM, "Planning Information for Intraoperative Usage") | 1 | U |  |
37 | >> | CONTAINS | TEXT | EV (121173, DCM, "Physician Note") | 1-n | U |  |
38 | >> | CONTAINS | COMPOSITE | EV (112359, DCM, "Supporting Information") | 1 | U |  | SOP Class UID shall be Encapsulated PDF Storage
39 | >> | CONTAINS | COMPOSITE | EV (112372, DCM, "Derived Planning Images") | 1-n | U |  |
40 | >> | CONTAINS | COMPOSITE | EV (112353, DCM, "Spatial Registration") | 1-n | U |  | References Spatial Registration SOP Instances and Deformable Spatial Registration SOP Instances
41 | >>> | HAS PROPERTIES | UIDREF | EV (112227, DCM, "Frame of Reference UID") | 1-n | U |  |
42 | >> | CONTAINS | COMPOSITE | EV (112373, DCM, "Derived Planning Data") | 1-n | U |  | References SOP Instances except Images and Spatial Registrations
43 | >>> | HAS PROPERTIES | UIDREF | EV (112357, DCM, "Derived Fiducial") | 1-n | MC | IFF row 42 references a Fiducial SOP Instance |
44 | >>>> | HAS CONCEPT MOD | TEXT | EV (112369, DCM, "Fiducial Intent") | 1 | U |  |
45 | >> | CONTAINS | COMPOSITE | EV (112364, DCM, "Related Patient Data Not Used During Planning") | 1-n | U |  |

Content Item Descriptions

Row 7 | If an Implant Assembly Template was used for the planning, it should be referenced here.
Row 9 | ID given to this Implant Component. Used to reference this specific Component within the Implantation Plan.
Row 10 | See description of Component Type Code Sequence (0076,0034) Attribute in Section C.29.2.1 “Implant Assembly Template Module” in PS3.3
Row 11 | Reference to the Template that describes that component. May be the same Implant Template as referenced in row 13. The target of the reference may not be needed or available during implantation. e.g., if the plan is opened in another hospital where those implant templates are not used.
Row 12 | This Frame of Reference is the Frame of Reference of the Implant Component (Frame of Reference UID (0020,0052) Attribute in the “Generic Implant Template Description Module” in PS3.3 ). This may help to find the right registration information (row 43).
Row 13 | References the Original Template that was the basis for the Derived Template. May be the same Implant Template as referenced in row 11. The target of the reference may not be needed or available during implantation, e.g., if the plan is opened in another hospital where those implant templates are not used.
Row 14 | If there is no Component Connection between sets of Implant Components, one Assembly must be used for each set.
Row 17 | The ID of a planned Component that is defined in this document and that is part of this Relation.
Row 18 | See description of Mating Feature Set ID (0068,63C0) Attribute in the “Generic Implant Template Mating Features Module” in PS3.3
Row 19 | See description of Mating Feature ID (0068,63F0) Attribute in the “Generic Implant Template Mating Features Module” in PS3.3
Row 21 | See description of Degree of Freedom ID (0068,6410) Attribute in the “Generic Implant Template Mating Features Module” in PS3.3
Row 22 - 27 | Defines the range or exact value that was selected or calculated by the planning application.
Row 31 | Defines the calibrated Horizontal Pixel Spacing that was used by the planning application, which may be different from the spacing encoded in the referenced Image SOP Instance.
Row 32 | Defines the calibrated Vertical Pixel Spacing that was used by the planning application, which may be different from the spacing encoded in the referenced Image SOP Instance.
Row 33 | Any patient data other than Image IEs used for the planning, e.g., Surface Segmentations.
Row 34 | Fiducials selected by the user for registration of implant components referenced in the parent Content Item.
Row 35 | User comment about the Fiducial. This may be the reason it was selected, the intended use, the anatomical or non-anatomical structure that the Fiducial represents, or any other intent.
Row 38 | All kinds of information in PDF form that are created by a planning application may be referenced here, e.g., drawings.
Row 39 | All kinds of images that are created by a planning application should be referenced here, e.g., images that show patient images overlaid with contour information of the Implant Component, or images that show how several implant components may be composed, or merged patient images.
Row 40 | References registration objects that contain registration data that is relevant for this Implantation Plan, e.g., registration of Implant Components.
Rows 5, 6 | Identifies one or more items within the sequence of referenced Frames of Reference (Registration Sequence (0070,0308) in the Spatial Registration Module or Deformable Registration Sequence (0064,0002) in the “Deformable Spatial Registration Module” in PS3.3 ) that are relevant for this Implantation Plan. See Figure A-13.
Row 42 | Any patient data created during the planning process that is not referenced in row 39 and 40, e.g., Surface Segmentation Instances created by the planning application.
Row 43 | These Fiducials are derived from the Fiducials identified in Row 34.
Row 45 | References to any relevant patient data containing IOD instances that were not used in planning or derived from it but belong to the patient model. Might be reports, images, surface segmentations, or other.

Figure A-13. References to Registration Objects

### TID 7001 Related Implantation Reports

This general Template provides a means to reference related Implantation Plan SR Document instances that are not previous Reports. Other Implantations that are planned to be done during the same intervention should be referenced here.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 7001. Related Implantation Reports

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (112365, DCM, "Related Implantation Reports") | 1 | M |  |
2 | > | CONTAINS | COMPOSITE |  | 1-n | M |  |

## Acquisition Context SR IOD Templates

The Templates that comprise the Acquisition Context SR are interconnected as follows:

- TID 8101 “Preclinical Small Animal Image Acquisition Context”

  - TID 1204 "Language of Content Item and Descendants"

  - TID 1001 "Observation Context"

  - TID 8110 “Biosafety Conditions”

  - TID 8121 “Animal Housing”

  - TID 8122 “Animal Feeding”

  - TID 8140 “Heating Conditions”

  - TID 8150 “Circadian Effects”

  - TID 8170 “Physiological Monitoring Performed During Procedure”

  - TID 8130 “Anesthesia”

    - TID 8131 “Medications and Mixture Medications”

  - TID 9002 “Medication, Substance, Environmental Exposure”

  - TID 8182 “Exogenous Substance Administration”

### TID 8101 Preclinical Small Animal Image Acquisition Context

This root template encodes a description of the conditions present during and related to data acquisition for a single imaging procedure.

### Note

1. It is not expected that a single instance be used to describe the entire life of an animal, unless it is sacrificed after a single procedure. Rather, separate instances will be used for separate procedures, though there may be some duplication of common information, such as about the home cage environment.

2. It is expected that an SR instance encoded using this template will be contained in the same Study as other instances created during the procedure, e.g., with a common Study Instance UID. If this is not practical, e.g., due to recording on a separate device without use of a shared Modality Worklist, then commonality of other Study level attributes may be necessary to link procedures (and possibly coerce the Study Instance UID to a common value).

Type:

  Extensible

Order:

  Non-Significant

Root:

  Yes

Table TID 8101. Preclinical Small Animal Image Acquisition Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (127001, DCM, "Preclinical Small Animal Imaging Acquisition Context") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 "Language of Content Item and Descendants" | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 "Observation Context" | 1 | M |  |
5 | > | CONTAINS | INCLUDE | DTID 8110 “Biosafety Conditions” | 1 | U |  |
6 | > | CONTAINS | CONTAINER | EV (127005, DCM, "Animal handling during specified phase") | 1-n | U |  |
7 | >> | HAS CONCEPT MOD | CODE | EV (127006, DCM, "Phase of animal handling") | 1 | M |  | DCID 634 “Phase of Animal Handling”
8 | >> | CONTAINS | DATETIME | EV (111526, DCM, "DateTime Started") | 1 | U |  |
9 | >> | CONTAINS | DATETIME | EV (111527, DCM, "DateTime Ended") | 1 | U |  |
10 | >> | CONTAINS | INCLUDE | DTID 8121 “Animal Housing” | 1 | U |  |
11 | >> | CONTAINS | INCLUDE | DTID 8122 “Animal Feeding” | 1-n | U |  |
12 | >> | CONTAINS | INCLUDE | DTID 8140 “Heating Conditions” | 1 | U |  |
13 | >> | CONTAINS | INCLUDE | DTID 8150 “Circadian Effects” | 1 | U |  |
14 | >> | CONTAINS | INCLUDE | DTID 8170 “Physiological Monitoring Performed During Procedure” | 1 | U |  |
15 | > | CONTAINS | INCLUDE | DTID 8130 “Anesthesia” | 1 | U |  |
16 | > | CONTAINS | INCLUDE | DTID 9002 “Medication, Substance, Environmental Exposure” | 1 | U |  | $ContainerConcept = EV (10160-0, LN, "History Of Medication Use") $CodeConcept = EV (111516, DCM, "Medication Type") $Route = DCID 11 “Route of Administration”
17 | > | CONTAINS | INCLUDE | DTID 8182 “Exogenous Substance Administration” | 1 | U |  | $ContainerConcept = EV (127400, DCM, "Exogenous substance") $CodeConcept = DCID 637 “Exogenous Substance Types” $CodeValue = DCID 638 “Exogenous Substance” $Route = DCID 11 “Route of Administration” $Site = DCID 644 $TissueOfOrigin = DCID 645 $TaxonomicRankOfOrigin = DCID 7454 “Animal Taxonomic Rank Values”

Content Item Descriptions

Row 3 | A single pre-coordinated code describing the general type of imaging procedure can be described using TID 1005 Row 9 Procedure Code (included in TID 1001). For small animal (as opposed to human) imaging, this will likely describe a whole body acquisition in a modality-specific manner and the use of contrast and/or radiopharmaceutical. E.g., whole body FDG PET, or whole body DCE-MRI. May be redundant with (or default to) the value present in the top level data set in Procedure Code Sequence (0008,1032) of the General Study Module. Species and strain identification is not described in TID 1001; rather it is encoded in DICOM Attributes in the top level data set.
Row 5 | The biosafety conditions are expected to be consistent across all phases of handling, so are not described separately per-phase.
Rows 8-9 | The period of time during which the phase is defined, i.e., during which the animal was managed in the specified conditions. This may be more important for interpretation for some phases (e.g., transport) than others (e.g., at rest in the home cage), and hence is optional.
Rows 10-14 | The outline of subordinate templates follows the pattern of categories of Animal Housing, Care, and Physiologic Monitoring information described in [Stout et al 2013].
Row 11 | Animal feeding is 1-n to allow encoding of dietary supplements and treats in addition to the regular diet.
Row 15 | A single anesthesia event is normally assumed for a single procedure, though the template included can include multiple pre-, intra- and post-procedure descriptions.
Row 16 | Used to describe pharmaceuticals administered that are not described elsewhere, in particular, those that are not described as anesthesia medications, and those that are not described in the images (e.g., contrast, radiopharmaceuticals). This includes therapy (such as chemotherapy, immunotherapy) and similar interventions that may be the subject of the research. The value set of $CodeValue is not defined, given the vast range of possible codes and coding schemes for drugs or medicaments that might be used. Nor are value sets for $Classification or $Site defined.
Row 17 | Used to describe non-pharmaceutical exogenous substances administered, such as cells or other tumor graft, fibrils, viruses, cytokines and toxins that describe the "model" upon which the research is being performed, as distinct from the "therapeutic intervention" (Row 16) that may be the purpose of the research. The Classification parameter is not constrained by any value set.

### TID 8110 Biosafety Conditions

This template encodes a description of the biosafety conditions applicable to research small animals.

Type:

  Extensible

Order:

  Non-Significant

Table TID 8110. Biosafety Conditions

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (127010, DCM, "Biosafety conditions") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (F-0061F, SRT, "Biosafety level") | 1 | U |  | DCID 601 “Biosafety Levels”
2 | > | CONTAINS | CODE | EV (127011, DCM, "Reason for biosafety controls") | 1 | U |  | DCID 602 “Biosafety Control Reasons”
4 | > | CONTAINS | TEXT | EV (121106, DCM,"Comment") | 1 | U |  |

Content Item Descriptions

Row 4 | A brief description of any pertinent or unusual biosafety requirements.

### TID 8121 Animal Housing

This template encodes a description of housing of animals, e.g., in home cages, holders for imaging, etc., over an interval during which environmental and handling conditions are relatively homogenous.

### Note

1. Only "static" parameters of the design and setup are recorded, and "nominal" values for environmental conditions such as humidity and temperature, but not "dynamic" parameters that might vary during one housing interval, and potentially be monitored, such as oxygen or ammonia levels, temperature, humidity, urine or fecal corticosterone levels, etc.

2. Values for product names and codes are expected to be accurate at the time the information is recorded, recognizing that products may evolve over time.

Type:

  Extensible

Order:

  Non-Significant

Table TID 8121. Animal Housing

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (127120, DCM, "Animal housing") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (127121, DCM, "Animal room type") | 1 | U |  | DCID 603 “Animal Room Types”
2b | > | CONTAINS | TEXT | EV (127122, DCM, "Animal room identifier") | 1 | U |  |
3 | > | CONTAINS | TEXT | EV (127125, DCM, "Housing manufacturer") | 1 | U |  |
4 | > | CONTAINS | TEXT | EV (127126, DCM, "Housing rack product name") | 1 | U |  |
5 | > | CONTAINS | TEXT | EV (127127, DCM, "Housing rack product code") | 1 | U |  |
6 | > | CONTAINS | TEXT | EV (127128, DCM, "Housing unit product name") | 1 | U |  |
7 | > | CONTAINS | TEXT | EV (127129, DCM, "Housing unit product code") | 1 | U |  |
8 | > | CONTAINS | TEXT | EV (127130, DCM, "Housing unit lid product name") | 1 | U |  |
9 | > | CONTAINS | TEXT | EV (127131, DCM, "Housing unit lid product code") | 1 | U |  |
10 | > | CONTAINS | NUM | EV (127140, DCM, "Number of racks per room") | 1 | U |  | UNITS = EV ({racks}, UCUM, "racks")
11 | > | CONTAINS | NUM | EV (127141, DCM, "Number of housing units per rack") | 1 | U |  | UNITS = EV ({housing units}, UCUM, "housing units") or EV ({cages}, UCUM, "cages")
12 | > | CONTAINS | TEXT | EV (127142, DCM, "Housing unit location in rack") | 1 | U |  |
13 | > | CONTAINS | NUM | EV (127143, DCM, "Number of animals within same housing unit") | 1 | U |  | UNITS = EV ({animals}, UCUM, "animals")
14 | > | CONTAINS | CODE | EV (127144, DCM, "Sex of animals within same housing unit") | 1 | U |  | DCID 7457 “Sex - Male Female or Both”
15 | > | CONTAINS | CODE | EV (127145, DCM, "Sex of handler") | 1 | U |  | DCID 7457 “Sex - Male Female or Both”
16 | > | CONTAINS | NUM | EV (127150, DCM, "Total duration in housing") | 1 | U |  | UNITS = EV (d, UCUM, "days")
17 | > | CONTAINS | NUM | EV (127151, DCM, "Housing change interval") | 1 | U |  | UNITS = EV (d, UCUM, "days")
18 | > | CONTAINS | NUM | EV (127152, DCM, "Manual handling interval") | 1 | U |  | UNITS = EV (h, UCUM, "hours")
19 | > | CONTAINS | TEXT | EV (127153, DCM, "Housing unit movement") | 1 | U |  |
20 | > | CONTAINS | NUM | EV (127160, DCM, "Housing unit width") | 1 | U |  | UNITS = EV (cm, UCUM, "cm")
21 | > | CONTAINS | NUM | EV (127161, DCM, "Housing unit height") | 1 | U |  | UNITS = EV (cm, UCUM, "cm")
22 | > | CONTAINS | NUM | EV (127162, DCM, "Housing unit length") | 1 | U |  | UNITS = EV (cm, UCUM, "cm")
23 | > | CONTAINS | CODE | EV (127170, DCM, "Housing individually ventilated") | 1 | U |  | DCID 231 “Yes-No Only”
24 | > | CONTAINS | NUM | EV (127172, DCM, "Air changes") | 1 | U |  | UNITS = EV (/h, UCUM, "/hour")
25 | > | CONTAINS | NUM | EV (C90380, NCIt, "Environmental temperature") | 1 | U |  | UNITS = EV (Cel, UCUM, "C")
26 | > | CONTAINS | NUM | EV (C90395, NCIt, "Housing humidity") | 1 | U |  | UNITS = EV (%, UCUM, "%")
27 | > | CONTAINS | CODE | EV (127175, DCM, "Housing unit reuse") | 1 | U |  | DCID 604 “Device Reuse”
28 | > | CONTAINS | CODE | EV (C90366, NCIt, "Bedding material") | 1 | U |  | DCID 605 “Animal Bedding Material”
29 | > | CONTAINS | TEXT | EV (C90366, NCIt, "Bedding material") | 1 | U |  |
30 | > | CONTAINS | TEXT | EV (127180, DCM, "Bedding manufacturer") | 1 | U |  |
31 | > | CONTAINS | TEXT | EV (127181, DCM, "Bedding product name") | 1 | U |  |
32 | > | CONTAINS | TEXT | EV (127182, DCM, "Bedding product code") | 1 | U |  |
33 | > | CONTAINS | NUM | EV (127183, DCM, "Bedding volume") | 1 | U |  | UNITS = EV (ml, UCUM, "ml")
34 | > | CONTAINS | NUM | EV (127184, DCM, "Bedding mass") | 1 | U |  | UNITS = EV (g, UCUM, "g")
34b | > | CONTAINS | NUM | EV (127185, DCM, "Bedding depth") | 1 | U |  | UNITS = EV (mm, UCUM, mm")
35 | > | CONTAINS | NUM | EV (C90365, NCIt, "Bedding change") | 1 | U |  | UNITS = EV (d, UCUM, "days")
36 | > | CONTAINS | CODE | EV (127192, DCM, "Enrichment material present") | 1 | U |  | DCID 241 “Present-Absent Only”
36b | > | CONTAINS | TEXT | EV (127191, DCM, "Enrichment manufacturer") | 1 | U |  |
37 | > | CONTAINS | TEXT | EV (127190, DCM, "Enrichment material") | 1 | U |  |
38 | > | CONTAINS | CODE | EV (127193, DCM, "Exerciser device present") | 1 | U |  | DCID 241 “Present-Absent Only”
39 | > | CONTAINS | TEXT | EV (A-17200, SRT, "Exerciser device") | 1 | U |  |
40 | > | CONTAINS | CODE | EV (127195, DCM, "Shelter type") | 1 | U |  | DCID 606 “Animal Shelter Types”
41 | > | CONTAINS | TEXT | EV (127196, DCM, "Shelter manufacturer") | 1 | U |  |
42 | > | CONTAINS | TEXT | EV (127197, DCM, "Shelter product name") | 1 | U |  |
43 | > | CONTAINS | TEXT | EV (127198, DCM, "Shelter product code") | 1 | U |  |
44 | > | CONTAINS | TEXT | EV (121106, DCM,"Comment") | 1 | U |  |

Content Item Descriptions

Row 2 and 2b | The type and identifier of the entire room in which, for example, one or more racks of housing units is located, not the housing unit itself.
Row 3 | The manufacturer is expected to be the same for all housing unit components, rack, bottom and lid.
Row 12 | The position in the rack is encoded as text since it may be an "identifier" or a description. It is not a set of numeric (e.g., row, column) or coded (e.g., top, bottom or middle) values, since there are too many possible arrangements.
Row 13 | The number of animals usually applies to a single housing unit, but may also be used to describe the number of animals imaged simultaneously in a multi-animal imaging carrier or support device ("chamber", "holder", etc.).
Rows 20-22 | These may be internal or external dimensions, and are intended to provide an approximation of the living space and shape available.
Rows 25-26 | Description of measured or monitored or nominal values of temperature and humidity. The means of maintaining these conditions, if relevant, is described elsewhere (e.g., in the case of peri-procedural temperature control, in TID 8140 “Heating Conditions”).
Rows 28-29 | The bedding material may be described as a code or text, or both. The codes do not distinguish between methods of sterilization of the bedding material (e.g., irradiation, autoclaving or other heat treatment), since that is not a relevant factor for image interpretation. The definition of the NCIt concept is "that which comprises the place where a subject sleeps".
Row 35 | The definition of the NCIt concept is "a replacement of the existing materials that make up the sleeping area of a subject", and is used here to specify the interval between bedding changes.
Row 36-37 | The presence or absence of enrichment material is coded, but the type is not, and may be described as text, e.g., "facial tissue", "cotton (nesting material)".
Row 38-39 | The presence or absence of an exercise device is coded, but the type is not, and may be described as text.

### TID 8122 Animal Feeding

This template encodes a description of feeding and watering of animals, over an interval during which conditions are relatively homogenous.

### Note

1. No specific time interval during which the diet is applicable is described.

2. Values for product names and codes are expected to be accurate at the time the information is recorded, recognizing that products may evolve over time.

Type:

  Extensible

Order:

  Non-Significant

Table TID 8122. Animal Feeding

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (PA-00600, SRT, "Feeding") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (C-F5000, SRT, "Animal feed") | 1 | U |  | DCID 607 “Animal Feed Types”
3 | > | CONTAINS | CODE | EV (127205, DCM, "Feed source") | 1 | U |  | DCID 608 “Animal Feed Sources”
4 | > | CONTAINS | TEXT | EV (127200, DCM, "Feed manufacturer") | 1 | U |  |
5 | > | CONTAINS | TEXT | EV (127201, DCM, "Feed product name") | 1 | U |  |
6 | > | CONTAINS | TEXT | EV (127202, DCM, "Feed product code") | 1 | U |  |
7 | > | CONTAINS | CODE | EV (C0015746, UMLS, "Feeding method") | 1 | U |  | DCID 609 “Animal Feeding Methods”
8 | > | CONTAINS | CODE | EV (C-10120, SRT, "Water") | 1 | U |  | DCID 610 “Water Types”
9 | > | CONTAINS | CODE | EV (C90486, NCIt, "Water delivery") | 1 | U |  | DCID 609 “Animal Feeding Methods”
10 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

### TID 8130 Anesthesia

This template encodes a description of the anesthesia applied during a procedure (e.g., imaging of research small animals).

### Note

This template combines selected concepts from the [AQI Schema] elements, their complex types, and their children:

- AnesthesiaMethodSet (type AnesthesiaMethodSetType). See http://www.aqihq.org/aqischdoc/AnesthesiaMethodSet.html and http://www.aqihq.org/aqischdoc/AnesthesiaMethodSetType.html.

- AirwayManagementSet (type AirwayManagementSetType). See http://www.aqihq.org/aqischdoc/AirwayManagementSet.html and http://www.aqihq.org/aqischdoc/AirwayManagementSetType.html.

- MedicationsSet (type MedicationsSetType). See http://www.aqihq.org/aqischdoc/MedicationsSet.html and http://www.aqihq.org/aqischdoc/MedicationsSetType.html.

Type:

  Extensible

Order:

  Non-Significant

Table TID 8130. Anesthesia

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (P1-0512A, SRT, "Administration of anesthesia") | 1 | M |  |
2 | > | CONTAINS | CONTAINER | EV (127300, DCM, "Anesthesia Method Set") | 1 | M |  |
3 | >> | CONTAINS | CONTAINER | EV (127301, DCM, "Anesthesia Method") | 1-n | M |  |
4 | >>> | CONTAINS | CODE | EV (127302, DCM, "Anesthesia Category") | 1 | M |  | DCID 611 “Anesthesia Category Code Type for Small Animal Anesthesia”
5 | >>> | CONTAINS | TEXT | EV (127303, DCM, "Anesthesia SubCategory") | 1 | U |  |
6 | >>> | CONTAINS | DATETIME | EV (DF-0068E, SRT, "Anesthesia Start Time") | 1 | U |  |
7 | >>> | CONTAINS | DATETIME | EV (DF-0070B, SRT, "Anesthesia Finish Time") | 1 | U |  |
8 | >>> | CONTAINS | CODE | EV (P1-C0012, SRT, "Anesthesia Induction") | 1 | U |  | DCID 613 “Anesthesia Induction Code Type for Small Animal Anesthesia”
9 | >>> | CONTAINS | CODE | EV (P1-C001A, SRT, "Anesthesia Maintenance") | 1 | U |  | DCID 615 “Anesthesia Maintenance Code Type for Small Animal Anesthesia”
10 | >>> | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
11 | > | CONTAINS | CONTAINER | EV (127310, DCM, "Airway Management Set") | 1 | M |  |
12 | >> | CONTAINS | CONTAINER | EV (P0-0409B, SRT, "Airway Management") | 1-n | M |  |
13 | >>> | CONTAINS | CODE | EV (127312, DCM, "Airway Management Method") | 1 | M |  | DCID 617 “Airway Management Method Code Type for Small Animal Anesthesia”
14 | >>> | CONTAINS | CODE | EV (127313, DCM, "Airway Sub-Management Method") | 1 | M |  | DCID 619 “Airway Management Sub-Method Code Type for Small Animal Anesthesia”
15 | > | CONTAINS | CONTAINER | EV (127320, DCM, "Medications Set") | 1-n | M |  |
16 | >> | CONTAINS | CODE | EV (G-7292, SRT, "Procedure Phase") | 1 | M |  | DCID 631 “Phase of Procedure Requiring Anesthesia”
17 | >> | CONTAINS | INCLUDE | DTID 8131 “Medications and Mixture Medications” | 1-n | M |  |

Content Item Descriptions

Rows 1-3 | If this template is used, at least one description of anesthesia method is required. Note that the specific agents used are described separately, as intra-operative medications, per the [AQI Schema].
Rows 6-7 | These correspond to AQI elements that are named "Time" rather than "DateTime", though their value is a DateTime; the DICOM naming convention is used here.
Row 9 | Only inhalational methods of maintenance are included in this row. Absence of this row implies that the (non-inhalational) induction method is used for maintenance.
Row 10 | The comment corresponds to AQI element "Anesthesia Notes".
Rows 11-13 | At least one description of airway management is required. The airway management method also serves as the description of the method of inhalational anesthesia delivery, even if it does not involve "management" of the "airway" per se (e.g., delivery via nose cone).
Rows 15-17 | In the AQI model, a single AQI MedicationsSet is used in the AQI IntraOp element to describe intra-operative medications. This template allows a more general usage, with one or more Medications Set containers, each of which may be qualified by the phase of the procedure (pre-operative, intra-operative or post-operative). The purpose of the medication (e.g., general anesthetic) is described in the (111516, DCM, "Medication Type") of the included TID 8131 “Medications and Mixture Medications”.

### TID 8131 Medications and Mixture Medications

This template encodes a description of medications (including but not limited to anesthetic agents) used during a procedure (e.g., anesthesia for imaging of research small animals).

### Note

This template combines selected concepts from the [AQI Schema] elements, their complex types, and their children:

- Medication (type MedicationType). See http://www.aqihq.org/aqischdoc/Medication.html and http://www.aqihq.org/aqischdoc/MedicationType.html.

- MixtureMedications (type MixtureMedicationType). See http://www.aqihq.org/aqischdoc/MixtureMedications.html and http://www.aqihq.org/aqischdoc/MixtureMedicationType.html.

Type:

  Extensible

Order:

  Non-Significant

Table TID 8131. Medications and Mixture Medications

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (F-04460, SRT, "Medication given") | 1 | M |  |
2 | > | CONTAINS | DATETIME | EV (122081, DCM, "Drug start") | 1 | U |  |
3 | > | CONTAINS | DATETIME | EV (122082, DCM, "Drug end") | 1 | U |  |
4 | > | CONTAINS | CODE | EV (G-C340, SRT, "Route of administration") | 1 | M |  | DCID 11 “Route of Administration”
5 | > | CONTAINS | CONTAINER | EV (R-40826, SRT, "Mixture") | 1-n | M |  |
6 | >> | CONTAINS | CODE | EV (122083, DCM, "Drug administered") | 1 | MC | XOR Row 7 | DCID 623 “Medication for Small Animal Anesthesia”
7 | >> | CONTAINS | TEXT | EV (122083, DCM, "Drug administered") | 1 | MC | XOR Row 6 |
8 | >> | CONTAINS | CODE | EV (111516, DCM, "Medication Type") | 1 | M |  | DCID 621 “Medication Type Code Type for Small Animal Anesthesia”
9 | >> | CONTAINS | NUM | EV (G-C0B7, SRT, "Dosage") | 1 | U |  | UNITS = DCID 82 “Units of Measurement”
10 | >> | CONTAINS | NUM | EV (122093, DCM, "Concentration") | 1 | U |  | UNITS = DCID 82 “Units of Measurement”

Content Item Descriptions

Row 1 | AQI Medication type and element correspond to (F-04460, SRT, "Medication given") (situation). (See TID 3806 Cath Procedure).
Rows 2-3 | AQI DoseStart and DoseEnd elements correspond to (122081, DCM, "Drug start") and (122082, DCM, "Drug end") respectively. (See CID 3409 Administration of Drugs/Contrast). If the medication is delivered as a bolus, the end time is omitted.
Row 4 | AQI MedicationRoute corresponds to (G-C340, SRT, "Route of administration"). The existing CID 11 “Route of Administration” contains a relevant subset of concepts for the enumerated values of AQI MedicationRouteCodeType.
Row 5 | The AQI schema allows the Medication type not only to describe medications with a single component, but also to add MixtureMedications children, each of which is encoded following a similar pattern to the contents of Medication, though the start and end time and route of administration are shared. This had been modeled by allowing every medication to have one or more mixture children. For medications that are not a mixture, a single instance of this row defines the medication (even though the mixture container is still used).
Rows 6, 7 | AQI MedicationName and MixtureMedicationName elements correspond to (122083, DCM, "Drug administered"). (See TID 3806 Cath Procedure). The medication (e.g., anesthesia agent) can be described with a code or text, e.g., (F-61B0A, SRT, "Isoflurane") or "isoflurane".
Row 9 | Both AQI MedDose (or MixtureMedDose) and DoseUnits (or MixtureDoseUnits) elements are combined in one content item. Units are required to be encoded as UCUM but are not otherwise constrained.
Row 10 | Both AQI MedConcentration (or MixtureMedConcentration) and MedConcentrationUnit (or MixtureMedConcentrationUnit) elements are combined in one content item. Units are required to be encoded as UCUM but are not otherwise constrained.

### TID 8140 Heating Conditions

This template encodes a description of the heating conditions applied prior to, during or after data acquisition (e.g., during imaging of research small animals).

Type:

  Extensible

Order:

  Non-Significant

Table TID 8140. Heating Conditions

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (127040, DCM, "Heating conditions") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (G-7292, SRT, "Procedure Phase") | 1 | U |  | DCID 631 “Phase of Procedure Requiring Anesthesia”
3 | > | CONTAINS | CODE | EV (C0018851, UMLS, "Heating") | 1 | U |  | DCID 635 “Heating Method”
4 | > | CONTAINS | CODE | EV (127210, DCM, "Feedback temperature regulation") | 1 | U |  | DCID 231 “Yes-No Only”
5 | > | CONTAINS | CODE | EV (C50304, NCIt, "Temperature sensor device component") | 1 | U |  | DCID 636 “Temperature Sensor Device Component Type for Small Animal Procedures”
6 | > | CONTAINS | NUM | EV (F-021FF, SRT, "Equipment Temperature") | 1 | U |  | UNITS = EV (Cel, UCUM, "C")

Content Item Descriptions

Row 2 | Phase during which the conditions are applicable may be implicit in the context of invocation of this template (e.g., TID 8101 “Preclinical Small Animal Image Acquisition Context” Row 7), or explicitly specified.
Row 3 | The definition (from MESH) is "The application of heat to raise the temperature of the environment, ambient or local, or systems for accomplishing this effect".
Row 6 | This is the nominal temperature of the heating device (e.g., heating pad) and/or the set point of the feedback regulation device.

### TID 8150 Circadian Effects

This template encodes a description of the Circadian effects relevant during data acquisition (e.g., during imaging of research small animals).

Type:

  Extensible

Order:

  Non-Significant

Table TID 8150. Circadian Effects

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (127050, DCM, "Circadian effects") | 1 | M |  |
2 | > | CONTAINS | NUM | EV (127214, DCM, "Total duration of light-dark cycle") | 1 | U |  | UNITS = EV (h, UCUM, "hours")
3 | > | CONTAINS | NUM | EV (C90419, NCIt, "Light cycle") | 1 | U |  | UNITS = EV (%, UCUM, "%")
4 | > | CONTAINS | TIME | EV (127215, DCM, "Lights on time of day") | 1-n | U |  |

Content Item Descriptions

Row 2 | Usually 24 hours.
Row 3 | The definition is "the amount of ambient light/darkness to which a subject is exposed in a period of time"; also mapped to CDISC "the period of light that a subject is exposed to in a period of time, usually expressed as the amount of time in a 24 hour cycle".
Row 4 | Can only be encoded if the light-dark cycles are aligned to a 24 hour clock. May be multiple if either multiple cycles occur during a 24 hour period, or if the cycle is longer than a 24 hour period and a multiple of 24 hours in duration.

### TID 8170 Physiological Monitoring Performed During Procedure

This template encodes a description of the physiological monitoring performed during a period of time during or related to a data acquisition procedure (e.g., imaging of research small animals).

Type:

  Extensible

Order:

  Non-Significant

Table TID 8170. Physiological Monitoring Performed During Procedure

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (P0-005ED, SRT, "Physiological monitoring") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (P2-31209, SRT, "Electrocardiographic monitoring") | 1 | U |  | DCID 231 “Yes-No Only”
3 | > | CONTAINS | CODE | EV (P2-22010, SRT, "Monitoring of respiration") | 1 | U |  | DCID 231 “Yes-No Only”

Content Item Descriptions

Row 2 | There is no non-surgical procedure non-specific variant of (P2-34122, SRT, "Monitoring of electrocardiogram at surgery"). (P2-31209, SRT, "Continuous electrocardiogram monitoring") is intended for non-procedural (e.g., 24-hour) monitoring. So a more generic code for any kind of monitoring is used.

### TID 8182 Exogenous Substance Administration

This template provides detailed information on a research subject's exposure to exogenous substances. It is a specialization of the more general template TID 9002 “Medication, Substance, Environmental Exposure”.

Table TID 8182. Parameters

Parameter Name | Parameter Usage
--- | ---
$ContainerConcept | Coded term for the concept name of the CONTAINER, identifying it as an exogenous substance.
$CodeConcept | Coded term for the concept name of the CODE, identifying the type of substance.
$CodeValue | Coded term or Context Group for value of the substance.
$Classification | Coded term or Context Group for classification of the substance.
$Route | Coded term or Context Group for the route of administration of the substance.
$Site | Coded term or Context Group for the anatomical site of administration of the substance
$TissueOfOrigin | Coded term or Context Group for the tissue of origin of the substance
$TaxonomicRankOfOrigin | Coded term or Context Group for the taxonomic rank (e.g., species) of origin of the substance

Type:

  Extensible

Order:

  Significant

Table TID 8182. Exogenous Substance Administration

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | $ContainerConcept | 1 | M |  |
2 | > | CONTAINS | CODE | $CodeConcept | 1-n | M |  | $CodeValue
3 | >> | HAS CONCEPT MOD | CODE | EV (G-C032, SRT, "Classification") | 1 | U |  | $Classification
4 | >> | HAS OBS CONTEXT | CODE | EV (111534, DCM, "Role of person reporting") | 1 | U |  | DCID 7450 “Person Roles”
5 | >> | HAS PROPERTIES | NUM | EV (111524, DCM, "Age Started") | 1 | U |  | UNITS = DCID 7456 “Units of Measure for Age”
6 | >> | HAS PROPERTIES | NUM | EV (111525, DCM, "Age Ended") | 1 | U |  | UNITS = DCID 7456 “Units of Measure for Age”
7 | >> | HAS PROPERTIES | DATETIME | EV (111526, DCM, "DateTime Started") | 1 | U |  |
8 | >> | HAS PROPERTIES | DATETIME | EV (111527, DCM, "DateTime Ended") | 1 | U |  |
9 | >> | HAS PROPERTIES | NUM | EV (G-7290, SRT, "Duration") | 1 | U |  | UNITS = DCID 6046 “Units of Follow-up Interval”
10 | >> | HAS PROPERTIES | CODE | EV (111528, DCM, "Ongoing") | 1 | U |  | DCID 230 “Yes-No”
11 | >> | HAS PROPERTIES | TEXT | EV (111529, DCM, "Brand Name") | 1 | U |  |
12 | >> | HAS PROPERTIES | NUM | DCID 6092 “Quantitative Concepts for Usage, Exposure” | 1 | U |  | The unit of measure shall be quantity per unit of time
13 | >> | HAS PROPERTIES | CODE | DCID 6093 “Qualitative Concepts for Usage, Exposure Amount” | 1 | U |  | DCID 6090 “Relative Usage, Exposure Amount”
14 | >> | HAS PROPERTIES | CODE | DCID 6094 “Qualitative Concepts for Usage, Exposure Frequency” | 1 | U |  | DCID 6091 “Relative Frequency of Event Values”
15 | >> | HAS PROPERTIES | CODE | EV (G-C340, SRT, "Route of administration") | 1 | U |  | $Route
16 | >>> | HAS PROPERTIES | CODE | EV (G-C581, SRT, "Site of") | 1 | U |  | $Site
17 | >>>> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IF Row 16 has laterality | DCID 244 “Laterality”
18 | >>> | HAS PROPERTIES | SCOORD3D | EV (127450, DCM, "Stereotactic coordinates") | 1 | U |  |
19 | >>> | HAS PROPERTIES | CODE | EV (127451, DCM, "Position reference indicator") | 1 | U |  | DCID 647 “Position Reference Indicator for Frame of Reference”
20 | >> | HAS PROPERTIES | CODE | EV (127401, DCM, "Tissue of origin") | 1 | U |  | $TissueOfOrigin
21 | >> | HAS PROPERTIES | CODE | EV (127402, DCM, "Taxonomic rank of origin") | 1 | U |  | $TaxonomicRankOfOrigin
22 | >> | HAS PROPERTIES | CODE | EV (127411, DCM, "Strain") | 1 | U |  |
23 | >> | HAS PROPERTIES | TEXT | EV (127412, DCM, "Strain description") | 1 | U |  |
24 | >>> | HAS CONCEPT MOD | TEXT | EV (127413, DCM, "Nomenclature") | 1 | U |  |
25 | >> | HAS PROPERTIES | TEXT | EV (127415, DCM, "Genetic modifications description") | 1-n | U |  |
26 | >>> | HAS CONCEPT MOD | TEXT | EV (127413, DCM, "Nomenclature") | 1 | U |  |
27 | >>> | HAS PROPERTIES | CODE | EV (127414, DCM, "Genetic modifications") | 1 | U |  |

Content Item Descriptions

Row 3 | Classification is inherited from the more general template TID 9002 “Medication, Substance, Environmental Exposure”, and may be supplied as a parameter, but is entirely generic and is not used as an alternative to the more specific information provided in other rows, for example, Rows 19 and 20, tissue and taxonomic rank of origin.
Row 11 | Brand name may be used for any type of descriptor or identifier. E.g., a particular cell line might have a designated name, such as "MDA-MB-468", which designates a particular human breast cancer cell line.
Rows 22-27 | These rows describe the strain and genetic modifications of the source of the graft using content items that correspond to the Attributes described in Section C.7.1.1.1.4 “Patient Strain and Genetic Modifications” in PS3.3 . The strain and genetic characteristics of the animal into which the exogenous substance is grafted are described in the Patient Module; see Section C.7.1.1.1.4 “Patient Strain and Genetic Modifications” in PS3.3 .

## Relevant Patient Information Templates

### TID 9000 Relevant Patient Information for Breast Imaging

This Template collects a patient's relevant information as it relates to breast imaging. This Template, together with its subordinate Templates, describes the history of a patient's reproductive system, hormone medications, past procedures, risk factors, and indicated problems as they relate to breast health.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 9000. Relevant Patient Information for Breast Imaging

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111511, DCM, "Relevant Patient Information for Breast Imaging") | 1 | M |  |
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | CONTAINS | INCLUDE | DTID 3114 “Patient Assessment” | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 9001 “Gynecological History” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 9002 “Medication, Substance, Environmental Exposure” | 1 | U |  | $ContainerConcept = EV (10160-0, LN, "History Of Medication Use") $CodeConcept = EV (111516, DCM, "Medication Type") $CodeValue = DCID 6080 “Gynecological Hormones”
6 | > | CONTAINS | INCLUDE | DTID 9003 “Previous Procedure” | 1 | U |  | $ProcedureList = DCID 6083 “Procedures for Breast” $ProcedureModifier = DCID 6058 “Procedure Modifiers for Breast” $NumConceptName = DCID 6095 “Numeric Properties of Procedures” $LateralityValue = DCID 6022 “Side” $ProcedureResult = DCID 6063 “Interventional Procedure Results” $ComplicationValue = DCID 6062 “Interventional Procedure Complications”
7 | > | CONTAINS | INCLUDE | DTID 9004 “Indicated Problem” | 1 | U |  | $ProblemList = DCID 6055 “Breast Clinical Finding or Indicated Problem” $LateralityValue = DCID 6022 “Side” $LocationValue = DCID 6018 “Clockface Location or Region”, DCID 6020 “Quadrant Location”
8 | > | CONTAINS | INCLUDE | DTID 9005 “Risk Factor” | 1 | U |  | $RiskList = DCID 6081 “Breast Cancer Risk Factors” $FamilyList = DCID 7451 “Family Member”

### TID 9001 Gynecological History

This general Template collects the details of a patient's reproductive system history, such as number of births, and gynecological surgery history.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 9001. Gynecological History

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (R-20767, SRT, "Gynecological History") | 1 | M |  |
2 | > | HAS OBS CONTEXT | CODE | EV (111534, DCM, "Role of person reporting") | 1 | U |  | DCID 7450 “Person Roles”
3 | > | CONTAINS | DATE | EV (11955-2, LN, "Date of last menstrual period") | 1 | U |  |
4 | > | CONTAINS | NUM | EV (111518, DCM, "Age when first menstrual period occurred") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
5 | > | CONTAINS | NUM | EV (111519, DCM, "Age at First Full Term Pregnancy") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
6 | > | CONTAINS | NUM | EV (11977-6, LN, "Para") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
7 | > | CONTAINS | NUM | EV (11639-2, LN, "Term") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
8 | > | CONTAINS | NUM | EV (11637-6, LN, "Preterm") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
9 | > | CONTAINS | NUM | EV (11636-8, LN, "Live Births") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
10 | > | CONTAINS | NUM | EV (111593, DCM, "LBW or IUGR") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
11 | > | CONTAINS | NUM | EV (11996-6, LN, "Gravida") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
12 | > | CONTAINS | NUM | EV (11612-9, LN, "Aborta") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
13 | > | CONTAINS | NUM | EV (33065-4, LN, "Ectopic Pregnancies") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
14 | > | CONTAINS | NUM | EV (111520, DCM, "Age at Menopause") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
15 | > | CONTAINS | NUM | EV (111521, DCM, "Age when hysterectomy performed") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
16 | >> | HAS CONCEPT MOD | CODE | EV (R-404ED, SRT, "Extent") | 1 | U |  | EV (R-404F1, SRT, "Complete") EV (R-404FE, SRT, "Partial")
17 | > | CONTAINS | NUM | EV (111522, DCM, "Age when left ovary removed") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
18 | > | CONTAINS | NUM | EV (111523, DCM, "Age when right ovary removed") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
19 | > | CONTAINS | CODE | EV (111543, DCM, "Breast feeding history") | 1 | U |  | DCID 230 “Yes-No”
20 | >> | HAS PROPERTIES | NUM | EV (111544, DCM, "Average breast feeding period") | 1 | U |  | UNITS = EV (wk, UCUM, "Week")
21 | > | CONTAINS | CODE | EV (364320009, SCT, "Pregnancy observable") | 1 | U |  | DCID 6096 “Pregnancy Status”
22 | > | CONTAINS | CODE | EV (111391, DCM, "Menstrual Cycle Phase") | 1 | U |  | DCID 6163 “Menstrual Cycle Phase”

### TID 9002 Medication, Substance, Environmental Exposure

This general Template provides detailed information on a patient's medication or substance use, or exposure to environmental factors, including type and duration of use or exposure.

Table TID 9002. Parameters

Parameter Name | Parameter Usage
--- | ---
$ContainerConcept | Coded term for the concept name of the CONTAINER, identifying it as medication, substance, or environmental exposure history.
$CodeConcept | Coded term for the concept name of the CODE, identifying it as medication, substance, or environmental exposure.
$CodeValue | Coded term or Context Group for value of the medication, substance, or environmental exposure.
$Classification | Coded term or Context Group for classification of the medication, substance, or environmental exposure.
$Route | Coded term or Context Group for the route of administration of the medication, substance, or route of environmental exposure.
$Site | Coded term or Context Group for the anatomical site of administration of the medication, substance, or anatomical site of environmental exposure.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 9002. Medication, Substance, Environmental Exposure

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | $ContainerConcept | 1 | M |  |
2 | > | CONTAINS | CODE | $CodeConcept | 1-n | M |  | $CodeValue
3 | >> | HAS CONCEPT MOD | CODE | EV (G-C032, SRT, "Classification") | 1 | U |  | $Classification
4 | >> | HAS OBS CONTEXT | CODE | EV (111534, DCM, "Role of person reporting") | 1 | U |  | DCID 7450 “Person Roles”
5 | >> | HAS PROPERTIES | NUM | EV (111524, DCM, "Age Started") | 1 | U |  | DCID 7456 “Units of Measure for Age”
6 | >> | HAS PROPERTIES | NUM | EV (111525, DCM, "Age Ended") | 1 | U |  | DCID 7456 “Units of Measure for Age”
7 | >> | HAS PROPERTIES | DATETIME | EV (111526, DCM, "DateTime Started") | 1 | U |  |
8 | >> | HAS PROPERTIES | DATETIME | EV (111527, DCM, "DateTime Ended") | 1 | U |  |
9 | >> | HAS PROPERTIES | NUM | EV (G-7290, SRT, "Duration") | 1 | U |  | UNITS = DCID 6046 “Units of Follow-up Interval”
10 | >> | HAS PROPERTIES | CODE | EV (111528, DCM, "Ongoing") | 1 | U |  | DCID 230 “Yes-No”
11 | >> | HAS PROPERTIES | TEXT | EV (111529, DCM, "Brand Name") | 1 | U |  |
12 | >> | HAS PROPERTIES | NUM | DCID 6092 “Quantitative Concepts for Usage, Exposure” | 1 | U |  | The unit of measure shall be quantity per unit of time
13 | >> | HAS PROPERTIES | CODE | DCID 6093 “Qualitative Concepts for Usage, Exposure Amount” | 1 | U |  | DCID 6090 “Relative Usage, Exposure Amount”
14 | >> | HAS PROPERTIES | CODE | DCID 6094 “Qualitative Concepts for Usage, Exposure Frequency” | 1 | U |  | DCID 6091 “Relative Frequency of Event Values”
15 | >> | HAS PROPERTIES | CODE | EV (G-C340, SRT, "Route of administration") | 1 | U |  | $Route
16 | >>> | HAS PROPERTIES | CODE | EV (G-C581, SRT, "Site of") | 1 | U |  | $Site
17 | >>>> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IF Row 16 has laterality | DCID 244 “Laterality”

Content Item Descriptions

Row 3 | Classification is mapped in UMLS to (C0008902, UMLS, "Classification"). Its definition is completely generic; i.e., it does not refer to any particular type of classification.
Rows 13 & 14 | If both of these Content Items are instantiated, the concept names selected for each should match. For example, use "Relative dose amount" as the concept name for row 13 with "Relative dose frequency" as the concept name for row 14.
Row 15 | Even though the concept name is route of administration, it is also used for route of exposure in the case of environmental exposure.
Rows 16 and 17 | This pattern of route with a site and laterality modifier follows that used in TID 10022 “Radiopharmaceutical Administration Event Data”.

### TID 9003 Previous Procedure

This general Template provides detailed information on a patient's previous procedure, surgery, or treatment.

Table TID 9003. Parameters

Parameter Name | Parameter Usage
--- | ---
$ProcedureList | Coded term or Context Group for value of Previous Procedure
$ProcedureModifier | Coded term or Context Group for value of Previous Procedure Modifier
$NumConceptName | Coded term or Context Group for the concept name of a numeric property of the Previous Procedure
$LateralityValue | Coded term or Context Group for value of Laterality
$ProcedureResult | Coded term or Context Group for value of Result of Procedure
$ComplicationValue | Coded term or Context Group for value of Complication

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 9003. Previous Procedure

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111513, DCM, "Relevant Previous Procedures") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (111531, DCM, "Previous Procedure") | 1-n | M |  | $ProcedureList
3 | >> | HAS CONCEPT MOD | CODE | EV (111464, DCM, "Procedure Modifier") | 1-n | U |  | $ProcedureModifier
4 | >> | HAS OBS CONTEXT | CODE | EV (111534, DCM, "Role of person reporting") | 1 | U |  | DCID 7450 “Person Roles”
5 | >> | HAS PROPERTIES | NUM | $NumConceptName | 1-n | U |  |
6 | >> | HAS PROPERTIES | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | $LateralityValue
7 | >> | HAS PROPERTIES | DATETIME | EV (122146, DCM, "Procedure DateTime") | 1 | UC | XOR row 8 |
8 | >> | HAS PROPERTIES | CODE | EV (111395, DCM, "Estimated Timeframe") | 1 | UC | XOR row 7 | BCID 6164 “Time Intervals”
9 | >> | HAS PROPERTIES | NUM | EV (R-42009, SRT, "Number of occurrences") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
10 | >> | HAS PROPERTIES | CODE | EV (DD-60002, SRT, "Complication of procedure") | 1-n | U |  | $ComplicationValue
11 | >>> | HAS PROPERTIES | CODE | EV (111466, DCM, "Severity of Complication") | 1 | U |  | DCID 251 “Severity of Complication”
12 | >> | HAS PROPERTIES | CODE | EV (122177, DCM, "Procedure Result") | 1 | U |  | $ProcedureResult
13 | >> | HAS PROPERTIES | INCLUDE | DTID 4207 “Breast Imaging Pathology Results” | 1-n | U |  |

### TID 9004 Indicated Problem

This general Template provides information about indicated problems presented by a patient. For example, indicated breast problems relating to the purpose for a mammographic examination.

Table TID 9004. Parameters

Parameter Name | Parameter Usage
--- | ---
$ProblemList | Coded term or Context Group for value of Indicated Problem
$LateralityValue | Coded term or Context Group for value of Laterality
$LocationValue | Coded term or Context Group for value of Location

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 9004. Indicated Problem

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (11450-4, LN, "Problem List") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (111533, DCM, "Indicated Problem") | 1-n | M |  | $ProblemList
3 | >> | HAS OBS CONTEXT | CODE | EV (111534, DCM, "Role of person reporting") | 1 | U |  | DCID 7450 “Person Roles”
4 | >> | HAS OBS CONTEXT | DATETIME | EV (111535, DCM, "DateTime problem observed") | 1 | U |  |
5 | >> | HAS PROPERTIES | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | $LateralityValue
6 | >> | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding site") | 1 | U |  | $LocationValue
7 | >> | HAS PROPERTIES | NUM | EV (G-7290, SRT, "Duration") | 1 | U |  |
8 | >> | HAS PROPERTIES | CODE | EV (R-407E7, SRT, "Frequency") | 1 | U |  | DCID 6091 “Relative Frequency of Event Values”
9 | >> | HAS PROPERTIES | DATETIME | EV (111536, DCM, "DateTime of last evaluation") | 1 | U |  |
10 | >> | HAS PROPERTIES | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

### TID 9005 Risk Factor

This general Template provides detailed information on the risk factors for a patient, related to medical history for themselves and family members.

Table TID 9005. Parameters

Parameter Name | Parameter Usage
--- | ---
$RiskList | Coded term or Context Group for value of Risk Factor
$FamilyList | Coded term or Context Group for value of Family Member with Risk Factor

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 9005. Risk Factor

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111515, DCM, "Relevant Risk Factors") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (F-01500, SRT, "Risk factor") | 1-n | M |  | $RiskList
3 | >> | HAS CONCEPT MOD | CODE | EV (111530, DCM, "Risk Factor modifier") | 1 | U |  | EV (G-0002, SRT, "Family history of")
4 | >> | HAS PROPERTIES | NUM | EV (18185-9, LN, "Gestational Age") | 1 | UC | IFF value of row 2 is (G-0305, SRT, "History of - premature delivery") |
5 | >> | HAS OBS CONTEXT | CODE | EV (111534, DCM, "Role of person reporting") | 1 | U |  | DCID 7450 “Person Roles”
6 | >> | HAS PROPERTIES | NUM | EV (111538, DCM, "Age at Occurrence") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
7 | >> | HAS PROPERTIES | NUM | EV (G-7290, SRT, "Duration") | 1 | U |  | UNITS = DCID 6046 “Units of Follow-up Interval”
8 | >> | HAS PROPERTIES | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
9 | >> | INFERRED FROM | CODE | EV (111537, DCM, "Family Member with Risk Factor") | 1-n | U |  | $FamilyList
10 | >>> | HAS PROPERTIES | NUM | EV (111538, DCM, "Age at Occurrence") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
11 | >>> | HAS CONCEPT MOD | CODE | EV (111539, DCM, "Menopausal phase") | 1 | U |  | DCID 6086 “Menopausal Phase”
12 | >>> | HAS CONCEPT MOD | CODE | EV (111540, DCM, "Side of Family") | 1 | U |  | DCID 6097 “Side of Family”

### TID 9006 Obstetric History

This general Template collects the details of a patient's obstetric history for a current pregnancy. Information regarding previous pregnancies is conveyed using the Gynecological History Template.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 9006. Obstetric History

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (R-20658, SRT, "Obstetric History") | 1 | M |  |
2 | > | CONTAINS | DATE | DCID 12003 “OB-GYN Dates” | 1-n | U |  |
3 | > | CONTAINS | NUM | EV (18185-9, LN, "Gestational Age") | 1 | U |  | UNITS = EV (d, UCUM, "day")
4 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1-n | U |  |

Content Item Descriptions

Row 3 "Gestational Age" | Observation DateTime (0040,A032) for Content Item shall be present, in order to convey the date and time at which this Gestational Age was established.

### TID 9007 General Relevant Patient Information

This Template collects a patient's relevant information for general purpose use. This Template, together with its subordinate Templates, describes the history of a patient's reproductive system, medications, substance use, environmental exposure, past procedures, risk factors, and indicated problems.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 9007. General Relevant Patient Information

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111517, DCM, "Relevant Patient Information") | 1 | M |  |
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | CONTAINS | INCLUDE | DTID 3114 “Patient Assessment” | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 9002 “Medication, Substance, Environmental Exposure” | 1 | U |  | $ContainerConcept = EV (10160-0, LN, "History Of Medication Use") $CodeConcept = EV (111516, DCM, "Medication Type")
5 | > | CONTAINS | INCLUDE | DTID 9002 “Medication, Substance, Environmental Exposure” | 1 | U |  | $ContainerConcept = EV (111545, DCM, "Substance Use History") $CodeConcept = EV (111546, DCM, "Used Substance Type") $CodeValue = BCID 6089 “Substances”
6 | > | CONTAINS | INCLUDE | DTID 9002 “Medication, Substance, Environmental Exposure” | 1 | U |  | $ContainerConcept = EV (111547, DCM, "Environmental Exposure History") $CodeConcept = EV (111548, DCM, "Environmental Factor")
7 | > | CONTAINS | INCLUDE | DTID 9003 “Previous Procedure” | 1 | U |  | $LateralityValue = BCID 244 “Laterality”
8 | > | CONTAINS | INCLUDE | DTID 9004 “Indicated Problem” | 1 | U |  | $LateralityValue = BCID 244 “Laterality”
9 | > | CONTAINS | INCLUDE | DTID 9005 “Risk Factor” | 1 | U |  | $RiskList = BCID 6087 “General Risk Factors” $FamilyList = DCID 7451 “Family Member”
10 | > | CONTAINS | INCLUDE | DTID 9001 “Gynecological History” | 1 | U |  |
11 | > | CONTAINS | INCLUDE | DTID 9006 “Obstetric History” | 1 | U |  |
12 | > | CONTAINS | INCLUDE | DTID 3802 “Cardiovascular Patient History” | 1 | U |  |
13 | > | CONTAINS | INCLUDE | DTID 351 “Previous Reports” | 1 | U |  |

## X-Ray Radiation Dose SR IOD Templates

The Templates that comprise the X-Ray Radiation Dose SR are interconnected as in Figure A-14.

Figure A-14. X-Ray Radiation Dose SR IOD Template Structure

### TID 10001 Projection X-Ray Radiation Dose

This Template defines a container (the root) with subsidiary Content Items, each of which represents a single projection X-Ray irradiation event entry or plane-specific dose accumulations. There is a defined recording observer (the system or person responsible for recording the log, generally the system). A Biplane irradiation event will be recorded as two individual events, one for each plane. Accumulated values will be kept separate for each plane.

Type:

  Extensible

Order:

  Non-Significant

Root:

  Yes

Table TID 10001. Projection X-Ray Radiation Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113701, DCM, "X-Ray Radiation Dose Report") | 1 | M |  | Root node
1b | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (113704, DCM, "Projection X-Ray") DT (P5-40010, SRT, "Mammography")
3 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E8, SRT, "Has Intent") | 1 | M |  | DCID 3629 “Procedure Intent”
4 | > | CONTAINS | CODE | EV (122142, DCM, "Acquisition Device Type") | 1 | U |  | DCID 10032 “Projection X-Ray Acquisition Device Types”
5 | > |  | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
6 | > | HAS OBS CONTEXT | CODE | EV (113705, DCM, "Scope of Accumulation") | 1 | M |  | DCID 10000 “Scope of Accumulation”
7 | >> | HAS PROPERTIES | UIDREF | DCID 10001 “UID Types” | 1 | M |  |
8 | > | CONTAINS | CODE | EV (113945, DCM, "X-Ray Detector Data Available") | 1 | U |  | DCID 230 “Yes-No”
9 | > | CONTAINS | CODE | EV (113943, DCM, "X-Ray Source Data Available") | 1 | U |  | DCID 230 “Yes-No”
10 | > | CONTAINS | CODE | EV (113944, DCM, "X-Ray Mechanical Data Available") | 1 | U |  | DCID 230 “Yes-No”
11 | > | CONTAINS | INCLUDE | DTID 10002 “Accumulated X-Ray Dose” | 1 | MC | IFF Single Plane system | $Plane = EV (113622, DCM, "Single Plane")
12 | > | CONTAINS | INCLUDE | DTID 10002 “Accumulated X-Ray Dose” | 1 | MC | IFF Biplane system | $Plane = EV (113620, DCM, "Plane A")
13 | > | CONTAINS | INCLUDE | DTID 10002 “Accumulated X-Ray Dose” | 1 | MC | IFF Biplane system | $Plane = EV (113621, DCM, "Plane B")
14 | > | CONTAINS | INCLUDE | DTID 10003 “Irradiation Event X-Ray Data” | 1-n | MC | IF any of the values of TID (10001) Row 18 are not (113858, DCM, "MPPS Content"), (113866, DCM, "Copied From Image Attributes") or (113867, DCM, "Computed From Image Attributes") |
15 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
16 | > | CONTAINS | IMAGE | EV (121342, DCM, "Dose Image") | 1-n | U |  |
17 | > | CONTAINS | INCLUDE | DTID 1020 “Person Participant” | 1 | U |  | $PersonProcedureRole = EV (113850, DCM, "Irradiation Authorizing")
18 | > | CONTAINS | CODE | EV (113854, DCM, "Source of Dose Information") | 1-n | M |  | DCID 10020 “Source of Projection X-Ray Dose Information”

Content Item Descriptions

Row 2 | "Projection X-Ray" refers to procedures performed on either integrated equipment (where information is passed between the X-Ray source (generator and tube), detector, and mechanical systems), or non-integrated equipment (where data might not be available for one or more components such as cassette-based systems). The data availability can be described in Rows 8, 9 and 10. The specific type of equipment can be described in Row 4. The coded term for "Mammography" is intended to encompass all types of projection X-Ray imaging of the breast. ### Note Mammography exams are distinguished by a different value in this attribute, Angiography exams are distinguished by the Irradiation Event Type attribute = Fluoroscopy, CR/DR exams are distinguished by one or more of the Data Availability Flags = No, and CT exams are distinguished by the use of a different Template.
Row 5 | The observer context may include both a Person Observer identification, as well as the identity of the equipment providing the values for the irradiation event (Device Observer identification), if not inherited.
Row 8 | A value of "No" indicates that details associated with the X-Ray Detector are not available to the device generating this report. For example, an X-Ray Source system might lack any communication with the associated cassette-based X-Ray detector or any method of entering such information. A value of "Yes" or the absence of this row means that the details are available.
Row 9 | A value of "No" indicates that details associated with the X-Ray Source are not available to the device generating this report. For example, a cassette-based X-Ray detector might lack any communication with the associated X-Ray Source or any method of entering such information. A value of "Yes" or the absence of this row means that the details are available.
Row 10 | A value of "No" indicates that details associated with the Gantry and/or Table are not available to the device generating this report. For example, a cassette-based X-Ray detector might lack any communication with the associated gantry/table or any method of entering such information. A value of "Yes" or the absence of this row means that the details are available.
Row 11 | Cassette-based X-Ray systems should consider themselves to be Single Plane systems.
Row 14 | Details of the underlying irradiation events. If Row 18 has a value of "MPPS Content" then a TID 10003 “Irradiation Event X-Ray Data” item may be generated for each item in the MPPS Exposure Dose Sequence (0040,030E), but since this is an optional element in MPPS, if it is absent, empty or incomplete, there may be no irradiation event level information available. Alternatively, the information may be copied or computed from the images.
Row 16 | The Dose Image references a graphic representation of the radiation dose distribution. This may be a Secondary Capture scan of a dosimetry film.
Row 17 | The physician responsible for determining that the irradiating procedure was appropriate for the indications. The value may come from Requesting Physician (0032,1032), Requesting Physician Identification Sequence (0032,1031) or somewhere else based on hospital policies.
Row 18 | The primary source of information from which this dose object was constructed. The Source of Dose Information (Row 18) is independent of the Scope Of Accumulation (Row 6); e.g., it would be typical to have a scope of (113016, DCM, "Performed Procedure Step"), but a source of (113856, DCM, "Automated Data Collection") rather than (113858, DCM, "MPPS Content").

### TID 10002 Accumulated X-Ray Dose

This general Template provides detailed information on projection X-Ray dose value accumulations over several irradiation events from the same equipment (typically a study or a performed procedure step).

Table TID 10002. Parameters

Parameter Name | Parameter Usage
--- | ---
$Plane | Coded term identifying to which acquisition plane the encoded information belongs.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 10002. Accumulated X-Ray Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113702, DCM, "Accumulated X-Ray Dose Data") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (113764, DCM, "Acquisition Plane") | 1 | M |  | $Plane
3 | > | CONTAINS | CONTAINER | EV (122505, DCM, "Calibration") | 1-n | MC | IFF Calibration Data is available |
4 | >> | HAS CONCEPT MOD | CODE | EV (113794, DCM, "Dose Measurement Device") | 1 | M |  | DCID 10010 “Dose Measurement Devices”
5 | >> | CONTAINS | DATETIME | EV (113723, DCM, "Calibration DateTime") | 1 | M |  |
6 | >> | CONTAINS | NUM | EV (122322, DCM, "Calibration Factor") | 1 | M |  | UNITS = EV (1, UCUM, "no units")
7 | >> | CONTAINS | NUM | EV (113763, DCM, "Calibration Uncertainty") | 1 | M |  | UNITS = EV (%, UCUM, "Percent")
8 | >> | CONTAINS | TEXT | EV (113724, DCM, "Calibration Responsible Party") | 1 | M |  |
9 | >> | CONTAINS | TEXT | EV (113720, DCM, "Calibration Protocol") | 1 | U |  |
10 | > | CONTAINS | INCLUDE | DTID 10004 “Accumulated Fluoroscopy and Acquisition Projection X-Ray Dose” | 1 | MC | IFF TID (10001) Row 4 = (113957, DCM, "Fluoroscopy-Guided Projection Radiography System") or TID (10001) Row 2 = (113704, DCM, "Projection X-Ray") and TID (10001) Row 4 is absent) |
11 | > | CONTAINS | INCLUDE | DTID 10005 “Accumulated Mammography X-Ray Dose” | 1 | MC | IFF TID (10001) Row 2 = (P5-40010, SRT, "Mammography") |
12 | > | CONTAINS | INCLUDE | DTID 10007 “Accumulated Total Projection Radiography Dose” | 1 | MC | IFF TID (10001) Row 4 = (113958, DCM, "Integrated Projection Radiography System") or TID (10001) Row 4 = (113957, DCM, "Fluoroscopy-Guided Projection Radiography System") or TID (10001) Row 2 = (113704, DCM, "Projection X-Ray") and TID (10001) Row 4 is absent) |
13 | > | CONTAINS | INCLUDE | DTID 10006 “Accumulated Cassette-based Projection Radiography Dose” | 1 | MC | IFF TID (10001) Row 4 = (113959, DCM, "Cassette-based Projection Radiography System") |
14 | > | CONTAINS | INCLUDE | DTID 1021 “Device Participant” | 1 | MC | Required if the irradiating device is not the recording device and the dose was accumulated on a single device. | $DeviceProcedureRole = EV (113859, DCM, "Irradiating Device")
15 | > | CONTAINS | CODE | EV (128750, DCM, "Equipment Landmark") | 1 | U |  | EV (128751, DCM, "Center of Table Head")
16 | >> | HAS PROPERTIES | NUM | EV (128752, DCM, "Equipment Landmark X Position") | 1 | M |  | UNITS = EV (mm, UCUM, "mm")
17 | >> | HAS PROPERTIES | NUM | EV (128753, DCM, "Equipment Landmark Z Position") | 1 | M |  | UNITS = EV (mm, UCUM, "mm")
18 | > | CONTAINS | CONTAINER | EV (128754, DCM, "Patient Location Fiducial") | 1-n | U |  |
19 | >> | CONTAINS | INCLUDE | DTID 400 “Reference Location” | 1 | M |  |
20 | >> | HAS PROPERTIES | NUM | EV (128756, DCM, "Equipment Landmark to Patient Fiducial Z Distance") | 1 | M |  | UNITS = EV (mm, UCUM, "mm")

Content Item Descriptions

Row 5 | Date that the calibration of the equipment's dose indicators was performed
Row 6 | Typically a value provided by the medical physicist. The recorded dose or dose area product values in this report can be multiplied by this factor to obtain estimated real-world values. ### Note It is important that this value must not be applied to the measured values before storing them in the report.
Row 7 | Value range from 0 to 100 percent. Uncertainty of the 'actual' value expressed as+/- of the mean.
Row 8 | Identifies Individual or organization responsible for calibration
Row 9 | Describes calibration protocol according to equipment standards or local guidelines.
Row 14 | The device that produced the irradiation accumulated in this Template. I.e., the X-Ray source. This is not required to be present if the information is the same as that already recorded in TID 1004 “Device Observer Identifying Attributes” encoded via the inclusion of TID 1002 “Observer Context” in TID 10001 “Projection X-Ray Radiation Dose” Row 5, which in turn may be absent if identical to the content in the Enhanced General Equipment Module, or if more than one device produced the accumulated irradiation.
Rows 16 and 17 | These coordinates relate a visible landmark on the X-Ray table to the Table Reference Point that is arbitrarily defined by the manufacturer and not necessarily visible to the operator. The Equipment Landmark Y Position is not recorded since it is, by definition, in the plane of the table as is the origin of the Table Coordinate System so the value would always be zero.
Row 19 | In many instances, the values will be either: EV (128772, DCM, "Reference Basis") = (T-D1120, SRT, "Vertex of Head") with EV (128773, DCM, "Reference Geometry") = (128120, DCM, "Plane through Superior Extent"), or EV (128772, DCM, "Reference Basis") = (T-D9700, SRT, "Foot") with EV (128773, DCM, "Reference Geometry") = (128121, DCM, "Plane through Inferior Extent")
Row 20 | This distance (likely recorded by the operator) locates the patient with respect to an X-Ray table landmark. The patient is assumed to be centered in the left-right axis of the X-Ray table.

### TID 10003 Irradiation Event X-Ray Data

This Template conveys the dose and equipment parameters of a single irradiation event.

The Template and requirements are structured to consider equipment with various levels of integration between the components (X-Ray Source, Plate or Detector, and Gantry/Table) of the equipment.

An irradiation event is the loading of X-Ray equipment caused by a single continuous actuation of the equipment's irradiation switch, from the start of the loading time of the first pulse until the loading time trailing edge of the final pulse. The irradiation event is the "smallest" information entity to be recorded in the realm of Radiation Dose reporting. Individual Irradiation Events are described by a set of accompanying physical parameters that are sufficient to understand the "quality" of irradiation that is being applied. This set of parameters may be different for the various types of equipment that are able to create irradiation events. Any automatic on-off switching of the irradiation source during the event shall not be treated as separate events, rather the event includes the time between start and stop of irradiation as triggered by the user. E.g., a pulsed fluoro X-Ray acquisition shall be treated as a single irradiation event.

As described in Section 6.2.4, measurement concepts may be post-coordinated, even though not explicitly specified in the Template. In particular, post-coordination using modifier concept (121401, DCM, "Derivation"), with modifier values drawn from CID 10009 “Measured/Calculated” would be appropriate to encode indications of measured or of calculated values.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 10003. Irradiation Event X-Ray Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113706, DCM, "Irradiation Event X-Ray Data") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (113764, DCM, "Acquisition Plane") | 1 | M |  | DCID 10003 “Equipment Plane Identification”
3 | > | CONTAINS | UIDREF | EV (113769, DCM, "Irradiation Event UID") | 1 | M |  |
4 | > | CONTAINS | TEXT | EV (113605, DCM, "Irradiation Event Label") | 1 | U |  |
5 | >> | HAS CONCEPT MOD | CODE | EV (113606, DCM, "Label Type") | 1 | MC | IF the value of Row 4 is the value of an Attribute in the images. | DCID 10022 “Label Types”
6 | > | CONTAINS | DATETIME | DT (111526, DCM, "DateTime Started") | 1 | M |  |
7 | > | CONTAINS | CODE | EV (113721, DCM, "Irradiation Event Type") | 1 | M |  | DCID 10002 “Irradiation Event Types”
8 | > | CONTAINS | TEXT | EV (125203, DCM, "Acquisition Protocol") | 1 | U |  |
11 | > | CONTAINS | CODE | EV (111031, DCM, "Image View") | 1 | U |  | DCID 4010 “DX View” DCID 4014 “View for Mammography”
12 | >> | HAS CONCEPT MOD | CODE | EV (111032, DCM, "Image View Modifier") | 1-n | U |  | DCID 4011 “DX View Modifier” DCID 4015 “View Modifier for Mammography”
13 | >> | CONTAINS | CODE | EV (113946, DCM, "Projection Eponymous Name") | 1 | U |  | DCID 4012 “Projection Eponymous Name”
14 | > | CONTAINS | CODE | EV (113745, DCM, "Patient Table Relationship") | 1 | U |  | DCID 21 “Patient Equipment Relationship”
15 | > | CONTAINS | CODE | EV (113743, DCM, "Patient Orientation") | 1 | U |  | DCID 19 “Patient Orientation”
16 | >> | HAS CONCEPT MOD | CODE | EV (113744, DCM, "Patient Orientation Modifier") | 1 | M |  | DCID 20 “Patient Orientation Modifier”
17 | > | CONTAINS | CODE | EV (123014, DCM, "Target Region") | 1 | M |  | DCID 4031 “Common Anatomic Regions”
17b | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | UC | If anatomy is bi-lateral | DCID 244 “Laterality”
18 | > | CONTAINS | NUM | EV (122130, DCM, "Dose Area Product") | 1 | MC | IFF TID (10001) Row 2 = (113704, DCM, "Projection X-Ray") | UNITS = EV (Gy.m2, UCUM, "Gy.m2")
19 | > | CONTAINS | NUM | EV (111634, DCM, "Half Value Layer") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
20 | > | CONTAINS | NUM | EV (111638, DCM, "Patient Equivalent Thickness") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
21 | > | CONTAINS | NUM | EV (111636, DCM, "Entrance Exposure at RP") | 1 | MC | IF TID (10001) Row 2 = (P5-40010, SRT, "Mammography") and (TID (10001) Row 9 is absent or value is (R-0038D, SRT, "Yes")) and (TID (10001) Row 10 is absent or value is (R-0038D, SRT, "Yes")) | UNITS = EV (mGy, UCUM, "mGy")
22 | > | CONTAINS | TEXT | EV (113780, DCM, "Reference Point Definition") | 1 | MC | IF Row 21 is present and Row 23 is not present |
23 | > | CONTAINS | CODE | EV (113780, DCM, "Reference Point Definition") | 1 | MC | IF Row 21 is present and Row 22 is not present | DCID 10025 “Radiation Dose Reference Points”
24 | > | CONTAINS | INCLUDE | DTID 4007 “Mammography CAD Breast Composition” | 1 | U |  |
25 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
26 | > | CONTAINS | INCLUDE | DTID 1020 “Person Participant” | 1-n | U |  | $PersonProcedureRole = EV (113851, DCM, "Irradiation Administering")
27 | > | CONTAINS | INCLUDE | DTID 10003A “Irradiation Event X-Ray Detector Data” | 1 | MC | IFF TID (10001) Row 8 is absent or has a value of (R-0038D, SRT, "Yes") |
28 | > | CONTAINS | INCLUDE | DTID 10003B “Irradiation Event X-Ray Source Data” | 1 | MC | IFF TID (10001) Row 9 is absent or has a value of (R-0038D, SRT, "Yes") |
29 | > | CONTAINS | INCLUDE | DTID 10003C “Irradiation Event X-Ray Mechanical Data” | 1 | MC | IFF TID (10001) Row 10 is absent or has a value of (R-0038D, SRT, "Yes") |

Content Item Descriptions

Row 3 | If the image generating entity does not assign a DICOM UID to the irradiation event (e.g., for non-digital imaging equipment), the application generating this report shall assign a UID. In the case of non-integrated cassette-based equipment, a standalone Detector will generate UIDs for the Events it observes. If the X-Ray Source component of the equipment also reports information, it too will generate UIDs for the Events it creates. A downstream system (e.g., a workstation or the Dose Information Reporter itself) may combine the two reports into a composite report, and match up the events based on details such as the time information, and use the UIDs of the X-Ray Source.
Row 6 | Provide DateTime the application of X-Ray started. This shall correspond to the start of the first irradiation in the Irradiation Event, which defines the starting point for the calculation of Row 36 "Irradiation Duration".
Row 17 | The target region is the anatomy exposed.
Row 17b | Previously, a CODE content item (T-D0005, SRT, "Anatomical structure") along with CODE concept modifier (G-C171, SRT, "Laterality") were used to identify bilateral anatomy. This duplicated the function of Row 17 and was retired. See PS3.16 2017c.
Row 21 | A text definition of the Reference Point (RP) used for RP-related dose values.
Row 22 | A coded definition of the Reference Point (RP) used for RP-related dose values
Row 26 | People responsible for the administration of the radiation reported in the irradiation event. May include values that would appear in Performing Physicians' Name (0008,1050), Performing Physician Identification Sequence (0008,1052), Operators' Name (0008,1070) and/or Operator Identification Sequence (0008,1072).

### TID 10003A Irradiation Event X-Ray Detector Data

This Template contains data that is expected to be available to the X-Ray detector or plate reader component of the equipment.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 10003A. Irradiation Event X-Ray Detector Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | EV (113845, DCM, "Exposure Index") | 1 | MC | IF the value is displayable to the X-Ray system operator. | UNITS = EV (1, UCUM, "no units")
2 |  |  | NUM | EV (113846, DCM, "Target Exposure Index") | 1 | MC | IF the value is displayable to the X-Ray system operator. | UNITS = EV (1, UCUM, "no units")
3 |  |  | NUM | EV (113847, DCM, "Deviation Index") | 1 | MC | IF the value is displayable to the X-Ray system operator. | UNITS = EV (1, UCUM, "no units")
4 |  |  | INCLUDE | DTID 1021 “Device Participant” | 1 | U |  | $DeviceProcedureRole = EV (113942, DCM, "X-Ray Reading Device")
5 |  |  | IMAGE | EV (113795, DCM, "Acquired Image") | 1-n | MC | IFF Image Object is created for this irradiation event |

Content Item Descriptions

Row 4 | The device that read the detector of this Irradiation Event. E.g., the CR Plate Reader.
Row 5 | Reference to Image instances created during this event, if any. The UID reference(s) provided here shall be the values at the time the images were initially created. (Note that image UIDs may be changed as the images are managed over a long term.)

### TID 10003B Irradiation Event X-Ray Source Data

This Template contains data that is expected to be available to the X-Ray source component of the equipment.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 10003B. Irradiation Event X-Ray Source Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | EV (113738, DCM, "Dose (RP) ") | 1 | MC | IF TID (10001) Row 2 = (113704, DCM, "Projection X-Ray") AND any of the values of TID (10001) Row 18 are not (113858, DCM, "MPPS Content") | UNITS = EV (Gy, UCUM, "Gy")
2 |  |  | TEXT | EV (113780, DCM, "Reference Point Definition") | 1 | MC | IF Row 1 is present and Row 3 is not present |
3 |  |  | CODE | EV (113780, DCM, "Reference Point Definition") | 1 | MC | IF Row 1 is present and Row 2 is not present | DCID 10025 “Radiation Dose Reference Points”
4 |  |  | NUM | EV (111631, DCM, "Average Glandular Dose") | 1 | MC | IFF TID (10001) Row 2 = (P5-40010, SRT, "Mammography") | UNITS = EV (mGy, UCUM, "mGy")
5 |  |  | CODE | EV (113732, DCM, "Fluoro Mode") | 1 | UC | IFF TID (10003) Row 7 value = (P5-06000, SRT, "Fluoroscopy") | DCID 10004 “Fluoro Modes”
6 |  |  | NUM | EV (113791, DCM, "Pulse Rate") | 1 | MC | IFF Row 5 value = (113631, DCM, "Pulsed") | UNITS = EV ({pulse}/s, UCUM, "pulse/s")
7 |  |  | NUM | EV (113768, DCM, "Number of Pulses") | 1 | MC | IFF Row 5 is not present or Row 5 is present and equals (113631, DCM, "Pulsed") | UNITS = EV (1, UCUM, "no units")
8 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | MC | IFF count of pulses in Row 7 is estimated | EV (R-10260, SRT, "Estimated")
9 |  |  | NUM | EV (113793, DCM, "Pulse Width") | 1-n | U |  | UNITS = EV (ms, UCUM, "ms")
10 |  |  | NUM | EV (113742, DCM, "Irradiation Duration") | 1 | U |  | UNITS = EV (s, UCUM, "s")
11 |  |  | NUM | EV (113733, DCM, "KVP") | 1-n | M |  | UNITS = EV (kV, UCUM, "kV")
12 |  |  | NUM | EV (113734, DCM, "X-Ray Tube Current") | 1-n | MC | IF Row 15 is not present | UNITS = EV (mA, UCUM, "mA")
13 |  |  | NUM | EV (113767, DCM, "Average X-Ray Tube Current") | 1 | U |  | UNITS = EV (mA, UCUM, "mA")
14 |  |  | NUM | EV (113824, DCM, "Exposure Time") | 1 | MC | IF Row 15 is not present | UNITS = EV (ms, UCUM, "ms")
15 |  |  | NUM | EV (113736, DCM, "Exposure") | 1-n | MC | IF Row 12 or 14 is not present | UNITS = EV (uA.s, UCUM, "uA.s")
16 |  |  | NUM | EV (113766, DCM, "Focal Spot Size") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
17 |  |  | CODE | EV (111632, DCM, "Anode Target Material") | 1 | U |  | DCID 10016 “Anode Target Material”
18 |  |  | CONTAINER | EV (113771, DCM, "X-Ray Filters") | 1-n | U |  |
19 | > | CONTAINS | CODE | EV (113772, DCM, "X-Ray Filter Type") | 1 | U |  | DCID 10007 “X-Ray Filter Types”
20 | > | CONTAINS | CODE | EV (113757, DCM, "X-Ray Filter Material") | 1 | U |  | DCID 10006 “X-Ray Filter Materials”
21 | > | CONTAINS | NUM | EV (113758, DCM, "X-Ray Filter Thickness Minimum") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
22 | > | CONTAINS | NUM | EV (113773, DCM, "X-Ray Filter Thickness Maximum") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
23 |  |  | NUM | EV (113790, DCM, "Collimated Field Area") | 1 | U |  | UNITS = EV (m2, UCUM, "m2")
24 |  |  | NUM | EV (113788, DCM, "Collimated Field Height") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
25 |  |  | NUM | EV (113789, DCM, "Collimated Field Width") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
26 |  |  | CODE | EV (111635, DCM, "X-Ray Grid") | 1-n | U |  | DCID 10017 “X-Ray Grid”
27 |  |  | INCLUDE | DTID 1021 “Device Participant” | 1 | MC | Required if the irradiating device is not the recording device. | $DeviceProcedureRole = EV (113859, DCM, "Irradiating Device")

Content Item Descriptions

Row 1 | Dose applied by this irradiation event, relative to defined reference point.
Row 7 | If a precise count of pulses is not available, an estimated number shall be provided, and the Row 8 Concept Modifier shall indicate "Estimated"
Row 9 | Pulse width as measured/recorded by the system, either as a single total value, or as multiple values. If multiple values are provided, their number shall match the value in Row 7 "Number of Pulses".
Row 11 | KVP value as measured/recorded by system, either as a single mean value, or as multiple values. If multiple values are provided, their number shall match the value in Row 7 "Number of Pulses".
Row 12 | Tube current as measured/recorded by system, either as a single mean value, or as multiple values. If multiple values are provided, their number shall match the value in Row 7 "Number of Pulses".
Row 14 | Exposure time as measured/recorded by the system.
Row 15 | Exposure as measured/recorded by system, either as a single total value, or as multiple values. If multiple values are provided, their number shall match the value in Row 7 "Number of Pulses". The Exposure will be affected by the shape of the pulse and other factors, and may not be a simple multiplication of tube current and exposure time.
Row 18 | If one or more Filter(s) were applied during this irradiation event
Row 23 | Collimated area at the receptor plane.
Row 27 | The device that produced the irradiation in this Irradiation Event. I.e., the X-Ray source. This is not required to be present if the information is the same as that already recorded in TID 1004 “Device Observer Identifying Attributes” encoded via the inclusion of TID 1002 “Observer Context” in TID 10001 “Projection X-Ray Radiation Dose” Row 5, which in turn may be absent if identical to the content in the Enhanced General Equipment Module.

### TID 10003C Irradiation Event X-Ray Mechanical Data

This Template contains data that is expected to be available to the gantry or mechanical component of the equipment.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 10003C. Irradiation Event X-Ray Mechanical Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (113956, DCM, "CR/DR Mechanical Configuration") | 1 | U |  | DCID 10031 “CR/DR Mechanical Configuration”
2 |  |  | NUM | EV (112011, DCM, "Positioner Primary Angle") | 1 | UC | XOR Row 6 | UNITS = EV (deg, UCUM, "deg")
3 |  |  | NUM | EV (112012, DCM, "Positioner Secondary Angle") | 1 | UC | XOR Row 6 | UNITS = EV (deg, UCUM, "deg")
4 |  |  | NUM | EV (113739, DCM, "Positioner Primary End Angle") | 1 | UC | IFF TID (10003) Row 7 value = (113613, DCM, "Rotational Acquisition") | UNITS = EV (deg, UCUM, "deg")
5 |  |  | NUM | EV (113740, DCM, "Positioner Secondary End Angle") | 1 | UC | IFF TID (10003) Row 7 value = (113613, DCM, "Rotational Acquisition") | UNITS = EV (deg, UCUM, "deg")
6 |  |  | NUM | EV (113770, DCM, "Column Angulation") | 1 | UC | XOR Rows 2, 3 | UNITS = EV (deg, UCUM, "deg")
7 |  |  | NUM | EV (113754, DCM, "Table Head Tilt Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
8 |  |  | NUM | EV (113755, DCM, "Table Horizontal Rotation Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
9 |  |  | NUM | EV (113756, DCM, "Table Cradle Tilt Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
10 |  |  | NUM | EV (111633, DCM, "Compression Thickness") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
11 |  |  | NUM | DCID 10008 “Dose Related Distance Measurements” | 1-n | U |  | UNITS = EV (mm, UCUM, "mm")
12 |  |  | NUM | EV (128757, DCM, "Positioner Isocenter Primary Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
13 |  |  | NUM | EV (128758, DCM, "Positioner Isocenter Secondary Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
14 |  |  | NUM | EV (128759, DCM, "Positioner Isocenter Detector Rotation Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
15 |  |  | NUM | EV (128760, DCM, "Positioner Isocenter Primary End Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
16 |  |  | NUM | EV (128761, DCM, "Positioner Isocenter Secondary End Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
17 |  |  | NUM | EV (128762, DCM, "Positioner Isocenter Detector Rotation End Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
18 |  |  | NUM | EV (128763, DCM, "Table Head Tilt End Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
19 |  |  | NUM | EV (128764, DCM, "Table Horizontal Rotation End Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
20 |  |  | NUM | EV (128765, DCM, "Table Cradle Tilt End Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")

Content Item Descriptions

Row 2 | Angle in patient's "equatorial" plane (LAO to RAO). For dynamically changing angle during the event, the start value shall be provided. Equivalent to (0018,1510) in an image instance.
Row 3 | Angle in patient's "sagittal" plane (CRAN to CAUD). For dynamically changing angle during the event, the start value shall be provided. Equivalent to (0018,1511) in an image instance.
Row 4 | In case of motion during irradiation event, Positioner Primary ending angle
Row 5 | In case of motion during irradiation event., Positioner Secondary ending angle
Row 6 | Column device Angle in equipment based coordinates
Rows 12 to 20 | Refer to the definitions of the X-Ray Isocenter Reference System (“X-Ray Isocenter Reference System Macro” in PS3.3 ).

### TID 10004 Accumulated Fluoroscopy and Acquisition Projection X-Ray Dose

This general Template provides detailed information on projection X-Ray dose value accumulations over several irradiation events from the same equipment (typically a study or a performed procedure step).

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 10004. Accumulated Fluoroscopy and Acquisition Projection X-Ray Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | EV (113726, DCM, "Fluoro Dose Area Product Total") | 1 | MC | IFF TID (10003) Row 7 value = (P5-06000, SRT, "Fluoroscopy") for at least one irradiation event | UNITS = EV (Gy.m2, UCUM, "Gy.m2")
2 |  |  | NUM | EV (113728, DCM, "Fluoro Dose (RP) Total") | 1 | MC | IFF TID (10003) Row 7 value = (P5-06000, SRT, "Fluoroscopy") for at least one irradiation event AND any of the values of TID (10001) Row 18 are not (113858, DCM, "MPPS Content"). | UNITS = EV (Gy, UCUM, "Gy")
3 |  |  | NUM | EV (113730, DCM, "Total Fluoro Time") | 1 | MC | IFF TID (10003) Row 7 value = (P5-06000, SRT, "Fluoroscopy") for at least one irradiation event. | UNITS = EV (s, UCUM, "s")
4 |  |  | NUM | EV (113727, DCM, "Acquisition Dose Area Product Total") | 1 | MC | IF any of the values of TID (10001) Row 18 are not (113858, DCM, "MPPS Content"). | UNITS = EV (Gy.m2, UCUM, "Gy.m2")
5 |  |  | NUM | EV (113729, DCM, "Acquisition Dose (RP) Total") | 1 | MC | IF any of the values of TID (10001) Row 18 are not (113858, DCM, "MPPS Content"). | UNITS = EV (Gy, UCUM, "Gy")
6 |  |  | NUM | EV (113855, DCM, "Total Acquisition Time") | 1 | MC | IF any of the values of TID (10001) Row 18 are not (113858, DCM, "MPPS Content"). | UNITS = EV (s, UCUM, "s")

Content Item Descriptions

Rows 1-3 | Fluoroscopic component only
Row 3 | Total clock time of Fluoroscopy accumulated over the defined scope of accumulation (i.e., the sum of the Irradiation Duration values for accumulated fluoroscopy irradiation events)
Rows 4-6 | Acquisition component only
Row 6 | Total clock time of acquisitions accumulated over the defined scope of accumulation (i.e., the sum of the Irradiation Duration values for accumulated acquisition irradiation events)

### TID 10005 Accumulated Mammography X-Ray Dose

This modality specific Template provides detailed information on breast imaging projection X-Ray dose value accumulations over several irradiation events from the same equipment (typically a study or a performed procedure step).

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 10005. Accumulated Mammography X-Ray Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | EV (111637, DCM, "Accumulated Average Glandular Dose") | 1-2 | M |  | UNITS = EV (mGy, UCUM, "mGy")
2 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | DCID 6022 “Side”

### TID 10006 Accumulated Cassette-based Projection Radiography Dose

This Template provides information on Projection Radiography dose values accumulated on Cassette-based systems over one or more irradiation events (typically a study or a performed procedure step) from the same equipment.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 10006. Accumulated Cassette-Based Projection Radiography Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (113947, DCM, "Detector Type") | 1 | MC | IF TID (10001) Row 8 is absent or value is (R-0038D, SRT, "Yes") | DCID 10030 “Detector Types”
2 |  |  | NUM | EV (113731, DCM, "Total Number of Radiographic Frames") | 1 | MC | IF TID (10001) Row 8 is absent or value is (R-0038D, SRT, "Yes") | UNITS = EV (1, UCUM, "no units")

Content Item Descriptions

Row 2 | The number of radiographic frames recorded by the X-Ray detector or the number of exposures recorded by the X-Ray source, whichever is known to be greater.

### TID 10007 Accumulated Total Projection Radiography Dose

This Template provides information on total Projection Radiography dose values accumulated on Integrated or combined fluoroscopy/acquisition systems over one or more irradiation events (typically a study or a performed procedure step) from the same equipment.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 10007. Accumulated Total Projection Radiography Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | EV (113722, DCM, "Dose Area Product Total") | 1 | M |  | UNITS = EV (Gy.m2, UCUM, "Gy.m2")
2 |  |  | NUM | EV (113725, DCM, "Dose (RP) Total") | 1 | MC | IF TID (10001) Row 4 = (113958, DCM, "Integrated Projection Radiography System") or any of the values of TID (10001) Row 18 are not (113858, DCM, "MPPS Content"). | UNITS = EV (Gy, UCUM, "Gy")
3 |  |  | NUM | EV (113737, DCM, "Distance Source to Reference Point") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
4 |  |  | NUM | EV (113731, DCM, "Total Number of Radiographic Frames") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
5 |  |  | CODE | EV (113780, DCM, "Reference Point Definition") | 1 | MC | IF any of (113725, DCM, "Dose (RP) Total"), (113728, DCM, "Fluoro Dose (RP) Total") or (113729, DCM, "Acquisition Dose (RP) Total") are present, and Row 6 is not present. | DCID 10025 “Radiation Dose Reference Points”
6 |  |  | TEXT | EV (113780, DCM, "Reference Point Definition") | 1 | MC | IF any of (113725, DCM, "Dose (RP) Total"), (113728, DCM, "Fluoro Dose (RP) Total") or (113729, DCM, "Acquisition Dose (RP) Total") are present, and Row 5 is not present. |

Content Item Descriptions

Row 1 | Accumulated Dose Area Product
Row 2 | Accumulated dose relative to reference point.
Row 3 | A single value for Radiography systems calculating reference point dose based on fixed distance.
Row 5 | A coded definition of the Reference Point (RP) used for RP-related dose values.
Row 6 | A text definition of the Reference Point (RP) used for RP-related dose values.

## CT Radiation Dose SR IOD Templates

The Templates that comprise the CT Radiation Dose SR are interconnected as in Figure A-15.

Figure A-15. CT Radiation Dose SR IOD Template Structure

### TID 10011 CT Radiation Dose

This Template defines a container (the root) with subsidiary Content Items, each of which corresponds to a single CT X-Ray irradiation event entry. There is a defined recording observer (the system or person responsible for recording the log, generally the system). Accumulated values shall be kept for a whole Study or at least a part of a Study, if the Study is divided in the workflow of the examination, or a performed procedure step. Multiple CT Radiation Dose objects may be created for one Study.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 10011. CT Radiation Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113701, DCM, "X-Ray Radiation Dose Report") | 1 | M |  | Root node
1b | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | EV (P5-08000, SRT, "Computed Tomography X-Ray")
3 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E8, SRT, "Has Intent") | 1 | M |  | DCID 3629 “Procedure Intent”
4 | > |  | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
5 | > | HAS OBS CONTEXT | DATETIME | EV (113809, DCM, "Start of X-Ray Irradiation") | 1 | M |  |
6 | > | HAS OBS CONTEXT | DATETIME | EV (113810, DCM, "End of X-Ray Irradiation") | 1 | M |  |
7 | > | HAS OBS CONTEXT | CODE | EV (113705, DCM, "Scope of Accumulation") | 1 | M |  | DCID 10000 “Scope of Accumulation”
8 | >> | HAS PROPERTIES | UIDREF | DCID 10001 “UID Types” | 1 | M |  |
9 | > | CONTAINS | INCLUDE | DTID 10012 “CT Accumulated Dose Data” | 1 | M |  |
10 | > | CONTAINS | INCLUDE | DTID 10013 “CT Irradiation Event Data” | 1-n | M |  |
11 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
12 | > | CONTAINS | CODE | EV (113854, DCM, "Source of Dose Information") | 1-n | M |  | DCID 10021 “Source of CT Dose Information”
13 | > | CONTAINS | INCLUDE | DTID 1020 “Person Participant” | 1 | U |  | $PersonProcedureRole = EV (113850, DCM, "Irradiation Authorizing")

Content Item Descriptions

Row 4 | The observer context may include both a Person Observer identification, as well as the identity of the equipment providing the values for the irradiation event (Device Observer identification), if not inherited.
Row 5 | Start, DateTime of the first CT Irradiation Event of the accumulation
Row 6 | End, DateTime of the last CT Irradiation Event of the accumulation
Row 12 | The primary source of information from which this dose object was constructed.
Row 13 | The physician responsible for determining that the irradiating procedure was appropriate for the indications. The value may come from Requesting Physician (0032,1032), Requesting Physician Identification Sequence (0032,1031) or somewhere else based on hospital policies.

### TID 10012 CT Accumulated Dose Data

This general Template provides detailed information on CT X-Ray dose value accumulations over several irradiation events from the same equipment and over the scope of accumulation specified for the report (typically a Study or a Performed Procedure Step).

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 10012. CT Accumulated Dose Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113811, DCM, "CT Accumulated Dose Data") | 1 | M |  |
2 | > | CONTAINS | NUM | EV (113812, DCM, "Total Number of Irradiation Events") | 1 | M |  | UNITS = EV ({events}, UCUM, "events")
3 | > | CONTAINS | NUM | EV (113813, DCM, "CT Dose Length Product Total") | 1 | M |  | UNITS = EV (mGy.cm, UCUM, "mGy.cm")
4 | > | CONTAINS | NUM | EV (113814, DCM, "CT Effective Dose Total") | 1 | U |  | UNITS = EV (mSv, UCUM, "mSv")
5 | >> | HAS PROPERTIES | TEXT | EV (121406, DCM, "Reference Authority") | 1 | MC | XOR row 6 |
6 | >> | HAS PROPERTIES | CODE | EV (121406, DCM, "Reference Authority") | 1 | MC | XOR row 5 | DCID 10015 “CT Dose Reference Authorities”
7 | >> | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | M |  | DCID 10011 “Effective Dose Evaluation Method”
8 | >> | HAS PROPERTIES | TEXT | EV (113815, DCM, "Patient Model") | 1 | MC | IF the value of row 7 equals (113800, DCM, "DLP to E conversion via MC computation") or equals (113801, DCM, "CTDIfreeair to E conversion via MC computation") |
9 | >> | HAS PROPERTIES | CONTAINER | EV (113816, DCM, "Condition Effective Dose measured") | 1 | MC | IF the value of row 7 equals (113802, DCM, "DLP to E conversion via measurement") or equals (113803, DCM, "CTDIfreeair to E conversion via measurement") |
10 | >>> | CONTAINS | TEXT | EV (113817, DCM, "Effective Dose Phantom Type") | 1 | M |  |
11 | >>> | CONTAINS | TEXT | EV (113818, DCM, "Dosimeter Type") | 1 | M |  |
12 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
13 | > | CONTAINS | INCLUDE | DTID 1021 “Device Participant” | 1 | MC | Required if the irradiating device is not the recording device and the dose was accumulated on a single device. | $DeviceProcedureRole = EV (113859, DCM, "Irradiating Device")

Content Item Descriptions

Rows 5, 6 | Total Number of CT irradiation events. A CT irradiation event is one continuous irradiation procedure and is defined through consistent acquisition parameters. In the case of dose modulation the calculations are based on the effective parameters (e.g., the effective mA recorded in the Mean X-Ray Tube Current), and these acquisition parameters are consistent.
Rows 5, 6 | The Dose Length Product (DLP) is calculated for every irradiation event. The Dose Length Product Total is the sum of the DLP values. The calculation is based on the CTDIvol result of each irradiation event.
Rows 5, 6 | Effective dose (E, in units of mSv) evaluated as a total over the scope is defined in Row 6 of Template TID 10011 “CT Radiation Dose”. Effective dose is defined by the reference in Rows 5 or 6 of this Template. It may be calculated from a product of DLP and an 'Effective Dose Conversion Factor' (E/DLP). Or it may be calculated from a product of the Mean CTDIfreeair and the ratio E/CTDIfreeair. The ratios E/DLP or E/CTDIfreeair may be evaluated either from computer simulations applying Monte Carlo (MC) sampling techniques or from dosimetric measurements in an anthropomorphic phantom, e.g., the Alderson-Rando phantom.. The specific method used is identified in Rows 7 through 11.
Row 5 - 6 | Reference of the base publication defining the Effective Dose, either as a coded value, or a textual bibliographic reference. ICRP Publications shall be referenced using their assigned coded values.
Row 7 | Description of the method used for Effective Dose evaluations.
Row 8 | Description of the reference-patient mathematical or computational model used when Effective Dose is derived via Monte Carlo simulations of radiation transport in such models. Examples of publications that specify particular reference patient models are NUREG/CR-1159, ORNL/NUREG/TM-367 (1980); NRPB-R186 (1985); GSF-Bericht S-885 (1986); Fill et al., Health Physics Vol. 86 (3): 253-272 (2004).
Row 9 | Description of the condition Effective Dose measured
Row 10 | Type of Effective Dose phantom used, e.g., Alderson-Rando
Row 11 | Type of dosimeter used, e.g., TLD (Thermo Luminescence Dosimeter)
Row 13 | The device that produced the irradiation accumulated in this Template. I.e., the CT Scanner. This is not required to be present if the information is the same as that already recorded in TID 1004 “Device Observer Identifying Attributes” encoded via the inclusion of TID 1002 “Observer Context” in TID 10011 “CT Radiation Dose” Row 4, which in turn may be absent if identical to the content in the Enhanced General Equipment Module, or if more than one device produced the accumulated irradiation.

### TID 10013 CT Irradiation Event Data

This Template conveys the dose and equipment parameters of a single irradiation event.

A CT irradiation event is the loading of X-Ray equipment caused by a single continuous actuation of the equipment's irradiation switch, from the start of the loading time of the first pulse until the loading time trailing edge of the final pulse. Any on-off switching of the radiation source during the event shall not be treated as separate events; rather the event includes the time between start and stop of radiation as triggered by the user, e.g., a single sequence of scanning comprised of multiple slices acquired with successive tube rotations and table increments shall be treated as a single irradiation event. Depending on the examination workflow and the anatomical target region the CT irradiation event data may split into multiple instances of this Template for better dose estimation. The irradiation event is the "smallest" information entity to be recorded in the realm of Radiation Dose reporting. Individual Irradiation Events are described by a set of accompanying physical parameters that are sufficient to understand the "quality" of irradiation that is being applied. This set of parameters may be different for the various types of equipment that are able to create irradiation events.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 10013. CT Irradiation Event Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113819, DCM, "CT Acquisition") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (125203, DCM, "Acquisition Protocol") | 1 | U |  |
3 | > | CONTAINS | CODE | EV (123014, DCM, "Target Region") | 1 | M |  | DCID 4030 “CT, MR and PET Anatomy Imaged”
4 | > | CONTAINS | CODE | EV (113820, DCM, "CT Acquisition Type") | 1 | M |  | DCID 10013 “CT Acquisition Type”
4b | >> | CONTAINS | CODE | EV (113961, DCM, "Reconstruction Algorithm") | 1-n | U |  | DCID 10033 “CT Reconstruction Algorithm”
5 | > | CONTAINS | CODE | EV (G-C32C, SRT, "Procedure Context") | 1 | U |  | DCID 10014 “Contrast Imaging Technique”
6 | > | CONTAINS | UIDREF | EV (113769, DCM, "Irradiation Event UID") | 1 | M |  |
6b | > | CONTAINS | TEXT | EV (113605, DCM, "Irradiation Event Label") | 1 | U |  |
6c | >> | HAS CONCEPT MOD | CODE | EV (113606, DCM, "Label Type") | 1 | MC | IF the value of Row 6b is the value of an Attribute in the images. | DCID 10022 “Label Types”
6d | > | CONTAINS | CODE | EV (128551, DCM, "Is Repeated Acquisition") | 1 | U |  | DCID 230 “Yes-No”
6e | >> | HAS CONCEPT MOD | CODE | EV (128552, DCM, "Reason for Repeating Acquisition") | 1 | M |  | DCID 10034 “Reason for Repeating Acquisition”
7 | > | CONTAINS | CONTAINER | EV (113822, DCM, "CT Acquisition Parameters") | 1 | M |  |
8 | >> | CONTAINS | NUM | EV (113824, DCM, "Exposure Time") | 1 | M |  | UNITS = EV (s, UCUM, "s")
9 | >> | CONTAINS | INCLUDE | DTID 10014 “Scanning Length” | 1 | M |  |
10 | >> | CONTAINS | NUM | EV (113826, DCM, "Nominal Single Collimation Width") | 1 | M |  | UNITS = EV (mm, UCUM, "mm")
11 | >> | CONTAINS | NUM | EV (113827, DCM, "Nominal Total Collimation Width") | 1 | M |  | UNITS = EV (mm, UCUM, "mm")
12 | >> | CONTAINS | NUM | EV (113828, DCM, "Pitch Factor") | 1 | MC | IF row 4 equals (P5-08001, SRT, "Spiral Acquisition") or equals (113804, DCM, "Sequenced Acquisition") | UNITS = EV ({ratio}, UCUM, "ratio")
13 | >> | CONTAINS | NUM | EV (113823, DCM, "Number of X-Ray Sources") | 1 | M |  | UNITS = EV ({X-Ray sources}, UCUM, "X-Ray sources")
14 | >> | CONTAINS | CONTAINER | EV (113831, DCM, "CT X-Ray Source Parameters") | 1-n | M |  |
15 | >>> | CONTAINS | TEXT | EV (113832, DCM, "Identification of the X-Ray Source") | 1 | M |  |
16 | >>> | CONTAINS | NUM | EV (113733, DCM, "KVP") | 1 | M |  | UNITS = EV (kV, UCUM, "kV")
17 | >>> | CONTAINS | NUM | EV (113833, DCM, "Maximum X-Ray Tube Current") | 1 | M |  | UNITS = EV (mA, UCUM, "mA")
18 | >>> | CONTAINS | NUM | EV (113734, DCM, "X-Ray Tube Current") | 1 | M |  | UNITS = EV (mA, UCUM, "mA")
19 | >>> | CONTAINS | NUM | EV (113834, DCM, "Exposure Time per Rotation") | 1 | MC | IF row 4 does not equal (113805, DCM, "Constant Angle Acquisition") | UNITS = EV (s, UCUM, "s")
20 | >>> | CONTAINS | NUM | EV (113821, DCM, "X-Ray Filter Aluminum Equivalent") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
21 | > | CONTAINS | CONTAINER | EV (113829, DCM, "CT Dose") | 1 | MC | IF row 4 does not equal (113805, DCM, "Constant Angle Acquisition") |
22 | >> | CONTAINS | NUM | EV (113830, DCM, "Mean CTDIvol") | 1 | M |  | UNITS = EV (mGy, UCUM, "mGy")
23 | >> | CONTAINS | CODE | EV (113835, DCM, "CTDIw Phantom Type") | 1 | M |  | DCID 4052 “Phantom Devices”
24 | >> | CONTAINS | NUM | EV (113836, DCM, "CTDIfreeair Calculation Factor") | 1 | U |  | UNITS = EV (mGy/mA.s, UCUM, "mGy/mA.s")
25 | >> | CONTAINS | NUM | EV (113837, DCM, "Mean CTDIfreeair") | 1 | U |  | UNITS = EV (mGy, UCUM, "mGy")
26 | >> | CONTAINS | NUM | EV (113838, DCM, "DLP") | 1 | M |  | UNITS = EV (mGy.cm, UCUM, "mGy.cm")
27 | >> | CONTAINS | NUM | EV (113839, DCM, "Effective Dose") | 1 | U |  | UNITS = EV (mSv, UCUM, "mSv")
28 | >>> | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | MC | IF row 27 is present | DCID 10011 “Effective Dose Evaluation Method”
29 | >>>> | HAS PROPERTIES | NUM | EV (113840, DCM, "Effective Dose Conversion Factor") | 1 | MC | IF row 28 is present and equals (113800, DCM, "DLP to E conversion via MC computation") or equals (113802, DCM, "DLP to E conversion via measurement") | UNITS = EV (mSv/mGy.cm, UCUM, "mSv/mGy.cm")
30 | >> | CONTAINS | NUM | EV (113930, DCM, "Size Specific Dose Estimation") | 1-n | U |  | UNITS = EV (mGy, UCUM, "mGy")
31 | >>> | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | M |  | DCID 10023 “Size Specific Dose Estimation Method for CT”
32 | >>>> | INFERRED FROM | NUM | EV (113931, DCM, "Measured Lateral Dimension") | 1 | MC | IF row 31 equals (113934, DCM, "AAPM 204 Lateral Dimension") or (113936, DCM, "AAPM 204 Sum of Lateral and AP Dimension") | UNITS = EV (mm, UCUM, "mm")
33 | >>>> | INFERRED FROM | NUM | EV (113932, DCM, "Measured AP Dimension") | 1 | MC | IF row 31 equals (113935, DCM, "AAPM 204 AP Dimension") or (113936, DCM, "AAPM 204 Sum of Lateral and AP Dimension") | UNITS = EV (mm, UCUM, "mm")
34 | >>>> | INFERRED FROM | NUM | EV (113933, DCM, "Derived Effective Diameter") | 1 | MC | IF row 31 equals (113934, DCM, "AAPM 204 Lateral Dimension") or (113935, DCM, "AAPM 204 AP Dimension") or (113936, DCM, "AAPM 204 Sum of Lateral and AP Dimension") or (113937, DCM, "AAPM 204 Effective Diameter Estimated From Patient Age") | UNITS = EV (mm, UCUM, "mm")
34b | >>>> | INFERRED FROM | NUM | EV (113980, DCM, "Water Equivalent Diameter") | 1 | MC | IF row 31 equals (113981, DCM, "Water Equivalent Diameter Representative Value") | UNITS = EV (mm, UCUM, "mm")
34c | >>>>> | HAS CONCEPT MOD | CODE | EV EV (G-C036, SRT, "Measurement Method") | 1 | M |  | DCID 10024 “Water Equivalent Diameter Method”
34d | >>>> | INFERRED FROM | UIDREF | EV (113985, DCM, "Series or Instance used for Water Equivalent Diameter estimation") | 1-n | MC | IF row 31 equals (113982, DCM, "Water Equivalent Diameter Integrated Across Scan Range") or (113984, DCM, "Water Equivalent Diameter From Localizer") or (row 31 equals (113983, DCM, "Water Equivalent Diameter From Raw Data") and the Raw Data is encoded in DICOM). |
34e | >>>> | INFERRED FROM | NUM | EV (113986, DCM, "Z value of location of Water Equivalent Diameter estimation") | 1 | MC | IF row 31 equals (113981, DCM, "Water Equivalent Diameter Representative Value") | UNITS = EV (mm, UCUM, "mm")
35 | >> | CONTAINS | INCLUDE | DTID 10015 “CT Dose Check Details” | 1 | M |  |
36 | > | CONTAINS | TEXT | EV (113842, DCM, "X-Ray Modulation Type") | 1 | U |  |
37 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
38 | > | CONTAINS | INCLUDE | DTID 1020 “Person Participant” | 1-n | U |  | $PersonProcedureRole = EV (113851, DCM, "Irradiation Administering")
39 | > | CONTAINS | INCLUDE | DTID 1021 “Device Participant” | 1 | MC | Required if the irradiating device is not the recording device. | $DeviceProcedureRole = EV (113859, DCM, "Irradiating Device")

Content Item Descriptions

Row 2 | User-defined type of clinical acquisition protocol for creating images or image-derived measurements. May be taken from Protocol Name (0018,1030) or from Performed Procedure Step Description (0040,0254).
Row 3 | The target region is the anatomy exposed.
Row 4 | Description of the method used during acquisition of this CT irradiation event, may be derived from Acquisition Type (0018,9302).
Row 4b | Though not a characteristic of the acquisition per se, the type of reconstruction intended has a bearing on the technique used. If multiple types of reconstruction are performed, multiple values can be listed. These values should correspond to the values of Reconstruction Algorithm (0018,9315) in the reconstructed images.
Row 5 | The acquisition was performed with or without contrast medium application.
Row 6d, 6e | If an acquisition is a repeat because an earlier acquisition was unsatisfactory, this may be recorded along with a coded reason. This is intended to help with subsequent analysis by providing a priori information about why the study might be flagged as an outlier with higher dose exposure values than usual for the type of study.
Row 8 | Total time the patient has received X-Ray exposure during the irradiation event.
Row 10 | The value of the nominal width (referenced to the location of the isocenter along the z axis) of a single collimated slice in mm.
Row 11 | The value of the nominal width (referenced to the location of the isocenter along the z axis) of the nominal total collimation in mm over the area of active X-Ray detection (z-coverage).
Row 12 | Pitch Factor: For Spiral Acquisition, the Pitch Factor is the ratio of the Table Feed per Rotation to the Nominal Total Collimation Width. For Sequenced Acquisition, the Pitch Factor is the ratio of the Table Feed per single sequenced scan to the Nominal Total Collimation Width.
Row 14 | CT X-Ray source parameters related to the acquisition. For each X-Ray source an item must be present. For multi-energy acquisitions, multiple items may be present for each X-Ray source, each item describing one energy level.
Row 15 | Identification of the X-Ray source. Identifies the particular X-Ray source (in a multi-source CT system) for which the set of X-Ray source parameter values is reported.
Row 16 | KVP value as measured/recorded by system.
Row 18 | Mean tube current as measured/recorded by system.
Row 19 | Exposure time as measured/recorded by the system per rotation.
Row 20 | Thickness of an equivalent filter constructed from aluminum, in case of multi-source CT systems AND if Row 4 is not present
Row 21 | CT Dose for one acquisition
Rows 5, 6 | "Mean CTDIvol" refers to the average value of the CTDIvol applied within this acquisition. CTDIvol is the volume CTDIw, where CTDIw is the weighted computed tomography dose index 100 as defined in IEC 60601-2-44. For Sequenced and Spiral scanning, CTDIvol = CTDIw / Pitch Factor. For Stationary and Free scanning, CTDIvol = CTDIw ✕ Cumulative Exposure Time/ Exposure Time Per Rotation. According to IEC 60601-2-44 Ed 3 for Constant Angle Acquisition may be calculated as CTDIvol = (CTDIw / Current Time Product (mAs)) ✕ X-Ray Tube Current (mA) x (Nominal Total Collimation Width (mm) / Table Speed (mm/s)). ### Note The ratio CTDIw / Current Time Product is evaluated independently of the Constant Angle Acquisition but with the same settings of tube voltage and Total Collimation Width as those of the Constant Angle Acquisition. See also CTDIvol (0018,9345) and Spiral Pitch Factor (0018,9311) in the “Enhanced Computed Tomography Image IOD” in PS3.3 .
Row 23 | The type of phantom used for CTDI measurement according to IEC 60601-2-44(e.g., Head 16 cm diameter PMMA, Body 32 cm diameter PMMA).
Rows 5, 6 | The CTDIfree air Calculation Factor is the CTDIfree air per mAs, expressed in units of mGy/mAs. The CTDIfree air Calculation Factor may be used in one method calculating Dose. For example, for this acquisition, Effective Dose = Mean X-Ray Tube Current ✕ Cumulative Exposure Time ✕ CTDIfree air Calculation Factor ✕ (Effective Dose/ CTDIfree air).
Rows 5, 6 | MeanCTDIfree air is the mean CTDI for this acquisition, evaluated free-in-air according to IEC 60601-2-44. MeanCTDIfree air = Mean X-Ray Tube Current ✕ Cumulative Exposure Time ✕ CTDIfree air Calculation Factor. The CTDIfree air may be used in one method of calculating Effective Dose.
Rows 5, 6 | For Spiral scanning, DLP = CTDIvol ✕ Scanning Length. For Sequenced scanning, DLP = CTDIvol ✕ Nominal Total Collimation Width ✕ Cumulative Exposure Time / Exposure Time per Rotation. For Stationary and Free scanning, DLP = CTDIvol ✕ Nominal Total Collimation Width (according to IEC 60601-2-44).
Row 27 | Effective Dose in mSv of the single continuous time-frame of the irradiation computed as described in TID 10012 “CT Accumulated Dose Data”.
Row 29 | The Effective Dose Conversion Factor is the ratio of the Effective Dose to the DLP, expressed in units of mSv/mGy.cm, and it is used as a factor in one method of estimating Effective Dose. Monte Carlo Simulations (or dosimetric measurements in an anthropomorphic phantom, e.g., the Alderson-Rando phantom) may be used as a basis for the evaluation of Effective Dose Conversion Factors.
Row 30 | More than one Size Specific Dose Estimation may be included, for example if different computation methods are used.
Row 31 | The methods of [AAPM Report 204] are listed in CID 10023 “Size Specific Dose Estimation Method for CT”; other methods may be used. The phantom size (16cm or 32cm) used for the calculation is available from the phantom type defined in Row 23.
Row 32 | The condition specifies inclusion of the Measured Lateral Dimension if it was used in the calculation.
Row 33 | The condition specifies inclusion of the Measured AP Dimension if it was used in the calculation.
Row 34 | The Derived Effective Diameter is conditionally included, whether it was derived from measurements or estimated from age, but may not be used for other (non-AAPM Report 204) methods.
Row 34b | A single value for Water Equivalent Diameter is encoded in Row 34b if the method uses a single value. It is required if the method uses a representative slice, but may also be present if the method used a Localizer or Raw Data at a single location rather than the entire scan range.
Row 34c | The modifier is intended to specify the family of methods and not the specific technique (e.g., for AAPM 220 (113987, DCM, "AAPM 220") is used, not (113981, DCM, "Water Equivalent Diameter Representative Value"), etc.).
Row 34d | If the method uses multiple slices across the scan range, the reconstructed image Series or (list of) Instances used is referenced; the values for Water Equivalent Diameter may or may not be recorded in the CT Image Module or CT Exposure Macro of those images. More than one Series may be referenced if the reconstructed images for this acquisition used for Water Equivalent Diameter estimation span multiple series. If the Water Equivalent Diameter was computed from raw views rather than reconstructed images, then the Raw Data is referenced, if it was encoded in DICOM (it is not required to be).
Row 34e | This location is patient (not table or gantry) relative, to allow it to be defined in the Patient Coordinate System and hence related to the Image Position (Patient) in the reconstructed images (see TID 10014 “Scanning Length”, included at Row 9). It is required if the method uses a representative slice, but may also be present if the method used a Localizer or Raw Data at a single location rather than the entire scan range.
Row 35 | Record of details associated with using the NEMA Dose Check Standard (NEMA XR-25-2010).
Row 36 | The type of exposure modulation. May use the value of Exposure Modulation Type (0018,9323) from CT Exposure Macro or from CT Image Module.
Row 38 | People responsible for the administration of the radiation reported in the irradiation event. May include values that would appear in Performing Physicians' Name (0008,1050), Performing Physician Identification Sequence (0008,1052), Operators' Name (0008,1070) and/or Operator Identification Sequence (0008,1072).
Row 39 | The device that produced the irradiation in this Irradiation Event. I.e., the CT scanner. This is not required to be present if the information is the same as that already recorded in the TID 1004 “Device Observer Identifying Attributes” encoded via the inclusion of TID 1002 “Observer Context” in TID 10011 “CT Radiation Dose” Row 4, which in turn may be absent if identical to the content in the Enhanced General Equipment Module.

### TID 10014 Scanning Length

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 10014. Scanning Length

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | EV (113825, DCM, "Scanning Length") | 1 | M |  | UNITS = EV (mm, UCUM, "mm")
2 |  |  | NUM | EV (113893, DCM, "Length of Reconstructable Volume") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
3 |  |  | NUM | EV (113899, DCM, "Exposed Range") | 1 | UC | IFF TID 10013 “CT Irradiation Event Data” row 4 CT Acquisition Type equals (P5-08001, SRT, "Spiral Acquisition") | UNITS = EV (mm, UCUM, "mm")
4 |  |  | NUM | EV (113895, DCM, "Top Z Location of Reconstructable Volume") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
5 |  |  | NUM | EV (113896, DCM, "Bottom Z Location of Reconstructable Volume") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
6 |  |  | NUM | EV (113897, DCM, "Top Z Location of Scanning Length") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
7 |  |  | NUM | EV (113898, DCM, "Bottom Z Location of Scanning Length") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
8 |  |  | UIDREF | EV (112227, DCM, "Frame of Reference UID") | 1 | MC | IF any of Rows 4 through 7 or Row 34e of TID 10013 are present. | If present, shall be the same UID as in the images reconstructed from this irradiation event.

Content Item Descriptions

Row 1 | For Spiral scanning, the scanning length is normally the table travel in mm during the tube loading (see Figure A-16). For Sequenced scanning, the scanning length is the table travel between consecutive scans times the number of scans. For Stationary and Free scanning, the scanning length is the nominal width of the total collimation.
Row 2 | The length of the reconstructable volume is the maximum z-range between the outermost edges of the top and bottom slices that can be reconstructed from the acquisition. For Spiral scanning, the length of reconstructable volume is the z-range between the outermost beginning of the first reconstructable slice and the outermost end of the last reconstructable slice (see Figure A-16). For Sequenced scanning, the length of reconstructable volume is the z-range between the outermost beginning of the first slice and the outermost end of the last slice (i.e., including any skip). For Stationary and Free scanning, the length of reconstructable volume is the nominal width of the total collimation.
Row 3 | For Spiral scanning, the exposed range is as defined in IEC 60601-2-44 (Ed. 3) 302.115(b) (see Figure A-16). Exposed range is not defined for other modes of scanning.
Rows 4-5 | The Top and Bottom Z Locations of the Reconstructable Volume are independent of the slice width of any actual reconstructed slices. They are measured from the edges of the volume, and hence are not equal to the Z locations encoded in the images of any actual reconstructed slices, which are recorded as the center of the slice.
Rows 4-7 | These locations are patient (not table or gantry) relative, to allow them to be defined in the Patient Coordinate System and hence related to the Image Position (Patient) in the reconstructed images. They are also defined in terms of the top (towards the patient's head), and bottom (towards the patient's feet) of the corresponding ranges, in order to make them independent of whether the scan starts at the top or the bottom or shuttles back and forth in between (see Figure A-16).

Figure A-16. Spiral Acquisition Parameters

### TID 10015 CT Dose Check Details

This Template records details related to the use of the NEMA Dose Check Standard (NEMA XR-25-2010).

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 10015. CT Dose Check Details

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113900, DCM, "Dose Check Alert Details") | 1 | MC | IF the scanning device has implemented dose alerts |
2 | > | CONTAINS | CODE | EV (113901, DCM, "DLP Alert Value Configured") | 1 | M |  | DCID 230 “Yes-No”
3 | > | CONTAINS | CODE | EV (113902, DCM, "CTDIvol Alert Value Configured") | 1 | M |  | DCID 230 “Yes-No”
4 | > | CONTAINS | NUM | EV (113903, DCM, "DLP Alert Value") | 1 | MC | IFF value of Row 2 is (R-0038D, SRT, "Yes") | UNITS = EV (mGy.cm, UCUM, "mGy.cm")
5 | > | CONTAINS | NUM | EV (113904, DCM, "CTDIvol Alert Value") | 1 | MC | IFF value of Row 3 is (R-0038D, SRT, "Yes") | UNITS = EV (mGy, UCUM, "mGy")
6 | > | CONTAINS | NUM | EV (113905, DCM, "Accumulated DLP Forward Estimate") | 1 | MC | IF Accumulated DLP Forward Estimate (Row 6) exceeds DLP Alert Value (Row 4) | UNITS = EV (mGy.cm, UCUM, "mGy.cm")
7 | > | CONTAINS | NUM | EV (113906, DCM, "Accumulated CTDIvol Forward Estimate") | 1 | MC | IF Accumulated CTDIvol Forward Estimate (Row 7) exceeds CTDIvol Alert Value (Row 5) | UNITS = EV (mGy, UCUM, "mGy")
8 | > | CONTAINS | TEXT | EV (113907, DCM, "Reason for Proceeding") | 1 | UC | IFF Accumulated DLP Forward Estimate (Row 6) exceeds DLP Alert Value (Row 4) or Accumulated CTDIvol Forward Estimate (Row 7) exceeds CTDIvol Alert Value (Row 5) |
9 | > | CONTAINS | INCLUDE | DTID 1020 “Person Participant” | 1 | MC | IF Accumulated DLP Forward Estimate (Row 6) exceeds DLP Alert Value (Row 4) or Accumulated CTDIvol Forward Estimate (Row 7) exceeds CTDIvol Alert Value (Row 5) | $PersonProcedureRole = EV (113850, DCM, "Irradiation Authorizing")
10 |  |  | CONTAINER | EV (113908, DCM, "Dose Check Notification Details") | 1 | MC | IF the scanning device has implemented dose notifications |
11 | > | CONTAINS | CODE | EV (113909, DCM, "DLP Notification Value Configured") | 1 | M |  | DCID 230 “Yes-No”
12 | > | CONTAINS | CODE | EV (113910, DCM, "CTDIvol Notification Value Configured") | 1 | M |  | DCID 230 “Yes-No”
13 | > | CONTAINS | NUM | EV (113911, DCM, "DLP Notification Value") | 1 | MC | IFF value of Row 11 is (R-0038D, SRT, "Yes") | UNITS = EV (mGy.cm, UCUM, "mGy.cm")
14 | > | CONTAINS | NUM | EV (113912, DCM, "CTDIvol Notification Value") | 1 | MC | IFF value of Row 12 is (R-0038D, SRT, "Yes") | UNITS = EV (mGy, UCUM, "mGy")
15 | > | CONTAINS | NUM | EV (113913, DCM, "DLP Forward Estimate") | 1 | MC | IF DLP Forward Estimate (Row 15) exceeds DLP Notification Value (Row 13) | UNITS = EV (mGy.cm, UCUM, "mGy.cm")
16 | > | CONTAINS | NUM | EV (113914, DCM, "CTDIvol Forward Estimate") | 1 | MC | IF CTDIvol Forward Estimate (Row 16) exceeds CTDIvol Notification Value (Row 14) | UNITS = EV (mGy, UCUM, "mGy")
17 | > | CONTAINS | TEXT | EV (113907, DCM, "Reason for Proceeding") | 1 | UC | IFF DLP Forward Estimate (Row 15) exceeds DLP Notification Value (Row 13) or CTDIvol Forward Estimate (Row 16) exceeds CTDIvol Notification Value (Row 14) |
18 | > | CONTAINS | INCLUDE | DTID 1020 “Person Participant” | 1 | UC | IFF DLP Forward Estimate (Row 15) exceeds DLP Notification Value (Row 13) or CTDIvol Forward Estimate (Row 16) exceeds CTDIvol Notification Value (Row 14) | $PersonProcedureRole = EV (113850, DCM, "Irradiation Authorizing")

Content Item Descriptions

Row 1 | Container for Dose Check Alert details.
Row 2 | Indicates whether a DLP Alert Value was configured (e.g., by the institution) for the exam to which this irradiation event belongs.
Row 3 | Indicates whether a CTDIvol Alert Value was configured (e.g., by the institution) for the exam to which this irradiation event belongs.
Row 4 | The configured value applicable to the current exam that would trigger an alert if the accumulated DLP were projected to exceed it.
Row 5 | The configured value applicable to the current exam that would trigger an alert if the Accumulated CTDIvol at any given location were projected to exceed it.
Row 6 | The value estimated prior to performing this irradiation event of the projected DLP accumulated during this exam, including this irradiation event. The estimate may include assumptions such as those described in NEMA XR 25-2010.
Row 7 | The value estimated prior to performing this irradiation event of the projected CTDIvol accumulated during this exam, including this irradiation event. The value is for the location with the highest estimated accumulation. The actual location is not recorded. The estimate may include assumptions such as those described in NEMA XR 25-2010.
Row 8 | The reason provided by the operator for proceeding with an irradiation event projected to exceed an alert value.
Row 9 | Person responsible for authorizing irradiation projected to exceed an alert value.
Row 10 | Container for Dose Check Notification details.
Row 11 | Indicates whether a DLP Notification Value was configured (e.g., by the institution) for the Protocol Element Group to which this irradiation event corresponds.
Row 12 | Indicates whether a CTDIvol Notification Value was configured (e.g., by the institution) for the Protocol Element Group to which this irradiation event corresponds.
Row 13 | The configured value applicable to the current irradiation event that would trigger a notification if the DLP were projected to exceed it.
Row 14 | The configured value applicable to the current irradiation event that would trigger a notification if the CTDIvol were projected to exceed it.
Row 15 | The value estimated prior to performing this irradiation event of the DLP for this irradiation event. The estimate may include assumptions such as those described in NEMA XR 25-2010.
Row 16 | The value estimated prior to performing this irradiation event of the CTDIvol for this irradiation event. The value is for the location with the highest estimated value. The actual location is not recorded. The estimate may include assumptions such as those described in NEMA XR 25-2010.
Row 17 | The reason provided by the operator for proceeding with an irradiation event projected to exceed a notification value.
Row 18 | Person responsible for authorizing irradiation projected to exceed a notification value.

## Radiopharmaceutical Radiation Dose SR IOD Templates

The Templates that comprise the Radiopharmaceutical Radiation Dose SR are interconnected as in Figure A-17.

Figure A-17. Radiopharmaceutical Radiation Dose SR IOD Template Structure

### TID 10021 Radiopharmaceutical Radiation Dose

This Template defines a container (the root) with subsidiary Content Items, each of which corresponds to a single Radiopharmaceutical Administration Dose event entry. There is a defined recording observer (the system and/or person responsible for recording the assay of the radiopharmaceutical, and the person administered the radiopharmaceutical). Multiple Radiopharmaceutical Radiation Dose objects may be created for one study. Radiopharmaceutical Start DateTime in TID 10022 “Radiopharmaceutical Administration Event Data” will convey the order of administrations.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 10021. Radiopharmaceutical Radiation Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113500, DCM, "Radiopharmaceutical Radiation Dose Report") | 1 | M |  | Root node
1b | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C2D0, SRT, "Associated Procedure") | 1 | M |  | DCID 3108 “NM/PET Procedures”
3 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E8, SRT, "Has Intent") | 1 | M |  | DCID 3629 “Procedure Intent”
4 | > | CONTAINS | INCLUDE | DTID 10022 “Radiopharmaceutical Administration Event Data” | 1 | M |  |
5 | > | CONTAINS | INCLUDE | DTID 10024 “Radiopharmaceutical Administration Patient Characteristics” | 1 | U |  |
6 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

Content Item Descriptions

Row 2 | The associated procedure is the procedure performed, or if no procedure was performed the procedure that was ordered.

### TID 10022 Radiopharmaceutical Administration Event Data

The Radiopharmaceutical Administration Event conveys the dose and assay and time information of a single radiopharmaceutical event. A Radiopharmaceutical Administration event is one radioactive pharmaceutical administered to a patient.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 10022. Radiopharmaceutical Administration Event Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113502, DCM, "Radiopharmaceutical Administration") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (F-61FDB, SRT, "Radiopharmaceutical agent") | 1 | M |  | DCID 25 “Radiopharmaceuticals” DCID 4021 “PET Radiopharmaceutical”
3 | >> | HAS PROPERTIES | CODE | EV (C-10072, SRT, "Radionuclide") | 1 | M |  | DCID 18 “Isotopes in Radiopharmaceuticals” DCID 4020 “PET Radionuclide”
4 | >> | HAS PROPERTIES | NUM | EV (R-42806, SRT, "Radionuclide Half Life") | 1 | M |  | UNITS = EV (s, UCUM, "seconds")
5 | > | CONTAINS | NUM | EV (123007, DCM, "Radiopharmaceutical Specific Activity") | 1 | U |  | UNITS = EV (Bq/mmol, UCUM, "Bq/mmol")
6 | > | CONTAINS | UIDREF | EV (113503, DCM, "Radiopharmaceutical Administration Event UID") | 1 | M |  |
7 | > | CONTAINS | CODE | EV (113505, DCM, "Intravenous Extravasation Symptoms") | 1-n | U |  | DCID 10043 “Intravenous Extravasation Symptoms”
8 | > | CONTAINS | NUM | EV (113506, DCM, "Estimated Extravasation Activity") | 1 | U |  | UNITS = EV(%, UCUM, "percent")
9 | > | CONTAINS | DATETIME | EV (123003, DCM, "Radiopharmaceutical Start DateTime") | 1 | M |  |
10 | > | CONTAINS | DATETIME | EV (123004, DCM, "Radiopharmaceutical Stop DateTime") | 1 | U |  |
11 | > | CONTAINS | NUM | EV (113507, DCM, "Administered activity") | 1 | M |  | UNITS = EV (MBq, UCUM, "MBq")
12 | > | CONTAINS | NUM | EV (123005, DCM, "Radiopharmaceutical Volume") | 1 | U |  | UNITS = EV (cm3, UCUM, "cm3")
13 | > | CONTAINS | NUM | EV (113508, DCM, "Pre-Administration Measured Activity") | 1 | U |  | UNITS = EV (MBq, UCUM, "MBq")
14 | >> | HAS OBS CONTEXT | CODE | EV (113540, DCM, "Activity Measurement Device") | 1 | U |  | DCID 10041 “Source of Radioisotope Activity Information”
15 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
16 | > | CONTAINS | NUM | EV (113509, DCM, "Post-Administration Measured Activity") | 1 | U |  | UNITS = EV (MBq, UCUM, "MBq")
17 | >> | HAS OBS CONTEXT | CODE | EV (113540, DCM, "Activity Measurement Device") | 1 | U |  | DCID 10041 “Source of Radioisotope Activity Information”
18 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
19 | > | CONTAINS | INCLUDE | DTID 10023 “Organ Dose” | 1-n | U |  |
20 | > | CONTAINS | CODE | EV (G-C340, SRT, "Route of administration") | 1 | M |  | BCID 11 “Route of Administration”
21 | >> | HAS PROPERTIES | CODE | EV (G-C581, SRT, "Site of") | 1 | MC | IF Row 20 equals (G-D101, SRT, "Intravenous route") or (G-D103, SRT, "Intramuscular route") | DCID 3746 “Percutaneous Entry Site”
22 | >>> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IF Row 21 has laterality | DCID 244 “Laterality”
23 | > | CONTAINS | INCLUDE | DTID 1020 “Person Participant” | 1-n | M |  | $PersonProcedureRole = EV (113851, DCM, "Irradiation Administering")
24 | > | CONTAINS | CODE | EV (121147, DCM, "Billing Code(s) ") | 1-n | U |  |
25 | > | CONTAINS | CODE | EV (113510, DCM, "Drug Product Identifier") | 1-n | U |  |
26 | > | CONTAINS | TEXT | EV (111529, DCM, "Brand Name") | 1 | U |  |
27 | > | CONTAINS | TEXT | EV (113511, DCM, "Radiopharmaceutical Dispense Unit Identifier") | 1 | U |  |
28 | >> | CONTAINS | TEXT | EV (113512, DCM, "Radiopharmaceutical Lot Identifier") | 1-n | U |  |
29 | >> | CONTAINS | TEXT | EV (113513, DCM, "Reagent Vial Identifier") | 1-n | U |  |
30 | >> | CONTAINS | TEXT | EV (113514, DCM, "Radionuclide Identifier") | 1-n | U |  |
31 | > | CONTAINS | TEXT | EV (113516, DCM, "Prescription Identifier") | 1 | U |  |
32 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

Content Item Descriptions

Row 4 | The value of Half-life that was used for computing the decay of the administered radiopharmaceutical. It is not intended for use by the receiver for any further computation.
Row 5 | Activity per unit mass of the radiopharmaceutical at Radiopharmaceutical Start Time
Row 6 | Unique identification of a single radiopharmaceutical administration event.
Row 8 | The estimated percentage of administered activity lost at the injection site. The estimation includes extravasation, paravenous administration and leakage at the injection site.
Row 9 | The time the radiopharmaceutical was administered to the patient for imaging purposes.
Row 11 | Total amount of radioactivity administered to the patient at Radiopharmaceutical Start Time. It is a computed field from the TID 10022 Pre-Administration Measured Activity Row 13, TID 10022 Post-Administration Measured Activity Row 17, Radionuclide Half Life Row 4 and Radiopharmaceutical Start Time Row 9. Does not include estimated extravasation activity.
Rows 13, 16 | Observation DateTime (0040,A032) shall be used to record when the measurement was taken.
Row 23 | Identifies the person administering the product.
Row 24 | The billing codes for the preparation and administration of the radiopharmaceutical. It does not include performance and interpretation of the imaging.
Row 25 | Registered drug establishment code for the product. A coding scheme example is NDC, WHO-DDE or RxNorm. Multiple entries can be used for equivalent drug product codes.
Row 27 | The human readable identification of the specific radiopharmaceutical quantity (dose) administered to the patient.
Row 28 | Identifies the vial, batch or lot number from which the individual radiopharmaceutical quantity (dose) was produced. Row 27 the Radiopharmaceutical Identifier records the identification for each individual dose.
Row 29 | Identifies the lot or unit serial number for the reagent component for the radiopharmaceutical identified in row 27.
Row 30 | Identifies the lot or unit serial number for the radionuclide component for the radiopharmaceutical identified in row 27.

### TID 10023 Organ Dose

This Template conveys the information about the dose to a single organ.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 10023. Organ Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113517, DCM, "Organ Dose Information") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 10044 “Radiosensitive Organs”
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IFF anatomy has laterality | DCID 244 “Laterality”
4 | > | CONTAINS | NUM | EV (G-D701, SRT, "Mass") | 1 | U |  | UNITS = EV (g, UCUM, "grams")
5 | >> | HAS CONCEPT MOD | TEXT | EV (G-C036, SRT, "Measurement Method") | 1 | M |  |
6 | > | CONTAINS | NUM | EV (113518, DCM, "Organ Dose") | 1 | M |  | UNITS = EV (mGy, UCUM, "mGy")
7 | >> | HAS PROPERTIES | CODE | EV (121406, DCM, "Reference Authority") | 1 | MC | XOR Row 8 | BCID 10040 “Radiopharmaceutical Organ Dose Reference Authority”
8 | >> | HAS PROPERTIES | TEXT | EV (121406, DCM, "Reference Authority") | 1 | MC | XOR Row 7 |

Content Item Descriptions

Row 3 | For paired organs, use (G-A102, SRT, "Right and Left") to report the estimated absorbed dose for both organs.
Row 4 | The estimated mass of organ in grams used when calculating the organ dose.
Row 5 | Method used to obtain the estimate. This could include a method that does not involve performing a measurement (e.g., Standard Organ Mass Tables).
Row 6 | Organ dose (in units of mGy). Organ is specified by row 2.

### TID 10024 Radiopharmaceutical Administration Patient Characteristics

This Template describes the characteristics of the patient that are specific to the current clinical presentation (visit). The characteristics noted may affect the activity received, and how dose is calculated for the patient. Patient Characteristic concepts in this Template, which may replicate attributes in the Patient Study Module, are included here as possible targets of by-reference relationships from other Content Items in the SR tree.

Type:

  Extensible

Order:

  Significant

Root:

  No

Table TID 10024. Radiopharmaceutical Administration Patient Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121118, DCM, "Patient Characteristics") |  |  |  |
2 | > | CONTAINS | CODE | EV (109054, DCM, "Patient state") | 1-n | U |  | DCID 10045 “Radiopharmaceutical Patient State”
3 | > | CONTAINS | NUM | EV (121033, DCM, "Subject Age") | 1 | U |  | UNITS = DCID 7456 “Units of Measure for Age”
4 | > | CONTAINS | CODE | EV (121032, DCM, "Subject Sex") | 1 | U |  | DCID 7455 “Sex”
5 | > | CONTAINS | NUM | EV (8302-2, LN, "Patient Height") | 1 | U |  | UNITS = EV (cm, UCUM, "cm")
6 | > | CONTAINS | NUM | EV (29463-7, LN, "Patient Weight") | 1 | U |  | UNITS = EV (kg, UCUM, "kg").
7 | > | CONTAINS | NUM | EV (8277-6, LN, "Body Surface Area") | 1 | U |  | UNITS = EV (m2, UCUM, "m^2")
8 | >> | INFERRED FROM | CODE | EV (8278-4, LN, "Body Surface Area Formula") | 1 | U |  | BCID 3663 “Body Surface Area Equations”
9 | > | CONTAINS | NUM | EV (F-01860, SRT, "Body Mass Index") | 1 | U |  | UNITS = EV (kg/m2, UCUM, "kg/m^2")
10 | >> | INFERRED FROM | CODE | EV (121420, DCM, "Equation") | 1 | U |  | DT (122265, DCM, "BMI = Wt/Ht^2")
11 | > | CONTAINS | NUM | EV (14749-6, LN, "Glucose") | 1 | U |  | UNITS = EV (mmol/l, UCUM, "mmol/l")
12 | > | CONTAINS | NUM | EV(113550, DCM, "Fasting Duration") | 1 | U |  | UNITS = DT (h, UCUM, "hours")
13 | > | CONTAINS | NUM | EV(113551, DCM, "Hydration Volume") | 1 | U |  | UNITS = DT (ml, UCUM, "ml")
14 | > | CONTAINS | TEXT | EV (113552, DCM, "Recent Physical Activity") | 1 | U |  |
15 | > | CONTAINS | NUM | EV (2160-0, LN, "Serum Creatinine") | 1 | U |  | UNITS = DT (mg/dl, UCUM, "mg/dl")
16 | > | CONTAINS | NUM | EV (F-70210, SRT, "Glomerular Filtration Rate") | 1-n | U |  | UNITS = DT (ml/min{1.73_m2}, UCUM, "ml/min/1.73m2")
17 | >> | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | DCID 10047 “GFR Measurement Methods”
18 | >> | HAS CONCEPT MOD | CODE | EV (121050, DCM, "Equivalent meaning of concept name") | 1 | M |  | DCID 10046 “GFR Measurements”

Content Item Descriptions

Row 3 | Defaults to value of Patient's Age (0010,1010) in Patient Study Module
Row 5 | Patient height may differ from Patient's Size (0010,1020). Row 4 is the height value used for any height based protocols. Observation DateTime (0040,A032) may be used to record when the measurement was taken.
Row 6 | Patient weight may differ from Patient's Weight (0010,1030). Row 5 is the weight value used for any weight based protocols. Observation DateTime (0040,A032) shall be used to record when the measurement was taken.
Row 11 | Patient's Blood Glucose level. Observation DateTime (0040,A032) shall be used to record when the measurement was taken.
Row 15 | Serum Creatinine level. Observation DateTime (0040,A032) shall be used to record when the measurement was taken.
Row 16 | Glomerular Filtration Rate Observation DateTime (0040,A032) shall be used to record when the measurement was taken. The formatting of the UCUM units is aligned with LOINC. See http://unitsofmeasure.org/trac/ticket/98

## Patient Radiation Dose SR IOD Templates

The Templates that comprise the Patient Radiation Dose SR are interconnected as in Figure A-18.

Figure A-18. Patient Radiation Dose Structured Report IOD Template Structure

### TID 10030 Patient Radiation Dose

This template defines a container (the root) with subsidiary content items for determining an estimated radiation dose to a patient.

Type:

  Extensible

Order:

  Non-Significant

Root:

  Yes

Table TID 10030. Patient Radiation Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (128401, DCM, "Patient Radiation Dose Report") | 1 | M |  | Root Node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
4 | > | CONTAINS | INCLUDE | DTID 10031 “Radiation Dose Estimate” | 1 | M |  |
5 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

Content Item Descriptions

Row 3 | Identify all observers and devices involved with creating the organ estimations included in this Patient Radiation Dose SR.

### TID 10031 Radiation Dose Estimate

The dose estimate is used to record the results from one analysis method from one or more radiation sources. Organ dose estimates are calculated from one or more irradiation events to a patient. The output from one or more sources of radiation can be used separately or combined to estimate the dose to a patient or individual organs.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 10031. Radiation Dose Estimate

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (128402, DCM, "Radiation Dose Estimate") | 1 | M |  |
2 | > | HAS CONCEPT MOD | TEXT | EV (128403, DCM, "Radiation Dose Estimate Name") | 1 | M |  |
3 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 10033 “Radiation Dose Estimate Methodology” | 1 | M |  |
5 | > | CONTAINS | INCLUDE | DTID 10032 “Radiation Dose Estimate Representation” | 1-n | U |  |
6 | > | CONTAINS | CONTAINER | EV (113517, DCM, "Organ Radiation Dose Information") | 1-n | M |  |
7 | >> | CONTAINS | CODE | EV (T-D0060, SRT, "Organ") | 1 | M |  | DCID 10060 “Organs for Radiation Dose Estimates”
8 | >>> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IF Row 7 has laterality | DCID 244 Laterality
9 | >> | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
10 | >> | CONTAINS | NUM | DCID 10061 “Absorbed Radiation Dose Types” | 1 | M |  | UNITS = EV (mGy, UCUM, "mGy")
11 | >>> | HAS PROPERTIES | NUM | DCID 225 “Measurement Uncertainty Concepts” | 1-n | U |  | UNITS = EV (mGy, UCUM, "mGy")
12 | >>>> | HAS PROPERTIES | TEXT | EV (128511, DCM, "Reference to Uncertainty Determination Method") | 1 | U |  |
13 | >> | CONTAINS | NUM | DCID 10062 “Equivalent Radiation Dose Types” | 1 | U |  | UNITS = EV (mSv, UCUM, "mSv")
14 | >>> | HAS PROPERTIES | NUM | DCID 225 “Measurement Uncertainty Concepts” | 1-n | U |  | UNITS = EV (mSv, UCUM, "mSv")
15 | >>>> | HAS PROPERTIES | TEXT | EV (128511, DCM, "Reference to Uncertainty Determination Method") | 1 | U |  |

Content Item Descriptions

Row 13 | Equivalent Dose is an international quantity and includes the use of a Radiation Weighting Factor to compensate for the radiation type, e.g., photon, neutron, alpha or beta particle, etc. Stating equivalent dose is not recommended in almost all dosimetry situations, except in Radiopharmaceutical dose. This is not Effective Dose.

### TID 10032 Radiation Dose Estimate Representation

Different representations (e.g., images) of the distribution of absorbed energy allow a better understanding of how this energy may affect tissue.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 10032. Radiation Dose Estimate Representation

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (128412, DCM, "Radiation Dose Estimate Representation") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (128413, DCM, "Distribution Representation") | 1 | M |  | DCID 10063 “Radiation Dose Estimate Distribution Representation”
3 | > | CONTAINS | IMAGE | EV (128414, DCM, "Radiation Dose Representation Data") | 1 | MC | XOR Row 4 |
4 | > | CONTAINS | COMPOSITE | EV (128414, DCM, "Radiation Dose Representation Data") | 1 | MC | XOR Row 3 |
5 | > | CONTAINS | CODE | EV (T-D0060, SRT, "Organ") | 1-n | M |  | DCID 10060 “Organs for Radiation Dose Estimates”
6 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IF Row 5 has laterality | DCID 244 “Laterality”
7 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

Content Item Descriptions

Roww 3 and 4 | Reference to an Instance that contains the dose representation, e.g., surface segmentation, mesh, parametric map, RT dose, Secondary Capture, etc.
Row 5 | The organs in the representation. The organs in this Row shall be present in Row 7 of TID 10031 “Radiation Dose Estimate”.

### TID 10033 Radiation Dose Estimate Methodology

This template includes the information specific to the organ dose calculation methodology used when estimating dose to individual organs, entire body or a phantom from imaging studies that use ionizing radiation.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 10033. Radiation Dose Estimate Methodology

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (128415, DCM, "Radiation Dose Estimate Methodology") | 1 | M |  |
2 | > | CONTAINS | COMPOSITE | EV (128416, DCM, "SR Instance Used") | 1-n | M |  |
3 | >> | HAS OBS CONTEXT | COMPOSITE | EV (128447, DCM, "Spatial Fiducials") | 1-n | U |  |
4 | >> | HAS PROPERTIES | UIDREF | EV (128429, DCM, "Event UID Used") | 1-n | MC | IFF some Events in the Structured Report were not used in calculating the dose. |
5 | > | CONTAINS | CONTAINER | EV (128500, DCM, "Patient Radiation Dose Model") | 1 | M |  |
6 | >> | CONTAINS | CODE | EV (128417, DCM, "Patient Model Type") | 1 | M |  | DCID 10064 “Patient Model Type”
7 | >> | CONTAINS | CODE | EV (128420, DCM, "Radiation Transport Model Type") | 1 | M |  | DCID 10065 “Radiation Transport Model Type”
8 | >> | CONTAINS | IMAGE | EV (128425, DCM, "Patient Radiation Dose Model Data") | 1 | UC | XOR Row 9, 10 |
9 | >> | CONTAINS | COMPOSITE | EV (128425, DCM, "Patient Radiation Dose Model Data") | 1 | UC | XOR Row 8, 10 |
10 | >> | CONTAINS | UIDREF | EV (128425, DCM, "Patient Radiation Dose Model Data") | 1 | UC | XOR Row 8, 9 |
11 | >> | CONTAINS | TEXT | EV (128426, DCM, "Patient Radiation Dose Model Reference") | 1 | U |  |
12 | >> | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
13 | >> | CONTAINS | CONTAINER | EV (128427, DCM, "Patient Model Demographics") | 1 | M |  |
14 | >>> | CONTAINS | NUM | EV (128428, DCM, "Model Minimum Age") | 1 | MC | IF model requires minimum age to be defined | DCID 7456 “Units of Measure for Age”
15 | >>> | CONTAINS | NUM | EV (128430, DCM, "Model Maximum Age") | 1 | MC | IF model requires maximum age to be defined | DCID 7456 “Units of Measure for Age”
16 | >>> | CONTAINS | CODE | EV (128437, DCM, "Model Patient Sex") | 1 | MC | IF model requires sex to be defined. | DCID 7455 “Sex”
17 | >>> | CONTAINS | NUM | EV (128438, DCM, "Model Minimum Weight") | 1 | MC | IF model requires minimum weight to be defined | UNITS = EV (kg, UCUM, "kg")
18 | >>> | CONTAINS | NUM | EV (128441, DCM, "Model Maximum Weight") | 1 | MC | IF model requires maximum weight to be defined | UNITS = EV (kg, UCUM, "kg")
19 | >>> | CONTAINS | NUM | EV (128439, DCM, "Model Minimum Height") | 1 | MC | IF model requires minimum height to be defined | UNITS = EV (cm, UCUM, "cm")
20 | >>> | CONTAINS | NUM | EV (128442, DCM, "Model Maximum Height") | 1 | MC | IF model requires maximum height to be defined | UNITS = EV (cm, UCUM, "cm")
21 | >> | CONTAINS | CONTAINER | EV (128456, DCM, "Patient Model Registration") | 1-n | UC | IF spatial information used from Radiation Dose SR or Patient Radiation Dose Model |
22 | >>> | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
23 | >>> | CONTAINS | CODE | EV (128446, DCM, "Registration Method") | 1 | M |  | DCID 7100 “RCS Registration Method Type”
24 | >>> | CONTAINS | COMPOSITE | EV (128444, DCM, "Spatial Registration Reference") | 1 | MC | IFF Row 8, 9 or 10 are present and Frame of Reference is defined |
25 | > | CONTAINS | CONTAINER | EV (128457, DCM, "X-Ray Beam Attenuator") | 1-n | MC | IF attenuators used in estimation |
26 | >> | CONTAINS | CODE | EV (128458, DCM, "Attenuator Category") | 1 | M |  | DCID 10066 “Attenuator Category”
27 | >> | CONTAINS | CODE | EV (128465, DCM, "Equivalent Attenuator Material") | 1 | M |  | DCID 10067 “Radiation Attenuator Materials”
28 | >> | CONTAINS | NUM | EV (128469, DCM, "Equivalent Attenuator Thickness") | 1 | MC | IF the attenuator is of uniform thickness | UNITS = EV (mm, UCUM, "mm")
29 | >> | CONTAINS | TEXT | EV (128468, DCM, "Attenuator Description") | 1 | U |  |
30 | >> | CONTAINS | CONTAINER | EV (128472, DCM, "X-Ray Beam Attenuator Model") | 1 | U |  |
31 | >>> | CONTAINS | CODE | EV (128420, DCM, "Radiation Transport Model Type") | 1 | U |  | DCID 10065 “Radiation Transport Model Type”
32 | >>> | CONTAINS | TEXT | EV (128474, DCM, "X-Ray Beam Attenuator Model Reference") | 1 | U |  |
33 | >>> | CONTAINS | IMAGE | EV (128470, DCM, "X-Ray Attenuator Model Data") | 1 | UC | XOR Row 34, 35 |
34 | >>> | CONTAINS | COMPOSITE | EV (128470, DCM, "X-Ray Attenuator Model Data") | 1 | UC | XOR Row 33, 35 |
35 | >>> | CONTAINS | UIDREF | EV (128470, DCM, "X-Ray Attenuator Model Data") | 1 | UC | XOR Row 33, 34 |
36 | >> | CONTAINS | CONTAINER | EV (128475, DCM, "X-Ray Beam Attenuator Model Registration") | 1-n | U |  |
37 | >>> | CONTAINS | CODE | EV (128446, DCM, "Registration Method") | 1 | M |  | DCID 7100 “RCS Registration Method Type”
38 | >>> | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
39 | >>> | CONTAINS | COMPOSITE | EV (128444, DCM, "Spatial Registration Reference") | 1 | MC | IFF Row 33, 34 or 35 are present and Frame of Reference is defined |
40 | > | CONTAINS | CONTAINER | EV (128476, DCM, "Radiation Dose Estimate Method") | 1-n | M |  |
41 | >> | CONTAINS | CODE | EV (128477, DCM, "Radiation Dose Estimate Method Type") | 1 | M |  | DCID 10068 “Estimate Method Types”
42 | >> | CONTAINS | INCLUDE | DTID 10034 “Radiation Dose Estimate Parameters” | 1 | U |  |
43 | >> | CONTAINS | TEXT | EV (128482, DCM, "Radiation Dose Estimate Method Reference") | 1 | U |  |

Content Item Descriptions

Row 2 | Reference to Radiation Dose SRs or Radiopharmaceutical Administration Dose SRs used in the dose estimation. At least one such SR shall be referenced. ### Note If an SR does not exist, one must be created from estimated data.
Row 3 | Reference to Fiducial SOP Instance that is used to register the Frame of Reference of the Radiation Dose SR.
Row 4 | Reference to Irradiation Event UIDs or Radiopharmaceutical Event UIDs used in the Radiation Dose Estimate Methodology. This shall not be present if all events in the SR are used.
Rows 8 and 9 | Reference to an instance that contains the model used when determining the radiation transport and deposition of energy within a patient, e.g., Surface Segmentation, Mesh, Parametric Map, etc.
Row 10 | Reference to the series of images that contains the model used when determining the radiation transport and deposition of energy within a patient,, e.g., CT, MRI, etc.
Row 11 | Reference to Publication describing the model used. If proprietary, reference the manufacturer model and version of software used.
Rows 13-20 | Provide the demographics used in the patient model to estimate dose. These are not necessarily the demographics of the actual patient.
Row 21 | Contains the Spatial Registration from each Source Radiation Dose SR Frame of Reference to the patient model Frame of Reference. The Frame of Reference of patient model is defined by the space of model coordinates. The Frame of Reference of the Source Radiation Dose SR is the Frame of Reference of the acquired patient images. If no Frame of Reference of the acquired patient images exists, fiducials can be used to define Frame of Reference in both the equipment space, i.e., Source Radiation Dose SR, and the Patient Model space and referenced in Row 5. If RCS Registration Method Type is Visual Alignment, it is assumed any translation/rotation information from the visual alignment is added to other alignment translation/rotation information and saved as a single Spatial Registration SOP Instance.
Row 25 | One content item per attenuator. This can be information about materials in the radiation beam that is used in the estimation method and that may or may not have been included in the Radiation Dose SR. If the beam Attenuator (e.g., filter) is included here and is also in the Radiation Dose SR it is assumed additional information relative to the beam Attenuator material, shape, size, location was needed and this information was not in the Radiation Dose SR or the Radiation Dose SR information is considered incorrect or incomplete.
Row 27 | The estimation method may use an equivalent material rather than the actual material, e.g., a plastic table may be use equivalent aluminum attenuation.
Row 28 | If the attenuator is not uniform, a thickness may still be provided and it is expected that Row 29 (Attenuator Description) will clarify how that thickness was determined. The specified equivalent material is identified in Row 27.
Row 29 | The attenuator characteristics may be described here. If the attenuator thickness was not provided in Row 28, the attenuator may still be described.
Row 30 | Complex attenuators are best described by a model.
Rows 33 and 34 | Reference to an Instance that contains the model e.g., Surface Segmentation, Mesh, Parametric Map, etc.
Row 35 | Reference to the Series of Images that contains the model, e.g., CT, MRI, etc. This can be a Spatial Fiducials SOP Instance.
Row 36 | Contains the Spatial Registration from each Source Radiation Dose SR Frame of Reference to the X-Ray attenuator model Frame of Reference. The Frame of Reference of the X-Ray attenuator model is defined by the space of model coordinates. The Frame of Reference of the Source Radiation Dose SR is the Frame of Reference of the acquired patient images. If no Frame of Reference of the acquired patient images exists, fiducials can be used to define Frame of Reference in both the equipment space, i.e., Source Radiation Dose SR, and X-Ray attenuator model space and referenced in Row 30. If RCS Registration Method Type is Visual Alignment it is assumed any translation/rotation information from the visual alignment is added to other alignment translation/rotation information and saved as a single Spatial Registration SOP Instance.
Row 32 and 43 | Provide name of method, reference to a publication or the manufacturer model and version

### TID 10034 Radiation Dose Estimate Parameters

This template includes the parameters that are specific to the Radiation Dose Estimate Method used in the algorithms when estimating dose to individual organs, phantoms, or the entire body from imaging studies that use ionizing radiation.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 10034. Radiation Dose Estimate Parameters

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (128434, DCM, "Radiation Dose Estimate Parameters") | 1 | M |  |
2 | > | CONTAINS | NUM | DCID 10069 “Radiation Dose Estimation Parameter ” | 1-n | MC | IF Row 4 absent | UNITS = DCID 82 “Units of Measurement”
4 | > | CONTAINS | COMPOSITE | EV (128436, DCM, "Radiation Dose Composite Parameters") | 1-n | MC | IF Row 2 absent |
5 | >> | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

Content Item Descriptions

Row 2 | These are the parameters of the method specified in Row 43 of TID 10033 “Radiation Dose Estimate Methodology”. The code meanings should correlate directly with the names of the parameters used in the methodology documentation.
Row 4 | References to Parametric Map Image, Mesh, encapsulated pdf, or other similar IOD.
Row 5 | Describes the contents of the IOD referenced in Row 4

# B DCMR Context Groups (Normative)

This Annex specifies the content of Context Groups required by DICOM IODs.

### Note

Section 7.1 of this Part defines the fields of Context Group tables.

## CID 2 Anatomic Modifier

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.1

Table CID 2. Anatomic Modifier

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A100 | Right | 24028007 | C0205090
SRT | G-A101 | Left | 7771000 | C0205091
SRT | G-A102 | Bilateral | 51440002 | C0238767
SRT | G-A103 | Unilateral | 66459002 | C0205092
SRT | G-A104 | Lateral | 49370004 | C0205093
SRT | R-404CC | Anterior | 255549009 | C1704448
SRT | R-404CE | Posterior | 255551008 | C0205095
SRT | G-A107 | Cephalic | 66787007 | C0205096
SRT | G-A108 | Caudal | 3583002 | C0205097
SRT | R-404D5 | Medial | 255561001 | C0205098
SRT | G-A110 | Central | 26216008 | C0205099
SRT | G-A111 | Peripheral | 14414005 | C0205100
SRT | R-40941 | External | 261074009 | C0205101
SRT | R-40819 | Internal | 260521003 | C0205102
SRT | G-A114 | Intermediate | 11896004 | C0205103
SRT | R-4094A | Inferior | 261089000 | C0542339
SRT | R-42191 | Superior | 264217000 | C1282910
SRT | G-A117 | Transverse | 62824007 | C0205106
SRT | G-A118 | Proximal | 40415009 | C0205107
SRT | G-A119 | Distal | 46053002 | C0205108
SRT | G-A120 | Postaxial | 60583000 | C0205109
SRT | G-A121 | Preaxial | 32400000 | C0205110
SRT | G-A122 | Apical | 43674008 | C0205111
SRT | G-A123 | Basal | 57195005 | C0205112
SRT | G-A127 | Afferent | 49530007 | C0205115
SRT | G-A128 | Efferent | 33843005 | C0205116
SRT | G-A138 | Coronal | 81654009 | C0205123
SRT | G-A139 | Superficial | 26283006 | C0205124
SRT | G-A140 | Deep | 795002 | C0205125
SRT | G-A142 | Horizontal | 24020000 | C0205126
SRT | G-A143 | Longitudinal | 38717003 | C0205127
SRT | G-A144 | Vertical | 33096000 | C0205128
SRT | G-A145 | Sagittal | 30730003 | C0205129
SRT | G-A147 | Axial | 24422004 | C0205131
SRT | G-A151 | Extra-articular | 87687004 | C0205135
SRT | G-A206 | Surface | 410679008 | C0205148
SRT | G-A169 | Gutter | 68493006 | C0205149
SRT | G-A170 | Hilar | 32381004 | C0205150
SRT | G-A171 | Capsular | 11070000 | C0205151
SRT | G-A172 | Subcapsular | 61397002 | C0205152
SRT | G-A174 | Edge | 57183005 | C0205154
SRT | G-A180 | Anterolateral | 37197008 | C0332194
SRT | G-A182 | Posterolateral | 90069004 | C0332195
SRT | G-A15A | Intra-articular | 131183008 | C0442108
SRT | G-A428 | Marginal | 112233002 | C0205284

## CID 4 Anatomic Region

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.2

Table CID 4. Anatomic Region

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 4030 “CT, MR and PET Anatomy Imaged” |  |
Include CID 4040 “Endoscopy Anatomic Regions” |  |
Include CID 4042 “XA/XRF Anatomy Imaged” |  |
SRT | T-32100 | Atrium | 59652004 | C0018792
SRT | T-D8104 | Axilla | 91470000 | C0004454
SRT | T-D2100 | Back | 77568009 | C0004600
SRT | T-D6500 | Broad ligament | 34411009 | C0006205
SRT | T-D1206 | Buccal region of face | 60819002 | C0007966
SRT | T-D2600 | Buttock | 46862004 | C0006497
SRT | T-72100 | Calyx | 2334006 | C0022651
SRT | T-D1206 | Cheek | 60819002 | C0007966
SRT | T-AA200 | Cornea | 28726007 | C0010031
SRT | T-AB001 | Ear | 117590005 | C0013443
SRT | T-41000 | Endo-arterial | 51114001 | C0003842
SRT | T-32000 | Endo-cardiac | 80891009 | C0018787
SRT | T-56000 | Endo-esophageal | 32849002 | C0014876
SRT | T-21300 | Endo-nasal | 53342003 | C0225425
SRT | T-23050 | Endo-nasopharyngeal | 18962004 | C0225497
SRT | T-59600 | Endo-rectal | 34402009 | C0034896
SRT | T-71000 | Endo-renal | 64033007 | C0022646
SRT | T-73000 | Endo-ureteric | 87953007 | C0041951
SRT | T-75000 | Endo-urethral | 13648007 | C0041967
SRT | T-82000 | Endo-vaginal | 76784001 | C0042232
SRT | T-40000 | Endo-vascular | 59820001 | C0005847
SRT | T-48000 | Endo-venous | 29092000 | C0042449
SRT | T-74250 | Endo-vesical | 48367006 | C0227710
SRT | T-D4200 | Epigastric region | 27947004 | C0230185
SRT | T-AA810 | Eyelid | 80243003 | C0015426
SRT | T-D1200 | Face | 89545001 | C0015450
SRT | T-D2310 | Flank | 58602004 | C0230171
SRT | T-15200 | Fontanel of skull | 79361005 | C0224548
SRT | T-D2600 | Gluteal region | 46862004 | C0006497
SRT | T-15710 | Hip joint | 24136001 | C0019558
SRT | T-D4240 | Hypogastric region | 11708003 | C0230189
SRT | T-D161E | Submental | 170887008 | C0931905
SRT | T-55300 | Hypopharynx | 81502006 | C0020629
SRT | T-D4010 | Intra-abdominal | 52731004 | C0230168
SRT | G-A15A | Intra-articular | 131183008 | C0442108
SRT | T-D1400 | Intracranial | 1101003 | C0230041
SRT | T-56000 | Intra-esophageal | 32849002 | C0014876
SRT | T-D6221 | Intra-pelvic | 21844003 | C0559769
SRT | T-D3000 | Intra-thoracic | 51185008 | C0817096
SRT | T-D4211 | Left hypochondriac region | 133945003 | C0738591
SRT | T-D7020 | Left inguinal region | 85119005 | C0230321
SRT | T-D4140 | Left lower quadrant of abdomen | 68505006 | C0230180
SRT | T-D2340 | Left lumbar region | 133943005 | C1297910
SRT | T-D4130 | Left upper quadrant of abdomen | 86367003 | C0230179
SRT | T-04003 | Lower inner quadrant of breast | 19100000 | C0222597
SRT | T-04005 | Lower outer quadrant of breast | 33564002 | C0222599
SRT | T-D2300 | Lumbar region | 52612000 | C0024090
SRT | T-28000 | Lung | 39607008 | C0024109
SRT | T-D0662 | Mouth | 123851003 | C0230028
SRT | T-21000 | Nose | 45206002 | C0028429
SRT | T-D4450 | Omental bursa | 113346000 | C0230212
SRT | T-D4600 | Omentum | 27398004 | C0028977
SRT | T-87000 | Ovary | 15497006 | C0029939
SRT | T-65010 | Pancreatic duct | 69930009 | C0030288
SRT | T-D3136 | Parasternal | 91691001 | C0458345
SRT | T-91000 | Penis | 18911002 | C0030851
SRT | T-D2700 | Perineum | 38864007 | C0031066
SRT | T-D9310 | Popliteal fossa | 32361000 | C0230436
SRT | T-72000 | Renal pelvis | 25990002 | C0227666
SRT | T-D4900 | Retroperitoneum | 82849001 | C0035359
SRT | T-D4212 | Right hypochondriac region | 133946002 | C0738590
SRT | T-D7010 | Right inguinal region | 37117007 | C0230318
SRT | T-D4120 | Right lower quadrant of abdomen | 48544008 | C0230178
SRT | T-D2342 | Right lumbar region | 133944004 | C1297911
SRT | T-D4110 | Right upper quadrant of abdomen | 50519007 | C0230177
SRT | T-D1160 | Scalp | 41695006 | C0036270
SRT | T-AA110 | Sclera | 18619003 | C0036410
SRT | T-98000 | Scrotum | 20233005 | C0036471
SRT | T-A7010 | Spinal cord | 2748008 | C0037925
SRT | T-D4210 | Subcostal | 19695001 | C0442184
SRT | T-D1603 | Submandibular area | 5713008 | C0230070
SRT | T-D3213 | Subxiphoid | 5076001 | C0230144
SRT | T-D1620 | Supraclavicular region of neck | 77621008 | C0230078
SRT | T-D4240 | Suprapubic region | 11708003 | C0230189
SRT | T-11218 | Suprasternal notch | 26493002 | C0222769
SRT | T-D9100 | Thigh | 68367000 | C0039866
SRT | T-D3000 | Thorax | 51185008 | C0817096
SRT | T-53000 | Tongue | 21974007 | C0040408
SRT | T-D4230 | Umbilical region | 90290004 | C0041638
SRT | T-04002 | Upper inner quadrant of breast | 77831004 | C0222596
SRT | T-04004 | Upper outer quadrant of breast | 76365002 | C0222598
SRT | T-82000 | Vagina | 76784001 | C0042232
SRT | A-04140 | Vascular graft | 118375008 | C1289794
SRT | T-32400 | Ventricle | 21814001 | C0018827
SRT | T-81000 | Vulva | 45292006 | C0042993
SRT | T-15460 | Wrist joint | 74670003 | C1322271

## CID 5 Transducer Approach

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.3

Table CID 5. Transducer Approach

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A100 | Right | 24028007 | C0205090
SRT | G-A101 | Left | 7771000 | C0205091
SRT | G-A104 | Lateral | 49370004 | C0205093
SRT | R-404CC | Anterior | 255549009 | C1704448
SRT | R-404CE | Posterior | 255551008 | C0205095
SRT | G-A108 | Caudal | 3583002 | C0205097
SRT | R-404D5 | Medial | 255561001 | C0205098
SRT | G-A110 | Central | 26216008 | C0205099
SRT | G-A111 | Peripheral | 14414005 | C0205100
SRT | R-40941 | External | 261074009 | C0205101
SRT | R-40819 | Internal | 260521003 | C0205102
SRT | R-4094A | Inferior | 261089000 | C0542339
SRT | R-42191 | Superior | 264217000 | C1282910
SRT | G-A117 | Transverse | 62824007 | C0205106
SRT | G-A118 | Proximal | 40415009 | C0205107
SRT | G-A119 | Distal | 46053002 | C0205108
SRT | G-A122 | Apical | 43674008 | C0205111
SRT | G-A206 | Surface | 410679008 | C0205148
SRT | G-A599 | Ascending | 79458005 | C0205385
SRT | G-A600 | Descending | 75294000 | C0205386
SRT | T-03000 | Subcutaneous tissue | 71966008 | C0278403
SRT | T-A1120 | Dura mater | 18545000 | C0013313
SRT | T-A1280 | Pia mater | 23180006 | C0031869
SRT | A-2C600 | External prosthesis for sonographic procedure [Stand-off] | 102322008 | C0522650
SRT | A-2C602 | Water bag prosthesis for imaging procedure | 102323003 | C0522651
SRT | A-2C604 | Saline bag prosthesis for imaging procedure | 102324009 | C0522652
SRT | A-2C606 | Gel prosthesis for imaging procedure | 102325005 | C0522653
SRT | G-A107 | Cranial | 66787007 | C0205096
SRT | G-A10A | Midline | 261129000 | C0441992
SRT | G-A188 | Mid-longitudinal | 103342007 | C0522490
SRT | G-A189 | Parasagittal | 103343002 | C0522491
SRT | R-42142 | Intraluminal | 264045001 | C0442115
SRT | G-A171 | Capsular | 11070000 | C0205151
SRT | T-D0048 | Lumen | 113342003 | C0524461
SRT | G-4022 | Contact with | 11723008 | C0332158
SRT | T-D0062 | Parenchyma | 91772007 | C0524464

### Note

In a prior version of this Context Group, the codes G-A11A, G-A11B, G-A12A, G-A16A, G-A16B, G-A16C, and G-A16D were specified for various concepts. The use of some of those codes conflicts with their assignment to other concepts in SNOMED, and their use in this context is deprecated. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 6 Transducer Orientation

Type:

  Extensible

Version:

  20040322

UID:

  1.2.840.10008.6.1.4

Table CID 6. Transducer Orientation

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A138 | Coronal | 81654009 | C0205123
SRT | G-A143 | Longitudinal | 38717003 | C0205127
SRT | G-A145 | Sagittal | 30730003 | C0205129
SRT | G-A189 | Parasagittal | 103343002 | C0522491
SRT | G-A472 | Oblique | 21114003 | C0205315
SRT | G-A185 | Long axis | 103339001 | C0522487
SRT | G-A13B | Off axis | 419161000 | C1635161
SRT | G-A186 | Short axis | 103340004 | C0522488
SRT | G-A191 | Five chamber | 398994001 | C1302157
SRT | G-A19B | Two chamber | 399232001 | C1302267
SRT | G-A19C | Four chamber | 399214001 | C1302256
SRT | G-A117 | Transverse | 62824007 | C0205106

### Note

In a prior version of this table, the code G-A11B was specified for the concept Parasagittal. The use of this code conflicts with its assignment to another concept in SNOMED, and its use in this context is deprecated. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 7 Ultrasound Beam Path

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.5

Table CID 7. Ultrasound Beam Path

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-D027 | Trans-hepatic | 103381007 | C0522516
SRT | G-A1B2 | Trans-gastric | 103353001 | C0442355
SRT | G-A1A5 | Trans-pleural | 103348006 | C0522494
SRT | G-A1B3 | Trans-mural | 103354007 | C0522497
SRT | G-D065 | Trans-orbital | 129226004 | C0442367
SRT | G-A1A6 | Trans-pancreatic | 103349003 | C0522495
SRT | G-A1A4 | Trans-renal | 103347001 | C0522493
SRT | G-D032 | Trans-temporal | 103382000 | C0522517
SRT | G-A1A2 | Trans-thecal | 103345009 | C0522492
SRT | G-A1A1 | Trans-vesical | 103344008 | C0442393
SRT | G-A1A3 | Trans-splenic | 103346005 | C0589466
SRT | G-D033 | Trans-esophageal | 103383005 | C0522518
SRT | G-D001 | Trans-abdominal | 66739002 | C0205496
SRT | G-D002 | Trans-vaginal (endovaginal) | 54300008 | C0175672

## CID 8 Angiographic Interventional Devices

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.6

Table CID 8. Angiographic Interventional Devices

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-25500 | Stent | 65818007 | C0038257
SRT | A-26800 | Catheter | 19923001 | C0085590
SRT | A-81080 | Laser | 38586004 | C0458142
SRT | C-20005 | Glue | 57126000 | C0017780
SRT | A-25600 | Atherectomy device | 102312002 | C0522642
SRT | A-25614 | Embolization ball | 102315000 | C0522645
SRT | A-26912 | Percutaneous transluminal angioplasty balloon | 102319006 | C0522648
SRT | A-25612 | Embolization coil | 102314001 | C0522644
SRT | A-25612 | Gianturco coil | 102314001 | C0522644
SRT | A-27322 | Detachable balloon | 102320000 | C0522649
SRT | A-26802 | Guiding catheter | 102317008 | C0221799
SRT | A-25616 | Embolization particulate | 102316004 | C0522646
SRT | A-25610 | Rotational atherectomy device | 102313007 | C0522643
SRT | A-10141 | Measuring ruler | 102304005 | C0522637
DCM | 122485 | Sphere |  |

## CID 9 Image Guided Therapeutic Procedures

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.7

Table CID 9. Image Guided Therapeutic Procedures

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-39780 | Vasoconstriction | 32318003 | C0042396
SRT | F-39800 | Vasodilatation | 30017007 | C0042401
SRT | P1-03100 | Biopsy | 86273004 | C0005558
SRT | P1-03176 | Removal of foreign body | 10849003 | C0184937
SRT | P1-05035 | Intra-arterial infusion of thrombolytic agent | 69245005 | C0184952
SRT | P1-05052 | Irrigation following insertion of catheter | 8592001 | C0022101
SRT | P1-05535 | Catheterization | 45211000 | C0007430
SRT | P1-30350 | Atherectomy | 6832004 | C0162513
SRT | P1-30351 | Atherectomy by rotary cutter | 65659003 | C0162655
SRT | P1-30352 | Atherectomy by laser | 76611008 | C0521229
SRT | P1-30530 | Selective embolization of artery | 57238002 | C0189632
SRT | P5-31500 | Percutaneous transluminal balloon angioplasty | 68457009 | C0411287
SRT | P5-39010 | Transcatheter therapy for embolization | 16736007 | C0203006
SRT | P0-05AFA | Percutaneous retrieval of intravascular foreign body | 240946003 | C0411305
SRT | P1-00018 | Failed attempted procedure | 103709008 | C0522770
SRT | P1-05550 | Stent placement | 103716009 | C0522776
SRT | P1-05536 | Catheter manipulation | 103712006 | C0522773
SRT | P1-05537 | Catheter replacement | 103713001 | C0522774
SRT | P1-05538 | Occlusion of catheter | 103714007 | C0522775
SRT | P1-05539 | Removal of catheter | 103715008 | C0394884
SRT | P5-39015 | Transcatheter deployment of detachable balloon | 105372003 | C0524313
SRT | P5-39191 | Percutaneous insertion of intravascular filter | 105373008 | C0524314
SRT | P1-86100 | Amniocentesis | 34536000 | C0002627
SRT | P5-B8310 | Ultrasonic guidance for amniocentesis | 65388005 | C0203432
SRT | P1-86520 | Amnioinfusion [injection of amnion] | 15415002 | C0521272
SRT | P1-86180 | Intrauterine cordocentesis | 6708002 | C0162650
SRT | P1-28160 | Thoracentesis | 91602002 | C0189477
SRT | P1-86E70 | Breech Version [Obstetrical Version] | 65240009 | C0195731
SRT | P2-68060 | Intrauterine transfusion | 45460008 | C0005843
SRT | P1-86C50 | Fetocide (selective reduction) | 133874006 | C1297889
SRT | P1-93506 | Prostaglandin injection | 133875007 | C1297890

## CID 10 Interventional Drug

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.8

Table CID 10. Interventional Drug

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-21047 | Ethanol | 419442005 | C0001962
SRT | C-22947 | Methylene blue | 6725000 | C0025746
SRT | C-51000 | Antihistamine | 6425004 | C0003360
SRT | C-67770 | Atropine | 73949004 | C0004259
SRT | C-72000 | Diuretic | 30492008 | C0012798
SRT | C-80110 | Antiarrhythmic drug | 67507000 | C0003195
SRT | C-80120 | Inotropic agent | 111139005 | C0304509
SRT | C-80123 | Cardiotonic drug | 69440003 | C0007209
SRT | F-6181D | Cardiac adrenergic blocking agent | 373263004 | C1277070
SRT | C-80131 | Alpha-adrenergic blocking agent | 67440007 | C0001641
SRT | C-80135 | beta-Adrenergic blocking agent | 33252009 | C0001645
SRT | C-80330 | Digoxin | 796001 | C0012265
SRT | C-80400 | Lidocaine | 82573000 | C0023660
SRT | C-80401 | Lidocaine hydrochloride | 61773008 | C0546869
SRT | C-80430 | Nifedipine | 85272000 | C0028066
SRT | C-80450 | Propranolol | 55745002 | C0033497
SRT | C-80460 | Quinidine | 31306009 | C0034414
SRT | C-80490 | Verapamil | 47898004 | C0042523
SRT | C-81100 | Hypotensive agent | 1182007 | C0003364
SRT | C-81120 | Centrally acting hypotensive agent | 4382004 | C0304523
SRT | C-81560 | Nitroglycerin | 71759000 | C0017887
SRT | C-A2010 | Glucagon preparation | 10712001 | C0017687
SRT | C-A6500 | Anticoagulant | 81839001 | C0003280
SRT | C-A6530 | Warfarin | 48603004 | C0043031
SRT | C-A6540 | Heparin | 84812008 | C0019134
SRT | C-A6700 | Anti-heparin agent | 3361000 | C0304941
SRT | C-A6710 | Protamine sulfate | 64520006 | C0033602
SRT | C-A6900 | Coagulant | 15117003 | C0009117
SRT | F-D7011 | Human fibrinogen | 418326009 | C2587184
SRT | C-A7000 | Hemostatic agent | 26370007 | C0019120
SRT | C-A7001 | Astringent drug | 60533005 | C0004110
SRT | C-A7021 | Antihemophilic factor preparation | 59057006 | C0301494
SRT | F-6ACA0 | Thrombin preparation | 36176003 | C0040018
SRT | F-D7B50 | Thromboplastin | 65265006 | C0040048
SRT | C-A7220 | Dextran | 13132007 | C0086140
SRT | C-50434 | Thrombolytic agent | 303960004 | C0016018
SRT | C-A7420 | Streptokinase preparation | 20847002 | C0038418
SRT | C-A7430 | Urokinase preparation | 59082006 | C0042071
SRT | C-A7440 | Injectable fibrinolysin | 87811005 | C0301485
SRT | C-815E1 | Tolazoline hydrocholoride | 19041007 | C0770500
SRT | F-B2135 | Epinephrine | 387362001 | C0014563

## CID 11 Route of Administration

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.9

Table CID 11. Route of Administration

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-D101 | Intravenous route | 47625008 | C1522726
SRT | G-D102 | Intra-arterial route | 58100008 | C1561451
SRT | G-D103 | Intramuscular route | 78421000 | C1556154
SRT | G-D104 | Subcutaneous route | 34206005 | C1522438
SRT | G-D17D | Intracutaneous route | 372464004 | C1522475
SRT | G-D106 | Intraperitoneal route | 38239002 | C1522583
SRT | G-D107 | Intramedullary route | 60213007 | C1512957
SRT | G-D108 | Intrathecal route | 72607000 | C0677897
SRT | G-D109 | Intra-articular route | 12130007 | C0205528
NCIt | C38244 | Intraepithelial route |  | C1512943
SRT | G-D112 | Topical route | 6064005 | C1522168
SRT | G-D140 | Oral route | 26643006 | C1527415
NCIt | C38306 | Transluminal route |  | C1522231
SRT | G-D144 | Intraluminal route | 37737002 | C1522217
NCIt | C38213 | Extraluminal route |  | C1517059
SRT | R-40B32 | By inhalation | 446406008 | C1998547
SRT | G-D160 | Per rectum | 37161004 | C1527425
SRT | G-D164 | Vaginal route | 16857009 | C1522570
SRT | G-D17C | Intracoronary route | 372463005 | C0595454
SRT | G-D173 | Intracardiac route | 372460008 | C1522207
SRT | R-F2C86 | Intraventricular route - cardiac | 420287000 | C1720462
DCM | 127070 | Retro-orbital route |  |
SRT | G-D172 | Nasal route | 46713006 | C1522019
SRT | G-D17D | Intradermal route | 372464004 | C1522475
SRT | R-F2CD4 | Intratumor route | 447122006 | C2960749

## CID 12 Radiographic Contrast Agent

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.10

Table CID 12. Radiographic Contrast Agent

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Trade Name (Informative)
--- | --- | --- | --- | --- | ---
SRT | A-80230 | Air | 15158005 | C0001861 |
SRT | C-10110 | Oxygen | 24099007 | C0030054 |
SRT | C-10120 | Water | 11713004 | C0043047 |
SRT | C-10520 | Carbon dioxide | 31811003 | C0007012 |
SRT | C-12217 | Barium Sulfate | 25419009 | C0004754 |
SRT | C-17800 | Gadolinium | 58281002 | C0016911 |
SRT | C-B0300 | Contrast agent | 7140000 | C2930749 |
SRT | C-B0300 | Radiopaque medium | 7140000 | C2930749 |
SRT | C-B0312 | Non radiopaque medium | 43538006 | C0301446 |
SRT | C-B0315 | Bunamiodyl | 90745007 | C0623554 |
SRT | C-B0316 | Chloriodized oil | 62442005 | C0301444 |
SRT | C-B0317 | Diatrizoate | 12335007 | C0012004 | Angiovist™(Berlex), Cardiografin™(Bracco), Cystografin™(Bracco), Gastrografin™(Bracco), Gastrovist™(Berlex), Hypaque™(GE), MD-nn™(Mallinckrodt), Reno-nn™(Bracco), Renografin™(Bracco), Renovist™(Bracco), Sinografin™(Bracco), Urovist™(Berlex)
SRT | C-B0318 | Iodipamide | 73212002 | C0021971 | Cholographin™(Bracco), Sinografin™(Bracco)
SRT | C-B0319 | Iodized oil | 89595000 | C0021972 |
SRT | C-B0323 | Iodoalphionic acid | 86584005 | C0063766 |
SRT | C-B0324 | Meglumine iodipamide | 69783005 | C0065885 | Cholographin Meglumine™(Bracco)
SRT | C-B0325 | Sodium iodipamide | 925002 | C0301445 | Cholographin Sodium™(Bracco)
SRT | C-B0326 | Iodamide meglumine | 12801003 | C0065884 | Renovue™(Bracco)
SRT | C-B0327 | Iodopyracet | 40710000 | C0021990 |
SRT | C-B0328 | Iopanoic acid | 76155001 | C0022028 | Telepaque™(GE)
SRT | C-B0331 | Iophendylate | 28121005 | C0022029 | Pantopaque™(Alcon)
SRT | C-B0333 | Iophenoxic acid | 23053002 | C0063816 |
SRT | C-B0335 | Ipodate | 87445005 | C0022049 | Bilivist™(Berlex), Oragrafin™(Bracco)
SRT | C-B0337 | Propyliodone | 111158001 | C0033509 | Dionosil™(GSK)
SRT | C-B0338 | Sodium acetrizoate | 32836007 | C0546847 | Salpix™(Ortho)
SRT | C-B0341 | Iodophthalein | 74554008 | C0163095 |
SRT | C-B0342 | Sodium diprotrizoate | 83423008 | C0301447 |
SRT | C-B0344 | Sodium iodomethamate | 38344006 | C0301448 |
SRT | C-B0345 | Meglumine diatrizoate | 47192000 | C0012005 | Angiovist™(Berlex), Cardiografin™(Bracco), Cystografin™(Bracco), Gastrografin™(Bracco), Gastrovist™(Berlex), Hypaque™(GE), MD-nn™(Mallinckrodt), Reno-nn™(Bracco), Renografin™(Bracco), Renovist™(Bracco), Sinografin™(Bracco), Urovist™(Berlex)
SRT | C-B0347 | Sodium diatrizoate | 24891006 | C0012007 | Angiovist™(Berlex), Gastrografin™(Bracco), Gastrovist™(Berlex), Hypaque™(GE), MD-nn™(Mallinckrodt), Renografin™(Bracco), Renovist™(Bracco), Urovist™(Berlex)
SRT | C-B0348 | Metrizamide | 90733003 | C0025869 | Amipaque™(GE)
SRT | C-B0314 | Sodium tyropanoate | 109212003 | C0936260 |
SRT | C-B0301 | Ionic iodinated contrast agent | 96387000 | C0361904 |
DCM | 127855 | Non-ionic iodinated contrast agent |  | C0521968 |
SRT | C-B0322 | Iohexol | 109218004 | C0022005 | Omnipaque™(GE)
SRT | C-B03BC | Iodixanol | 353962003 | C0063757 | Visipaque™(GE)
SRT | C-B03C3 | Gadodiamide | 354088005 | C0082646 | Omniscan™(GE)
SRT | C-B05A3 | Mangafodipir trisodium | 410873007 | C0067297 | Teslascan™(GE)
SRT | C-B038B | Iothalamate | 353912008 | C0022032 | Conray™(Mallinckrodt), Cysto-Conray™(Mallinckrodt), Vascoray™(Mallinckrodt)
SRT | C-B0339 | Ioxaglate | 109223004 | C0205807 | Hexbrix™(Mallinckrodt)
SRT | C-B03C9 | Metrizoate | 354094002 | C0025870 | Isopaque™(GE)
SRT | C-B014D | Gadopentetate dimeglumine | 404846007 | C0060934 | Magnevist™(Berlex)
SRT | C-B0329 | Iopamidol | 109219007 | C0022026 | Isovue™(Bracco)
SRT | C-B0332 | Ioversol | 109222009 | C0063828 | Optiray™(Mallinckrodt)
SRT | C-B0382 | Iopromide | 353903006 | C0063817 | Ultravist or Imeron
SRT | C-B0303 | Ioxilan | 409484007 | C0063829 | Imagenil

### Note

- The codes drawn from SNOMED are recommended to be those from the concept hierarchy of "radiographic contrast media" in the hierarchy "pharmaceutical/biological product", and secondarily from the hierarchy "substance".

- Trade names are from http://www.accessdata.fda.gov/scripts/cder/drugsatfda/index.cfm.

## CID 13 Radiographic Contrast Agent Ingredient

Type:

  Extensible

Version:

  20051101

UID:

  1.2.840.10008.6.1.11

Table CID 13. Radiographic Contrast Agent Ingredient

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-11400 | Iodine | 44588005 | C0021968
SRT | C-17800 | Gadolinium | 58281002 | C0016911
SRT | C-10520 | Carbon Dioxide | 31811003 | C0007012
SRT | C-12200 | Barium | 39290007 | C0004749
SRT | C-17200 | Xenon | 83598005 | C0043339
SRT | A-80230 | Air | 15158005 | C0001861
SRT | C-10110 | Oxygen | 24099007 | C0030054
SRT | C-10120 | Water | 11713004 | C0043047
SRT | C-130F9 | Iron | 105840005 | C0303213

## CID 18 Isotopes in Radiopharmaceuticals

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.12

Table CID 18. Isotopes in Radiopharmaceuticals

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-105A2 | ^14^Carbon | 71647005 | C0302945
SRT | C-111A1 | ^18^Fluorine | 77004003 | C0302995
SRT | C-155A1 | ^22^Sodium | 71633006 | C0303511
SRT | C-155A2 | ^24^Sodium | 58541008 | C0303512
SRT | C-106A1 | ^32^Phosphorus | 32505007 | C0851287
SRT | C-135A2 | ^42^Potassium | 59844004 | C0303277
SRT | C-135A3 | ^43^Potassium | 8202008 | C0303278
SRT | C-129A2 | ^51^Chromium | 52745005 | C0303212
SRT | C-144A3 | ^57^Cobalt | 27054007 | C0303392
SRT | C-144A4 | ^58^Cobalt | 89272005 | C0303393
SRT | C-130A3 | ^59^Iron | 68580003 | C0303220
SRT | C-144A6 | ^60^Cobalt | 5405008 | C0303395
SRT | C-127A2 | ^64^Copper | 3932008 | C0303190
SRT | C-127A3 | ^67^Copper | 53700003 | C0303191
SRT | C-131A2 | ^67^Gallium | 2008008 | C0303225
SRT | C-116A3 | ^75^Selenium | 43239002 | C0303048
SRT | C-173A5 | ^81m^Krypton | 61716009 | C0303689
SRT | C-173A7 | ^85^Krypton | 34127007 | C0303691
SRT | C-158A3 | ^85^Strontium | 111084009 | C0303544
SRT | C-158A5 | ^87m^Strontium | 78023008 | C0303546
SRT | C-158A6 | ^89^Strontium | 7770004 | C0281385
SRT | C-162A7 | ^90^Yttrium | 14691008 | C0303596
SRT | C-180A2 | ^97^Ruthenium | 23788009 | C0303730
SRT | C-163A8 | ^99m^Technetium | 72454006 | C0303611
SRT | C-145A4 | ^111^Indium | 56609000 | C0303403
SRT | C-145A5 | ^113m^Indium | 48895003 | C0303404
SRT | C-114A4 | ^123^Iodine | 21572004 | C0303023
SRT | C-114A6 | ^125^Iodine | 68630002 | C0796396
SRT | C-172A5 | ^127^Xenon | 27081007 | C0303677
SRT | C-114B1 | ^131^Iodine | 1368003 | C0303029
SRT | C-122A5 | ^133^Barium | 3027009 | C0303126
SRT | C-172A8 | ^133^Xenon | 80751004 | C0872916
SRT | C-178A8 | ^153^Gadolinium | 14529005 | C0303714
SRT | C-B1134 | ^153^Samarium | 419804008 | C0677942
SRT | C-181A3 | ^169^Ytterbium | 41758004 | C0303739
SRT | C-101ED | ^177^Lutetium | 447553000 | C2959378
SRT | C-156A6 | ^178^Tantalum | 6301006 | C0303521
SRT | C-11906 | ^186^Rhenium | 395865006 | C1273039
SRT | C-1018D | ^188^Rhenium | 423578007 | C1828331
DCM | 126604 | ^191m^Iridium |  |
SRT | C-146A9 | ^198^Gold | 24301009 | C0303420
SRT | C-146B1 | ^199^Gold | 70544003 | C0303421
SRT | C-138A9 | ^201^Thallium | 60057003 | C0303322
SRT | C-132A8 | ^203^Lead | 47588004 | C0303240
SRT | C-136A2 | ^223^Radium | 24853006 | C0303282

### Note

The use of this Context Group in the Radionuclide Code Sequence (0054,0300) of the “Nuclear Medicine Image IOD” in PS3.3 requires a Coding Scheme Designator value of "99SDM".

## CID 19 Patient Orientation

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.13

Table CID 19. Patient Orientation

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
NCIt | C86043 | erect |  | C0522015
SRT | F-10450 | recumbent | 102538003 | C0444334
SRT | F-10460 | semi-erect | 102539006 | C0522018

### Note

1. The use of this Context Group in the Patient Orientation Code Sequence (0054,0410) of the “Nuclear Medicine Image IOD” in PS3.3 and the “Positron Emission Tomography Image IOD” in PS3.3 requires a Coding Scheme Designator value of "99SDM".

2. In a prior version of this Context Group (F-10440, SRT, "Upright body position (finding)") was specified for the concept "erect" but has been inactivated as being ambiguous, with no suitable replacement. Accordingly the NCIt concept of "upright" defined as "in a vertical position or posture" is used as a replacement.

## CID 20 Patient Orientation Modifier

Type:

  Extensible

Version:

  20070524

UID:

  1.2.840.10008.6.1.14

Table CID 20. Patient Orientation Modifier

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-10310 | prone | 1240000 | C0033422
SRT | F-10316 | semi-prone | 34026001 | C0150435
SRT | F-10318 | lateral decubitus | 32185000 | C0444379
SRT | F-10320 | standing | 10904000 | C0231472
SRT | F-10326 | anatomical | 51845000 | C0277809
SRT | F-10330 | kneeling | 55864004 | C1260920
SRT | F-10336 | knee-chest | 23242002 | C0277810
SRT | F-10340 | supine | 40199007 | C0038846
SRT | F-10346 | lithotomy | 14205002 | C0150665
SRT | F-10348 | Trendelenburg | 34106002 | C0277812
SRT | F-10349 | inverse Trendelenburg | 26527006 | C0277813
SRT | F-10380 | frog | 34296003 | C0426962
SRT | F-10390 | stooped-over | 87068006 | C0231478
SRT | F-103A0 | sitting | 33586001 | C0277814
SRT | F-10410 | curled-up | 34108001 | C0277815
SRT | F-10317 | right lateral decubitus | 102535000 | C0559228
SRT | F-10319 | left lateral decubitus | 102536004 | C0559227
SRT | R-40799 | lordotic | 260450008 | C0442217

### Note

The use of this Context Group in the Patient Orientation Modifier Code Sequence (0054,0412) of the “Nuclear Medicine Image IOD” in PS3.3 and the “Positron Emission Tomography Image IOD” in PS3.3 requires a Coding Scheme Designator value of "99SDM".

## CID 21 Patient Equipment Relationship

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.15

Table CID 21. Patient Equipment Relationship

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-10516 | oblique | 399366008 | C1302343
SRT | F-10470 | headfirst | 102540008 | C0522020
SRT | F-10480 | feet-first | 102541007 | C0522022
SRT | R-10515 | transverse | 399220000 | C1302259
DCM | 126830 | left first |  |
DCM | 126831 | right first |  |
DCM | 126832 | posterior first |  |
DCM | 126833 | anterior first |  |

### Note

1. The use of this Context Group in the Patient Orientation Modifier Code Sequence (0054,0412) of the “Nuclear Medicine Image IOD” in PS3.3 and the “Positron Emission Tomography Image IOD” in PS3.3 requires a Coding Scheme Designator value of "99SDM".

2. In a prior version of this Context Group, the codes G-5190 and G-5191 were specified for the concepts "headfirst" and "feet-first". The use of these codes is deprecated as they are not actually in SNOMED. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

3. For devices that do not have a conventional gantry geometry, the vendor of that device may describe in the Conformance Statement how the codes are interpreted with respect to the device geometry.

4. (126830, DCM, "left first"), (126831, DCM, "right first), (126832, DCM, "posterior first) and (126833, DCM, "anterior first) are more specific than (R-10515, SRT, "transverse") in that they specify which side of the patient is towards the front of the equipment.

5. For quadrupeds, separate concepts for ventral and dorsal are not introduced, rather it is expected that anterior and posterior will be considered synonymous as they are when applied to the trunk.

## CID 23 Cranio-Caudad Angulation

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.16

Table CID 23. Cranio-Caudad Angulation

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A107 | Cephalic | 66787007 | C0205096
SRT | G-A108 | Caudal | 3583002 | C0205097

## CID 25 Radiopharmaceuticals

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.17

Table CID 25. Radiopharmaceuticals

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Trade name (Informative)
--- | --- | --- | --- | --- | ---
SRT | C-B1302 | Carbon^14^ D-xylose | 2942001 | C0305043 |
SRT | C-B1300 | Carbon^14^ triolein | 42417005 | C0305042 |
SRT | C-B1304 | Cholyl-carbon^14^ glycine | 70086001 | C0305044 |
SRT | C-B1140 | Chromic phosphate P^32^ | 17069007 | C0392428 |
SRT | C-B1012 | Chromium^51^ albumin | 4693006 | C0304956 |
SRT | C-B1013 | Chromium^51^ chloride | 6973004 | C0304957 |
SRT | C-B1051 | Colloidal gold Au^198^ | 37947008 | C0304966 |
SRT | C-B1063 | Colloidal Indium^111^ | 30825005 | C0304969 |
SRT | C-B1017 | Copper^64^ acetate | 78686003 | C0304959 |
SRT | C-B1016 | Copper^64^ versenate | 88166005 | C0304958 |
SRT | C-B1018 | Copper^67^ ceruloplasmin | 29460005 | C0304960 |
SRT | C-B1021 | Cyanocobalamin Co^57^ | 187006 | C0304961 |
SRT | C-B1022 | Cyanocobalamin Co^58^ | 5692007 | C0304962 |
SRT | C-B1023 | Cyanocobalamin Co^60^ | 72159005 | C0304963 |
SRT | C-B1000 | Diagnostic radioisotope | 17600005 | C0360048 |
SRT | C-B1092 | Diiodofluorecein I^131^ | 53207004 | C0304989 |
SRT | C-B1062 | Disodium indium^111^ | 56475001 | C0304968 |
SRT | C-B1122 | Ferrous chloride Fe^59^ | 31192007 | C0305004 |
SRT | C-B1121 | Ferrous citrate Fe^59^ | 87958003 | C0305003 |
SRT | C-B1123 | Ferrous sulfate Fe^59^ | 125001 | C0305005 |
SRT | C-B1082 | Fibrinogen I^123^ | 71636003 | C0304978 |
SRT | C-B1031 | Fluorodeoxyglucose F^18^ | 35321007 | C0046056 |
SRT | C-B1041 | Gallium^67^ citrate | 73065000 | C0893383 |
SRT | C-145AB | Indium^111 Capromab Pendetide | 446871009 | C2959379 | Prostascint
SRT | C-14512 | Indium^111 Chloride | 395742005 | C0087296 | Zevalin
SRT | C-145AA | Indium^111 Pentetreotide | 446800006 | C0379955 | Octreoscan
SRT | C-B1061 | Indium^111^ pentetate | 29218008 | C0304967 |
SRT | C-B1066 | Indium^111^ red cell label | 81621007 | C0304971 |
SRT | C-B1067 | Indium^111^ transferrin | 78570003 | C0936259 |
SRT | C-B1065 | Indium^111^-Fe(OH) >3< | 6516008 | C0304970 |
SRT | C-B1135 | Indium^111^oxyquinoline | 424570009 | C1827660 |
SRT | C-B1068 | Indium^113m^ bleomycin | 90617008 | C0304972 |
SRT | C-B1069 | Indium^113m^ chloride | 21451004 | C0361440 |
SRT | C-B1072 | Indium^113m^ oxoquinoline platelet label | 56006008 | C0304975 |
SRT | C-B1073 | Indium^113m^ oxoquinoline RBC label | 56867003 | C0304976 |
SRT | C-B1071 | Indium^113m^ oxoquinoline WBC label | 77510008 | C0304974 |
SRT | C-B1070 | Indium^113m^ pentetate | 42728008 | C0304973 |
SRT | C-B1084 | Iodinated I^125^ albumin | 72015003 | C0304980 |
SRT | C-B1100 | Iodinated I^125^ human serum albumin | 64488003 | C0304997 |
SRT | C-B1094 | Iodinated I^125^ levothyroxine | 80260008 | C0304991 |
SRT | C-B1093 | Iodinated I^125^ oleic acid and triolein | 73745003 | C0304990 |
SRT | C-B1096 | Iodinated I^125^ povidone | 69839009 | C0304993 |
SRT | C-B1097 | Iodinated I^125^ Rose Bengal | 21378001 | C0304994 |
SRT | C-B1098 | Iodinated I^125^ sealed source | 37437001 | C0304995 |
SRT | C-B1099 | Iodinated I^125^ sodium iodine | 70154008 | C0304996 |
SRT | C-B1090 | Iodinated I^131^ aggregated albumin | 55814006 | C0304986 |
SRT | C-B1089 | Iodinated I^131^ albumin | 39200002 | C0304985 |
SRT | C-B1111 | Iodinated I^131^ gamma globulin | 52408003 | C0305002 |
SRT | C-114AB | Iodine^123 15-(4-Iodophenyl)-3(R,S)-Methylpentadecanoic Acid | 447134003 | C2959625 | Cardiodine
SRT | C-B110E | Iodine^123 3-Iodobenzylguanidine MIBG | 395787009 | C0887719 |
SRT | C-B112D | Iodine^131 3-Iodobenzylguanidine MIBG | 395789007 | C0524959 |
SRT | C-114B6 | Iodine^131 Methylnorcholestenol | 446531006 | C2960809 | Adosterol
SRT | C-B1109 | Iodine^131^ polyvinylpyrrolidone | 35884005 | C0305001 |
SRT | C-B1087 | Iodocholesterol I^131^ | 68967007 | C0304983 |
SRT | C-B1095 | Iodohippurate I^123^ sodium | 33785000 | C0304992 |
SRT | C-B1105 | Iodohippurate I^125^ sodium | 36900006 | C0304998 |
SRT | C-B1091 | Iodohippurate I^131^ sodium | 33271006 | C0304987 |
SRT | C-B1108 | Iofetamine I^123^ hydrochloride | 78481003 | C0305000 |
SRT | C-B1088 | Iothalamate sodium I^125^ | 55673009 | C0304984 |
SRT | C-B1124 | Iron Fe^59^ labeled dextran | 60459006 | C0305006 |
SRT | C-173A5 | Krypton^81m | 61716009 | C0303689 |
SRT | C-B1083 | Oleic acid I^125^ | 22979004 | C0304979 |
SRT | C-B1251 | Pentetate calcium trisodium Yb^169^ | 29348008 | C0305041 |
SRT | C-B1151 | Potassium carbonate K^42^ | 111161000 | C0305009 |
SRT | C-B1152 | Potassium chloride K^42^ | 36641004 | C0305010 |
SRT | C-B1150 | Potassium chloride K^43^ | 47729008 | C0305008 |
SRT | C-B1085 | Rose Bengal sodium I^131^ | 111159009 | C0282340 |
SRT | C-B1172 | Selenium^75^ HCAT | 13626001 | C0046666 |
SRT | C-B1171 | Selenomethionione Se^75^ | 88473009 | C0034616 |
SRT | C-B1176 | Sodium chloride Na^22^ | 6257000 | C0205951 |
SRT | C-B1175 | Sodium chloride Na^24^ | 31527000 | C0305013 |
SRT | C-B1011 | Sodium chromate Cr^51^ | 62517004 | C0304955 |
SRT | C-B1032 | Sodium fluoride F^18^ | 129501009 | C0304965 |
SRT | C-B1081 | Sodium iodide I^123^ | 67690002 | C0304977 |
SRT | C-B1086 | Sodium iodide I^131^ | 111160004 | C0304982 |
SRT | C-B1206 | Sodium pertechnetate Tc^99m^ | 19495007 | C0039418 |
SRT | C-B1142 | Sodium phosphate P^32^ | 10781003 | C0305007 |
SRT | C-B1180 | Strontium chloride Sr^85^ | 69076006 | C0305015 |
SRT | C-B1181 | Strontium chloride Sr^87^ | 38424001 | C0305016 |
SRT | C-B1182 | Strontium nitrate Sr^85^ | 8858006 | C0305017 |
SRT | C-B1183 | Strontium nitrate Sr^87^ | 31953001 | C0305018 |
SRT | C-B1205 | Technetium Tc^99c^ albumin microspheres | 55494003 | C0305022 |
SRT | C-B1200 | Technetium Tc^99m^ aggregated albumin | 85693008 | C0039415 |
SRT | C-B1204 | Technetium Tc^99m^ albumin colloid | 16011006 | C0305021 |
SRT | C-B1133 | Technetium Tc^99m^ depreotide | 415704007 | C1100674 |
SRT | C-B1207 | Technetium Tc^99m^ disofenin | 3040004 | C0075932 |
SRT | C-B1223 | Technetium Tc^99m^ exametazine | 77313009 | C0145055 |
SRT | C-B1210 | Technetium Tc^99m^ iron ascorbate | 87853006 | C0305027 |
SRT | C-B1209 | Technetium Tc^99m^ lidofenin | 7281000 | C0075958 |
SRT | C-B1208 | Technetium Tc^99m^ mebrofenin | 4832001 | C0075962 |
SRT | C-B1218 | Technetium Tc^99m^ medronate | 96390006 | C0039416 |
SRT | C-B1203 | Technetium Tc^99m^ microaggregated albumin | 81761004 | C0305020 |
SRT | C-B1225 | Technetium Tc^99m^ N-substituted iminodiacetate | 87410002 | C0305039 |
SRT | C-B1213 | Technetium Tc^99m^ oxidronate | 53951001 | C0305030 |
SRT | C-163B0 | Technetium Tc^99m^ pentetate | 430276001 | C0080212 |
SRT | C-B1215 | Technetium Tc^99m^ pyro and polyphosphates | 65156006 | C0305032 |
SRT | C-B1216 | Technetium Tc^99m^ serum albumin | 79610008 | C0665175 |
SRT | C-163AB | Technetium Tc^99m^ sestamibi | 424299003 | C0162680 |
SRT | C-B1220 | Technetium Tc^99m^ sodium glucoheptonate | 45849009 | C0305034 |
SRT | C-B1211 | Technetium Tc^99m^ stannous etidronate | 111162007 | C0305028 |
SRT | C-B1221 | Technetium Tc^99m^ succimer | 24511001 | C0075928 |
SRT | C-B1222 | Technetium Tc^99m^ sulfur colloid | 5931004 | C0039419 |
SRT | C-B1224 | Technetium Tc^99m^ tagged red cells | 89818005 | C0305038 |
SRT | C-163AC | Technetium Tc^99m^Teboroxime | 424318009 | C0076030 |
SRT | C-163AD | Technetium Tc^99m^Tetrofosmin | 424118002 | C1828125 |
SRT | C-163BD | Technetium^99m Dimercaptosuccinic Acid DMSA | 447201007 | C0075928 | Kidneyscinti
SRT | C-163B6 | Technetium^99m Galactosyl Human Serum Albumin Diethylenetriamine | 446534003 | C2960066 | Asialoscinti
SRT | C-163B7 | Technetium^99m Hydroxymethylene diphosphonate HMDP | 446535002 | C0075953 |
SRT | C-163B9 | Technetium^99m labeled carbon | 447125008 | C2960082 | Technegas
SRT | C-163B8 | Technetium^99m Mercaptoacetyl triglycine MAG3 | 446536001 | C2960081 | MAGscinti
SRT | C-163BA | Technetium^99m N-pyridoxyl-5-methyltryptophan | 447126009 | C2960810 | Hepatimage
SRT | C-163BB | Technetium^99m Phytate | 447127000 | C2960676 |
SRT | C-163BC | Technetium^99m Stannous Colloid | 447128005 | C2960677 |
SRT | C-B1231 | Thallous chloride Tl^201^ | 73685002 | C0305040 |
SRT | C-B1010 | Therapeutic radioisotope | 439007 | C0358509 |
SRT | C-B1251 | Yb^169^-DTPA - pentetate | 29348008 | C0305041 |

## CID 26 Nuclear Medicine Projections

Type:

  Extensible

Version:

  20040322

UID:

  1.2.840.10008.6.1.18

Table CID 26. Nuclear Medicine Projections

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A138 | Coronal | 81654009 | C0205123
SRT | G-A145 | Sagittal | 30730003 | C0205129
SRT | G-A147 | Axial | 24422004 | C0205131
SRT | G-5206 | Right anterior oblique | 399108003 | C1275818
SRT | G-5207 | Left anterior oblique | 399074003 | C1275814
SRT | G-5208 | Right posterior oblique | 399075002 | C1275815
SRT | G-5209 | Left posterior oblique | 399136008 | C1275824
SRT | G-5210 | Oblique axial | 399089007 | C1275817
SRT | G-5212 | Sagittal-oblique axial | 399273000 | C1275844
SRT | G-5220 | Medial-lateral | 399012007 | C1275804
SRT | G-5221 | Lateral-medial | 399300004 | C1275847
SRT | G-5222 | Right lateral projection | 399297009 | C1261185
SRT | G-5223 | Left lateral projection | 399118008 | C1306031
SRT | G-5224 | Medio-lateral oblique | 399268006 | C1275843
SRT | G-5225 | Latero-medial oblique | 399159002 | C1275827
SRT | G-A117 | Transverse | 62824007 | C0205106
SRT | G-A104 | Lateral | 49370004 | C0205093
Include CID 27 “Basic Cardiac Views” |  |
SRT | G-5215 | Anterior projection | 399321004 | C1275849
SRT | G-5216 | Posterior projection | 399001007 | C1275801

### Note

1. In a prior version of this table, the code G-A117 was specified for the concept Transaxial, and R-11300 was specified for the concept Transverse. Since these concepts are synonymous in nuclear projections, and since SNOMED assigns G-A117 to the concept Transverse, the use of R-11300 is deprecated. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

2. The following Code Values were formerly included in CID 26 “Nuclear Medicine Projections” and are retired:

   SRT | G-5200 | Antero-posterior
   SRT | G-5201 | Postero-anterior
   SRT | G-5203 | Frontal oblique
   SRT | G-5204 | Antero-posterior Oblique
   SRT | G-5205 | Postero-anterior Oblique
   SRT | G-5211 | Frontal-oblique axial
   SRT | G-5213 | Submento-vertex axial
   SRT | G-5214 | Oblique submento-vertex
   SRT | G-5226 | Right to left oblique
   SRT | G-5227 | Left to right oblique

## CID 27 Basic Cardiac Views

Type:

  Extensible

Version:

  20120822

UID:

  1.2.840.10008.6.1.957

Table CID 27. Basic Cardiac Views

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A186 | Short Axis | 103340004 | C0522488
SRT | G-A18A | Vertical Long Axis | 131185001 | C1295721
SRT | G-A18B | Horizontal Long Axis | 131186000 | C1295722

## CID 29 Acquisition Modality

This Context Group includes codes that may be used to identify an image or waveform acquisition modality, as used in Attribute Modality (0008,0060) of a Modality Worklist Scheduled Procedure Step or a Composite SOP Instance (see PS3.3). It generally corresponds to a class of diagnostic equipment, or to a specific acquisition function or technique in a device. This Context Group may be used as the value set for HL7 v2 Table 0259 (see HL7 v2.6 Chapter 8 Section 8.8.8.47).

### Note

This Context Group is not the complete set of codes that may appear in the Attribute Modality (0008,0060); these are only the codes associated with orderable acquisition processes (not post-processing).

Type:

  Extensible

Version:

  20121129

UID:

  1.2.840.10008.6.1.19

Table CID 29. Acquisition Modality

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | AR | Autorefraction
DCM | BMD | Bone Mineral Densitometry
DCM | BDUS | Ultrasound Bone Densitometry
DCM | EPS | Cardiac Electrophysiology
DCM | CR | Computed Radiography
DCM | CT | Computed Tomography
DCM | DX | Digital Radiography
DCM | ECG | Electrocardiography
DCM | ES | Endoscopy
DCM | XC | External-camera Photography
DCM | GM | General Microscopy
DCM | HD | Hemodynamic Waveform
DCM | IO | Intra-oral Radiography
DCM | IVOCT | Intravascular Optical Coherence Tomography
DCM | IVUS | Intravascular Ultrasound
DCM | KER | Keratometry
DCM | LEN | Lensometry
DCM | MR | Magnetic Resonance
DCM | MG | Mammography
DCM | NM | Nuclear Medicine
DCM | OAM | Ophthalmic Axial Measurements
DCM | OCT | Optical Coherence Tomography
DCM | OPM | Ophthalmic Mapping
DCM | OP | Ophthalmic Photography
DCM | OPR | Ophthalmic Refraction
DCM | OPT | Ophthalmic Tomography
DCM | OPV | Ophthalmic Visual Field
DCM | OSS | Optical Surface Scanner
DCM | PX | Panoramic X-Ray
DCM | PT | Positron emission tomography
DCM | RF | Radiofluoroscopy
DCM | RG | Radiographic imaging
DCM | SM | Slide Microscopy
DCM | SRF | Subjective Refraction
DCM | US | Ultrasound
DCM | VA | Visual Acuity
DCM | XA | X-Ray Angiography

## CID 30 DICOM Devices

This Context Group includes codes that may be used to identify a class of equipment that uses DICOM.

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.20

Table CID 30. DICOM Devices

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 29 “Acquisition Modality”
DCM | ARCHIVE | Archive
DCM | COMP | Computation Server
DCM | CAD | Computer Assisted Detection/Diagnosis
DCM | DSS | Department System Scheduler
DCM | FILMD | Film Digitizer
DCM | MCD | Media Creation Device
DCM | PRINT | Hard Copy Print Server
DCM | CAPTURE | Image Capture
DCM | LOG | Procedure Logging
DCM | RT | Radiation Therapy Device
DCM | WSD | Workstation

## CID 31 Abstract Priors

This Context Group includes codes that may be used to identify imaging procedures that may be referred to as priors for the purpose of image set selection in Hanging Protocols.

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.21

Table CID 31. Abstract Priors

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-40553 | On admission | 278307001 | C0457453
SRT | R-400B2 | Intraoperative | 277671009 | C0456904
SRT | R-41FD9 | Pre-admission | 281379000 | C0559269
SRT | R-411C0 | Pre-dose | 255235001 | C0439565
SRT | R-404DA | Post-dose | 255566006 | C0439568
SRT | R-413C5 | Pre-operative | 262068006 | C0445204
SRT | R-413B7 | Post-operative | 262061000 | C0032790
DCM | 109120 | On admission to unit |  |
DCM | 109121 | On discharge |  |
DCM | 109122 | On discharge from unit |  |
DCM | 109123 | Pre-intervention |  |
DCM | 109124 | Post-intervention |  |
DCM | 109125 | At last appointment |  |

## CID 42 Numeric Value Qualifier

Type:

  Extensible

Version:

  20020114

UID:

  1.2.840.10008.6.1.22

Table CID 42. Numeric Value Qualifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 114000 | Not a number
DCM | 114001 | Negative Infinity
DCM | 114002 | Positive Infinity
DCM | 114003 | Divide by zero
DCM | 114004 | Underflow
DCM | 114005 | Overflow
DCM | 114006 | Measurement failure
DCM | 114007 | Measurement not attempted
DCM | 114008 | Calculation failure
DCM | 114009 | Value out of range
DCM | 114010 | Value unknown
DCM | 114011 | Value indeterminate

## CID 50 Instance Availability Status

Type:

  Non-Extensible

Version:

  20090616

UID:

  1.2.840.10008.6.1.811

Table CID 50. Instance Availability Status

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | NEARLINE | Nearline
DCM | OFFLINE | Offline
DCM | ONLINE | Online
DCM | UNAVAILABLE | Unavailable

## CID 82 Units of Measurement

Context Group ID 82 comprises the case-sensitive codes of UCUM. See Section 7.2.2.

### Note

Equivalent to the HL7 Value Set "Units of Measure case sensitive" 2.16.840.1.113883.11.12839.

## CID 83 Units for Real World Value Mapping

Type:

  Extensible

Version:

  20080123

UID:

  1.2.840.10008.6.1.24

Table CID 83. Units for Real World Value Mapping

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 84 “PET Units”
UCUM | [hnsf'U] | Hounsfield unit

## CID 84 PET Units

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1030

Table CID 84. PET Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 85 “SUV Units”
UCUM | {counts} | Counts
UCUM | {counts}/s | Counts per second
UCUM | {propcounts} | Proportional to counts
UCUM | {propcounts}/s | Proportional to counts per second
UCUM | cm2 | Centimeter**2
UCUM | cm2/ml | Centimeter**2/milliliter
UCUM | % | Percent
UCUM | Bq/ml | Becquerels/milliliter
UCUM | mg/min/ml | Milligrams/minute/milliliter
UCUM | umol/min/ml | Micromole/minute/milliliter
UCUM | ml/min/g | Milliliter/minute/gram
UCUM | ml/g | Milliliter/gram
UCUM | /cm | /Centimeter
UCUM | umol/ml | Micromole/milliliter

## CID 85 SUV Units

Type:

  Extensible

Version:

  20161106

UID:

  1.2.840.10008.6.1.984

Table CID 85. SUV Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | g/ml{SUVbw} | Standardized Uptake Value body weight
UCUM | g/ml{SUVlbm} | Standardized Uptake Value lean body mass (James)
UCUM | g/ml{SUVlbm(James128)} | Standardized Uptake Value lean body mass (James 128 multiplier)
UCUM | g/ml{SUVlbm(Janma)} | Standardized Uptake Value lean body mass (Janma)
UCUM | cm2/ml{SUVbsa} | Standardized Uptake Value body surface area
UCUM | g/ml{SUVibw} | Standardized Uptake Value ideal body weight

### Note

The formulas for the determination of SUVbw, SUVbsa, SUVlbm (James) and SUVibw are defined in Sugawara et al. Reevaluation of the Standardized Uptake Value for FDG: Variations with Body Weight and Methods for Correction.Radiology, 1999 at http://radiology.rsna.org/content/213/2/521.

Unfortunately, Sugawara used a parameter of 120 rather than 128 for males, propagating an error in Morgan DJ, Bray KM. Lean Body Mass as a Predictor of Drug Dosage: Implications for Drug Therapy. Clinical Pharmacokinetics. 1994;26(4):292–307, which misquoted the original LBM definition that used 128 in James WPT, Waterlow JC. Research on Obesity: A Report of the DHSS/MRC Group. London: Her Majesty’s Stationery Office; 1976. Implementations differ in whether they have used 120 or 128 for ({SUVlbm}g/ml{SUVlbm}, UCUM, "Standardized Uptake Value lean body mass (James)"). See Kelly M. SUV: Advancing Comparability and Accuracy. Siemens; 2009. Available from: http://www.mpcphysics.com/documents/SUV_Whitepaper_Final_11.17.09_59807428_2.pdf.

The Janmahasatian LBM formula is defined in Janmahasatian et al. Quantification of Lean Bodyweight. Clin Pharmacokinet. 2005 Oct 1;44(10):1051–65. at http://dx.doi.org/10.2165/00003088-200544100-00004 and its role in SUVlbm(Janma) calculation is discussed in Tahari et al. Optimum Lean Body Formulation for Correction of Standardized Uptake Value in PET Imaging. Journal of Nuclear Medicine. 2014 Sep 1;55(9):1481–4. at http://jnm.snmjournals.org/content/55/9/1481. The patient size correction factors are summarized here, where weight is in kg and height is in cm:

SUVbw: males & females: weight

SUVlbm (James): males : 1.10 * weight - 120 * (weight/height) ^2

SUVlbm (James): females: 1.07 * weight - 148 * (weight/height) ^2

SUVlbm(Janma): males: 9.27E3 * weight / (6.68E3 + 216 * weight / (height^2))

SUVlbm(Janma): females: 9.27E3 * weight / (8.78E3 + 244 * weight / (height^2))

SUVbsa: males & females: weight^ 0.425 * height^0.725 * 0.007184

SUVibw: males: 48.0 + 1.06 * (height - 152)

females: 45.5 + 0.91 * (height - 152)

## CID 91 Functional Condition Present During Acquisition

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.891

Table CID 91. Functional Condition Present During Acquisition

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 3271 “Hemodynamic Physiological Challenges” |  |
SRT | F-F7100 | Phonation | 43914001 | C0031577
SRT | F-12300 | Weight bearing | 87731000 | C0231573
DCM | 109137 | During voiding |  |
DCM | 109134 | Prior to voiding |  |
DCM | 109135 | Post voiding |  |

## CID 92 Joint Position During Acquisition

Type:

  Extensible

Version:

  20100625

UID:

  1.2.840.10008.6.1.892

Table CID 92. Joint Position During Acquisition

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 109136 | Neutral musculoskeletal position |  |
SRT | F-10110 | Flexion | 9964006 | C0231452
SRT | F-10100 | Extension | 24154002 | C0522009
SRT | F-10120 | Abduction | 60074003 | C0231456
SRT | F-10130 | Adduction | 11554009 | C0231457
SRT | F-10210 | Internal rotation | 12852001 | C0231459
SRT | F-10220 | External rotation | 52019005 | C0231462
SRT | F-10226 | Supination | 14502000 | C0038845
SRT | F-10216 | Pronation | 88241000 | C0033421
SRT | F-10240 | Torsion | 51795009 | C0040480

## CID 93 Joint Positioning Method

Type:

  Extensible

Version:

  20100625

UID:

  1.2.840.10008.6.1.893

Table CID 93. Joint Positioning Method

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-13060 | Passive movement | 21278004 | C0079991
SRT | P0-05083 | Manipulation of joint | 118745001 | C1292923

## CID 94 Physical Force Applied During Acquisition

Type:

  Extensible

Version:

  20100625

UID:

  1.2.840.10008.6.1.894

Table CID 94. Physical Force Applied During Acquisition

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P0-02160 | Traction - action | 129411004 | C0040597
SRT | P0-021B2 | Compression - action | 263720003 | C0565514
SRT | P0-021AB | Rotation - action | 257912008 | C0677597

## CID 100 Quantitative Diagnostic Imaging Procedures

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.998

Table CID 100. Quantitative Diagnostic Imaging Procedures

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P5-09051 | Magnetic resonance imaging guidance | 258177008 | C0442974
DCM | 126020 | Multiparametric MRI |  |
DCM | 126021 | Multiparametric MRI of prostate |  |
DCM | 126022 | Multiparametric MRI of whole body |  |
SRT | P5-0907F | Dynamic magnetic resonance imaging of knee | 433139009 | C2315346
SRT | P5-70694 | Dynamic magnetic resonance imaging of pelvis | 446315002 | C2960816
LN | 44139-4 | PET whole body |  | C1715409
SRT | P5-080FF | PET/CT FDG imaging of whole body | 443271005 | C2732676
SRT | P5-08118 | PET/CT MET imaging of whole body | 443844003 | C2732956
RADLEX | RPID96 | CT head perfusion with IV contrast |  |
RADLEX | RPID5258 | NM head perfusion brain SPECT |  |
RADLEX | RPID5427 | NM head perfusion brain PET-CT AV-45 |  |

## CID 220 Level of Significance

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.25

Table CID 220. Level of Significance

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-00333 | Most significant | 371925005 | C1299394
SRT | R-0030C | Highly significant | 371926006 | C1299395
SRT | R-10045 | Significant | 386134007 | C0750502
SRT | R-00345 | Not significant | 371928007 | C1273937
SRT | R-10046 | Significance Undetermined | 386135008 | C1272585

## CID 221 Measurement Range Concepts

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.26

Table CID 221. Measurement Range Concepts

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 226 “Population Statistical Descriptors”
Include CID 227 “Sample Statistical Descriptors”

## CID 222 Normality Codes

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.27

Table CID 222. Normality Codes

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A460 | Normal | 17621005 | C0205307
SRT | R-42037 | Abnormal | 263654008 | C0205161
SRT | R-002C4 | Abnormally High | 371879000 | C1299351
SRT | R-002C5 | Abnormally Low | 371880002 | C1299352
SRT | R-0039B | Normality Undetermined | 371934000 | C1299401

## CID 223 Normal Range Values

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.28

Table CID 223. Normal Range Values

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-0038B | Normal Range Upper Limit | 371933006 | C1299400
SRT | R-41F90 | Normal Range Lower Limit | 385524004 | C1272773

## CID 224 Selection Method

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.29

Table CID 224. Selection Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121410 | User chosen value
DCM | 121411 | Most recent value chosen
DCM | 121412 | Mean value chosen

## CID 225 Measurement Uncertainty Concepts

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.30

Table CID 225. Measurement Uncertainty Concepts

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-00363 | +/-, range of measurement uncertainty | 371884006 | C1299354
SRT | R-00364 | +, range of upper measurement uncertainty | 371886008 | C1299356
SRT | R-00362 | -, range of lower measurement uncertainty | 371885007 | C1299355

## CID 226 Population Statistical Descriptors

Type:

  Extensible

Version:

  20121101

UID:

  1.2.840.10008.6.1.31

Table CID 226. Population Statistical Descriptors

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-00337 | 95th Percentile Value of population | 371889001 | C1299358
SRT | R-00338 | 90th Percentile Value of population | 371887004 | C1276309
SRT | R-00346 | 1 Sigma Upper Value of population | 371917008 | C1299386
SRT | R-00387 | 2 Sigma Upper Value of population | 371920000 | C1299389
SRT | R-00317 | Mean Value of population | 373098007 | C1298794
SRT | R-00319 | Median Value of population | 373099004 | C1298795
SRT | R-00377 | 10th Percentile Value of population | 371890005 | C1299359
SRT | R-00397 | 5th Percentile Value of population | 371888009 | C1299357
SRT | R-00347 | 1 Sigma Lower Value of population | 371919006 | C1299388
SRT | R-00388 | 2 Sigma Lower Value of population | 371918003 | C1299387
DCM | 121414 | Standard deviation of population |  |
DCM | 121417 | 2 Sigma deviation of population |  |

### Note

The SNOMED meaning for R-00317 is "Mean - numeric estimation technique", but in the context of its use here, a more specific meaning has been used.

## CID 227 Sample Statistical Descriptors

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.32

Table CID 227. Sample Statistical Descriptors

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121415 | Percentile Ranking of measurement
DCM | 121416 | Z-Score of measurement

## CID 228 Equation or Table

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.33

Table CID 228. Equation or Table

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121420 | Equation
DCM | 121421 | Equation Citation
DCM | 121424 | Table of Values
DCM | 121422 | Table of Values Citation
DCM | 121423 | Method Citation

## CID 230 Yes-No

Type:

  Non-Extensible

Version:

  20060613

UID:

  1.2.840.10008.6.1.34

Table CID 230. Yes-No

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-0038D | Yes | 373066001 | C1298907
SRT | R-00339 | No | 373067005 | C1298908
SRT | R-0038A | Undetermined | 373068000 | C3536725

## CID 231 Yes-No Only

Type:

  Non-Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1064

Table CID 231. Yes-No Only

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-0038D | Yes | 373066001 | C1298907
SRT | R-00339 | No | 373067005 | C1298908

### Note

This context group is intended for use rather than CID 230 "Yes-No" when the value (R-0038A, SRT, "Undetermined") is not permissible.

## CID 240 Present-Absent

Type:

  Non-Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.35

Table CID 240. Present-Absent

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A203 | Present | 52101004 | C0150312
SRT | R-4089B | Absent | 272519000 | C0442733
SRT | R-0038A | Undetermined | 373068000 | C3536725

### Note

In a previous version of this Context Group (R-40271, SRT, "Findings values") was used incorrectly to mean "Presence Undetermined"; there is no SNOMED CT concept that specifically means that the "presence" (of a finding) is undetermined, so the more general "undetermined" concept is used.

## CID 241 Present-Absent Only

Type:

  Non-Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1113

Table CID 241. Present-Absent Only

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A203 | Present | 52101004 | C0150312
SRT | R-4089B | Absent | 272519000 | C0442733

### Note

This context group is intended for use rather than CID 240 "Normal-Abnormal" when the value (R-0038A, SRT, "Undetermined") is not permissible.

## CID 242 Normal-Abnormal

This Context Group is a subset of CID 222 “Normality Codes”.

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.36

Table CID 242. Normal-Abnormal

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A460 | Normal | 17621005 | C0205307
SRT | R-42037 | Abnormal | 263654008 | C0205161
SRT | R-0039B | Normality Undetermined | 371934000 | C1299401

## CID 244 Laterality

Type:

  Non-Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.37

Table CID 244. Laterality

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A100 | Right | 24028007 | C0205090
SRT | G-A101 | Left | 7771000 | C0205091
SRT | G-A102 | Right and left | 51440002 | C0238767
SRT | G-A103 | Unilateral | 66459002 | C0205092

## CID 250 Positive-Negative

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.38

Table CID 250. Positive-Negative

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A200 | Positive | 10828004 | C1446409
SRT | R-40759 | Negative | 260385009 | C0205160

## CID 251 Severity of Complication

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.39

Table CID 251. Severity of Complication

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-404F9 | Major | 255603008 | C0205164
SRT | R-404FC | Minor | 255606000 | C0205165

## CID 252 S-M-L Size Descriptor

CID 6118 is a superset of this Context Group.

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.735

Table CID 252. S-M-L Size Descriptor

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-404A8 | Small | 255507004 | C0700321
SRT | R-404A9 | Medium | 255508009 | C0439536
SRT | R-404AA | Large | 255509001 | C0549177

## CID 270 Observer Type

Type:

  Non-Extensible

Version:

  20040920

UID:

  1.2.840.10008.6.1.40

Table CID 270. Observer Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121006 | Person
DCM | 121007 | Device

## CID 271 Observation Subject Class

Type:

  Non-Extensible

Version:

  20071102

UID:

  1.2.840.10008.6.1.41

Table CID 271. Observation Subject Class

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121025 | Patient
DCM | 121026 | Fetus
DCM | 121027 | Specimen
DCM | 121192 | Device Subject

## CID 280 Longitudinal Temporal Event Types

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1184

Table CID 280. Longitudinal Temporal Event Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
NCIt | C37948 | Enrollment |  | C1516879
DCM | 121079 | Baseline |  |

## CID 400 Audit Event ID

Type:

  Extensible

Version:

  20170416

UID:

  1.2.840.10008.6.1.903

Table CID 400. Audit Event ID

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110100 | Application Activity
DCM | 110101 | Audit Log Used
DCM | 110102 | Begin Transferring DICOM Instances
DCM | 110103 | DICOM Instances Accessed
DCM | 110104 | DICOM Instances Transferred
DCM | 110105 | DICOM Study Deleted
DCM | 110106 | Export
DCM | 110107 | Import
DCM | 110108 | Network Entry
DCM | 110109 | Order Record
DCM | 110110 | Patient Record
DCM | 110111 | Procedure Record
DCM | 110112 | Query
DCM | 110113 | Security Alert
DCM | 110114 | User Authentication

## CID 401 Audit Event Type Code

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.904

Table CID 401. Audit Event Type Code

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110120 | Application Start
DCM | 110121 | Application Stop
DCM | 110122 | Login
DCM | 110123 | Logout
DCM | 110124 | Attach
DCM | 110125 | Detach
DCM | 110126 | Node Authentication
DCM | 110127 | Emergency Override Started
DCM | 110128 | Network Configuration
DCM | 110129 | Security Configuration
DCM | 110130 | Hardware Configuration
DCM | 110131 | Software Configuration
DCM | 110132 | Use of Restricted Function
DCM | 110133 | Audit Recording Stopped
DCM | 110134 | Audit Recording Started
DCM | 110135 | Object Security Attributes Changed
DCM | 110136 | Security Roles Changed
DCM | 110137 | User Security Attributes Changed
DCM | 110138 | Emergency Override Stopped
DCM | 110139 | Remote Service Operation Started
DCM | 110140 | Remote Service Operation Stopped
DCM | 110141 | Local Service Operation Started
DCM | 110142 | Local Service Operation Stopped
DCM | 110143 | Authentication Decision
DCM | 110144 | Authorization Decision
DCM | 110145 | Session start
DCM | 110146 | Session stop
DCM | 110147 | Access Control Decision

## CID 402 Audit Active Participant Role ID Code

Type:

  Extensible

Version:

  20100826

UID:

  1.2.840.10008.6.1.905

Table CID 402. Audit Active Participant Role ID Code

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110150 | Application
DCM | 110151 | Application Launcher
DCM | 110152 | Destination Role ID
DCM | 110153 | Source Role ID
DCM | 110154 | Destination Media
DCM | 110155 | Source Media

## CID 403 Security Alert Type Code

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.906

Table CID 403. Security Alert Type Code

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110120 | Application Start
DCM | 110121 | Application Stop
DCM | 110122 | Login
DCM | 110123 | Logout
DCM | 110124 | Attach
DCM | 110125 | Detach
DCM | 110126 | Node Authentication
DCM | 110127 | Emergency Override Started
DCM | 110128 | Network Configuration
DCM | 110129 | Security Configuration
DCM | 110130 | Hardware Configuration
DCM | 110131 | Software Configuration
DCM | 110132 | Use of Restricted Function
DCM | 110133 | Audit Recording Stopped
DCM | 110134 | Audit Recording Started
DCM | 110135 | Object Security Attributes Changed
DCM | 110136 | Security Roles Changed
DCM | 110137 | User Security Attributes Changed
DCM | 110138 | Emergency Override Stopped
DCM | 110139 | Remote Service Operation Started
DCM | 110140 | Remote Service Operation Stopped
DCM | 110141 | Local Service Operation Started
DCM | 110142 | Local Service Operation Stopped
DCM | 110143 | Authentication Decision
DCM | 110144 | Authorization Decision
DCM | 110145 | Session start
DCM | 110146 | Session stop
DCM | 110147 | Access Control Decision

## CID 404 Audit Participant Object ID Type Code

Type:

  Extensible

Version:

  20100621

UID:

  1.2.840.10008.6.1.907

Table CID 404. Audit Participant Object ID Type Code

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110180 | Study Instance UID
DCM | 110181 | SOP Class UID
DCM | 110182 | Node ID

## CID 405 Media Type Code

Type:

  Extensible

Version:

  20100824

UID:

  1.2.840.10008.6.1.908

Table CID 405. Media Type Code

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110030 | USB Disk Emulation
DCM | 110031 | Email
DCM | 110032 | CD
DCM | 110033 | DVD
DCM | 110034 | Compact Flash
DCM | 110035 | Multi-media Card
DCM | 110036 | Secure Digital Card
DCM | 110037 | URI
DCM | 110010 | Film
DCM | 110038 | Paper Document

## CID 501 Volumetric View Description

Type:

  Extensible

Version:

  20150915

UID:

  1.2.840.10008.6.1.1057

Table CID 501. Volumetric View Description

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 6 “Transducer Orientation”
Include CID 26 “Nuclear Medicine Projections”
Include CID 4010 “DX View”
Include CID 12226 “Echocardiography Image View”

## CID 502 Volumetric View Modifier

Type:

  Extensible

Version:

  20150915

UID:

  1.2.840.10008.6.1.1058

Table CID 502. Volumetric View Modifier

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 6 “Transducer Orientation”
Include CID 23 “Cranio-Caudad Angulation”
Include CID 4011 “DX View Modifier”

## CID 601 Biosafety Levels

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1065

Table CID 601. Biosafety Levels

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-41E4D | Biosafety level 1 | 409600007 | C1443934
SRT | R-41E4E | Biosafety level 2 | 409603009 | C1443937
SRT | R-41E4F | Biosafety level 3 | 409604003 | C1443938
SRT | R-41E50 | Biosafety level 4 | 409605002 | C1443939

## CID 602 Biosafety Control Reasons

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1066

Table CID 602. Biosafety Control Reasons

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-61E79 | Biohazardous material | 409595003 | C0079021
SRT | C-29000 | Carcinogen | 88376000 | C0007090
SRT | F-00D5F | Patient immunocompromised | 370388006 | C0085393
UMLS | C0003069 | Transgenic animal |  | C0003069

## CID 603 Animal Room Types

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1068

Table CID 603. Animal Room Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 127370 | Animal housing room |  |
DCM | 127371 | Preparation room |  |
DCM | 127372 | Imaging procedure room |  |
SRT | R-305D6 | Induction room | 414485004 | C1532289
SRT | R-305C3 | Recovery room | 398161000 | C0198828
SRT | R-305D3 | Isolation room | 409688003 | C1443994

### Note

1. Only rooms appropriate for animals in the context of in vivo imaging are described (e.g., not necropsy rooms, etc.)

2. (R-305C3, SRT, "Recovery room" (synonym of "postoperative anesthesia care unit ") is reused here even though its parent is "Location within hospital premises (environment)", which is arguably specifically human. The same is true for (R-305D6, SRT, "Induction room") and (R-305D3, SRT, "Isolation room").

## CID 604 Device Reuse

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1069

Table CID 604. Device Reuse

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 127177 | Unused |  |
DCM | 127178 | Reused |  |

## CID 605 Animal Bedding Material

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1070

Table CID 605. Animal Bedding Material

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 127230 | Aspen chip bedding |  |
DCM | 127231 | Aspen shaving bedding |  |
DCM | 127232 | Corn cob bedding |  |
DCM | 127233 | Paper-based bedding |  |
DCM | 127234 | Pine chip bedding |  |
DCM | 127235 | Pine shaving bedding |  |

## CID 606 Animal Shelter Types

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1071

Table CID 606. Animal Shelter Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-40775 | None | 260413007 | C0549184
DCM | 127220 | Igloo |  |
DCM | 127221 | Red translucent igloo |  |

## CID 607 Animal Feed Types

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1072

Table CID 607. Animal Feed Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 127271 | NIH07 |  |
DCM | 127270 | NIH31 |  |
DCM | 127272 | AIN76 |  |
DCM | 127273 | AIN93G |  |
DCM | 127274 | AIN93M |  |

### Note

This context group includes the open source diets described in Barnard DE et al. Open- and Closed-Formula Laboratory Animal Diets and Their Importance to Research. Journal of the American Association for Laboratory Animal Science : JAALAS (2009), 48(6), 709–713. http://www.ncbi.nlm.nih.gov/pmc/articles/PMC2786923/.

A more extensive list of NIH-specified diets for small animals (not just mice and rats) can be found at https://web.archive.org/web/20100527090853/http://dvrnet.ors.od.nih.gov/diets_info.asp.

## CID 608 Animal Feed Sources

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1073

Table CID 608. Animal Feed Sources

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
UMLS | C1547887 | Commercial product |  | C1547887
DCM | 127390 | Locally manufactured product |  |

### Note

(C1547887, UMLS, "Commercial product") originates from the HL7 V2.5 Chapter 04 blood products description as an attribute name rather than a value, but in UMLS is not expressly constrained and has as a parent generic semantic type of "Manufactured Object".

## CID 609 Animal Feeding Methods

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1074

Table CID 609. Animal Feeding Methods

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
NCIt | C64636 | ad libitum |  | C1879743
UMLS | C0425422 | Restricted diet |  | C0425422
DCM | 127391 | Food treat |  |
SRT | PA-00620 | Gavage | 61420007 | C0041281

### Note

(C0425422, UMLS, "Restricted diet") corresponds to the inactive SNOMED concept of "Dietary restriction NOS"; SNOMED currently does not seem to have an active generic concept of a restricted diet, as opposed to many specific types of restricted diet. In this context, the intent is to convey that the diet is controlled and restricted to finite quantities (e.g., as opposed to ad libitum) without requiring a detailed classification of what components are restricted.

## CID 610 Water Types

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1075

Table CID 610. Water Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-101E9 | Tap water | 444923006 | C2919405
SRT | C-101E8 | Distilled water | 444883009 | C0790233
DCM | 127290 | Reverse osmosis purified water |  |
DCM | 127291 | Reverse osmosis purified, HCl acidified water |  |

## CID 611 Anesthesia Category Code Type for Small Animal Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1076

Table CID 611. Anesthesia Category Code Type for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 612 “Anesthesia Category Code Type from Anesthesia Quality Initiative (AQI)”

## CID 612 Anesthesia Category Code Type from Anesthesia Quality Initiative (AQI)

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1077

Table CID 612. Anesthesia Category Code Type from Anesthesia Quality Initiative (AQI)

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P1-C0010 | General anesthesia | 50697003 | C0002915
SRT | P1-C0B00 | Sedation | 72641008 | C0344106
SRT | P1-C0208 | Spinal anesthesia | 231249005 | C0002928
SRT | P1-C0220 | Epidural anesthesia | 18946005 | C0002913
SRT | P1-C0200 | Regional anesthesia | 27372005 | C0002911
SRT | P1-C0037 | Topical local anesthesia | 386760001 | C0472473
SRT | P1-C0038 | Local anesthesia | 386761002 | C0002921
SRT | P1-0512E | Monitored Anesthesia Care (MAC) | 398239001 | C1301902

### Note

This context group contains SNOMED procedure code equivalents of enumerated string concepts for the [AQI Schema] element AnesthesiaCategoryCodeType. See http://www.aqihq.org/aqischdoc/AnesthesiaCategoryCodeType.html.

## CID 613 Anesthesia Induction Code Type for Small Animal Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1078

Table CID 613. Anesthesia Induction Code Type for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 614 “Anesthesia Induction Code Type from Anesthesia Quality Initiative (AQI)”
SRT | G-D106 | Intraperitoneal route | 38239002 | C1522583

### Note

The intraperitoneal route is added to the AQI value set, since that route is used for small animal imaging.

## CID 614 Anesthesia Induction Code Type from Anesthesia Quality Initiative (AQI)

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1079

Table CID 614. Anesthesia Induction Code Type from Anesthesia Quality Initiative (AQI)

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-40B32 | By inhalation | 446406008 | C1998547
SRT | G-D101 | Intravenous route | 47625008 | C1522726
SRT | G-D160 | Per rectum | 37161004 | C1527425
SRT | G-D103 | Intramuscular route | 78421000 | C1556154

### Note

This context group contains SNOMED administration route code equivalents of enumerated string concepts for the [AQI Schema] element AnesthesiaInductionCodeType. See http://www.aqihq.org/aqischdoc/AnesthesiaInductionCodeType.html.

## CID 615 Anesthesia Maintenance Code Type for Small Animal Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1080

Table CID 615. Anesthesia Maintenance Code Type for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 616 “Anesthesia Maintenance Code Type from Anesthesia Quality Initiative (AQI)”

## CID 616 Anesthesia Maintenance Code Type from Anesthesia Quality Initiative (AQI)

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1081

Table CID 616. Anesthesia Maintenance Code Type from Anesthesia Quality Initiative (AQI)

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P1-C0020 | Inhalation anesthesia system closed rebreathing primary agent | 112987001 | C0198795
SRT | P1-C0030 | Inhalation anesthesia system closed no rebreathing primary agent | 44812007 | C0198796

### Note

This context group contains SNOMED procedure code equivalents of enumerated string concepts for the [AQI Schema] element AnesthesiaMaintenanceCodeType. See http://www.aqihq.org/aqischdoc/AnesthesiaMaintenanceCodeType.html.

The AQI value "circle system" corresponds to (P1-C0020, SRT, "Inhalation anesthesia system closed rebreathing primary agent"). The SNOMED code meaning has been abbreviated to fit within the allowed DICOM Value Representation.

The AQI value "non-rebreathing" corresponds to (P1-C0030, SRT, "Inhalation anesthesia system closed no rebreathing primary agent"). The SNOMED code meaning has been abbreviated to fit within the allowed DICOM Value Representation.

## CID 617 Airway Management Method Code Type for Small Animal Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1082

Table CID 617. Airway Management Method Code Type for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 618 “Airway Management Method Code Type from Anesthesia Quality Initiative (AQI)”
DCM | 127060 | Nose cone |  |

## CID 618 Airway Management Method Code Type from Anesthesia Quality Initiative (AQI)

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1083

Table CID 618. Airway Management Method Code Type from Anesthesia Quality Initiative (AQI)

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P2-2290D | Controlled Ventilation | 243147009 | C0419011
SRT | P2-22902 | Artificial Respiration | 40617009 | C0035205
SRT | P2-22500 | Oxygen Therapy | 57485005 | C0184633
SRT | P0-05DE2 | Laryngeal Mask Airway (LMA) | 424979004 | C0396618
SRT | P0-06211 | Intubation of respiratory tract | 447996002 | C3164350
SRT | A-00BA2 | Anesthetic face mask | 297120004 | C0573976
SRT | A-00BA2 | Anesthetic face mask | 297120004 | C0573976
DCM | 127061 | Nasal cannula |  | C0179574
SRT | G-D13E | Via tracheostomy | 180640008 | C0393370

### Note

This context group contains SNOMED procedure or physical object or qualifier value code equivalents of enumerated string concepts for the [AQI Schema] element AirwayManagementMethodCodeType. See http://www.aqihq.org/aqischdoc/AirwayManagementMethodCodeType.html. Used by permission of the Anesthesia Quality Institute (AQI) (http://aqihq.org/).

## CID 619 Airway Management Sub-Method Code Type for Small Animal Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1084

Table CID 619. Airway Management Sub-Method Code Type for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 620 “Airway Management Sub-Method Code Type from Anesthesia Quality Initiative (AQI)”

## CID 620 Airway Management Sub-Method Code Type from Anesthesia Quality Initiative (AQI)

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1085

Table CID 620. Airway Management Sub-Method Code Type from Anesthesia Quality Initiative (AQI)

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
UMLS | C2223982 | Inverse ratio ventilation |  | C2223982
SRT | P2-22914 | High frequency ventilation | 243154003 | C0019540
SRT | P2-22933 | Transtracheal jet ventilation | 448442005 | C3164603
SRT | P2-22916 | Continuous flow ventilation | 243156001 | C0419018

### Note

This context group contains SNOMED procedure code equivalents of enumerated string concepts for the [AQI Schema] element AirwayManagementSubMethodCodeType. See http://www.aqihq.org/aqischdoc/AirwayManagementSubMethodCodeType.html.

## CID 621 Medication Type Code Type for Small Animal Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1086

Table CID 621. Medication Type Code Type for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 622 “Medication Type Code Type from Anesthesia Quality Initiative (AQI)”
DCM | 127330 | Carrier gas |  |

## CID 622 Medication Type Code Type from Anesthesia Quality Initiative (AQI)

Type:

  Extensible

Version:

  20160212

UID:

  1.2.840.10008.6.1.1087

Table CID 622. Medication Type Code Type from Anesthesia Quality Initiative (AQI)

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-68000 | Adrenergic agent | 86308005 | C0001637
SRT | C-52500 | Aminoglycoside antibiotic | 14443002 | C0002556
SRT | F-6181F | Analgesic | 373265006 | C0002771
SRT | F-6196E | Antiarrhythmic | 372813008 | C0003195
SRT | C-5008C | Antibiotic | 255631004 | C0003232
SRT | F-6180B | Anticholinergic | 373246003 | C0242896
SRT | F-6180B | Anticholinergic agent | 373246003 | C0242896
SRT | F-6199A | Anticoagulant | 372862008 | C0003280
SRT | R-F1216 | Anticonvulsant | 255632006 | C0003286
SRT | R-F1216 | Anticonvulsant | 255632006 | C0003286
SRT | F-B1810 | Antidiuretic hormone | 77671006 | C1705480
SRT | C-85800 | Antiemetic | 52017007 | C0003297
SRT | F-617EF | Antifungal | 373219008 | C0003308
SRT | F-618BA | Anti-heparin agent | 372708000 | C0304941
SRT | F-61969 | Antihistamine | 372806008 | C0003360
SRT | C-81100 | Antihypertensive | 1182007 | C0003364
UMLS | C1579431 | Antihypoglycemic |  | C1579431
SRT | R-F2B23 | Barbiturate | 372798009 | C0004745
SRT | R-F2B1D | Benzodiazepine | 372664007 | C0005064
SRT | F-619EF | Benzodiazepine antagonist | 372906009 | C0360298
SRT | F-61814 | Beta-blocker | 373254001 | C0001645
SRT | C-00231 | Beta-Lactam antibiotic | 373297006 | C0026458
SRT | R-005B3 | Blood product | 410652009 | C0456388
SRT | F-616EB | Bronchodilator | 372580007 | C0006280
SRT | C-14300 | Calcium | 5540006 | C0006675
SRT | F-61878 | Calcium channel blocker | 373304005 | C0006684
SRT | F-618D8 | Caloric agent | 373530005 | C0280082
SRT | C-002B1 | Carbapenem antibiotic | 396345004 | C0006968
SRT | C-0021C | Cephalosporin antibiotic | 373262009 | C3536856
SRT | F-620E8 | Cholinergic agent | 421148003 | C1720711
SRT | F-618AF | Diuretic | 372695000 | C0012798
SRT | C-50013 | Drug diluent | 74626007 | C0304221
SRT | F-B2700 | Estrogen | 41598000 | C0014939
SRT | C-84989 | Gastrointestinal prokinetic | 116532005 | C1268865
SRT | F-6186A | General anesthetic | 373288007 | C0017302
UMLS | C0019593 | H2 antagonist |  | C0019593
SRT | F-618A5 | Hemostatic agent | 372681003 | C0019120
SRT | C-50309 | Hypoglycemic | 312064005 | C0020616
SRT | C-80120 | Inotropic agent | 111139005 | C0304509
SRT | C-0023B | Lincomycin antibiotic | 372677003 | C0023726
SRT | C-00286 | Linezolid antibiotic | 387056004 | C0663241
SRT | F-6183D | Local anesthetic | 373477003 | C0002934
SRT | F-6186F | Low Molecular Weight Heparin | 373294004 | C0019139
SRT | C-00211 | Macrolide antibiotic | 372480009 | C0003240
SRT | C-14800 | Magnesium | 72717003 | C0024467
SRT | F-616FE | Metronidazole antibiotic | 372602008 | C0025872
SRT | F-6188F | Narcotic (opiate) antagonist | 372656001 | C0027410
UMLS | C0027409 | Narcotic analgesic |  | C0027409
SRT | C-97302 | Nasal decongestant | 96329004 | C0042398
SRT | F-6180F | NeuroMuscular Blocking (NMB) - depolarizing | 373250005 | C0027867
SRT | F-61959 | NeuroMuscular Blocking (NMB) - non depolarizing | 372790002 | C0304435
SRT | F-61898 | NSAID | 372665008 | C0003211
SRT | F-61D70 | Ocular Lubricant | 398828005 | C0717951
SRT | F-61E2A | Oxytocic | 410937004 | C0030094
SRT | C-0021D | Penicillin antibiotic | 373270004 | C0030842
SRT | F-616E7 | Plasma Expander | 372578001 | C1268852
SRT | C-13500 | Potassium | 88480006 | C0032821
SRT | C-0024C | Quinolone antibiotic | 372722000 | C1533693
SRT | F-6205D | Respiratory stimulant | 418760000 | C0282685
SRT | F-61899 | Skeletal muscle relaxant | 372666009 | C0037250
SRT | C-10098 | Steroid | 116566001 | C0038317
SRT | C-00257 | Sulfonamide antibiotic | 372788003 | C0599503
SRT | C-00216 | Tetracycline antibiotic | 373206009 | C1744619
SRT | F-B3000 | Thyroid hormone | 18220004 | C0040135
SRT | C-0024E | Vancomycin antibiotic | 372735009 | C0042313
SRT | F-619AA | Vasoconstrictor | 372881000 | C0042397
SRT | F-61957 | Vasodilator | 372787008 | C0042402
SRT | F-BB000 | Vitamin | 87708000 | C0042890

### Note

This context group contains SNOMED substance or product code equivalents of enumerated string concepts for the [AQI Schema] element MedicationTypeCodeType. See http://www.aqihq.org/aqischdoc/MedicationTypeCodeType.html and http://www.aqihq.org/aqischdoc/MedicationTypeCodeTypeExampleList.html.

The AQI value "ABX-Miscellaneous" corresponds to (C-5008C, SRT, "Antibiotic") product, since there is no substance code in SNOMED.

The AQI value "Vasopressor" corresponds to (F-619AA, SRT, "Vasoconstrictor").

No equivalent concepts are included for MedicationTypeCodeType values of NonFormulary antibiotic, Dye, Indigo Carmine Red, and Non-Formulary.

## CID 623 Medication for Small Animal Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1088

Table CID 623. Medication for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 624 “Inhalational Anesthesia Agents for Small Animal Anesthesia”
Include CID 625 “Injectable Anesthesia Agents for Small Animal Anesthesia”
Include CID 626 “Premedication Agents for Small Animal Anesthesia”
Include CID 627 “Neuromuscular Blocking Agents for Small Animal Anesthesia”
Include CID 628 “Ancillary Medications for Small Animal Anesthesia”
Include CID 629 “Carrier Gases for Small Animal Anesthesia”
Include CID 630 “Local Anesthetics for Small Animal Anesthesia”

## CID 624 Inhalational Anesthesia Agents for Small Animal Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1089

Table CID 624. Inhalational Anesthesia Agents for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-10520 | Carbon dioxide | 31811003 | C0007012
SRT | C-20830 | Chloroform | 259153006 | C0008238
SRT | F-61AC9 | Desflurane | 386841003 | C0063252
SRT | C-21216 | Diethyl ether | 259170003 | C0014994
SRT | F-61A3F | Enflurane | 387176008 | C0014277
SRT | F-61AFE | Halothane | 387351001 | C0018549
SRT | F-61B0A | Isoflurane | 387368002 | C0022180
SRT | C-6A16A | Methoxyflurane | 11136004 | C0025688
SRT | C-6A118 | Nitrous oxide | 111132001 | C0028215
SRT | F-61ACA | Sevoflurane | 386842005 | C0074414

### Note

In this context group, SNOMED substance codes are used in preference to product codes, since there is no need to refer to specific products or preparations. SNOMED codes are used in preference to other potential sources of pharmaceutical related codes, such as from the National Drug Code (NDC) directory.

## CID 625 Injectable Anesthesia Agents for Small Animal Anesthesia

Type:

  Extensible

Version:

  20160212

UID:

  1.2.840.10008.6.1.1090

Table CID 625. Injectable Anesthesia Agents for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-23805 | Alphachloralose | 277016007 | C0008162
SRT | C-6A161 | Alphadolone | 125707004 | C0051481
UMLS | C0051482 | Alphaxalone |  | C0051482
SRT | C-640A0 | Azaperone | 96229001 | C0004477
SRT | R-F2B27 | Butabarbital | 372901004 | C0006464
SRT | R-F6E36 | Chloral hydrate | 273948005 | C0008161
SRT | R-F2B2C | Diazepam | 387264003 | C0012010
SRT | F-61A26 | Droperidol | 387146001 | C0013136
SRT | F-61A66 | Etomidate | 387218008 | C0015131
UMLS | C0060473 | Fluanisone |  | C0060473
SRT | F-6182F | Ketamine | 373464007 | C0022614
SRT | F-61848 | Methohexital | 373488009 | C0025668
UMLS | C0025856 | Metomidate |  | C0025856
SRT | F-6183C | Midazolam | 373476007 | C0026056
SRT | R-F2B1F | Pentobarbital | 372703009 | C0030883
SRT | F-61B48 | Propofol | 387423006 | C0033487
SRT | C-6A16B | Thiamylal | 40342009 | C0039855
SRT | F-61BB2 | Thiopental | 387448009 | C0936073
SRT | C-6A190 | Tiletamine | 96265006 | C0242522
SRT | C-6A16E | Tribromoethanol | 84386009 | C0084847
SRT | C-29020 | Urethane (ethyl carbamate) | 873008 | C0041964
SRT | C-640B0 | Xylazine | 96230006 | C0242544
SRT | C-64090 | Zolazepam | 96227004 | C0917859

### Note

In this context group, SNOMED substance codes are used in preference to product codes, since there is no need to refer to specific products or preparations. SNOMED codes are used in preference to other potential sources of pharmaceutical related codes, such as from the National Drug Code (NDC) directory.

## CID 626 Premedication Agents for Small Animal Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1091

Table CID 626. Premedication Agents for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-62960 | Acepromazine | 96218000 | C0000959
SRT | F-61A7F | Chlorpromazine | 387258005 | C0008286

## CID 627 Neuromuscular Blocking Agents for Small Animal Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1092

Table CID 627. Neuromuscular Blocking Agents for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-61916 | Succinylcholine | 372724004 | C0038627
SRT | F-61639 | Pancuronium | 373738000 | C0030310

### Note

In this context group, SNOMED substance codes are used in preference to product codes, since there is no need to refer to specific products or preparations. SNOMED codes are used in preference to other potential sources of pharmaceutical related codes, such as from the National Drug Code (NDC) directory.

## CID 628 Ancillary Medications for Small Animal Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1093

Table CID 628. Ancillary Medications for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---

### Note

This context group is currently empty since no ancillary medications have been identified for this use case yet.

## CID 629 Carrier Gases for Small Animal Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1094

Table CID 629. Carrier Gases for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-6A102 | Oxygen gas | 320917000 | C0350411
SRT | C-6A148 | Medical air | 417696007 | C3536832
UMLS | C3846005 | Room air |  | C3846005

### Note

In this context group, though SNOMED substance codes are normally used in preference to product codes, in the case of (C-6A102, SRT, "Oxygen gas") there is no corresponding substance that is specifically the gaseous form of oxygen.

## CID 630 Local Anesthetics for Small Animal Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1095

Table CID 630. Local Anesthetics for Small Animal Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-61A28 | Bupivacaine | 387150008 | C0006400
SRT | C-80477 | Lidocaine + Prilocaine | 346553009 | C0617623
SRT | F-61BD0 | Lidocaine | 387480006 | C0023660

### Note

- In this context group, SNOMED substance codes are used in preference to product codes, since there is no need to refer to specific products or preparations. SNOMED codes are used in preference to other potential sources of pharmaceutical related codes, such as from the National Drug Code (NDC) directory.

- For Lidocaine + Prilocaine, since it is a mixture of two substances, the code for the product concept is used. The code for a mixture of unspecified type is used, rather than a more specific code, e.g., for the so-called "Eutectic Mixture of Local Anesthetics (EMLA)", which consists of Lidocaine + Prilocaine. UMLS contains three distinct concepts, (C0059079, UMLS, "EMLA"), (C0617623, UMLS, "Lidocaine/Prilocaine") and (C0950567, UMLS, "Eutectic Lidocaine-Prilocaine").

## CID 631 Phase of Procedure Requiring Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1096

Table CID 631. Phase of Procedure Requiring Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 632 “Phase of Surgical Procedure Requiring Anesthesia”
Include CID 633 “Phase of Imaging Procedure Requiring Anesthesia”

## CID 632 Phase of Surgical Procedure Requiring Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1097

Table CID 632. Phase of Surgical Procedure Requiring Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-413C5 | Preoperative | 262068006 | C0445204
SRT | R-400B2 | Intraoperative | 277671009 | C0456904
SRT | R-413B7 | Postoperative | 262061000 | C0032790

## CID 633 Phase of Imaging Procedure Requiring Anesthesia

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1098

Table CID 633. Phase of Imaging Procedure Requiring Anesthesia

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-40FB9 | Before procedure | 307153007 | C0585032
SRT | R-40FBA | During procedure | 307154001 | C0585033
SRT | R-422A4 | After procedure | 303110006 | C0580203

### Note

The concepts used in this context group are more general than those for a specific procedure, e.g., surgery, radiotherapy, chemotherapy. In SNOMED, the concepts used in this context group are the parent concepts of the surgically-specific concepts used in CID 631 “Phase of Procedure Requiring Anesthesia”. There are no concepts defined specifically for periods related to an imaging procedure so the general concepts suffice (in context).

## CID 634 Phase of Animal Handling

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1099

Table CID 634. Phase of Animal Handling

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 127101 | In home cage |  |
DCM | 127102 | During transport |  |
DCM | 127103 | Staging prior to imaging |  |
DCM | 127104 | Preparation for imaging |  |
SRT | P1-C0012 | Anesthesia induction | 241687005 | C0473960
SRT | P0-0099A | Imaging procedure | 363679005 | C0011923
UMLS | C0002908 | Anesthesia recovery period |  | C0002908

## CID 635 Heating Method

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1100

Table CID 635. Heating Method

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-FDB79 | Air heating pad | 468192005 | C3877351
SRT | A-18041 | Electric blanket | 79811009 | C0336614
SRT | A-2C140 | Electric heating pad | 27812008 | C0181157
DCM | 127250 | Forced air heater |  |
SRT | A-17454 | Forced air warming blanket | 420572009 | C1719899
DCM | 127251 | Heated imaging device |  |
DCM | 127252 | Heated patient support |  |
DCM | 127253 | Heated water blanket |  |
UMLS | C0181514 | Heat lamp |  | C0181514
SRT | A-2C141 | Non-electric heating pad | 39790008 | C0521200
DCM | 127254 | Pre-heated pad |  |
DCM | 127255 | Unheated |  |
SRT | A-17450 | Warmer device | 71384000 | C0184348
SRT | A-17452 | Warming blanket | 421335007 | C0184351

## CID 636 Temperature Sensor Device Component Type for Small Animal Procedures

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1101

Table CID 636. Temperature Sensor Device Component Type for Small Animal Procedures

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-00BB8 | Rectal temperature | 307047009 | C0489749
UMLS | C0039810 | Thermography |  | C0039810
DCM | 127240 | Carrier temperature sensor |  |

### Note

(C0039810, UMLS, "Thermography") is a general concept that also encompasses diagnostic uses of thermography, in addition to simple temperature measurement; only the latter meaning is used here, as is implicit from the context of invocation.

## CID 637 Exogenous Substance Types

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1102

Table CID 637. Exogenous Substance Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-C1F9 | Graft material | 246345001 | C0181074
DCM | 127460 | Tumor Graft |  |
SRT | T-1A080 | Fibril | 88921000 | C0225328
SRT | L-30000 | Virus | 49872002 | C0042776
SRT | F-CB250 | Cytokine | 75777003 | C0079189
SRT | C-00224 | Toxin | 80917008 | C0040549

### Note

The specific concept (C22490, NCIt, "Tumor Cell Graft") (UMLS:C1519674) is not used, since grafts may not be cell suspensions, but rather entire tumors, fragments of tumor tissue, etc. Whether the graft is a xenograft or homograft is not specified, and may be encoded elsewhere (e.g., by encoding the species of origin). The non-tumor specific concept (G-C1F9, SRT, "Graft material") may be used when the graft is not a tumor (though strictly speaking, it is a SNOMED attribute rather than substance; UMLS:C0181074 does not make such a distinction).

## CID 638 Exogenous Substance

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1103

Table CID 638. Exogenous Substance

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 639 “Tumor Graft Histologic Type”
Include CID 640 “Fibrils”
Include CID 641 “Viruses”
Include CID 642 “Cytokines”
Include CID 643 “Toxins”

## CID 639 Tumor Graft Histologic Type

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1104

Table CID 639. Tumor Graft Histologic Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-81403 | Adenocarcinoma | 35917007 | C0001418
SRT | M-87303 | Amelanotic melanoma | 70594002 | C0206735
SRT | M-94003 | Astrocytoma | 38713004 | C0004114
NCIt | C2923 | Bronchioloalveolar adenocarcinoma |  | C0007120
SRT | M-80103 | Carcinoma | 68453008 | C0007097
SRT | M-89803 | Carcinosarcoma | 63264007 | C0007140
SRT | M-84403 | Cystadenocarcinoma | 21008007 | C0010631
SRT | M-94403 | Glioblastoma | 63634009 | C0017636
SRT | D0-F0369 | Infiltrating ductal carcinoma of breast | 408643008 | C1134719
SRT | M-80123 | Large cell carcinoma | 22687000 | C0206704
SRT | DC-F4113 | Leukemia | 93143009 | C0023418
SRT | M-87203 | Melanoma | 2092003 | C0025202
SRT | M-90503 | Mesothelioma | 62064005 | C0025500
SRT | M-80453 | Mixed small cell carcinoma | 21326004 | C0334240
SRT | M-80463 | Non-small cell carcinoma | 128632008 | C1266002
SRT | M-91803 | Osteosarcoma | 21708004 | C0029463
SRT | R-FB83F | Renal cell carcinoma | 702391001 | C0007134
SRT | M-88003 | Sarcoma | 2424003 | C1261473
SRT | M-80413 | Small cell carcinoma | 74364000 | C0262584
SRT | M-80323 | Spindle cell carcinoma | 65692009 | C0205697
SRT | M-80703 | Squamous cell carcinoma | 28899001 | C0007137

## CID 640 Fibrils

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1105

Table CID 640. Fibrils

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 127851 | Human alpha synuclein preformed fibrils |  |
DCM | 127852 | Mouse alpha synuclein preformed fibrils |  |
DCM | 127853 | Human Tau preformed fibrils |  |
DCM | 127854 | Mouse Tau preformed fibrils |  |

## CID 641 Viruses

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1106

Table CID 641. Viruses

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | L-30606 | Theiler's murine encephalomyelitis virus | 42024000 | C0206425
SRT | L-35500 | Adeno-associated virus group | 112381006 | C0001417

## CID 642 Cytokines

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1107

Table CID 642. Cytokines

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-CB962 | Tumor necrosis factor alpha | 39525005 | C1456820
SRT | F-C0101 | Interferon gamma | 420303002 | C0021745
SRT | F-CB902 | Vascular endothelial growth factor | 417324009 | C0078058

## CID 643 Toxins

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1108

Table CID 643. Toxins

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-63750 | Lysophosphatidylcholine | 54446009 | C0024360
UMLS | C0019873 | Ethidium Bromide |  | C0019873
PUBCHEM_CID | 4624 | 6-hydroxydopamine |  |
SRT | F-63390 | Lipopolysaccharide | 3325005 | C0023810

## CID 644 Exogenous Substance Administration Sites

Type:

  Extensible

Version:

  20170124

UID:

  1.2.840.10008.6.1.1109

Table CID 644. Exogenous Substance Administration Sites

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-D2310 | Flank | 58602004 | C0230171
SRT | T-A0100 | Brain | 12738006 | C0006104
SRT | M-8FFFF | Tumor | 108369006 | C0027651
SRT | T-62000 | Liver | 10200004 | C0023884
SRT | T-C1000 | Bone Marrow | 14016003 | C0005953
NCIt | C22550 | Mouse mammary fat pad |  | C1512979

### Note

Since this context group defines the sites, rather than routes of administration, if the exogenous substance is administered into a tumor, the code for the morphologic abnormality (M-8FFFF, SRT, "Tumor") is used, rather than the specific concept for the route (R-F2CD4, SRT, "Intratumor route") (which may also be present as the value for the separately encoded route of administration, if present).

## CID 645 Exogenous Substance Tissue of Origin

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1110

Table CID 645. Exogenous Substance Tissue of Origin

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-D04AC | Ascitic fluid | 409615008 | C0003964
SRT | T-D016E | Bone | 272673000 | C0262950
SRT | T-A0100 | Brain | 12738006 | C0006104
SRT | T-04000 | Breast | 76752008 | C0006141
SRT | T-A0090 | Central nervous system | 21483005 | C0927232
SRT | T-59300 | Colon | 71854001 | C0009368
DCM | 127801 | Embryonic kidney |  |
SRT | T-71000 | Kidney | 64033007 | C0022646
SRT | T-28000 | Lung | 39607008 | C0024109
SRT | T-C6020 | Lymph | 38000004 | C0024202
SRT | T-D03C2 | Lymphatic tissue | 181768009 | C0024296
SRT | T-C4000 | Lymph Node | 59441001 | C0024204
SRT | DF-00436 | Metastasis | 128462008 | C2939419
SRT | T-87000 | Ovary | 15497006 | C0029939
SRT | D2-80100 | Pleural effusion | 60046008 | C0032227
SRT | T-92000 | Prostate | 41216001 | C0033572
SRT | D2-F1106 | Pulmonary metastasis | 94391008 | C0153676

## CID 646 Preclinical Small Animal Imaging Procedures

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1111

Table CID 646. Preclinical Small Animal Imaging Procedures

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
LN | 46305-9 | Whole body CT |  | C1830206
LN | 24725-4 | Head CT | 303653007 | C0202691
LN | 46358-8 | MRI whole body | 426252008 | C1830259
LN | 24590-2 | Brain MRI |  | C0881827
LN | 44139-4 | PET whole body | 702767007 | C1715409
LN | 44138-6 | Brain PET | 241434002 | C0412493
LN | 42175-0 | Radionuclide scan of whole body | 229033006 | C1626178
LN | 24730-4 | Radionuclide brain scan | 41440006 | C0581582
DCM | 127901 | SPECT of whole body |  |
LN | 39632-5 | Brain SPECT |  | C1543694
DCM | 127902 | SPECT CT of whole body |  |
SRT | P5-B0008 | Ultrasonography of total body | 24135002 | C0203309

### Note

1. The inconsistent pattern of modality and anatomy in the code meaning is present in the source coding scheme (e.g., "Whole body CT" versus "PET whole body"), and not changed, except where necessary (e.g., (42175-0, LN, "Radionuclide scan of whole body") is actually just "scan of whole body" in the source scheme, which is insufficient, so "radionuclide" has been added).

2. The UMLS codes that map to the SNOMED concepts, when present, are shown, in the cases when UMLS is lacking a mapping between the LOINC and SNOMED codes. E.g., (44138-6, LN, "Brain PET") maps directly to (C1715408, UMLS, "Multisection:Find:Pt:Brain:Doc:Radnuc.PET"), but (P5-0A001, SRT, "PET Brain Study") maps to (C0412493, UMLS, "PET Brain Study"), which is used instead. In general, UMLS does not unify the mappings from LOINC and SNOMED, presumably due to the lexical dissimilarity of the terms (i.e., the LOINC mapping seems to be based on the fully-specified name rather than the long common name).

## CID 647 Position Reference Indicator for Frame of Reference

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1112

Table CID 647. Position Reference Indicator for Frame of Reference

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
FMA | 264776 | Bregma |  | C0934419
FMA | 264773 | Lambda |  | C0926407

### Note

An FMA code is used for bregma since SNOMED only contains fetal bregma.

## CID 701 Content Assessment Types

Type:

  Extensible

Version:

  20160318

UID:

  1.2.840.10008.6.1.1116

Table CID 701. Content Assessment Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 702 “RT Content Assessment Types”

## CID 702 RT Content Assessment Types

Type:

  Extensible

Version:

  20160318

UID:

  1.2.840.10008.6.1.1117

Table CID 702. RT Content Assessment Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121373 | RT Pre-Treatment Dose Check
DCM | 121374 | RT Pre-Treatment Consistency Check

## CID 703 Basis of Assessment

Type:

  Extensible

Version:

  20160318

UID:

  1.2.840.10008.6.1.1118

Table CID 703. Basis of Assessment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121375 | Assessment By Comparison
DCM | 121376 | Assessment By Rules

## CID 800 Protocol Assertion Codes

Type:

  Extensible

Version:

  20160626

UID:

  1.2.840.10008.6.1.1176

Table CID 800. Protocol Assertion Codes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128601 | Appropriate for the indications
DCM | 128621 | Inappropriate for the indications
DCM | 128602 | Consistent with labeling of the device
DCM | 128622 | Inconsistent with labeling of the device
DCM | 128603 | Approved for use at the institution
DCM | 128623 | Disapproved for use at the institution
DCM | 128604 | Approved for use in the clinical trial
DCM | 128624 | Disapproved for use in the clinical trial
DCM | 128611 | Approved for experimental use
DCM | 128612 | Disapproved for experimental use
DCM | 128605 | Approved for use on pregnant patients
DCM | 128617 | Disapproved for use on pregnant patients
DCM | 128609 | Disapproved for any use
DCM | 128613 | Eligible for reimbursement
DCM | 128614 | Eligible for reimbursement on per patient basis
DCM | 128615 | Ineligible for reimbursement
DCM | 128606 | Appropriate for the device
DCM | 128618 | Inappropriate for the device
DCM | 128607 | Inside operational limits of the device
DCM | 128619 | Outside operational limits of the device
DCM | 128608 | Optimized for the device instance
DCM | 128620 | Not optimized for the device instance
DCM | 128610 | Deprecated protocol

## CID 1000 CT Transverse Plane Reference Basis

The items in this context group provide the basis for defining transverse planes associated with the limits of CT acquisitions and reconstructions. It includes body structures, morphologic abnormalities and physical objects that may be the subject or serve as points of reference for imaging.

Type:

  Extensible

Version:

  20160905

UID:

  1.2.840.10008.6.1.1121

Table CID 1000. CT Transverse Plane Reference Basis

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 1001 “Anatomical Reference Basis”
SRT | M-01000 | Morphologically Abnormal Structure | 49755003 | C0332447
SRT | A-12000 | Orthopedic device | 16349000 | C0029352
SRT | A-11100 | Cardiac pacemaker | 14106009 | C0030163
SRT | A-04010 | Implant, device | 40388003 | C0021102
SRT | A-25500 | Stent, device | 65818007 | C0038257
DCM | 128160 | Acquired Volume |  |

## CID 1001 Anatomical Reference Basis

The items in this context group are body structures commonly used as a reference basis for imaging.

Type:

  Extensible

Version:

  20160905

UID:

  1.2.840.10008.6.1.1122

Table CID 1001. Anatomical Reference Basis

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 1002 “Anatomical Reference Basis - Head”
Include CID 1003 “Anatomical Reference Basis - Spine”
Include CID 1004 “Anatomical Reference Basis - Chest”
Include CID 1005 “Anatomical Reference Basis - Abdomen/Pelvis”
Include CID 1006 “Anatomical Reference Basis - Extremities”

## CID 1002 Anatomical Reference Basis - Head

The items in this context group are body structures in the head commonly used as a reference basis for imaging.

Type:

  Extensible

Version:

  20160905

UID:

  1.2.840.10008.6.1.1123

Table CID 1002. Anatomical Reference Basis - Head

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-45010 | Carotid Artery | 69105007 | C0007272
SRT | T-AB200 | External Auditory Meatus | 84301002 | C0013444
SRT | T-11106 | Foramen Magnum | 24532009 | C0016519
SRT | T-22200 | Frontal sinus | 55060009 | C0016734
SRT | T-11134 | Internal Auditory Meatus | 61671002 | C0222711
SRT | T-AA813 | Lateral Canthus | 61242005 | C0229246
SRT | T-11180 | Mandible | 91609006 | C0024687
SRT | T-11133 | Mastoid bone | 59066005 | C0446908
SRT | T-AB500 | Mastoid cells and antra | 91716001 | C0229422
SRT | T-22100 | Maxillary sinus | 15924003 | C0024957
FMA | 264779 | Nasion |  | C0934420
SRT | T-D14AE | Orbital structure | 363654007 | C0029180
SRT | T-D1460 | Pituitary Fossa | 42575006 | C0036609
SRT | T-11100 | Skull | 89546000 | C0037303
SRT | T-11130 | Temporal Bone | 60911003 | C0039484
SRT | T-D1120 | Vertex of Head | 88986008 | C0230003

### Note

1. (T-11134, SRT, "Internal Auditory Meatus") is also known as the "Internal Auditory Canal".

2. (T-AB200, SRT, "External Auditory Meatus") is also known as the "External Auditory Canal".

3. (T-D1460, SRT, "Pituitary Fossa") is also known as the "Sella Turcica".

## CID 1003 Anatomical Reference Basis - Spine

The items in this context group are body structures in the spine commonly used as a reference basis for imaging.

Type:

  Extensible

Version:

  20160905

UID:

  1.2.840.10008.6.1.1124

Table CID 1003. Anatomical Reference Basis - Spine

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-11610 | C1 vertebra | 14806007 | C0004170
SRT | T-D005D | Level of C2/C3 intervertebral disc | 243902007 | C0446383
SRT | T-D005E | Level of C3/C4 intervertebral disc | 243903002 | C0446384
SRT | T-D005F | Level of C4/C5 intervertebral disc | 243904008 | C0446385
SRT | T-D007C | Level of C5/C6 intervertebral disc | 243905009 | C0446386
SRT | T-D007D | Level of C6/C7 intervertebral disc | 243906005 | C0446387
SRT | T-D009C | Level of C7/T1 intervertebral disc | 243925008 | C0446406
SRT | T-D0097 | Level of L1/L2 intervertebral disc | 243920003 | C0446401
SRT | T-D0098 | Level of L2/L3 intervertebral disc | 243921004 | C0446402
SRT | T-D0099 | Level of L3/L4 intervertebral disc | 243922006 | C0446404
SRT | T-D009A | Level of L4/L5 intervertebral disc | 243923001 | C0446403
SRT | T-D009E | Level of L5/S1 intervertebral disc | 243927000 | C0446408
SRT | T-D007F | Level of T1/T2 intervertebral disc | 243908006 | C0446389
SRT | T-D0094 | Level of T10/T11 intervertebral disc | 243917006 | C0446398
SRT | T-D0095 | Level of T11/T12 intervertebral disc | 243918001 | C0446399
SRT | T-D009D | Level of T12/L1 intervertebral disc | 243926009 | C0446407
SRT | T-D008B | Level of T2/T3 intervertebral disc | 243909003 | C0446390
SRT | T-D008C | Level of T3/T4 intervertebral disc | 243910008 | C0446391
SRT | T-D008D | Level of T4/T5 intervertebral disc | 243911007 | C0446392
SRT | T-D008E | Level of T5/T6 intervertebral disc | 243912000 | C0446393
SRT | T-D008F | Level of T6/T7 intervertebral disc | 243913005 | C0446394
SRT | T-D0091 | Level of T7/T8 intervertebral disc | 243914004 | C0446395
SRT | T-D0092 | Level of T8/T9 intervertebral disc | 243915003 | C0446396
SRT | T-D0093 | Level of T9/T10 intervertebral disc | 243916002 | C0446397

## CID 1004 Anatomical Reference Basis - Chest

The items in this context group are body structures in the chest commonly used as a reference basis for imaging.

Type:

  Extensible

Version:

  20160905

UID:

  1.2.840.10008.6.1.1125

Table CID 1004. Anatomical Reference Basis - Chest

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-15420 | Acromioclavicular Joint | 85856004 | C0001208
SRT | T-42300 | Aortic Arch | 57034009 | C0003489
SRT | T-32004 | Apex of heart | 13383001 | C0225811
SRT | T-25201 | Carina | 28700002 | C0225594
SRT | T-41065 | Coronary artery graft | 264293000 | C0440761
SRT | T-D3400 | Diaphragm | 5798000 | C0011980
SRT | T-32000 | Heart | 80891009 | C0018787
SRT | T-28000 | Lung | 39607008 | C0024109
SRT | T-12280 | Scapula | 79601000 | C0036277
SRT | T-D2220 | Shoulder region structure | 16982005 | C0037004
SRT | T-15610 | Sternoclavicular Joint | 7844006 | C0038291
SRT | T-11210 | Sternum | 56873002 | C0038293
SRT | T-11218 | Suprasternal Notch | 26493002 | C0222769
SRT | T-D3160 | Thoracic Inlet | 42973007 | C0230137
SRT | T-11227 | Xiphoid Process | 20298003 | C0043356

### Note

(T-11218, SRT, "Suprasternal Notch") is also known as the "Jugular Notch (of Sternum) " and the "Sternal Notch".

## CID 1005 Anatomical Reference Basis - Abdomen/Pelvis

The items in this context group are body structures in the abdomen and pelvis commonly used as a reference basis for imaging.

Type:

  Extensible

Version:

  20160905

UID:

  1.2.840.10008.6.1.1126

Table CID 1005. Anatomical Reference Basis - Abdomen/Pelvis

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-12390 | Acetabulum | 37783008 | C0000962
SRT | T-B3000 | Adrenal gland | 23451007 | C0001625
SRT | R-10258 | Common iliac artery bifurcation | 413896006 | C1531837
SRT | T-12711 | Femoral head | 2812003 | C0015813
SRT | T-15710 | Hip joint | 24136001 | C0019558
SRT | T-1234A | Iliac Crest | 29850006 | C0223651
SRT | T-12350 | Ischium | 85710004 | C0022122
SRT | T-71000 | Kidney | 64033007 | C0022646
SRT | T-12714 | Lesser trochanter | 55499008 | C0223866
SRT | T-62000 | Liver | 10200004 | C0023884
SRT | T-65000 | Pancreas | 15776009 | C0030274
SRT | T-11AD0 | Sacrum | 54735007 | C0036037
SRT | T-15690 | Symphysis pubis | 82561000 | C0034015

### Note

(T-15690, SRT, "Symphysis pubis") is also known as the "Pubic Symphysis".

## CID 1006 Anatomical Reference Basis - Extremities

The items in this context group are body structures in the extremities commonly used as a reference basis for imaging.

Type:

  Extensible

Version:

  20160905

UID:

  1.2.840.10008.6.1.1127

Table CID 1006. Anatomical Reference Basis - Extremities

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-15750 | Ankle joint | 70258002 | C0003087
SRT | T-15430 | Elbow joint | 16953009 | C0013770
SRT | T-D9700 | Foot | 56459004 | C0016504
SRT | T-15720 | Knee joint | 49076000 | C0022745
SRT | T-127A7 | Malleolar structure of tibia | 314796009 | C1282383
SRT | T-12540 | Metacarpal | 36455000 | C0025526
SRT | T-12730 | Patella | 64234005 | C0030647
SRT | T-12450 | Scaphoid | 30518006 | C0223724
SRT | T-12780 | Talus | 67453005 | C0039277
SRT | T-1273F | Tibial Plateau | 306783000 | C0584640
SRT | T-D9800 | Toe | 29707007 | C0040357
SRT | T-15460 | Wrist joint | 74670003 | C1322271

### Note

(T-12450, SRT, "Scaphoid") is also known as the "Radial Carpal".

## CID 1010 Reference Geometry - Planes

The items in this context group identify a specific plane associated with a reference basis (see CID 1000 “CT Transverse Plane Reference Basis”). The plane is defined by the intersection of the scan plane with the specified extent of the reference basis.

Type:

  Extensible

Version:

  20160905

UID:

  1.2.840.10008.6.1.1128

Table CID 1010. Reference Geometry - Planes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128128 | Plane through Anterior Extent
DCM | 128123 | Plane through Distal Extent
DCM | 128121 | Plane through Inferior Extent
DCM | 128125 | Plane through Lateral Extent
DCM | 128126 | Plane through Leftmost Extent
DCM | 128124 | Plane through Medial Extent
DCM | 128130 | Plane through Center
DCM | 128129 | Plane through Posterior Extent
DCM | 128122 | Plane through Proximal Extent
DCM | 128127 | Plane through Rightmost Extent
DCM | 128120 | Plane through Superior Extent

## CID 1011 Reference Geometry - Points

The items in this context group identify a specific point associated with a reference basis (see CID 1000 “CT Transverse Plane Reference Basis”).

Type:

  Extensible

Version:

  20160905

UID:

  1.2.840.10008.6.1.1129

Table CID 1011. Reference Geometry - Points

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128137 | Geometric Centerpoint
DCM | 128138 | Center of Mass

## CID 1015 Patient Alignment Methods

The items in this context group identify methods for aligning a patient (or other imaging subject) in a scanner.

Type:

  Extensible

Version:

  20160905

UID:

  1.2.840.10008.6.1.1130

Table CID 1015. Patient Alignment Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128151 | Laser Cross-hairs

## CID 1200 Contraindications For CT Imaging

The items in this context group identify possible contraindications for specific CT imaging protocols. Contraindications for CT imaging in general, irrespective of the Protocol used, are not included here.

Type:

  Extensible

Version:

  20160905

UID:

  1.2.840.10008.6.1.1131

Table CID 1200. Contraindications For CT Imaging

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | DF-10F42 | X-ray Contrast Media Allergy | 293638001 | C0570563
SRT | DF-10F41 | Contrast Media Allergy | 293637006 | C0570562
SRT | F-84000 | Patient currently pregnant | 77386006 | C0549206
SRT | D7-11007 | Impaired Renal Function | 236423003 | C1565489

## CID 3000 Audio Channel Source

Type:

  Extensible

Version:

  20040326

UID:

  1.2.840.10008.6.1.42

Table CID 3000. Audio Channel Source

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109110 | Voice
DCM | 109111 | Operator's narrative
DCM | 109112 | Ambient room environment
DCM | 109113 | Doppler audio
DCM | 109114 | Phonocardiogram
DCM | 109115 | Physiological audio signal

## CID 3001 ECG Leads

This Context Group comprises the ECG lead identifiers of ISO/IEEE 11073-10101, including human and canine leads. The terms included in the table below may not constitute the complete list; see the ISO/IEEE Standard.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10101 available through http://standards.ieee.org.

Type:

  Extensible

Version:

  20130613

UID:

  1.2.840.10008.6.1.43

Table CID 3001. ECG Leads

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:71 | Lead A (Nehb - Anterior) | MDC_ECG_LEAD_A
MDC | 2:75 | Auxiliary unipolar lead 1 | MDC_ECG_LEAD_A1
MDC | 2:76 | Auxiliary unipolar lead 2 | MDC_ECG_LEAD_A2
MDC | 2:77 | Auxiliary unipolar lead 3 | MDC_ECG_LEAD_A3
MDC | 2:78 | Auxiliary unipolar lead 4 | MDC_ECG_LEAD_A4
MDC | 2:127 | Auxiliary bipolar lead 1 | MDC_ECG_LEAD_AB1
MDC | 2:128 | Auxiliary bipolar lead 2 | MDC_ECG_LEAD_AB2
MDC | 2:129 | Auxiliary bipolar lead 3 | MDC_ECG_LEAD_AB3
MDC | 2:130 | Auxiliary bipolar lead 4 | MDC_ECG_LEAD_AB4
MDC | 2:133 | EASI Lead AI | MDC_ECG_LEAD_AI
MDC | 2:132 | EASI Lead AS | MDC_ECG_LEAD_AS
MDC | 2:64 | aVF, augmented voltage, foot | MDC_ECG_LEAD_AVF
MDC | 2:63 | aVL, augmented voltage, left | MDC_ECG_LEAD_AVL
MDC | 2:62 | aVR, augmented voltage, right | MDC_ECG_LEAD_AVR
MDC | 2:65 | −aVR | MDC_ECG_LEAD_AVRneg
MDC | 2:86 | Chest lead | MDC_ECG_LEAD_C
MDC | 2:124 | negative: low right scapula Lead | MDC_ECG_LEAD_CB5
MDC | 2:98 | Chest lead (symmetric placement) | MDC_ECG_LEAD_CC
MDC | 2:99 | Chest lead per V1 and V1R placement | MDC_ECG_LEAD_CC1
MDC | 2:100 | Chest lead per V2 and V2R placement | MDC_ECG_LEAD_CC2
MDC | 2:101 | Chest lead per V3 and V3R placement | MDC_ECG_LEAD_CC3
MDC | 2:102 | Chest lead per V4 and V4R placement | MDC_ECG_LEAD_CC4
MDC | 2:19 | Chest lead per V5 and V5R placement | MDC_ECG_LEAD_CC5
MDC | 2:103 | Chest lead per V6 and V6R placement | MDC_ECG_LEAD_CC6
MDC | 2:104 | Chest lead per V7 and V8R placement | MDC_ECG_LEAD_CC7
MDC | 2:122 | Lead CH5 | MDC_ECG_LEAD_CH5
MDC | 2:105 | Chest-manubrium lead | MDC_ECG_LEAD_CM
MDC | 2:106 | Chest-manubrium lead per V1 placement | MDC_ECG_LEAD_CM1
MDC | 2:107 | Chest-manubrium lead per V2 placement | MDC_ECG_LEAD_CM2
MDC | 2:108 | Chest-manubrium lead per V3 placement | MDC_ECG_LEAD_CM3
MDC | 2:109 | Chest-manubrium lead per V4 placement | MDC_ECG_LEAD_CM4
MDC | 2:20 | Chest-manubrium lead per V5 placement | MDC_ECG_LEAD_CM5
MDC | 2:110 | Chest-manubrium lead per V6 placement | MDC_ECG_LEAD_CM6
MDC | 2:121 | Chest-manubrium lead per V7 placement | MDC_ECG_LEAD_CM7
MDC | 2:125 | Lead CR5 | MDC_ECG_LEAD_CR5
MDC | 2:123 | negative: right infraclavicular fossa | MDC_ECG_LEAD_CS5
MDC | 2:148 | Canine, fifth right intercostal space near edge of sternum | MDC_ECG_LEAD_CV5RL
MDC | 2:149 | Canine, sixth left intercostal space near edge of sternum | MDC_ECG_LEAD_CV6LL
MDC | 2:150 | Canine, sixth left intercostal space at costochondral junction | MDC_ECG_LEAD_CV6LU
MDC | 2:70 | Lead D (Nehb - Dorsal) | MDC_ECG_LEAD_D
MDC | 2:114 | Derived Lead aVF | MDC_ECG_LEAD_dAVF
MDC | 2:113 | Derived Lead aVL | MDC_ECG_LEAD_dAVL
MDC | 2:112 | Derived Lead aVR | MDC_ECG_LEAD_dAVR
MDC | 2:73 | Defibrillator lead: anterior-lateral | MDC_ECG_LEAD_DEFIB
MDC | 2:31 | Derived Lead I | MDC_ECG_LEAD_dI
MDC | 2:32 | Derived Lead II | MDC_ECG_LEAD_dII
MDC | 2:111 | Derived Lead III | MDC_ECG_LEAD_dIII
MDC | 2:33 | Derived Lead V1 | MDC_ECG_LEAD_dV1
MDC | 2:34 | Derived Lead V2 | MDC_ECG_LEAD_dV2
MDC | 2:35 | Derived Lead V3 | MDC_ECG_LEAD_dV3
MDC | 2:36 | Derived Lead V4 | MDC_ECG_LEAD_dV4
MDC | 2:37 | Derived Lead V5 | MDC_ECG_LEAD_dV5
MDC | 2:38 | Derived Lead V6 | MDC_ECG_LEAD_dV6
MDC | 2:131 | EASI Lead ES | MDC_ECG_LEAD_ES
MDC | 2:74 | External pacing lead: anterior-posterior | MDC_ECG_LEAD_EXTERN
MDC | 2:27 | Frank Lead A | MDC_ECG_LEAD_fA
MDC | 2:26 | Frank Lead C | MDC_ECG_LEAD_fC
MDC | 2:25 | Frank Lead E | MDC_ECG_LEAD_fE
MDC | 2:29 | Frank Lead F | MDC_ECG_LEAD_fF
MDC | 2:30 | Frank Lead H | MDC_ECG_LEAD_fH
MDC | 2:24 | Frank Lead I | MDC_ECG_LEAD_fI
MDC | 2:28 | Frank Lead M | MDC_ECG_LEAD_fM
MDC | 2:1 | Lead I | MDC_ECG_LEAD_I
MDC | 2:2 | Lead II | MDC_ECG_LEAD_II
MDC | 2:61 | Lead III | MDC_ECG_LEAD_III
MDC | 2:72 | Lead J (Nehb - Inferior) | MDC_ECG_LEAD_J
MDC | 2:21 | Left Arm Lead | MDC_ECG_LEAD_LA
MDC | 2:23 | Left Leg Lead | MDC_ECG_LEAD_LL
MDC | 2:91 | Modified chest lead (left arm indifferent) | MDC_ECG_LEAD_MCL
MDC | 2:92 | Modified chest lead per V1 placement | MDC_ECG_LEAD_MCL1
MDC | 2:93 | Modified chest lead per V2 placement | MDC_ECG_LEAD_MCL2
MDC | 2:94 | Modified chest lead per V3 placement | MDC_ECG_LEAD_MCL3
MDC | 2:95 | Modified chest lead per V4 placement | MDC_ECG_LEAD_MCL4
MDC | 2:96 | Modified chest lead per V5 placement | MDC_ECG_LEAD_MCL5
MDC | 2:97 | Modified chest lead per V6 placement | MDC_ECG_LEAD_MCL6
MDC | 2:126 | Modified limb lead | MDC_ECG_LEAD_ML
MDC | 2:22 | Right Arm Lead | MDC_ECG_LEAD_RA
MDC | 2:147 | Right Leg Lead | MDC_ECG_LEAD_RL
MDC | 2:134 | EASI upper sternum lead | MDC_ECG_LEAD_S
MDC | 2:87 | Precordial lead | MDC_ECG_LEAD_V
MDC | 2:3 | Lead V1 | MDC_ECG_LEAD_V1
MDC | 2:151 | Canine, over dorsal spinous process of 7th thoracic vertebra | MDC_ECG_LEAD_V10
MDC | 2:4 | Lead V2 | MDC_ECG_LEAD_V2
MDC | 2:10 | Lead V2R | MDC_ECG_LEAD_V2R
MDC | 2:5 | Lead V3 | MDC_ECG_LEAD_V3
MDC | 2:11 | Lead V3R | MDC_ECG_LEAD_V3R
MDC | 2:6 | Lead V4 | MDC_ECG_LEAD_V4
MDC | 2:12 | Lead V4R | MDC_ECG_LEAD_V4R
MDC | 2:7 | Lead V5 | MDC_ECG_LEAD_V5
MDC | 2:13 | Lead V5R | MDC_ECG_LEAD_V5R
MDC | 2:8 | Lead V6 | MDC_ECG_LEAD_V6
MDC | 2:14 | Lead V6R | MDC_ECG_LEAD_V6R
MDC | 2:9 | Lead V7 | MDC_ECG_LEAD_V7
MDC | 2:15 | Lead V7R | MDC_ECG_LEAD_V7R
MDC | 2:66 | Lead V8 | MDC_ECG_LEAD_V8
MDC | 2:68 | Lead V8R | MDC_ECG_LEAD_V8R
MDC | 2:67 | Lead V9 | MDC_ECG_LEAD_V9
MDC | 2:69 | Lead V9R | MDC_ECG_LEAD_V9R
MDC | 2:90 | Lead VF, nonaugmented voltage, vector of LL | MDC_ECG_LEAD_VF
MDC | 2:89 | Lead VL, nonaugmented voltage, vector of LA | MDC_ECG_LEAD_VL
MDC | 2:88 | Lead VR, nonaugmented voltage, vector of RA | MDC_ECG_LEAD_VR
MDC | 2:16 | Lead X | MDC_ECG_LEAD_X
MDC | 2:17 | Lead Y | MDC_ECG_LEAD_Y
MDC | 2:18 | Lead Z | MDC_ECG_LEAD_Z
MDC | 2:0 | Unspecified lead | MDC_ECG_LEAD_CONFIG

### Note

1. A prior version of this context group used codes from the SCP-ECG vocabulary.

2. In a prior version of this table, the code 2:26 was specified for the concept Chest lead and the code 2:19 was specified for the concept Chest-manubrium lead per V5 placement.

## CID 3003 Hemodynamic Waveform Sources

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.44

Table CID 3003. Hemodynamic Waveform Sources

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-DB22 | Aortic pressure waveform | 128444004 | C1264738
SRT | G-DB31 | Aortic valve pullback pressure waveform | 128453006 | C1264746
SRT | G-DB24 | Arterial pressure waveform | 128446002 | C0444695
SRT | G-DB23 | Central venous pressure waveform | 128445003 | C1264739
SRT | G-DB33 | Dye dilution cardiac output waveform | 128455004 | C1264748
SRT | G-DB20 | Femoral artery pressure waveform | 128442000 | C1264737
SRT | G-DB12 | Hemodynamic flow waveform | 128434001 | C1264729
SRT | G-DB34 | Hemodynamic impedance waveform | 128552003 | C1264749
SRT | G-DB13 | Hemodynamic oxygen saturation waveform | 128435000 | C1264730
SRT | G-DB11 | Hemodynamic pressure waveform | 128433007 | C1264728
SRT | G-DB10 | Hemodynamic waveform | 128432002 | C1264727
SRT | G-DB19 | Left atrium pressure waveform | 128441007 | C1264736
SRT | G-DB16 | Left ventricle pressure waveform | 128438003 | C1264733
SRT | G-DB28 | Mitral valve pullback pressure waveform | 128450009 | C1264743
SRT | G-DB25 | Pulmonary artery oxygen saturation waveform | 128447006 | C1264740
SRT | G-DB21 | Pulmonary artery pressure waveform | 128443005 | C0428729
SRT | G-DB27 | Pulmonary artery wedge pressure waveform | 128449009 | C1264742
SRT | G-DB26 | Pulmonary capillary wedge pressure waveform | 128448001 | C1264741
SRT | G-DB30 | Pulmonary valve pullback pressure waveform | 128452001 | C1264745
SRT | G-DB14 | Respiration impedance waveform | 128436004 | C1264731
SRT | G-DB18 | Right atrium pressure waveform | 128440008 | C1264735
SRT | G-DB17 | Right ventricle pressure waveform | 128439006 | C1264734
SRT | G-DB15 | Temperature waveform | 128437008 | C1264732
SRT | G-DB32 | Thermal cardiac output waveform | 128454000 | C1264747
SRT | G-DB29 | Tricuspid valve pullback pressure waveform | 128451008 | C1264744

## CID 3004 Arterial Pulse Waveform

Type:

  Extensible

Version:

  20090409

UID:

  1.2.840.10008.6.1.803

Table CID 3004. Arterial Pulse Waveform

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109116 | Arterial Pulse Waveform

## CID 3005 Respiration Waveform

Type:

  Extensible

Version:

  20090409

UID:

  1.2.840.10008.6.1.804

Table CID 3005. Respiration Waveform

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109117 | Respiration Waveform

## CID 3010 Cardiovascular Anatomic Locations

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.45

Table CID 3010. Cardiovascular Anatomic Locations

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-42500 | Abdominal aorta | 7832008 | C0003484
SRT | T-48503 | Anomalous pulmonary vein | 128585006 | C0265914
SRT | T-49215 | Antecubital vein | 128553008 | C1276271
SRT | T-48403 | Anterior cardiac vein | 194996006 | C0226662
SRT | T-45530 | Anterior communicating artery | 8012006 | C0149562
SRT | T-45730 | Anterior spinal artery | 17388009 | C0149603
SRT | T-47700 | Anterior tibial artery | 68053000 | C0085816
SRT | T-42000 | Aorta | 15825003 | C0003483
SRT | T-42300 | Aortic arch | 57034009 | C0003489
SRT | D3-81922 | Aortic fistula | 128551005 | C1290392
SRT | T-32602 | Apex of left ventricle | 128564006 | C0580781
SRT | T-32502 | Apex of right ventricle | 128565007 | C0445242
SRT | T-41000 | Artery | 51114001 | C0003842
SRT | T-42100 | Ascending aorta | 54247002 | C0003956
SRT | T-47100 | Axillary Artery | 67937003 | C0004455
SRT | T-49110 | Axillary vein | 68705008 | C0004456
SRT | T-48340 | Azygos vein | 72107004 | C0004526
SRT | A-00203 | Baffle | 128981007 | C1289790
SRT | T-45800 | Basilar artery | 59011009 | C0004811
SRT | T-D00AB | Body conduit | 91830000 | C1735317
SRT | T-49424 | Boyd's perforating vein | 128548003 | C1267522
SRT | T-47160 | Brachial artery | 17137000 | C0006087
SRT | T-49350 | Brachial vein | 20115005 | C0226812
SRT | T-45010 | Carotid Artery | 69105007 | C0007272
SRT | T-49240 | Cephalic vein | 20699002 | C0226802
SRT | T-45510 | Cerebral artery | 88556005 | C0007770
SRT | D4-31005 | Common atrium | 253276007 | C0392482
SRT | T-45100 | Common carotid artery | 32062004 | C0162859
SRT | T-46710 | Common iliac artery | 73634005 | C1261084
SRT | T-48920 | Common iliac vein | 46027005 | C0226758
SRT | D4-31120 | Common ventricle | 45503006 | C0152424
SRT | D4-32504 | Congenital coronary artery fistula to left atrium | 128555001 | C1290487
SRT | D4-32506 | Congenital coronary artery fistula to left ventricle | 128556000 | C1290488
SRT | D4-32509 | Congenital coronary artery fistula to right atrium | 128557009 | C1290489
SRT | D4-32510 | Congenital coronary artery fistula to right ventricle | 128558004 | C1290490
SRT | D3-40208 | Congenital pulmonary arteriovenous fistula | 111289009 | C0155675
SRT | T-43000 | Coronary artery | 41801008 | C0205042
SRT | T-48410 | Coronary sinus | 90219004 | C0456944
SRT | T-42400 | Descending aorta | 32672002 | C3163626
SRT | T-49429 | Dodd's perforating vein | 128554002 | C1267525
SRT | T-45200 | External carotid artery | 22286001 | C0007275
SRT | T-46910 | External iliac artery | 113269004 | C0226398
SRT | T-48930 | External iliac vein | 63507001 | C0226761
SRT | T-45240 | Facial artery | 23074001 | C0226109
SRT | T-47400 | Femoral artery | 7657000 | C0015801
SRT | T-49410 | Femoral vein | 83419000 | C0015809
SRT | T-48820 | Gastric vein | 110568007 | C0750610
SRT | T-47490 | Genicular artery | 128559007 | C0447108
SRT | T-48420 | Great cardiac vein | 5928000 | C0226659
SRT | T-46420 | Hepatic artery | 76015000 | C0019145
SRT | T-48720 | Hepatic vein | 8993003 | C0019155
SRT | T-4942A | Hunterian perforating vein | 128560002 | C1267526
SRT | T-46700 | Iliac artery | 10293006 | C0020887
SRT | T-484A4 | Inferior cardiac vein | 195416006 | C0226664
SRT | T-48540 | Inferior left pulmonary vein | 51249003 | C0226686
SRT | T-46520 | Inferior mesenteric artery | 33795007 | C0162860
SRT | T-48520 | Inferior right pulmonary vein | 113273001 | C0226676
SRT | T-48710 | Inferior vena cava | 64131007 | C0042458
SRT | T-46010 | Innominate artery | 12691009 | C0006094
SRT | T-48620 | Innominate vein | 8887007 | C0006095
SRT | T-45300 | Internal carotid artery | 86117002 | C0007276
SRT | T-48170 | Internal jugular vein | 12123001 | C0226550
SRT | T-46740 | Internal iliac artery | 90024005 | C0226364
SRT | T-46200 | Internal mammary artery | 69327007 | C0226276
SRT | D4-31052 | Juxtaposed atrial appendage | 128563000 | C1290478
SRT | T-45410 | Lacrimal artery | 59749000 | C0226171
SRT | T-45416 | Lacrimal artery of right eye | 128979005 | C0923299
SRT | T-32300 | Left atrium | 82471001 | C0225860
SRT | T-32310 | Left auricular appendage | 33626005 | C0225861
SRT | T-47420 | Left femoral artery | 113270003 | C0226448
SRT | T-44400 | Left pulmonary artery | 50408007 | C0226069
SRT | T-32600 | Left ventricle | 87878005 | C0225897
SRT | T-32640 | Left ventricle inflow | 70238003 | C0225911
SRT | T-32650 | Left ventricle outflow tract | 13418002 | C0225912
SRT | T-45230 | Lingual artery | 113264009 | C0226104
SRT | T-46960 | Lumbar artery | 34635009 | C0226408
SRT | T-46500 | Mesenteric artery | 86570000 | C0025465
SRT | T-4884A | Mesenteric vein | 128583004 | C0025473
SRT | T-45250 | Occipital artery | 31145008 | C0226117
SRT | T-48214 | Occipital vein | 32114007 | C0226579
SRT | T-45400 | Ophthalmic artery | 53549008 | C0029078
SRT | D4-32012 | Patent ductus arteriosus | 83330001 | C0013274
SRT | T-47630 | Peroneal artery | 8821006 | C0226476
SRT | T-47500 | Popliteal artery | 43899006 | C0032649
SRT | T-48810 | Portal vein | 32764006 | C0032718
SRT | T-45320 | Posterior communication artery | 43119007 | C0149559
SRT | T-49535 | Posterior medial tributary | 128569001 | C1267527
SRT | T-47600 | Posterior tibial artery | 13363002 | C0086835
SRT | T-F7001 | Primitive aorta | 14944004 | C0231136
SRT | T-F7040 | Primitive pulmonary artery | 91707000 | C0231157
SRT | T-44000 | Pulmonary artery | 81040000 | C0034052
SRT | D4-33142 | Pulmonary artery conduit | 128584005 | C1290491
SRT | T-32190 | Pulmonary chamber of cor triatriatum | 128586007 | C1267246
SRT | T-48581 | Pulmonary vein | 122972007 | C0034090
SRT | D4-33512 | Pulmonary vein confluence | 128566008 | C1290492
SRT | D4-33514 | Pulmonary venous atrium | 128567004 | C1290493
SRT | T-47300 | Radial artery | 45631007 | C0162857
SRT | T-46600 | Renal artery | 2841007 | C0035065
SRT | T-48740 | Renal vein | 56400007 | C0035092
SRT | T-32200 | Right atrium | 73829009 | C0225844
SRT | T-32210 | Right auricular appendage | 68300000 | C0225845
SRT | T-47410 | Right femoral artery | 69833005 | C0226447
SRT | T-44200 | Right pulmonary artery | 78480002 | C0226054
SRT | T-32500 | Right ventricle | 53085002 | C0225883
SRT | T-32540 | Right ventricle inflow | 8017000 | C0225891
SRT | T-32550 | Right ventricle outflow tract | 44627009 | C0225892
SRT | T-D930A | Saphenofemoral junction | 128587003 | C0447132
SRT | T-49530 | Saphenous vein | 60734001 | C0036186
SRT | T-46460 | Splenic artery | 22083002 | C0037996
SRT | T-48890 | Splenic vein | 35819009 | C0038001
SRT | T-46100 | Subclavian artery | 36765005 | C0038530
SRT | T-48330 | Subclavian vein | 9454009 | C0038532
SRT | T-45270 | Superficial temporal artery | 15672000 | C0226130
SRT | T-48530 | Superior left pulmonary vein | 43863001 | C0226682
SRT | T-46510 | Superior mesenteric artery | 42258001 | C0162861
SRT | T-48510 | Superior right pulmonary vein | 8629005 | C0226671
SRT | T-45210 | Superior thyroid artery | 72021004 | C0226093
SRT | T-48610 | Superior vena cava | 48345005 | C0042459
SRT | T-44007 | Systemic collateral artery to lung | 128589000 | C0345096
SRT | D4-33516 | Systemic venous atrium | 128568009 | C1290494
SRT | T-42070 | Thoracic aorta | 113262008 | C1522460
SRT | D4-31400 | Truncus arteriosus communis | 61959006 | C0041207
SRT | T-46400 | Truncus coeliacus | 57850000 | C0007569
SRT | T-47200 | Ulnar artery | 44984001 | C0162858
SRT | T-F1810 | Umbilical artery | 50536004 | C0041632
SRT | T-48832 | Umbilical vein | 284639000 | C0226734
SRT | T-48000 | Vein | 29092000 | C0042449
SRT | T-48003 | Venous network | 34340008 | C0226503
SRT | T-45700 | Vertebral artery | 85234005 | C0042559

## CID 3011 Electrophysiology Anatomic Locations

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.46

Table CID 3011. Electrophysiology Anatomic Locations

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-32850 | Accessory atrioventricular bundle | 8225009 | C0006383
SRT | T-32602 | Apex of left ventricle | 128564006 | C0580781
SRT | T-32502 | Apex of right ventricle | 128565007 | C0445242
SRT | T-32830 | Atrioventricular bundle | 345000 | C0006382
SRT | T-32820 | Atrioventricular node | 25943004 | C0004247
SRT | D4-31120 | Common ventricle | 45503006 | C0152424
SRT | T-48410 | Coronary sinus | 90219004 | C0456944
SRT | T-39010 | Epicardium | 6871001 | C0225968
SRT | T-48420 | Great cardiac vein | 5928000 | C0226659
SRT | G-DE02 | High right atrium | 128591008 | C0456955
SRT | T-48540 | Inferior left pulmonary vein | 51249003 | C0226686
SRT | T-48520 | Inferior right pulmonary vein | 113273001 | C0226676
SRT | G-DE04 | Lateral high right atrium | 128592001 | C1264751
SRT | T-32833 | Left anterior division of left branch atrioventricular bundle | 84654008 | C0225918
SRT | T-32300 | Left Atrium | 82471001 | C0225860
SRT | T-32310 | Left auricular appendage | 33626005 | C0225861
SRT | T-32832 | Left branch of atrioventricular bundle | 74031005 | C0459156
SRT | T-32834 | Left posterior division of left branch atrioventricular bundle | 91085002 | C0225919
SRT | T-32600 | Left ventricle | 87878005 | C0225897
SRT | T-32640 | Left ventricle inflow | 70238003 | C0225911
SRT | T-32650 | Left ventricle outflow tract | 13418002 | C0225912
SRT | G-DE08 | Low right atrium | 128594000 | C0456956
SRT | G-DE06 | Mid right atrium | 128593006 | C0225856
SRT | T-48430 | Middle cardiac vein | 73580002 | C0226660
SRT | T-35310 | Mitral ring | 65197004 | C0225947
SRT | T-48411 | Ostium of coronary sinus | 71271007 | C0226656
SRT | T-48581 | Pulmonary vein | 122972007 | C0034090
SRT | T-35210 | Pulmonic ring | 90318009 | C0225935
SRT | T-32840 | Purkinje fibers | 13050003 | C0034144
SRT | T-35120 | Right atrioventricular ostium | 90561006 | C0225927
SRT | T-32200 | Right Atrium | 73829009 | C0225844
SRT | T-32210 | Right auricular appendage | 68300000 | C0225845
SRT | T-32831 | Right branch of Atrioventricular bundle | 57383004 | C0225916
SRT | T-32500 | Right ventricle | 53085002 | C0225883
SRT | T-32540 | Right ventricle inflow | 8017000 | C0225891
SRT | T-32550 | Right ventricle outflow tract | 44627009 | C0225892
SRT | T-32810 | Sino-atrial node | 88210001 | C0037189
SRT | T-48530 | Superior left pulmonary vein | 43863001 | C0226682
SRT | T-48510 | Superior right pulmonary vein | 8629005 | C0226671
SRT | T-32202 | Tendon of Todaro | 128595004 | C0456939
SRT | T-35110 | Tricuspid ring | 113259005 | C0225926

### Note

In a prior version of this Context Group the code T-48500 rather than T-48581 was defined for the concept Pulmonary Vein; this was inconsistent with the DICOM approach of selecting the "structure of" rather than "entire" concept. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 3014 Coronary Artery Segments

Type:

  Extensible

Version:

  20130403

UID:

  1.2.840.10008.6.1.47

Table CID 3014. Coronary Artery Segments

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Equivalent BARI Code
--- | --- | --- | --- | --- | ---
SRT | T-43117 | 1st Diagonal Coronary Artery | 91750005 | C0524430 | 15
SRT | T-4312B | 1st Left Posterolateral Coronary Artery | 91757008 | C0524437 | 24
SRT | T-43128 | 1st Marginal Coronary Artery | 91754001 | C0524434 | 20
SRT | T-43213 | 1st Right posterolateral Coronary Artery | 91761002 | C0524441 | 6
SRT | T-43002 | 1st Septal Coronary Artery | 244251006 | C0447058 | 17
SRT | T-43118 | 2nd Diagonal Coronary Artery | 91751009 | C0524431 | 16
SRT | T-4312C | 2nd Left Posterolateral Coronary Artery | 91758003 | C0524438 | 25
SRT | T-43129 | 2nd Marginal Coronary Artery | 91755000 | C0524435 | 21
SRT | T-43214 | 2nd Right posterolateral Coronary Artery | 91762009 | C0524442 | 7
SRT | T-43119 | 3rd diagonal Coronary Artery | 91752002 | C0524432 | 29
SRT | T-4312D | 3rd Left Posterolateral Coronary Artery | 91759006 | C0524439 | 26
SRT | T-4312A | 3rd Marginal Coronary Artery | 91756004 | C0524436 | 22
SRT | T-43215 | 3rd Right posterolateral Coronary Artery | 91763004 | C0524443 | 8
SRT | T-43230 | Marginal Coronary Artery | 22765000 | C0226050 | 10
SRT | T-43124 | AV groove continuation of Circumflex Artery | 75902001 | C0226041 | 23
SRT | T-43122 | Distal Circumflex Coronary Artery | 6511003 | C0226039 | 19A
SRT | T-43112 | Distal Left Anterior Descending Coronary Artery | 36672000 | C0226034 | 14
SRT | T-43202 | Distal Right Coronary Artery | 41879009 | C0226044 | 3
BARI | 15A | 1st Diagonal Coronary Artery Laterals |  |  | 15A
BARI | 20A | 1st Marginal Coronary Artery Laterals |  |  | 20A
BARI | 16A | 2nd Diagonal Coronary Artery Laterals |  |  | 16A
BARI | 21A | 2nd Marginal Coronary Artery Laterals |  |  | 21A
BARI | 29A | 3rd Diagonal Coronary Artery Laterals |  |  | 29A
BARI | 22A | 3rd Marginal Coronary Artery Laterals |  |  | 22A
BARI | 28A | Ramus Laterals |  |  | 28A
SRT | T-43107 | Left Main Coronary Artery | 3227004 | C0226031 | 11
SRT | T-43105 | Left Main Coronary Artery Ostium | 76862008 | C0226030 | 11A
SRT | T-43126 | Left Posterior Descending Artery | 56322004 | C0278433 | 27
SRT | T-43127 | Mid Circumflex Coronary Artery | 91753007 | C0524433 | 19
SRT | T-43115 | Mid Left Anterior Descending Coronary Artery | 91748002 | C0524428 | 13
SRT | T-D6515 | Mid Right Coronary Artery | 450960006 | C3472627 | 2
SRT | T-43210 | Posterior Descending Right Coronary Artery | 53655008 | C0226047 | 4
BARI | 9 | Posterior descending septal perforators |  |  | 9
SRT | T-43121 | Proximal Circumflex Coronary Artery | 52433000 | C0226038 | 18
SRT | T-43111 | Proximal Left Anterior Descending Coronary Artery | 68787002 | C0226033 | 12
SRT | T-43201 | Proximal Right Coronary Artery | 91083009 | C0226043 | 1
SRT | T-43003 | Intermediate Artery (Ramus) | 244252004 | C0447059 | 28
SRT | T-43205 | Right Coronary Artery Ostium | 56789007 | C0226045 | 1A
SRT | T-43212 | Right posterior AV Coronary Artery | 12800002 | C0226048 | 5

### Note

In prior editions, this Context Group included BARI [1992] codes as the primary set. These have been replaced with equivalent SNOMED codes for the major artery segments (see PS3.16-2011).

## CID 3015 Coronary Arteries

Type:

  Extensible

Version:

  20130403

UID:

  1.2.840.10008.6.1.48

Table CID 3015. Coronary Arteries

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 3014 “Coronary Artery Segments”
SRT | T-43110 | Left Anterior Descending Coronary Artery | 59438005 | C0226032
SRT | T-43200 | Right Coronary Artery | 13647002 | C1261316
SRT | T-43120 | Circumflex Coronary Artery | 57396003 | C0226037
SRT | T-43125 | Left Posterolateral Circumflex Coronary Artery | 57823005 | C0278432
SRT | T-4312E | Left Posterior Descending Circumflex Coronary Artery | 91760001 | C0524440
SRT | T-41065 | Coronary Artery Graft | 264293000 | C0440761

## CID 3016 Major Coronary Arteries

Type:

  Extensible

Version:

  20110818

UID:

  1.2.840.10008.6.1.736

Table CID 3016. Major Coronary Arteries

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-43110 | Left Anterior Descending Coronary Artery | 59438005 | C0226032
SRT | T-43200 | Right Coronary Artery | 13647002 | C1261316
SRT | T-43120 | Circumflex Coronary Artery | 57396003 | C0226037
SRT | T-43107 | Left Main Coronary Artery | 3227004 | C0226031

## CID 3019 Cardiovascular Anatomic Location Modifiers

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.49

Table CID 3019. Cardiovascular Anatomic Location Modifiers

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-404CC | Anterior | 255549009 | C0205094
SRT | G-D873 | Arterial graft to cited segment | 128950003 | C1264698
SRT | G-A110 | Central | 26216008 | C0205099
SRT | G-A119 | Distal | 46053002 | C0205108
SRT | G-D870 | Graft to cited segment, body | 128947001 | C1264695
SRT | G-D872 | Graft to cited segment, distal anastomosis | 128948006 | C1264697
SRT | G-D871 | Graft to cited segment, proximal anastomosis | 128949003 | C1264696
SRT | R-4094A | Inferior | 261089000 | C0205104
SRT | G-A104 | Lateral | 49370004 | C0205093
SRT | G-A101 | Left | 7771000 | C0205091
SRT | R-4215C | Ostium | 264114003 | C0444567
SRT | R-404CE | Posterior | 255551008 | C0205095
SRT | G-A118 | Proximal | 40415009 | C0205107
SRT | G-A100 | Right | 24028007 | C0205090
SRT | R-42191 | Superior | 264217000 | C1282910
SRT | G-D874 | Venous graft to cited segment | 128951004 | C1264699
SRT | T-40003 | Entire Vessel | 361097006 | C1283786
DCM | 122101 | Aneurysm on cited vessel |  |
DCM | 122102 | Graft to cited segment, proximal section |  |
DCM | 122103 | Graft to cited segment, mid section |  |
DCM | 122104 | Graft to cited segment, distal section |  |

## CID 3082 Cardiology Units of Measurement (Retired)

This Context Group was a subset of CID 82 “Units of Measurement”, and is retired. See PS3.16-2011.

## CID 3083 Units of Radioactivity

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.737

Table CID 3083. Units of Radioactivity

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | Bq | becquerel
UCUM | MBq | megabecquerel
UCUM | mCi | millicurie

## CID 3090 Time Synchronization Channel Types

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.51

Table CID 3090. Time Synchronization Channel Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109001 | Digital timecode (NOS)
DCM | 109002 | ECG-based gating signal, processed
DCM | 109003 | IRIG-B timecode
DCM | 109004 | X-Ray Fluoroscopy On Signal
DCM | 109005 | X-Ray On Trigger

## CID 3101 Cardiac Procedural State Values

Type:

  Extensible

Version:

  20091021

UID:

  1.2.840.10008.6.1.52

Table CID 3101. Cardiac Procedural State Values

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01604 | Resting State | 128975004 | C0679218
SRT | F-05019 | Cardiac Stress State | 432655005 | C2317276
DCM | 109092 | Reinjection State |  |
DCM | 109093 | Redistribution State |  |
DCM | 109094 | Delayed Redistribution State |  |

## CID 3102 Rest-Stress

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.738

Table CID 3102. Rest-Stress

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01604 | Resting State | 128975004 | C0679218
DCM | 109091 | Cardiac Stress State |  |

## CID 3104 Cardiac Synchronization Technique

This Context Group corresponds to the Enumerated Values of Cardiac Synchronization Technique (0018,9037) (see PS3.3).

Type:

  Non-Extensible

Version:

  20111028

UID:

  1.2.840.10008.6.1.943

Table CID 3104. Cardiac Synchronization Technique

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109080 | Real time acquisition
DCM | 109081 | Prospective gating
DCM | 109082 | Retrospective gating
DCM | 109083 | Paced

## CID 3106 PET Cardiology Protocols

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.739

Table CID 3106. PET Cardiology Protocols

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122791 | PET Myocardial Perfusion, Rest only
DCM | 122792 | PET Myocardial Perfusion, Stress only
DCM | 122793 | PET Myocardial Perfusion, Rest and Stress
DCM | 122795 | PET Myocardial Viability, Rest only
DCM | 122796 | PET Myocardial Viability, Stress only
DCM | 122797 | PET Myocardial Viability, Rest and Stress

## CID 3107 PET Cardiology Radiopharmaceuticals

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.740

Table CID 3107. PET Cardiology Radiopharmaceuticals

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-B1031 | Fluorodeoxyglucose F^18^ | 35321007 | C0046056
SRT | C-107A1 | ^13^Nitrogen | 21576001 | C0302959
SRT | C-159A2 | ^82^Rubidium | 79197006 | C0303554

## CID 3108 NM/PET Procedures

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.741

Table CID 3108. NM/PET Procedures

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P5-D30F8 | Nuclear medicine cardiovascular study | 108294005 | C0581579
SRT | P5-0A006 | PET heart study | 241439007 | C0412498
SRT | P5-D6000 | Radioisotope study of endocrine system | 7562007 | C0203777
SRT | P5-D6500 | Radioisotope study of hematopoietic system | 41842006 | C0203797
SRT | P5-D5000 | Radioisotope study of gastrointestinal system | 53585008 | C0412377
SRT | P5-D0063 | Radionuclide study for localization of inflammatory disease | 252680004 | C0474787
SRT | P5-D1000 | Radioisotope study of musculoskeletal system | 68796002 | C0412452
SRT | P5-D90F8 | Nuclear medicine diagnostic procedure on nervous system | 108300008 | C0412330
SRT | P5-D0040 | Radionuclide localization of tumor | 45316007 | C0203651
SRT | P5-D2000 | Radioisotope study of respiratory system | 19086005 | C0203681
SRT | P5-D7000 | Radioisotope study of genitourinary system | 76927004 | C0203833
SRT | P5-0A001 | PET brain study | 241434002 | C0412493
SRT | P5-0A00D | PET breast study | 416323006 | C1562778
SRT | P5-0A00A | PET study for localization of tumor | 241443006 | C0473941

## CID 3110 Nuclear Cardiology Protocols

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.742

Table CID 3110. Nuclear Cardiology Protocols

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P5-D300B | Stress thallium procedure | 431511008 | C2316301
DCM | 122781 | Rest thallium/stress technetium procedure |  |
DCM | 122782 | Rest technetium/stress technetium 1 day procedure |  |
DCM | 122783 | Rest technetium/stress technetium 2 day procedure |  |
DCM | 122784 | Stress technetium/rest technetium 1 day procedure |  |
DCM | 122785 | NM Myocardial Viability procedure |  |

## CID 3111 Nuclear Cardiology Radiopharmaceuticals

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.743

Table CID 3111. Nuclear Cardiology Radiopharmaceuticals

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-B1130 | Thallium-201 | 353842007 | C0303322
SRT | C-B10A2 | Tc-99m sestamibi | 404706008 | C0361361
SRT | C-B10A4 | Tc-99m tetrofosmin | 404707004 | C0361363

## CID 3112 Attenuation Correction

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.744

Table CID 3112. Attenuation Correction

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122726 | Algorithmic Attenuation Correction
DCM | 122727 | NM Transmission Attenuation Correction
DCM | 122728 | CT-based Attenuation Correction
DCM | 122729 | No Attenuation Correction

## CID 3113 Types of Perfusion Defects

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.745

Table CID 3113. Types of Perfusion Defects

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-3014D | Reversible myocardial perfusion defect | 251055003 | C0428859
SRT | F-3014F | Fixed myocardial perfusion defect | 251057006 | C0428861
SRT | F-3014E | Partially Reversible myocardial perfusion defect | 251056002 | C0428860
DCM | 122748 | False Positive defect finding |  |

## CID 3114 Study Quality

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.746

Table CID 3114. Study Quality

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122740 | Excellent image quality
DCM | 122741 | Good image quality
DCM | 122742 | Poor image quality
DCM | 111235 | Unusable - Quality renders image unusable

## CID 3115 Stress Imaging Quality Issues

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.747

Table CID 3115. Stress Imaging Quality Issues

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111210 | Motion blur |  |
DCM | 122743 | Body habitus attenuation |  |
DCM | 122744 | Breast attenuation |  |
DCM | 122745 | Diaphragmatic attenuation |  |
SRT | F-04FD3 | Subdiaphragmatic uptake | 429382003 | C1997338

## CID 3116 NM Extracardiac Findings

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.748

Table CID 3116. NM Extracardiac Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-04FA0 | Normal extracardiac uptake | 428552000 | C1997656
SRT | F-04FB8 | Increased lung uptake | 428920008 | C1997679
SRT | F-04FE3 | Abnormal extracardiac uptake | 429576000 | C1998057

## CID 3117 Attenuation Correction Methods

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.749

Table CID 3117. Attenuation Correction Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122720 | OSEM algorithm
DCM | 122721 | Chang method

## CID 3118 Level of Risk

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.750

Table CID 3118. Level of Risk

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-4044 | Normal risk | 427986001 | C1998074
SRT | G-4041 | Low risk | 75976002 | C0332165
SRT | G-4045 | Low to moderate risk | 429551001 | C1998307
SRT | G-4042 | Moderate risk | 25594002 | C0332166
SRT | G-4046 | Moderate to high risk | 429557002 | C1998133
SRT | G-4043 | High risk | 15508007 | C0332167
SRT | G-A648 | Uncertain risk | 64957009 | C0087130

## CID 3119 LV Function

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.751

Table CID 3119. LV Function

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A460 | Normal | 17621005 | C0205307
SRT | F-300FA | Impaired left ventricular function | 275514001 | C0553982

## CID 3120 Perfusion Findings

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.752

Table CID 3120. Perfusion Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-30172 | Myocardial perfusion normal | 301121007 | C0577811
SRT | G-A466 | Equivocal | 42425007 | C0332241
SRT | R-42037 | Abnormal | 263654008 | C0205161

## CID 3121 Perfusion Morphology

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.753

Table CID 3121. Perfusion Morphology

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D3-1070D | Myocardial ischemia | 414795007 | C0151744
SRT | D3-15000 | Myocardial Infarction | 22298006 | C0027051
SRT | D3-10711 | Mixed Ischemia and Infarction | 428196007 | C1997401

## CID 3122 Ventricular Enlargement

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.754

Table CID 3122. Ventricular Enlargement

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-00343 | Normal size cardiac chamber | 373124004 | C1298811
SRT | R-0032A | Mildly enlarged cardiac chamber | 373126002 | C1298813
SRT | R-00331 | Moderately enlarged cardiac chamber | 373127006 | C1298814
SRT | R-00316 | Markedly enlarged cardiac chamber | 373128001 | C1298815

## CID 3200 Stress Test Procedure

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.755

Table CID 3200. Stress Test Procedure

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P0-006E4 | Exercise stress test | 165079009 | C0015260
SRT | P2-31107 | Pharmacologic stress test | 424064009 | C1827946
SRT | P2-31011 | Pharmacologic and exercise stress test | 428813002 | C1998158
SRT | P2-3110B | Paced stress test | 428685003 | C1997441

## CID 3201 Indications for Stress Test

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.756

Table CID 3201. Indications for Stress Test

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-37000 | Chest Pain | 29857009 | C0008031
SRT | R-413C5 | Pre-operative | 262068006 | C0445204
SRT | D3-13040 | Coronary Artery Disease | 53741008 | C0010054
SRT | D3-16000 | Heart failure | 84114007 | C0018801
SRT | F-03C97 | Heart disease risk factors | 171224000 | C0420044
SRT | F-201B3 | Dyspnea | 267036007 | C0013404
SRT | R-00357 | Post PTCA | 373108000 | C1269832
SRT | G-03A5 | History of CABG | 399261000 | C1275842
SRT | F-00103 | Abnormal exercise tolerance test | 165084003 | C0149612
SRT | F-38002 | Abnormal ECG | 102594003 | C0522055
SRT | D3-30000 | Arrhythmia | 44808001 | C0264886
SRT | D3-13012 | Angina pectoris | 194828000 | C0002962
SRT | D3-02000 | Hypertension | 38341003 | C0020538
SRT | F-37150 | Palpitations | 80313002 | C0030252
SRT | D3-31290 | Supraventricular tachycardia | 6456007 | C0039240
SRT | D3-00006 | Syncope | 271594007 | C0039070
SRT | G-03AA | History of Myocardial Infarction | 399211009 | C1275835
SRT | D3-33120 | Left bundle branch block | 63467002 | C0023211
SRT | D3-10800 | Valvular heart disease | 368009 | C0018824
SRT | P7-00044 | Occupational requirement | 429060002 | C1997084

## CID 3202 Chest Pain

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.757

Table CID 3202. Chest Pain

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D3-13020 | Stable Angina | 233819005 | C0340288
SRT | D3-12700 | Unstable Angina | 4557003 | C0002965
SRT | R-0038F | Atypical Angina | 371807002 | C0741026
SRT | F-37015 | Noncardiac Chest Pain | 274668005 | C0476281
SRT | F-A265A | Chest pain not present | 161971004 | C0423635
SRT | D3-13037 | Typical Angina | 429559004 | C1998435
DCM | 122799 | Anginal Equivalent |  |

## CID 3203 Exerciser Device

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.758

Table CID 3203. Exerciser Device

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-17230 | Bicycle ergometer | 739006 | C0180749
SRT | A-17222 | Treadmill | 1211003 | C0184069
SRT | A-1002A | Arm ergometer | 429560009 | C1996977

## CID 3204 Stress Agents

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.759

Table CID 3204. Stress Agents

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Trade Name (Informative)
--- | --- | --- | --- | --- | ---
SRT | C-81590 | Dipyridamole | 66859009 | C0012582 | Persantine
SRT | C-68030 | Dobutamine | 26523005 | C0012963 |
SRT | C-80349 | Adenosine | 108502004 | C0001443 |
SRT | C-67770 | Atropine | 73949004 | C0004259 |
SRT | C-80012 | Adenosine A2 receptor agonist | 432062000 | C1998062 | Regadenoson

## CID 3205 Indications for Pharmacological Stress Test

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.760

Table CID 3205. Indications for Pharmacological Stress Test

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D3-33120 | Left bundle branch block | 63467002 | C0023211
SRT | R-00728 | Patient has pacemaker | 441509002 | C2712998
SRT | DA-26000 | Paralytic syndrome | 29426003 | C0270788
SRT | F-A4580 | Ataxia or incoordination | 20262006 | C0004134
SRT | D3-8005B | Peripheral vascular disease | 400047006 | C0085096
SRT | D2-50000 | Pulmonary disease | 19829001 | C0024115
SRT | F-18002 | Gait problem | 22325002 | C0575081
SRT | F-A0846 | Transient limb paralysis | 274662006 | C0159034
SRT | F-01380 | Asthenia (debility) | 13791008 | C0004093
SRT | F-029F7 | Cachexia | 238108007 | C0006625
SRT | DD-13000 | Fracture of lower limb | 46866001 | C1542178
SRT | DD-33500 | Open wound of lower limb | 26947005 | C0178323
SRT | G-02BD | Lower limb amputation | 161622006 | C0455616
SRT | G-0202 | Request by Physician | 103321005 | C0686901
SRT | S-20570 | Dependence on enabling machine or device | 105501005 | C0524375
SRT | G-044D | Recent Myocardial infarction | 428752002 | C1998297
SRT | F-33019 | Cannot reach target heart rate | 429733000 | C1997932
DCM | 122764 | Patient weight exceeds equipment limit |  |

## CID 3206 Non-invasive Cardiac Imaging Procedures

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.761

Table CID 3206. Non-invasive Cardiac Imaging Procedures

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P5-D30F8 | Nuclear medicine cardiovascular study | 108294005 | C0581579
SRT | P5-D3304 | Cardiac blood pool imaging (nuclear) | 35621002 | C0203725
SRT | P5-0A006 | PET heart study | 241439007 | C0412498
SRT | P5-0A100 | SPECT | 105371005 | C0040399
SRT | P5-B3000 | Echocardiography | 40701008 | C0013516
SRT | P5-09011 | Cardiac MRI | 241620005 | C0412692

## CID 3207 Stress Test Procedure Phases

Type:

  Extensible

Version:

  20081031

UID:

  1.2.840.10008.6.1.637

Table CID 3207. Stress Test Procedure Phases

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01604 | Resting State | 128975004 | C0679218
SRT | F-05019 | Cardiac stress state | 432655005 | C2317276
SRT | F-05028 | Peak cardiac stress state | 434161005 | C2316487
SRT | F-05018 | Cardiac stress recovery state | 432554001 | C2316793
SRT | F-25040 | Hyperventilation | 68978004 | C0020578

## CID 3208 Summary Codes Exercise ECG

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.763

Table CID 3208. Summary Codes Exercise ECG

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-00101 | Exercise ECG normal | 165082004 | C0231162
SRT | F-00103 | Exercise ECG abnormal | 165084003 | C0149612
SRT | F-201B6 | Exercise ECG equivocal | 370367002 | C1299965
SRT | R-4135B | Not performed | 262008008 | C0445106

## CID 3209 Summary Codes Stress Imaging

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.764

Table CID 3209. Summary Codes Stress Imaging

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-04AB2 | Imaging result normal | 408573005 | C1319347
SRT | F-04AB3 | Imaging result abnormal | 408574004 | C1319348
SRT | F-04A13 | Imaging result equivocal | 408379005 | C1319511
SRT | R-4135B | Not performed | 262008008 | C0445106

## CID 3210 Speed of Response

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.765

Table CID 3210. Speed of Response

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A460 | normal | 17621005 | C0205307
SRT | R-40AA8 | accentuated | 428691001 | C1997416
SRT | R-40AA7 | blunted | 428247006 | C1997138

## CID 3211 BP Response

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.766

Table CID 3211. BP Response

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A460 | normal | 17621005 | C0205307
SRT | D3-04000 | Hypotensive | 45007003 | C0020649
SRT | D3-02000 | Hypertensive | 38341003 | C0020538
SRT | R-40AA7 | blunted | 428247006 | C1997138

## CID 3212 Treadmill Speed

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.767

Table CID 3212. Treadmill Speed

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | km/h | km/h
UCUM | [mi_i]/h | mph

## CID 3213 Stress Hemodynamic Findings

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.768

Table CID 3213. Stress Hemodynamic Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D3-0400A | Exertional hypotension | 429561008 | C1998376
SRT | D3-0200B | Exertional hypertension | 429198000 | C1997276
SRT | F-380B2 | Chronotropic incompetence | 427989008 | C1997984

## CID 3215 Perfusion Finding Method

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.769

Table CID 3215. Perfusion Finding Method

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-41D8B | ECG analysis | 258181008 | C0442977
SRT | P3-41910 | Image analysis | 24587005 | C0200765

## CID 3217 Comparison Finding

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.770

Table CID 3217. Comparison Finding

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122775 | Agreement with prior findings
DCM | 122776 | Disagreement with prior findings

## CID 3220 Stress Symptoms

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.771

Table CID 3220. Stress Symptoms

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-201B3 | Dyspnea | 267036007 | C0013404
SRT | F-18010 | Claudication | 16973004 | C1456822
SRT | D3-00006 | Syncope | 271594007 | C0039070
SRT | D0-30017 | Flushing | 238810007 | C0016382
SRT | F-04E95 | Nausea | 422587007 | C0027497
SRT | F-06017 | Dizziness | 404640003 | C0012833
SRT | F-01360 | Fatigue | 84229001 | C0015672
SRT | F-37000 | Chest pain | 29857009 | C0008031
SRT | F-37006 | Chest discomfort | 279084009 | C0235710
Include CID 3202 “Chest Pain” |  |

## CID 3221 Stress Test Termination Reasons

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.772

Table CID 3221. Stress Test Termination Reasons

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-37000 | Chest pain | 29857009 | C0008031
SRT | F-38002 | Abnormal ECG | 102594003 | C0522055
SRT | F-01360 | Fatigue | 84229001 | C0015672
SRT | F-201B3 | Dyspnea | 267036007 | C0013404
SRT | R-214DD | Patient Refused exercise test | 408551003 | C1319325
SRT | F-021E1 | Target Heart Rate Achieved | 258153002 | C0432605
SRT | D3-04001 | Hypotensive episode | 67763001 | C0520541
SRT | D3-02004 | Hypertensive episode | 62275004 | C0520539
SRT | D3-30000 | Arrhythmia | 44808001 | C0264886
SRT | F-18010 | Claudication | 16973004 | C1456822
SRT | R-4038D | End of Protocol | 255253007 | C0444496
SRT | D3-00006 | Syncope | 271594007 | C0039070

## CID 3227 QTc Measurements

This Context Group include both global and per lead corrected QT measurements specified in the ISO/IEEE 11073-10102 MDC nomenclature. Note that the MDC code for the per lead measurement is a base code for post-coordination with separately encoded lead identifiers. MDC also defines pre-coordinated codes that include both the measurement and the lead, which may be used in the same context as this Context Group; see the ISO/IEEE Standard.

While this Context Group includes distinct codes for the various QT correction algorithms, Templates using this Context Group may allow post-coordination using the QTc algorithm codes of CID 3678 “QT Correction Algorithms”.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

UID:

  1.2.840.10008.6.1.773

Table CID 3227. QTc Measurements

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:15876 | QTc interval global | MDC_ECG_TIME_PD_QTC
MDC | 2: 33792 | QTc interval per lead | MDC_ECG_TIME_PD_QTC_<lead>
MDC | 2:15880 | QTc global using Bazett formula | MDC_ECG_TIME_PD_QTC_BAZETT
MDC | 2:15880 | QTc global using Framingham formula | MDC_ECG_TIME_PD_QTC_FRAMINGHAM
MDC | 2:15892 | QTc global using Fredericia formula | MDC_ECG_TIME_PD_QTC_FREDERICA
MDC | 2:15892 | QTc global using Hodges formula | MDC_ECG_TIME_PD_QTC_HODGES
MDC | 2:34048 | QTc per lead using Bazett formula | MDC_ECG_TIME_PD_QTcB_<lead>
MDC | 2:34304 | QTc per lead using Fredericia formula | MDC_ECG_TIME_PD_QTcF_<lead>

## CID 3228 ECG Timing Measurements

This Context Group include both global and per lead ECG measurements specified in the ISO/IEEE 11073-10102 MDC nomenclature. Note that the MDC codes for "per lead" measurements are base codes for post-coordination with separately encoded lead identifiers. MDC also defines pre-coordinated codes that include both the measurement and the lead, which may be used in the same context as this Context Group; see the ISO/IEEE Standard.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.774

Table CID 3228. ECG Timing Measurements

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:15872 | PR interval global | MDC_ECG_TIME_PD_PR
MDC | 2:16160 | QT interval global | MDC_ECG_TIME_PD_QT
MDC | 2:16156 | QRS duration global | MDC_ECG_TIME_PD_QRS
MDC | 2:16184 | P duration global | MDC_ECG_TIME_PD_P
MDC | 2:16140 | PP interval global | MDC_ECG_TIME_PD_PP
MDC | 2:16168 | RR interval global | MDC_ECG_TIME_PD_RR
MDC | 2:7168 | PR interval per lead | MDC_ECG_TIME_PD_PR_<lead>
MDC | 2:8192 | QT interval per lead | MDC_ECG_TIME_PD_QT_<lead>
MDC | 2:7936 | QRS duration per lead | MDC_ECG_TIME_PD_QRS_<lead>
MDC | 2:6656 | P duration per lead | MDC_ECG_TIME_PD_P_<lead>
MDC | 2:32768 | PP interval per lead | MDC_ECG_TIME_PD_PP_<lead>
MDC | 2:33024 | RR interval per lead | MDC_ECG_TIME_PD_RR_<lead>

## CID 3229 ECG Axis Measurements

This Context Group comprises the ECG axis measurements of ISO/IEEE 11073-10102. The terms included in the table below may not constitute the complete list; see the ISO/IEEE Standard.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.775

Table CID 3229. ECG Axis Measurements

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:16132 | QRS axis | MDC_ECG_ANGLE_QRS_FRONT
MDC | 2:16128 | P Axis | MDC_ECG_ANGLE_P_FRONT
MDC | 2:16136 | T axis | MDC_ECG_ANGLE_T_FRONT

## CID 3230 ECG Findings

### Note

MDC Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.776

Table CID 3230. ECG Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | --- | --- | ---
SRT | F-000B7 | Normal | 164854000 | C0522054 | MDC_ECG_BEAT_NORMAL
SRT | D3-30A03 | Atrial premature contraction | 284470004 | C0033036 | MDC_ECG_BEAT_ATR_P_C
SRT | D3-31740 | Ventricular premature contraction | 17338001 | C0151636 | MDC_ECG_BEAT_V_P_C
SRT | D3-31520 | Atrial Fibrillation | 49436004 | C0004238 | MDC_ECG_RHY_ATR_FIB
SRT | D3-31290 | Supraventricular Tachycardia | 6456007 | C0039240 | MDC_ECG_RHY_SV_TACHY
SRT | D3-31710 | Non-sustained ventricular tachycardia | 66657009 | C0030591 | MDC_ECG_RHY_V_TACHY_PAROX
SRT | D3-31700 | Ventricular tachycardia | 25569003 | C0042514 | MDC_ECG_RHY_V_TACHY
SRT | D3-31720 | Ventricular fibrillation | 71908006 | C0042510 | MDC_ECG_RHY_V_FIB
SRT | D3-33000 | Intraventricular conduction disturbance | 4554005 | C0264909 | MDC_ECG_BEAT_BLK_IVCD
SRT | D3-33120 | Left bundle branch block | 63467002 | C0023211 | MDC_ECG_BEAT_LBB_BLK_COMP
SRT | D3-33110 | Right bundle branch block | 59118001 | C0085615 | MDC_ECG_BEAT_RBB_BLK_COMP
SRT | D3-33122 | Incomplete Left bundle branch block | 251120003 | C0281878 | MDC_ECG_BEAT_LBB_BLK_INCOMP
SRT | D3-33112 | Incomplete Right bundle branch block | 251124007 | C0262525 | MDC_ECG_BEAT_RBB_BLK_INCOMP
SRT | D3-33200 | Bifascicular Block | 74021003 | C0264914 | MDC_ECG_BEAT_BLK_BIFASC
SRT | D3-33140 | Left anterior fascicular block | 37760005 | C0264912 | MDC_ECG_BEAT_BLK_ANT_L_HEMI
SRT | D3-33150 | Left posterior fascicular block | 62026008 | C0264913 | MDC_ECG_BEAT_BLK_POS_L_HEMI
SRT | D3-30001 | First degree Atrioventricular block | 270492004 | C0085614 | MDC_ECG_RHY_AV_HEART_BLK_DEG_1
SRT | R-F81AE | Second degree Atrioventricular block | 195042002 | C0264906 | MDC_ECG_RHY_AV_HEART_BLK_DEG_2
SRT | D3-32102 | Third degree Atrioventricular block | 27885002 | C0151517 | MDC_ECG_RHY_AV_HEART_BLK_DEG_3
SRT | D3-31351 | Ventricular pre-excitation | 195060002 | C0559106 | MDC_ECG_BEAT_PREX
SRT | F-38278 | ST depression | 26141007 | C0520887 |
SRT | F-38277 | ST elevation | 76388001 | C0520886 |
SRT | F-380B3 | Early repolarization | 428417006 | C1997354 |
SRT | F-38794 | Nonspecific ST-T abnormality | 428750005 | C1997940 |
SRT | F-38793 | Secondary ST-T abnormality | 428549008 | C1998291 |

## CID 3231 ST Segment Findings

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.777

Table CID 3231. ST Segment Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-000C3 | ST Interval Normal | 164929001 | C0438164
DCM | 122750 | Non-diagnostic - low heart rate |  |
DCM | 122751 | Non-diagnostic - resting ST abnormalities |  |
DCM | 122752 | Non-diagnostic - ventricular pacing or LBBB |  |
SRT | G-A205 | Weakly positive | 260408008 | C0442730
SRT | G-A200 | Positive | 10828004 | C1446409
DCM | 122755 | Strongly positive |  |
DCM | 122756 | Strongly positive - ST elevation |  |

## CID 3232 ST Segment Location

Type:

  Extensible

Version:

  20091021

UID:

  1.2.840.10008.6.1.778

Table CID 3232. ST Segment Location

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-3260A | Left ventricle anterior segment | 284355001 | C0562222
SRT | T-3260C | Left ventricle inferior segment | 284357009 | C0562224
SRT | T-3260D | Left ventricle lateral segment | 284358004 | C0562225
SRT | T-3260B | Left ventricle septal segment | 284356000 | C0562223
SRT | T-32602 | Left ventricle apical segment | 128564006 | C0580781

## CID 3233 ST Segment Morphology

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.779

Table CID 3233. ST Segment Morphology

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 122757 | ST Depression - Horizontal |  |
DCM | 122758 | ST Depression - Upsloping |  |
DCM | 122759 | ST Depression - Downsloping |  |
SRT | F-38277 | ST Elevation | 76388001 | C0520886
SRT | F-38278 | ST Depression | 26141007 | C0520887

## CID 3234 Ectopic Beat Morphology

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.780

Table CID 3234. Ectopic Beat Morphology

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D3-31700 | Ventricular tachycardia | 25569003 | C0042514
SRT | F-33750 | Ventricular bigeminy | 11157007 | C0262662
SRT | D3-31744 | Multifocal PVCs | 10626002 | C0264903
SRT | D3-31742 | Unifocal PVCs | 27337007 | C0264902
SRT | D3-31704 | Ventricular tachycardia, polymorphic | 251159007 | C0344432

## CID 3235 Perfusion Comparison Findings

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.781

Table CID 3235. Perfusion Comparison Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-4075C | No change | 260388006 | C0442739
SRT | R-215D9 | New ischemia | 428927006 | C1997666
SRT | R-215DE | Less ischemia | 429232006 | C1998148
SRT | R-215D5 | Resolution of ischemia | 428824000 | C1996952
SRT | R-215E1 | More ischemia | 429477006 | C1997854
SRT | R-215E0 | New infarction | 429391004 | C1997076

## CID 3236 Tolerance Comparison Findings

Type:

  Extensible

Version:

  20080827

UID:

  1.2.840.10008.6.1.782

Table CID 3236. Tolerance Comparison Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-4075C | No change | 260388006 | C0442739
SRT | F-00454 | Decreased tolerance | 102460003 | C0151955
SRT | F-00453 | Increased tolerance | 102459008 | C0151956

## CID 3237 Wall Motion Comparison Findings

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.783

Table CID 3237. Wall Motion Comparison Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-4075C | No change | 260388006 | C0442739
SRT | R-215DC | New wall motion abnormality | 429058004 | C1997943
SRT | R-215D6 | Improvement of wall motion | 428825004 | C1997106

## CID 3238 Stress Scoring Scales

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.784

Table CID 3238. Stress Scoring Scales

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-E002 | Duke treadmill score | 304915008 | C0582804
DCM | 122770 | Ratio of achieved to predicted maximal oxygen consumption |  |
DCM | 122771 | Ratio of achieved to predicted functional capacity |  |
DCM | 122772 | Aerobic index |  |
DCM | 122773 | ST/HR Index |  |

## CID 3239 Perceived Exertion Scales

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.785

Table CID 3239. Perceived Exertion Scales

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122734 | Borg RPE Scale
DCM | 122735 | Borg CR10 Scale

## CID 3240 Electrophysiology Measurement Functions and Techniques

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.53

Table CID 3240. Electrophysiology Measurement Functions and Techniques

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109006 | Differential signal
DCM | 109007 | His bundle electrogram
DCM | 109008 | Monopole signal
DCM | 109009 | Pacing (electrical) stimulus, voltage
DCM | 109010 | Radio frequency ablation, power
DCM | 109011 | Voltage measurement by basket catheter
DCM | 109012 | Voltage measurement by mapping catheter
DCM | 109013 | Voltage measurement

## CID 3241 Hemodynamic Measurement Techniques

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.54

Table CID 3241. Hemodynamic Measurement Techniques

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | PA-50038 | Averaged hemodynamic measurement method | 128580001 | C1266842
SRT | PA-50035 | Composite hemodynamic measurement method | 128577002 | C1266839
SRT | PA-50034 | Computed hemodynamic measurement method | 128576006 | C1266838
SRT | PA-5003B | Conductance catheter method | 133910006 | C1297901
SRT | PA-5003C | Doppler catheter method | 133911005 | C1297902
SRT | PA-50031 | Dual catheter method | 128573003 | C1266836
SRT | PA-50039 | Fluid filled catheter method | 128581002 | C1266843
SRT | PA-5003D | Fiberoptic catheter method | 133912003 | C1297903
SRT | PA-5003E | Hall catheter method | 133913008 | C1297904
SRT | PA-50033 | Pullback method | 128575005 | C1276411
SRT | G-DB26 | Pulmonary capillary wedge method | 128448001 | C1264741
SRT | PA-50036 | Static catheter method | 128578007 | C1266840
SRT | PA-5003F | Thermistor catheter method | 133914002 | C1297905
SRT | PA-5003A | Tip manometer method | 128582009 | C1266844
SRT | PA-50037 | Wedge method | 128579004 | C1266841

## CID 3250 Catheterization Procedure Phase

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.55

Table CID 3250. Catheterization Procedure Phase

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-7299 | Cardiac catheterization bailout phase | 128961006 | C1292438
SRT | G-7293 | Cardiac catheterization baseline phase | 128955008 | C1292432
SRT | G-7294 | Cardiac catheterization image acquisition phase | 128956009 | C1292433
SRT | G-7295 | Cardiac catheterization intervention phase | 128957000 | C1292434
SRT | G-729B | Cardiac catheterization post contrast phase | 129083002 | C1292440
SRT | G-7298 | Cardiac catheterization post-intervention phase | 128960007 | C1292437
SRT | G-7296 | Cardiac catheterization pre-intervention phase | 128958005 | C1292435
SRT | R-002E4 | Cardiac catheterization test/challenge phase | 373105002 | C1300063
SRT | G-7297 | Cardiac catheterization therapy phase | 128959002 | C1292436
SRT | P1-3160A | Catheterization of both left and right heart with graft | 128952006 | C1293383
SRT | P1-3160B | Catheterization of both left and right heart without graft | 128953001 | C1293384
SRT | P1-31604 | Catheterization of left heart | 67629009 | C0189897
SRT | P1-31602 | Catheterization of right heart | 40403005 | C0189896
SRT | P1-31612 | Transseptal catheterization | 67338003 | C0189901
SRT | P2-71317 | Drug Infusion Challenge | 133882006 | C1297891
SRT | P2-71310 | Exercise challenge | 128967005 | C1293901
SRT | F-01604 | Resting State | 128975004 | C0679218

## CID 3254 Electrophysiology Procedure Phase

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.56

Table CID 3254. Electrophysiology Procedure Phase

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-729D | Atrial Effective Refractory Period, evaluation of | 129087001 | C0428938
SRT | G-7304 | Carotid Sinus Massage procedure phase | 129090007 | C1292445
SRT | G-7406 | Electrophysiology Mapping phase | 129092004 | C1292447
SRT | G-729A | Electrophysiology procedure baseline phase | 129082007 | C1292439
SRT | G-7408 | Post-ablation phase | 129093009 | C1292448
SRT | G-7305 | Post-defibrillation procedure phase | 129091006 | C1292446
SRT | G-729F | Radiofrequency Ablation procedure phase | 129089003 | C1292442
SRT | G-729C | Sinus Node Recovery Time, evaluation of | 129086005 | C1292441
SRT | G-729E | Ventricular Effective Refractory Period, evaluation of | 129088006 | C0428940

## CID 3261 Stress Protocols

Type:

  Extensible

Version:

  20081027

UID:

  1.2.840.10008.6.1.57

Table CID 3261. Stress Protocols

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P2-7131C | Balke protocol | 129097005 | C0442712
SRT | P2-7131A | Bruce protocol | 129095002 | C0442713
SRT | P2-7131D | Ellestad protocol | 129098000 | C1276407
SRT | P2-7131B | Modified Bruce protocol | 129096001 | C0442714
SRT | P2-713A1 | Modified Naughton protocol | 129102008 | C1293907
SRT | P2-713A0 | Naughton protocol | 129101001 | C0442715
SRT | P2-7131F | Pepper protocol | 129100000 | C1276409
SRT | P2-7131E | Ramp protocol | 129099008 | C1276408
SRT | P2-31010 | Exercise stress ECG test | 46136006 | C1304755
SRT | P2-31102 | Stress test using Bicycle Ergometer | 26046004 | C0430459
SRT | P2-31107 | Pharmacologic Stress protocol | 424064009 | C1827946
SRT | P2-3110A | Dipyridamole Stress protocol | 422685009 | C1827789
SRT | P2-31109 | Adenosine Stress protocol | 424444005 | C1827363
SRT | P2-31108 | Dobutamine Stress protocol | 424225000 | C1828348
SRT | P2-31011 | Pharmacologic and exercise stress test | 428813002 | C1998158
SRT | P2-3110B | Stress test using cardiac pacing | 428685003 | C1997441

## CID 3262 ECG Patient State Values

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.58

Table CID 3262. ECG Patient State Values

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01602 | Baseline state | 128974000 | C1290922
SRT | F-01606 | Exercise state | 128976003 | C1290923
SRT | F-01608 | Post-exercise state | 128977007 | C1290924
SRT | F-01604 | Resting state | 128975004 | C0679218
SRT | F-10340 | Supine body position | 40199007 | C0038846

## CID 3263 Electrode Placement Values

This Context Group comprises the ECG lead placement system identifiers of ISO/IEEE 11073-10102. The terms included in the table below may not constitute the complete list; see the ISO/IEEE Standard.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.59

Table CID 3263. Electrode Placement Values

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 10:11264 | Unspecified 12-lead system | MDC_ECG_LDSYS_12LD_UNSPECIFIED
MDC | 10:11265 | Standard 12-lead positions, electrodes placed individually | MDC_ECG_LDSYS_12LD_STD
MDC | 10:11266 | Mason-Likar lead positions, electrodes placed individually | MDC_ECG_LDSYS_12LD_MASON_LIKAR
MDC | 10:11267 | Mason-Likar lead positions, V1-V6 in electrode pad | MDC_ECG_LDSYS_12LD_VPAD
MDC | 10:11268 | 12-lead electrode pad | MDC_ECG_LDSYS_12LD_PAD
MDC | 10:11269 | 12-lead derived from Frank XYZ leads | MDC_ECG_LDSYS_12LD_FROM_FRANK
MDC | 10:11270 | 12-lead derived from non-standard leads | MDC_ECG_LDSYS_12LD_NON_STANDARD
MDC | 10:11271 | 12-lead for bicycle exercise testing, limb leads on back of patient | MDC_ECG_LDSYS_12LD_BICYCLE
MDC | 10:11272 | Standard 12-lead positions one intercostal space higher | MDC_ECG_LDSYS_12LD_RAISED_INTERCOSTAL
MDC | 10:11273 | Unspecified XYZ lead system | MDC_ECG_LDSYS_XYZ_UNSPECIFIED
MDC | 10:11274 | Frank XYZ lead system | MDC_ECG_LDSYS_XYZ_FRANK
MDC | 10:11275 | McFee-Parungao XYZ lead system | MDC_ECG_LDSYS_XYZ_MCFEE_PARUNAGO
MDC | 10:11276 | Cube XYZ lead system | MDC_ECG_LDSYS_XYZ_CUBE
MDC | 10:11277 | Bipolar uncorrected XYZ lead system | MDC_ECG_LDSYS_XYZ_BIPOLAR
MDC | 10:11278 | Pseudo-orthogonal XYZ lead system | MDC_ECG_LDSYS_XYZ_PSEUDO_ORTH
MDC | 10:11279 | XYZ leads derived from standard 12-lead | MDC_ECG_LDSYS_XYZ_FROM_12LD
MDC | 10:11280 | NEHB lead system | MDC_ECG_LDSYS_3LD_NEHB
MDC | 10:11281 | 3-lead system, CC5-CM5-ML | MDC_ECG_LDSYS_3LD_CC5_CM5_ML
MDC | 10:11282 | 3-lead system, CC5-CM5-CH5 | MDC_ECG_LDSYS_3LD_CM5_CC5_CH5
MDC | 10:11283 | 12-lead from Frank leads XYZ leads by Dower transformation | MDC_ECG_LDSYS_12LD_FROM_DOWER
MDC | 10:11284 | 12-lead from EASI leads (ES, AS, AI) by Dower/EASI transformation | MDC_ECG_LDSYS_12LD_FROM_EASI
MDC | 10:11285 | 12-lead from Limb Leads (I, II) and one or more V leads | MDC_ECG_LDSYS_12LD_FROM_LIMB
MDC | 10:11286 | Standard 12-lead and XYZ | MDC_ECG_LDSYS_12LD_STD_AND_XYZ
MDC | 10:11287 | Standard 12-lead and NEHB | MDC_ECG_LDSYS_12LD_STD_AND_NEHB
MDC | 10:11288 | Standard 12-lead and CC5-CM5-ML | MDC_ECG_LDSYS_12LD_STD_AND_CC5_CM5_ML
MDC | 10:11289 | Standard 12-lead and CM5-CC5-CH5 | MDC_ECG_LDSYS_12LD_STD_AND_CM5_CC5_CH5
MDC | 10:11290 | Standard 12-lead with extra leads to the right and/or left sides | MDC_ECG_LDSYS_12LD_STD_EXTD
MDC | 10:11291 | Standard 12-lead extended to the right by V5R, V4R, V3R | MDC_ECG_LDSYS_12LD_STD_EXTD_RIGHT
MDC | 10:11292 | Standard 12-lead extended to the left by V7, V8, V9 | MDC_ECG_LDSYS_12LD_STD_EXTD_LEFT

### Note

A prior version of this context group used codes from the SCP-ECG vocabulary.

## CID 3264 XYZ Electrode Placement Values (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3271 Hemodynamic Physiological Challenges

Type:

  Extensible

Version:

  20100625

UID:

  1.2.840.10008.6.1.61

Table CID 3271. Hemodynamic Physiological Challenges

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P2-71317 | Drug infusion | 133882006 | C1297891
SRT | P2-71310 | Exercise challenge | 128967005 | C1293901
SRT | P2-71306 | Handgrip | 128965002 | C1293900
SRT | P2-71302 | Head up | 128963009 | C1293898
SRT | P2-71314 | Held inspiration | 128969008 | C1293904
SRT | P2-71316 | Held ventilation | 128970009 | C1293905
SRT | P2-71304 | Leg up | 128964003 | C1293899
SRT | P2-71308 | Negative lower body pressure | 128966001 | C0024047
SRT | P2-35000 | Pacing | 18590009 | C0199640
SRT | P2-71318 | Post volume challenge | 128971008 | C1293906
SRT | P2-71312 | Vagal stimulation | 128968000 | C1293903
SRT | R-40928 | Valsalva maneuver | 261039008 | C0042293

## CID 3335 ECG Annotations

This Context Group comprises the nomenclature of ISO/IEEE 11073-10102, limited to the hierarchies under Reference IDs MDC_ECG_WAVEC, MDC_ECG_WAVEP, MDC_ECG_BEAT, and MDC_ECG_NOISE.

The base terms from those hierarchies are included in the table below for reference. Note that these base terms are pre-coordinated with a variety of concept discriminators, and the code values for those pre-coordinated terms are arithmetically derived from the code values of the base terms. For the complete current list of terms and discriminator values, see the ISO/IEEE Standard. All pre-coordinated terms (annotation plus discriminators) within the identified hierarchies are part of this Context Group.

### Note

1. This Context Group is used in the Concept Name Code Sequence of the Waveform Annotation Sequence (0040,B020). See PS3.3.

2. A prior version of this context group used codes from the SCP-ECG coding system.

3. Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20130613

UID:

  1.2.840.10008.6.1.62

Table CID 3335. ECG Annotations

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 10:256 | P wave | MDC_ECG_WAVC_PWAVE
MDC | 10:320 | P' wave (second deflection in P wave) | MDC_ECG_WAVC_PPWAVE
MDC | 10:384 | P'' wave (third deflection in P wave) | MDC_ECG_WAVC_PPPWAVE
MDC | 10:448 | Q wave | MDC_ECG_WAVC_QWAVE
MDC | 10:512 | QS wave | MDC_ECG_WAVC_QSWAVE
MDC | 10:576 | R wave | MDC_ECG_WAVC_RWAVE
MDC | 10:640 | R' wave (second deflection in R Wave) | MDC_ECG_WAVC_RRWAVE
MDC | 10:704 | R'' wave (third deflection in R Wave) | MDC_ECG_WAVC_RRRWAVE
MDC | 10:768 | Notch | MDC_ECG_WAVC_NOTCH
MDC | 10:832 | S wave | MDC_ECG_WAVC_SWAVE
MDC | 10:896 | S' wave (second deflection in S Wave) | MDC_ECG_WAVC_SSWAVE
MDC | 10:960 | S'' wave (third deflection in S Wave) | MDC_ECG_WAVC_SSSWAVE
MDC | 10:1024 | T wave | MDC_ECG_WAVC_TWAVE
MDC | 10:1088 | T' wave (second deflection in T Wave) | MDC_ECG_WAVC_TTWAVE
MDC | 10:1152 | U wave | MDC_ECG_WAVC_UWAVE
MDC | 10:1216 | Delta wave | MDC_ECG_WAVC_DELTA
MDC | 10:1280 | Isoelectric region from global QRS onset to specific lead onset | MDC_ECG_WAVC_IWAVE
MDC | 10:1344 | Isoelectric region from specific lead QRS Offset to global offset | MDC_ECG_WAVC_KWAVE
MDC | 10:1408 | Osborne wave | MDC_ECG_WAVC_JWAVE
MDC | 10:1472 | Entire Beat (Pon to Toff, excluding U) | MDC_ECG_WAVC_PQRSTWAVE
MDC | 10:1536 | Entire Beat (Qon to Toff, excluding P and U) | MDC_ECG_WAVC_QRSTWAVE
MDC | 10:1600 | Entire QRS (excluding P, T and U) | MDC_ECG_WAVC_QRSWAVE
MDC | 10:1664 | TU fused wave | MDC_ECG_WAVC_TUWAVE
MDC | 10:1728 | Ventricular flutter wave | MDC_ECG_WAVC_VFLWAVE
MDC | 10:1792 | Atrial flutter wave | MDC_ECG_WAVC_AFLWAVE
MDC | 10:1856 | Isoelectric point or segment | MDC_ECG_WAVC_ISO
MDC | 10:1920 | PR Segment | MDC_ECG_WAVC_PRSEG
MDC | 10:1984 | ST Segment | MDC_ECG_WAVC_STSEG
MDC | 10:2048 | J-point | MDC_ECG_WAVC_STJ
MDC | 10:2112 | ST measurement point | MDC_ECG_WAVC_STM
MDC | 10:2176 | Isolated QRS-like artifact | MDC_ECG_WAVC_ARFCT
MDC | 10:2240 | Calibration pulse (individual pulse) | MDC_ECG_WAVC_CALP
MDC | 10:2304 | ST change | MDC_ECG_WAVC_STCH
MDC | 10:2368 | T-wave change | MDC_ECG_WAVC_TCH
MDC | 10:2432 | Ventricular Activation Time | MDC_ECG_WAVC_VAT
MDC | 10:4096 | Antibradycardia pace spike | MDC_ECG_WAVP_PACE
MDC | 10:4352 | atrium Antibradycardia pace spike | MDC_ECG_WAVP_PACE_ATR
MDC | 10:4608 | right atrium Antibradycardia pace spike | MDC_ECG_WAVP_PACE_ATR_R
MDC | 10:4864 | left atrium Antibradycardia pace spike | MDC_ECG_WAVP_PACE_ATR_L
MDC | 10:5120 | ventricular Antibradycardia pace spike | MDC_ECG_WAVP_PACE_V
MDC | 10:5376 | right ventricle Antibradycardia pace spike | MDC_ECG_WAVP_PACE_V_R
MDC | 10:5632 | left ventricle Antibradycardia pace spike | MDC_ECG_WAVP_PACE_V_L
MDC | 10:5888 | transthoracic Antibradycardia pace spike | MDC_ECG_WAVP_PACE_EXT
MDC | 10:6144 | Antitachycardia pace spike | MDC_ECG_WAVP_ATPACE
MDC | 10:6400 | atrium Antitachycardia pace spike | MDC_ECG_WAVP_ATPACE_ATR
MDC | 10:6656 | ventricle Antitachycardia pace spike | MDC_ECG_WAVP_ATPACE_V
MDC | 10:6912 | transthoracic Antitachycardia pace spike | MDC_ECG_WAVP_ATPACE_EXT
MDC | 10:7168 | Cardioversion spike | MDC_ECG_WAVP_CDVS
MDC | 10:7424 | atrium Cardioversion spike | MDC_ECG_WAVP_CDVS_ATR
MDC | 10:7680 | ventricle Cardioversion spike | MDC_ECG_WAVP_CDVS_V
MDC | 10:7936 | transthoracic Cardioversion spike | MDC_ECG_WAVP_CDVS_EXT
MDC | 10:8192 | Defibrillation spike | MDC_ECG_WAVP_DEFIB
MDC | 10:8448 | atrium Defibrillation spike | MDC_ECG_WAVP_DEFIB_ATR
MDC | 10:8704 | ventricle Defibrillation spike | MDC_ECG_WAVP_DEFIB_V
MDC | 10:8960 | transthoracic Defibrillation spike | MDC_ECG_WAVP_DEFIB_EXT
MDC | 10:8192 | Heart beat | MDC_ECG_BEAT
MDC | 10:8208 | Normal beat (sinus beat, normal conduction) | MDC_ECG_BEAT_NORMAL
MDC | 10:8224 | Abnormal beat | MDC_ECG_BEAT_ABNORMAL
MDC | 10:8240 | Dominant beat | MDC_ECG_BEAT_DOMINANT
MDC | 10:8256 | Supraventricular premature contraction | MDC_ECG_BEAT_SV_P_C
MDC | 10:8272 | Atrial premature contraction (beat) | MDC_ECG_BEAT_ATR_P_C
MDC | 10:8288 | Junctional (nodal) premature contraction | MDC_ECG_BEAT_JUNC_P_C
MDC | 10:8304 | Aberrated atrial premature beat (Ashman beat) | MDC_ECG_BEAT_ATR_P_C_ABERR
MDC | 10:8320 | Non-conducted p-wave (blocked) | MDC_ECG_BEAT_ATR_PWAVE_BLK
MDC | 10:8336 | Ventricular premature contraction beat | MDC_ECG_BEAT_V_P_C
MDC | 10:8352 | Fusion of ventricular and normal beat | MDC_ECG_BEAT_V_P_C_FUSION
MDC | 10:8368 | R-on-T premature ventricular beat | MDC_ECG_BEAT_V_P_C_RonT
MDC | 10:8384 | Supraventricular escape beat | MDC_ECG_BEAT_SV_ESC
MDC | 10:8400 | Atrial escape beat | MDC_ECG_BEAT_ATR_ESC
MDC | 10:8416 | Junctional (nodal) escape beat | MDC_ECG_BEAT_JUNC_ESC
MDC | 10:8432 | Ventricular escape beat | MDC_ECG_BEAT_V_ESC
MDC | 10:8448 | Bundle branch block beat | MDC_ECG_BEAT_BB_BLK
MDC | 10:8464 | Left bundle branch block beat | MDC_ECG_BEAT_LBB_BLK_COMP
MDC | 10:8480 | Incomplete left bundle branch block beat | MDC_ECG_BEAT_LBB_BLK_INCOMP
MDC | 10:8496 | Right bundle branch block beat | MDC_ECG_BEAT_RBB_BLK_COMP
MDC | 10:8512 | Incomplete right bundle branch block beat | MDC_ECG_BEAT_RBB_BLK_INCOMP
MDC | 10:8528 | Left anterior fascicular block beat | MDC_ECG_BEAT_BLK_ANT_L_HEMI
MDC | 10:8544 | Left posterior fascicular block beat | MDC_ECG_BEAT_BLK_POS_L_HEMI
MDC | 10:8560 | bifascicular block beat | MDC_ECG_BEAT_BLK_BIFASC
MDC | 10:8576 | trifascicular block beat | MDC_ECG_BEAT_BLK_TRIFASC
MDC | 10:8592 | bilateral bundle-branch block beat | MDC_ECG_BEAT_BLK_BILAT
MDC | 10:8608 | intraventricular conduction disturbance | MDC_ECG_BEAT_BLK_IVCD
MDC | 10:8624 | pre-excitation | MDC_ECG_BEAT_PREX
MDC | 10:8640 | Wolf-Parkinson-White syndrome | MDC_ECG_BEAT_WPW_UNK
MDC | 10:8656 | Wolf-Parkinson type A | MDC_ECG_BEAT_WPW_A
MDC | 10:8672 | Wolf-Parkinson type B | MDC_ECG_BEAT_WPW_B
MDC | 10:8688 | Lown-Ganong-Levine syndrome | MDC_ECG_BEAT_LGL
MDC | 10:8704 | Paced beat | MDC_ECG_BEAT_PACED
MDC | 10:8720 | Pacemaker Fusion beat | MDC_ECG_BEAT_PACED_FUS
MDC | 10:8736 | Unclassifiable beat | MDC_ECG_BEAT_UNKNOWN
MDC | 10:8752 | Pacemaker Learning beat | MDC_ECG_BEAT_LEARN
MDC | 10:11200 | No Noise | MDC_ECG_NOISE_CLEAN
MDC | 10:11216 | Moderate Noise, beats can be detected but cannot be classified | MDC_ECG_NOISE_MODERATE
MDC | 10:11232 | Severe Noise, beats cannot be detected or classified | MDC_ECG_NOISE_SEVERE
MDC | 10:11248 | No ECG signal is available | MDC_ECG_NOISE_NOSIGNAL

### Note

In a prior version of this table, the code 10:608 was specified for the concept R wave.

## CID 3337 Hemodynamic Annotations

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.63

Table CID 3337. Hemodynamic Annotations

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 109014 | 35% of thermal/dye dilution CO |  |
DCM | 109015 | 70% of thermal/dye dilution CO |  |
DCM | 109016 | A wave peak pressure |  |
DCM | 109017 | A wave pressure, average |  |
DCM | 109018 | Beat detected (accepted) |  |
DCM | 109019 | Beat detected (rejected) |  |
SRT | F-00E22 | Average diastolic blood pressure | 314453003 | C1282163
SRT | F-00E1F | Minimum diastolic blood pressure | 314451001 | C1282161
SRT | R-FAB5C | End diastole | 416190007 | C1562146
DCM | 109023 | End of expiration |  |
DCM | 109024 | End of inspiration |  |
DCM | 109070 | End of systole |  |
DCM | 109071 | Indicator mean transit time |  |
DCM | 109025 | Max dp/dt |  |
DCM | 109026 | Max neg dp/dt |  |
SRT | F-31150 | Mean blood pressure | 6797001 | C0428886
DCM | 109028 | Peak of thermal cardiac output bolus |  |
DCM | 109029 | Start of expiration |  |
DCM | 109030 | Start of inspiration |  |
DCM | 109031 | Start of thermal CO |  |
SRT | F-00E14 | Average systolic blood pressure | 314440001 | C1282151
SRT | F-00E11 | Maximum systolic blood pressure | 314439003 | C1282150
DCM | 109072 | Tau |  |
DCM | 109073 | V max myocardial |  |
DCM | 109034 | V wave peak pressure |  |
DCM | 109035 | V wave pressure, average |  |
DCM | 109036 | Valve close |  |
DCM | 109037 | Valve open |  |

## CID 3339 Electrophysiology Annotations

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.64

Table CID 3339. Electrophysiology Annotations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109038 | Ablation off
DCM | 109039 | Ablation on
DCM | 109040 | HIS bundle wave
DCM | 109041 | P wave
DCM | 109042 | Q wave
DCM | 109043 | R wave
DCM | 109044 | S wave
DCM | 109045 | Start of atrial contraction
DCM | 109046 | Start of atrial contraction (subsequent)
DCM | 109047 | Stimulation at rate 1 interval
DCM | 109048 | Stimulation at rate 2 interval
DCM | 109049 | Stimulation at rate 3 interval
DCM | 109050 | Stimulation at rate 4 interval
DCM | 109051 | T wave
DCM | 109052 | V wave
DCM | 109053 | V wave of next beat

## CID 3400 Procedure Log Titles

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.65

Table CID 3400. Procedure Log Titles

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121120 | Cath Lab Procedure Log

## CID 3401 Types of Log Notes

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.66

Table CID 3401. Types of Log Notes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121171 | Tech Note
DCM | 121172 | Nursing Note
DCM | 121173 | Physician Note
DCM | 121174 | Procedure Note
DCM | 121123 | Patient Status or Event

## CID 3402 Patient Status and Events

Type:

  Extensible

Version:

  20090615

UID:

  1.2.840.10008.6.1.67

Table CID 3402. Patient Status and Events

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122001 | Patient called to procedure room
DCM | 122002 | Patient admitted to procedure room
DCM | 122003 | Patient given pre-procedure instruction
DCM | 122004 | Patient informed consent given
DCM | 122005 | Patient advance directive given
DCM | 122006 | Nil Per Os (NPO) status confirmed
DCM | 122007 | Patient assisted to table
DCM | 122008 | Patient prepped and draped
DCM | 122009 | Patient connected to continuous monitoring
DCM | 122010 | Patient transferred to holding area
DCM | 122011 | Patient transferred to surgery
DCM | 122012 | Patient transferred to CCU
DCM | 122020 | Patient disoriented
DCM | 122021 | Patient reports nausea
DCM | 122022 | Patient reports discomfort
DCM | 122023 | Patient reports chest pain
DCM | 122024 | Patient reports no pain
DCM | 122025 | Patient alert
DCM | 122026 | Patient restless
DCM | 122027 | Patient sedated
DCM | 122028 | Patient asleep
DCM | 122029 | Patient unresponsive
DCM | 122030 | Patient has respiratory difficulty
DCM | 122031 | Patient coughed
DCM | 122032 | Patient disconnected from continuous monitoring
DCM | 122033 | Hemostasis achieved
DCM | 122034 | Hemostasis not achieved - oozing
DCM | 122035 | Hemostasis not achieved - actively bleeding
DCM | 122036 | Patient given post-procedure instruction
DCM | 122038 | Patient pronounced dead
DCM | 122039 | Patient transferred to morgue
DCM | 122037 | Patient discharged from department

## CID 3403 Percutaneous Entry

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.68

Table CID 3403. Percutaneous Entry

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3746 “Percutaneous Entry Site”
Include CID 3747 “Percutaneous Closure”

## CID 3404 Staff Actions

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.69

Table CID 3404. Staff Actions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122041 | Personnel Arrived
DCM | 122042 | Personnel Departed
DCM | 122043 | Page Sent To
DCM | 122044 | Consultation With
DCM | 122045 | Office called

## CID 3405 Procedure Action Values

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.70

Table CID 3405. Procedure Action Values

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P1-30350 | Atherectomy | 6832004 | C0162513
SRT | P1-30351 | Atherectomy by rotary cutter | 65659003 | C0162655
SRT | P1-30352 | Atherectomy by laser | 76611008 | C0521229
SRT | P1-30530 | Selective embolization of artery | 57238002 | C0189632
SRT | P5-31500 | Percutaneous transluminal balloon angioplasty | 68457009 | C0411287
SRT | P5-39010 | Transcatheter therapy for embolization | 16736007 | C0203006
SRT | P5-39050 | Percutaneous retrieval of intravascular foreign body | 37630009 | C0203013
SRT | P1-05550 | Stent placement | 103716009 | C0522776
SRT | P5-39015 | Transcatheter deployment of detachable balloon | 105372003 | C0524313
SRT | P5-39191 | Percutaneous insertion of intravascular filter | 105373008 | C0524314
Include CID 3250 “Catheterization Procedure Phase” |  |
Include CID 3406 “Non-coronary Transcatheter Interventions” |  |
Include CID 3428 “Imaging Procedures” |  |

## CID 3406 Non-coronary Transcatheter Interventions

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.71

Table CID 3406. Non-coronary Transcatheter Interventions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122053 | Valvular Intervention
DCM | 122054 | Aortic Intervention
DCM | 122055 | Septal Defect Intervention
DCM | 122056 | Vascular Intervention
DCM | 122057 | Myocardial biopsy

## CID 3407 Purpose of Reference to Object

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.72

Table CID 3407. Purpose of Reference to Object

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122072 | Pre-procedure log
DCM | 122073 | Analysis or measurements for current procedure
DCM | 122075 | Prior report for current patient

## CID 3408 Actions With Consumables

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.73

Table CID 3408. Actions With Consumables

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122076 | Consumable taken from inventory
DCM | 122077 | Consumable returned to inventory
DCM | 122078 | Remaining consumable disposed
DCM | 122079 | Consumable unusable

## CID 3409 Administration of Drugs/Contrast

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.74

Table CID 3409. Administration of Drugs/Contrast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122081 | Drug start
DCM | 122082 | Drug end
DCM | 122083 | Drug administered
DCM | 122084 | Contrast start
DCM | 122085 | Contrast end
DCM | 122086 | Contrast administered
DCM | 122087 | Infusate start
DCM | 122088 | Infusate end

## CID 3410 Numeric Parameters of Drugs/Contrast

Type:

  Extensible

Version:

  20070124

UID:

  1.2.840.10008.6.1.75

Table CID 3410. Numeric Parameters of Drugs/Contrast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122091 | Volume administered
DCM | 122092 | Undiluted dose administered
DCM | 122093 | Concentration
DCM | 122094 | Rate of administration
DCM | 122095 | Duration of administration
DCM | 122096 | Volume unadministered or discarded
DCM | 121382 | Quantity administered
DCM | 121383 | Mass administered

## CID 3411 Intracoronary Devices

Type:

  Extensible

Version:

  20110609

UID:

  1.2.840.10008.6.1.76

Table CID 3411. Intracoronary Devices

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR Equivalent | Trade Name (Informative)
--- | --- | --- | --- | --- | --- | ---
SRT | A-26912 | Percutaneous Transluminal Angioplasty Balloon | 102319006 | C0522648 | 113-1 |
SRT | R-002F0 | Cutting Balloon Angioplasty (CBA) Device | 371794009 | C1269809 | 113-2 |
SRT | A-25500 | Stent | 65818007 | C0038257 | 113-3 |
SRT | R-002FD | Directional Coronary Atherectomy (DCA) Device | 371796006 | C1269811 | 113-4 |
SRT | A-25610 | Rotational Atherectomy Device | 102313007 | C0522643 | 113-5 | Rotablator™
SRT | R-0036F | Saline Thrombectomy | 371797002 | C1299427 | 113-6 | AngioJet™
SRT | A-26920 | Transluminal Extraction Catheter (TEC) | 21870002 | C0521199 | 113-7 |
SRT | A-81080 | Laser | 38586004 | C0458142 | 113-8 |
SRT | R-00312 | Intravascular Ultrasound (IVUS) Device | 371795005 | C1269810 | 113-9 |
SRT | R-00310 | Intracoronary Doppler guide wire | 371788001 | C1269808 | 113-10 | Flowire™
SRT | R-00311 | Intracoronary pressure guide wire | 371789009 | C1299422 | 113-11 |
SRT | A-040ED | Brachytherapy Device | 228748004 | C0454156 |  |
SRT | R-00361 | Radiofrequency Ablation Device | 371791001 | C1299424 |  |
SRT | A-00D87 | Intravascular Optical Coherence Tomography Device | 445282004 | C2919367 |  |
SRT | A-00927 | Guide Wire | 272224001 | C0181089 |  |
SRT | A-26802 | Guiding Catheter | 102317008 | C0221799 |  |

## CID 3412 Intervention Actions and Status

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.77

Table CID 3412. Intervention Actions and Status

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122301 | Guidewire crossing lesion unsuccessful
DCM | 122302 | Guidewire crossing lesion successful
DCM | 122303 | Angioplasty balloon inflated
DCM | 122304 | Angioplasty balloon deflated
DCM | 122305 | Device deployed
DCM | 122306 | Stent re-expanded
DCM | 122307 | Object removed
DCM | 122308 | Radiation applied
DCM | 122309 | Radiation removed
DCM | 122310 | Interventional device placement unsuccessful
DCM | 122311 | Interventional device placed
DCM | 122312 | Intervention performed
DCM | 122313 | Interventional device withdrawn

## CID 3413 Adverse Outcomes

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.78

Table CID 3413. Adverse Outcomes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122167 | Death During Catheterization
Include CID 3754 “Vascular Complications”
Include CID 3755 “Cath Complications”

## CID 3414 Procedure Urgency

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.79

Table CID 3414. Procedure Urgency

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR Equivalents
--- | --- | --- | --- | --- | ---
SRT | G-D210 | Elective Procedure | 103390000 | C0439608 | 21-1, 78-1, 92-1
SRT | G-D216 | Urgent Procedure | 103391001 | C0439609 | 21-2, 78-2, 92-2
SRT | G-D209 | Emergent Procedure | 25876001 | C0175673 | 21-3, 78-3, 92-3
SRT | R-41C8D | Salvage Procedure | 257950002 | C0442967 | 21-4, 78-4, 92-4

## CID 3415 Cardiac Rhythms

This Context Group comprises the ECG rhythm annotations of ISO/IEEE 11073-10102. The terms included in the table below may not constitute the complete list; see the ISO/IEEE Standard.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.80

Table CID 3415. Cardiac Rhythms

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 10:9216 | Sinus Rhythm | MDC_ECG_RHY_SINUS_RHY
MDC | 10:9232 | Normal Sinus Rhythm | MDC_ECG_RHY_SINUS_NORMAL_RHY
MDC | 10:9248 | Sinus Bradycardia | MDC_ECG_RHY_SINUS_BRADY
MDC | 10:9264 | Sinus Tachycardia | MDC_ECG_RHY_SINUS_TACHY
MDC | 10:9280 | Sinus Arrhythmia | MDC_ECG_RHY_SINUS_ARRHY
MDC | 10:9296 | Respiratory Sinus Arrhythmia | MDC_ECG_RHY_RESP_ARRHY
MDC | 10:9312 | Non-Respiratory Sinus Arrhythmia | MDC_ECG_RHY_NON_RESP_ARRHY
MDC | 10:9328 | Wandering Sinus Pacemaker within the sinus node | MDC_ECG_RHY_WANDP_ARRHY
MDC | 10:9344 | Wandering Pacemaker between the sinus node and the A-V node | MDC_ECG_RHY_WANDPAV_ARRHY
MDC | 10:9360 | Atrial Ectopic Rhythm | MDC_ECG_RHY_ATR_ECT_RHY
MDC | 10:9376 | Atrial Bigeminy | MDC_ECG_RHY_ATR_BIGEM
MDC | 10:9392 | Atrial Tachycardia | MDC_ECG_RHY_ATR_TACHY
MDC | 10:9408 | Paroxsysmal Atrial Tachycardia | MDC_ECG_RHY_ATR_TACHY_PAROX
MDC | 10:9424 | Multifocal Atrial Tachycardia | MDC_ECG_RHY_ATR_TACHY_MF
MDC | 10:9440 | Automatic Atrial Tachycardia | MDC_ECG_RHY_ATR_TACHY_AUTO
MDC | 10:9456 | Atrial flutter | MDC_ECG_RHY_ATR_FLUT
MDC | 10:9472 | Atrial fibrillation | MDC_ECG_RHY_ATR_FIB
MDC | 10:9488 | Supraventricular (atrial or junctional) Ectopic Rhythm | MDC_ECG_RHY_SV_ECT_RHY
MDC | 10:9504 | Supraventricular Tachycardia (atrial or junctional) | MDC_ECG_RHY_SV_TACHY
MDC | 10:9520 | Supraventricular Paroxsysmal Tachycardia | MDC_ECG_RHY_SV_TACHY_PAROX
MDC | 10:9536 | AV junctional (nodal) rhythm | MDC_ECG_RHY_JUNC_RHY
MDC | 10:9552 | AV junctional (nodal) escape rhythm | MDC_ECG_RHY_JUNC_ESC_BEATS
MDC | 10:9568 | Accelerated AV junctional (nodal) rhythm | MDC_ECG_RHY_JUNC_ACCEL
MDC | 10:9584 | Junctional Tachycardia | MDC_ECG_RHY_JUNC_TACHY
MDC | 10:9600 | AV reciprocating tachycardia | MDC_ECG_RHY_AV_TACHY_RECIP
MDC | 10:9616 | Reentrant AV nodal tachycardia | MDC_ECG_RHY_AV_TACHY_REENTRANT
MDC | 10:9632 | First Degree AV Block | MDC_ECG_RHY_AV_HEART_BLK_DEG_1
MDC | 10:9648 | Second Degree AV Block | MDC_ECG_RHY_AV_HEART_BLK_DEG_2
MDC | 10:9664 | Second Degree AV Block Type I (Wenckebach, or Mobitz Type I) | MDC_ECG_RHY_AV_HEART_BLK_DEG_2_I
MDC | 10:9680 | Second Degree AV Block Type II (Mobitz Type II) | MDC_ECG_RHY_AV_HEART_BLK_DEG_2_II
MDC | 10:9696 | Third Degree AV Block (complete AV block) | MDC_ECG_RHY_AV_HEART_BLK_DEG_3
MDC | 10:9712 | AV Dissociation | MDC_ECG_RHY_AV_DISSOC
MDC | 10:9728 | AV dissociation with interference | MDC_ECG_RHY_AV_DISSOC_INT
MDC | 10:9744 | Isorhythmic AV dissociation | MDC_ECG_RHY_AV_DISSOC_ISO
MDC | 10:9760 | Complete AV dissociation | MDC_ECG_RHY_AV_DISSOC_COMP
MDC | 10:9776 | First Degree SA Block | MDC_ECG_RHY_SA_HEART_BLK_DEG_1
MDC | 10:9792 | Second Degree SA Block Type I (Wenckebach) | MDC_ECG_RHY_SA_HEART_BLK_DEG_2_I
MDC | 10:9808 | Second Degree SA Block Type II | MDC_ECG_RHY_SA_HEART_BLK_DEG_2_II
MDC | 10:9824 | Third Degree SA Block (complete SA block) | MDC_ECG_RHY_SA_HEART_BLK_DEG_3
MDC | 10:9840 | Ventricular rhythm | MDC_ECG_RHY_V_RHY
MDC | 10:9856 | Idioventricular (ventricular escape) rhythm | MDC_ECG_RHY_V_IDIO_RHY
MDC | 10:9872 | Ventricular Parasystole | MDC_ECG_RHY_V_PARA
MDC | 10:9888 | Accelerated idioventricular rhythm | MDC_ECG_RHY_V_AIVR
MDC | 10:9904 | Slow Ventricular Tachycardia (Idioventricular Tachycardia) | MDC_ECG_RHY_V_IDIO_TACHY
MDC | 10:9920 | Ventricular Bigeminy | MDC_ECG_RHY_V_BIGEM
MDC | 10:9936 | Ventricular Trigeminy | MDC_ECG_RHY_V_TRIGEM
MDC | 10:9952 | Ventricular Couplet | MDC_ECG_RHY_V_P_C_CPLT
MDC | 10:9968 | Ventricular Run | MDC_ECG_RHY_V_P_C_RUN
MDC | 10:9984 | Ventricular Tachycardia (nonparoxysmal) | MDC_ECG_RHY_V_TACHY
MDC | 10:10000 | Ventricular Flutter | MDC_ECG_RHY_V_FLUT
MDC | 10:10016 | Ventricular Fibrillation | MDC_ECG_RHY_V_FIB
MDC | 10:10032 | Nonsustained Ventricular Tachycardia (paroxysmal) | MDC_ECG_RHY_V_TACHY_PAROX
MDC | 10:10048 | Sustained Monomorphic Ventricular Tachycardia | MDC_ECG_RHY_V_TACHY_MONO
MDC | 10:10064 | Polymorphic Ventricular Tachycardia | MDC_ECG_RHY_V_TACHY_POLY
MDC | 10:10080 | Torsades de Pointes Ventricular Tachycardia | MDC_ECG_RHY_V_TACHY_TDP
MDC | 10:10096 | pre-excitation | MDC_ECG_RHY_PREX
MDC | 10:10112 | Wolf-Parkinson-White syndrome | MDC_ECG_RHY_WPW_UNK
MDC | 10:10128 | Wolf-Parkinson type A | MDC_ECG_RHY_WPW_A
MDC | 10:10144 | Wolf-Parkinson type B | MDC_ECG_RHY_WPW_B
MDC | 10:10160 | Lown-Ganong-Levine syndrome | MDC_ECG_RHY_LGL
MDC | 10:10336 | Asystole | MDC_ECG_RHY_ASYSTOLE
MDC | 10:10352 | Irregular rhythm | MDC_ECG_RHY_IRREG
MDC | 10:10368 | Low Heart Rate Variability | MDC_ECG_RHY_LHRV
MDC | 10:10416 | T-wave alternans | MDC_ECG_RHY_TALT
MDC | 10:10432 | Bradycardia | MDC_ECG_RHY_BRADY
MDC | 10:10448 | Calibration signal (sustained) | MDC_ECG_RHY_CALS
MDC | 10:10176 | Atrial Demand Mode Pacing | MDC_ECG_RHY_EPADM
MDC | 10:10240 | Ventricular Demand Mode Pacing | MDC_ECG_RHY_EPVDM
MDC | 10:10304 | Anti-Tachycardia Pacing | MDC_ECG_RHY_EPAVT

### Note

A prior version of this context group used codes from the SCP-ECG vocabulary.

## CID 3416 Respiration Rhythms

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.81

Table CID 3416. Respiration Rhythms

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-21301 | normal respiratory rhythm | 5467003 | C0231843
SRT | F-21303 | irregular breathing | 248585001 | C0425492
SRT | F-20130 | gasping respiration | 23141003 | C0425449
SRT | F-21334 | abnormal respiratory rhythm | 248584002 | C0425491
SRT | F-21331 | respiration intermittent | 271824009 | C1313952

## CID 3418 Lesion Risk

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.82

Table CID 3418. Lesion Risk

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 111-1 | Low Risk Lesion
NCDR | 2.0b | 111-2 | Moderate Risk Lesion
NCDR | 2.0b | 111-3 | High Risk Lesion

## CID 3419 Findings Titles

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.83

Table CID 3419. Findings Titles

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121071 | Finding
DCM | 121073 | Impression
DCM | 121075 | Recommendation

## CID 3421 Procedure Action

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.84

Table CID 3421. Procedure Action

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121130 | Start Procedure Action
DCM | 121131 | End Procedure Action
DCM | 121132 | Suspend Procedure Action
DCM | 121133 | Resume Procedure Action

## CID 3422 Device Use Actions

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.85

Table CID 3422. Device Use Actions

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-002F8 | Device inserted into sheath | 371877003 | C1299350
SRT | R-002F7 | Device at site of interest | 371876007 | C1299349
SRT | R-002FB | Device withdrawn / removed | 371875006 | C1299348
SRT | R-002F6 | Device applied to patient | 373061006 | C1298903
SRT | R-002FA | Device used | 373062004 | C1298904
SRT | R-10042 | Device crossed septum | 386125002 | C1272581
DCM | 122089 | Device crossed lesion |  |

## CID 3423 Numeric Device Characteristics

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.86

Table CID 3423. Numeric Device Characteristics

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-D7FE | Length | 410668003 | C1444754
SRT | M-02550 | Diameter | 81827009 | C1301886
DCM | 122097 | Catheter Curve |  |
DCM | 122098 | Transmit Frequency |  |
SRT | G-D705 | Volume | 118565006 | C0449468
DCM | 121208 | Inter-Marker Distance |  |

## CID 3425 Intervention Parameters

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.87

Table CID 3425. Intervention Parameters

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-002D0 | Angioplasty Inflation pressure | 371851006 | C1299326
SRT | R-002CF | Angioplasty Inflation duration | 371852004 | C1299327
SRT | R-0036C | Rotational Atherectomy Speed | 371854003 | C1299329
SRT | R-002F2 | Delivered Radiation Dose | 371892002 | C1299361
SRT | R-10043 | Ablation power | 386131004 | C1272583
SRT | R-10044 | Ablation frequency | 386132006 | C1272584

## CID 3426 Consumables Parameters

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.88

Table CID 3426. Consumables Parameters

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121145 | Description of Material
DCM | 121148 | Unit Serial Identifier
DCM | 121149 | Lot Identifier

## CID 3427 Equipment Events

Type:

  Extensible

Version:

  20100608

UID:

  1.2.840.10008.6.1.89

Table CID 3427. Equipment Events

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110501 | Equipment failure
DCM | 122047 | Equipment brought to procedure room
DCM | 122048 | Equipment ready
DCM | 122049 | Equipment removed

## CID 3428 Imaging Procedures

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.90

Table CID 3428. Imaging Procedures

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P5-009A0 | Angiography | 77343006 | C0002978
SRT | P5-32130 | Aortography | 54640009 | C0003515
SRT | P5-30100 | Coronary Arteriography | 33367005 | C0085532
SRT | P5-3003A | Cardiac ventriculography | 252426003 | C0596683
SRT | P5-30041 | Left Ventriculography | 265484009 | C0412219
SRT | P5-3003F | Right Ventriculography | 265483003 | C0412220
SRT | P5-30107 | Bypass graft angiography | 252427007 | C0430469
DCM | 122058 | Arterial conduit angiography |  |
SRT | P5-B3002 | Transesophageal echocardiography | 105376000 | C0206054
SRT | P5-B3003 | Transthoracic echocardiography | 169242007 | C0013516
SRT | P5-B3004 | Epicardial echocardiography | 252418006 | C0430462
SRT | P5-B001D | Intravascular ultrasound | 241466007 | C0412530
SRT | P5-B3006 | Intracardiac echocardiography | 252421008 | C0430464

## CID 3429 Catheterization Devices

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.91

Table CID 3429. Catheterization Devices

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-28051 | Intra-Aortic Balloon Pump (IABP) | 129113006 | C0702122
SRT | R-00306 | Fluid filled catheter | 371798007 | C1299428
SRT | R-00304 | Fiberoptic catheter | 371801001 | C1300076
SRT | R-0030A | Hall catheter | 371799004 | C1299429
SRT | R-00379 | Thermistor catheter | 371800000 | C1299430
SRT | R-00383 | Tip manometer | 371802008 | C1299431
SRT | A-26860 | Swann-Ganz catheter | 79952001 | C0179790
SRT | F-9B75C | Sheath | 268461001 | C0419524
SRT | R-10041 | Transseptal catheter | 386124003 | C1272580
DCM | 122052 | Bioptome |  |
Include CID 3411 “Intracoronary Devices” |  |

## CID 3430 DateTime Qualifiers

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.92

Table CID 3430. DateTime Qualifiers

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121136 | DateTime Unsynchronized
DCM | 121137 | DateTime Estimated

## CID 3440 Peripheral Pulse Locations

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.93

Table CID 3440. Peripheral Pulse Locations

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-47160 | Brachial Artery | 17137000 | C0006087
SRT | T-45010 | Carotid Artery | 69105007 | C0007272
SRT | T-47740 | Dorsalis Pedis Artery | 86547008 | C0226492
SRT | T-47400 | Femoral Artery | 7657000 | C0015801
SRT | T-47500 | Popliteal Artery | 43899006 | C0032649
SRT | T-47600 | Posterior Tibial Artery | 13363002 | C0086835
SRT | T-47300 | Radial Artery | 45631007 | C0162857
SRT | T-47200 | Ulnar Artery | 44984001 | C0162858

## CID 3441 Patient Assessments

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.94

Table CID 3441. Patient Assessments

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
LN | 8884-9 | Cardiac Rhythm |  | C0488795
LN | 9304-7 | Respiration Rhythm |  | C0489261
SRT | F-046D8 | Skin condition assessment | 364528001 | C1286230
SRT | F-043E6 | Respiration assessment | 364062005 | C1285809
SRT | F-04317 | Patient mental state assessment | 363871006 | C0278060

## CID 3442 Peripheral Pulse Methods

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.95

Table CID 3442. Peripheral Pulse Methods

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P2-01510 | Palpation | 113011001 | C0030247
SRT | P1-30022 | Doppler | 83422003 | C0189575

## CID 3446 Skin Condition

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.96

Table CID 3446. Skin Condition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122271 | skin condition Warm
DCM | 122272 | skin condition Cool
DCM | 122273 | skin condition Cold
DCM | 122274 | skin condition Dry
DCM | 122275 | skin condition Clammy
DCM | 122276 | skin condition Diaphoretic
DCM | 122277 | skin condition Flush
DCM | 122278 | skin condition Mottled
DCM | 122279 | skin condition Pale

## CID 3448 Airway Assessment

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.97

Table CID 3448. Airway Assessment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122281 | airway unobstructed
DCM | 122282 | airway partially obstructed
DCM | 122283 | airway severely obstructed

## CID 3451 Calibration Objects

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.98

Table CID 3451. Calibration Objects

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-26800 | Catheter | 19923001 | C0085590
SRT | A-10141 | Measuring Ruler | 102304005 | C0522637
DCM | 122485 | Sphere |  |

## CID 3452 Calibration Methods

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.99

Table CID 3452. Calibration Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122486 | Geometric Isocenter
DCM | 122487 | Geometric Non-Isocenter
DCM | 122488 | Calibration Object Used

## CID 3453 Cardiac Volume Methods

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.100

Table CID 3453. Cardiac Volume Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122558 | Area Length Kennedy
DCM | 122559 | Area Length Dodge
DCM | 122560 | Area Length Wynne
DCM | 122562 | Multiple Slices
DCM | 122563 | Boak
DCM | 122564 | TS Pyramid
DCM | 122565 | Two Chamber
DCM | 122566 | Parallelepiped

## CID 3455 Index Methods

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.101

Table CID 3455. Index Methods

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID
--- | --- | --- | ---
LN | 8277-6 | BSA | C0487992
DCM | 122572 | BSA^1.219 |

## CID 3456 Sub-segment Methods

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.102

Table CID 3456. Sub-segment Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122574 | Equidistant method
DCM | 122575 | User selected method

## CID 3458 Contour Realignment

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.103

Table CID 3458. Contour Realignment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122475 | Center of Gravity
DCM | 122476 | Long Axis Based
DCM | 122477 | No Realignment

## CID 3460 Circumferential Extent

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.104

Table CID 3460. Circumferential Extent

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122464 | LAD Region in RAO Projection
DCM | 122465 | RCA Region in RAO Projection

## CID 3461 Regional Extent

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.105

Table CID 3461. Regional Extent

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122466 | Single LAD Region in RAO Projection
DCM | 122467 | Single RCA Region in RAO Projection
DCM | 122468 | Multiple LAD Region in RAO Projection
DCM | 122469 | Multiple RCA Region in RAO Projection
DCM | 122470 | LAD Region in LAO Projection
DCM | 122471 | RCA Region in LAO Projection
DCM | 122472 | CFX Region in LAO Projection

## CID 3462 Chamber Identification

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.106

Table CID 3462. Chamber Identification

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-32600 | Left Ventricle | 87878005 | C0225897
SRT | T-32500 | Right Ventricle | 53085002 | C0225883
SRT | T-32300 | Left Atrium | 82471001 | C0225860
SRT | T-32200 | Right Atrium | 73829009 | C0225844

## CID 3463 Ventricle Identification

Type:

  Extensible

Version:

  20080927

UID:

  1.2.840.10008.6.1.786

Table CID 3463. Ventricle Identification

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-32600 | Left Ventricle | 87878005 | C0225897
SRT | T-32500 | Right Ventricle | 53085002 | C0225883

## CID 3465 QA Reference Methods

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.107

Table CID 3465. QA Reference Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122489 | Curve Fitted Reference
DCM | 122490 | Interpolated Local Reference
DCM | 122491 | Mean Local Reference

## CID 3466 Plane Identification

Type:

  Extensible

Version:

  20130806

UID:

  1.2.840.10008.6.1.108

Table CID 3466. Plane Identification

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-40985 | Right Anterior Oblique | 399356000 | C1275852
SRT | R-10220 | Left Anterior Oblique | 399135007 | C1275823
SRT | R-10206 | Antero-posterior | 399348003 | C0442212
SRT | R-10236 | Left Lateral | 399173006 | C0442198
SRT | R-101C3 | Cranial LAO | 408723005 | C1443272
SRT | R-101C5 | Cranial RAO | 408725003 | C1443274
SRT | R-101C4 | Caudal LAO | 408724004 | C1443273
SRT | R-101C6 | Caudal RAO | 408726002 | C1443275

### Note

In a prior version of this Context Group, "right anterior oblique" was assigned the code R-10218, which in SNOMED is actually "Indirect iris transillumination"; this code has been replaced with the correct code R-40985.

## CID 3467 Ejection Fraction

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.109

Table CID 3467. Ejection Fraction

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID
--- | --- | --- | ---
LN | 8808-8 | Left Ventricular Ejection Fraction by Angiography | C0488723
LN | 8815-3 | Right Ventricular Ejection Fraction by Angiography | C0488731
DCM | 122406 | Left Atrial Ejection Fraction by Angiography |

## CID 3468 ED Volume

Type:

  Extensible

Version:

  20110124

UID:

  1.2.840.10008.6.1.110

Table CID 3468. ED Volume

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID
--- | --- | --- | ---
LN | 8821-1 | Left Ventricular ED Volume | C0488738
LN | 8822-9 | Right Ventricular ED Volume | C0488739
DCM | 122407 | Left Atrial ED Volume |

## CID 3469 ES Volume

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.111

Table CID 3469. ES Volume

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID
--- | --- | --- | ---
LN | 8823-7 | Left Ventricular ES Volume | C0488740
LN | 8824-5 | Right Ventricular ES Volume | C0488741
DCM | 122408 | Left Atrial ES Volume |

## CID 3470 Vessel Lumen Cross-sectional Area Calculation Methods

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.112

Table CID 3470. Vessel Lumen Cross-sectional Area Calculation Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122473 | Circular method
DCM | 122474 | Densitometric method

## CID 3471 Estimated Volumes

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.113

Table CID 3471. Estimated Volumes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121216 | Volume estimated from single 2D region
DCM | 121218 | Volume estimated from two non-coplanar 2D regions

## CID 3472 Cardiac Contraction Phase

Type:

  Extensible

Version:

  20071031

UID:

  1.2.840.10008.6.1.114

Table CID 3472. Cardiac Contraction Phase

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-32020 | Systolic | 111973004 | C0039155
SRT | F-32010 | Diastolic | 90892000 | C0012000

## CID 3480 IVUS Procedure Phases

This context group outlines the phases of a catheterization procedure in which measurements are performed.

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.115

Table CID 3480. IVUS Procedure Phases

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-7298 | Cardiac catheterization post-intervention phase | 128960007 | C1292437
SRT | G-7296 | Cardiac catheterization pre-intervention phase | 128958005 | C1292435

## CID 3481 IVUS Distance Measurements

This context group is the set of distance measurements made in an IVUS procedure.

Type:

  Extensible

Version:

  20050110

UID:

  1.2.840.10008.6.1.116

Table CID 3481. IVUS Distance Measurements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 122330 | EEM Diameter |  |
SRT | G-0364 | Vessel lumen diameter | 397413000 | C1301408
SRT | R-101AD | Stent Diameter | 408706001 | C1443256
DCM | 122331 | Plaque Plus Media Thickness |  |
DCM | 122332 | Lumen Perimeter |  |

## CID 3482 IVUS Area Measurements

This context group is the set of cross-sectional area measurements made in an IVUS procedure.

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.117

Table CID 3482. IVUS Area Measurements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 122333 | EEM Cross-Sectional Area |  |
SRT | G-0366 | Vessel lumen cross-sectional area | 397415007 | C1301410
SRT | R-101AF | Stent Cross-Sectional Area | 408705002 | C1443255
DCM | 122334 | Plaque plus Media Cross-Sectional Area |  |
DCM | 122335 | In-Stent Neointimal Cross-Sectional Area |  |

## CID 3483 IVUS Longitudinal Measurements

This context group is a set of measurements that are made on a longitudinal image. A longitudinal image is a perpendicular cut plane reconstructed from an IVUS pullback multi-frame image.

Type:

  Extensible

Version:

  20050110

UID:

  1.2.840.10008.6.1.118

Table CID 3483. IVUS Longitudinal Measurements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-101B0 | Stent Length | 408703009 | C1443253
SRT | R-101BC | Stenotic Lesion Length | 408716009 | C1443266
DCM | 122341 | Calcium Length |  |
DCM | 122364 | Stent Gap |  |

## CID 3484 IVUS Indices and Ratios

This context group is the set of index and ratio calculations made in an IVUS procedure.

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.119

Table CID 3484. IVUS Indices and Ratios

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122343 | Lumen Eccentricity Index
DCM | 122344 | Plaque plus Media Eccentricity Index
DCM | 122345 | Remodeling Index
DCM | 122346 | Stent Symmetry Index
DCM | 122347 | Stent Expansion Index
DCM | 122348 | Lumen Shape Index
DCM | 122350 | Lumen Diameter Ratio
DCM | 122351 | Stent Diameter Ratio
DCM | 122352 | EEM Diameter Ratio

## CID 3485 IVUS Volume Measurements

This context group is the set of volume measurements made from an IVUS procedure.

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.120

Table CID 3485. IVUS Volume Measurements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 122371 | EEM Volume |  |
DCM | 122372 | Lumen Volume |  |
SRT | R-101B2 | Stent Volume | 408704003 | C1443254
DCM | 122374 | In-Stent Neointimal Volume |  |
DCM | 122375 | Native Plaque Volume |  |
DCM | 122376 | Total Plaque Volume |  |

## CID 3486 Vascular Measurement Sites

This context group is the set of sites where vascular measurements can be made.

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.121

Table CID 3486. Vascular Measurement Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122380 | Proximal Reference
DCM | 122381 | Distal Reference
DCM | 122382 | Site of Lumen Minimum
DCM | 122687 | Site of Lumen Maximum

## CID 3487 Intravascular Volumetric Regions

This context group is the set of regions where intravascular volumetric measurements can be made.

Type:

  Extensible

Version:

  20170413

UID:

  1.2.840.10008.6.1.122

Table CID 3487. Intravascular Volumetric Regions

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 122383 | Stented Region |  |
DCM | 122384 | Entire Pullback |  |
DCM | 122385 | Proximal Stent Margin |  |
DCM | 122386 | Distal Stent Margin |  |
SRT | M-01000 | Morphologically Abnormal Structure | 49755003 | C0332447
SRT | M-01100 | Lesion | 52988006 | C0221198
SRT | R-002EF | Culprit Lesion | 371895000 | C1299364

### Note

(M-01000, SRT, "Morphologically Abnormal Structure") was previously described with a Code Meaning of "Lesion", but that synonym has been retired as "inappropriate" in SNOMED. The Code Meaning has been replaced with the preferred SNOMED term, and the separate concept (M-01100, SRT, "Lesion") added.

## CID 3488 Min/Max/Mean

This context group contains modifiers that indicate whether the measurement is a minimum, maximum or averaged value.

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.123

Table CID 3488. Min/Max/Mean

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A437 | Maximum | 56851009 | C0205289
SRT | R-404FB | Minimum | 255605001 | C0547040
SRT | R-00317 | Mean | 373098007 | C1298794

## CID 3489 Calcium Distribution

This context group is a set of modifiers specifying the distribution of a calcium deposit in an arc of calcium measurement.

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.124

Table CID 3489. Calcium Distribution

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A139 | Superficial | 26283006 | C0205124
SRT | G-A140 | Deep | 795002 | C0205125

## CID 3491 IVUS Lesion Morphologies

This context group is a set of qualitative assessments for lesion morphology.

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.125

Table CID 3491. IVUS Lesion Morphologies

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 3495 “IVUS Plaque Composition” |  |
DCM | 122356 | Soft plaque |  |
DCM | 122357 | In-Stent Neointima |  |
SRT | D3-80027 | Arterial (True) Aneurysm | 233981004 | C0340613
SRT | M-32390 | Pseudo Aneurysm | 22036004 | C1510412
DCM | 122361 | False Lumen |  |
SRT | R-4047B | Concentric | 255465008 | C0439744
SRT | R-40416 | Eccentric | 255380003 | C0439740
SRT | M-52103 | Plaque Ulceration | 62189002 | C0333481
DCM | 122363 | Plaque Rupture |  |
DCM | 122389 | Vulnerable Plaque |  |
DCM | 122390 | Eroded Plaque |  |

## CID 3492 Vascular Dissection Classifications

This context group is a set of dissection classifications commonly detected with IVUS or CT/MR angiography.

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.126

Table CID 3492. Vascular Dissection Classifications

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-101B7 | Medial Dissection | 408711004 | C1443261
SRT | R-101B8 | Intimal Dissection | 409509002 | C1443854
SRT | R-101B9 | Adventitial Dissection | 409510007 | C1443855
SRT | M-35063 | Intramural hematoma | 54493002 | C0333200
DCM | 122388 | Intra-stent Dissection |  |

## CID 3493 IVUS Relative Stenosis Severities

This context group is a set of stenosis severity classifications for multiple lesions within a segment. There will always be a worst stenosis (T-1), the stenosis with the smallest lumen size.

There can be multiple secondary stenoses (T-2, T-3, etc.), which are lesions meeting the definition of a stenosis, but with lumen sizes larger than the worst stenosis. Reference "American College of Cardiology Clinical Expert Consensus Document on Standards for Acquisition, Measurement and Reporting of Intravascular Ultrasound Studies (IVUS) ".

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.127

Table CID 3493. IVUS Relative Stenosis Severities

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122367 | T-1 Worst
DCM | 122368 | T-2 Secondary
DCM | 122369 | T-3 Secondary
DCM | 122370 | T-4 Secondary

## CID 3494 IVUS Non Morphological Findings

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.128

Table CID 3494. IVUS Non Morphological Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 122360 | True Lumen |  |
SRT | R-101B3 | Arterial Blood Stasis | 408707005 | C1443257
SRT | R-101B5 | Incomplete Stent apposition | 408709008 | C1443259
SRT | R-101B6 | Acquired Incomplete stent apposition | 408710003 | C1443260

## CID 3495 IVUS Plaque Composition

This context group is a set of qualitative assessments defining the composition of plaque.

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.129

Table CID 3495. IVUS Plaque Composition

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-78260 | Fibrous Plaque | 40772000 | C0334146
SRT | D6-34737 | Vascular Calcification | 237897009 | C0342649
SRT | M-35001 | Thrombus | 396339007 | C0087086
DCM | 122394 | Fibro-Lipidic Plaque |  |
DCM | 122395 | Necrotic-Lipidic Plaque |  |

## CID 3496 IVUS Fiducial Points

This context group is a set of fiducial points (anatomical markers). Fiducial points are used as identifiable axial landmarks in determining the location of a measurement in a vessel.

Type:

  Extensible

Version:

  20040614

UID:

  1.2.840.10008.6.1.130

Table CID 3496. IVUS Fiducial Points

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-035D | Collateral Branch of vessel | 397406000 | C1275670
SRT | A-25500 | Stent | 65818007 | C0038257
SRT | D6-34737 | Vascular Calcification | 237897009 | C0342649
SRT | M-78260 | Fibrous Plaque | 40772000 | C0334146
SRT | T-48000 | Vein | 29092000 | C0042449
SRT | G-036A | Vessel Origin | 397421006 | C1301415

## CID 3497 IVUS Arterial Morphology

Type:

  Extensible

Version:

  20050110

UID:

  1.2.840.10008.6.1.131

Table CID 3497. IVUS Arterial Morphology

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-41100 | Lumen of artery | 67170007 | C0225997
SRT | R-102AE | External Elastic Membrane | 414165007 | C1532733
Include CID 3495 “IVUS Plaque Composition” |  |

## CID 3500 Pressure Units

Type:

  Non-Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.132

Table CID 3500. Pressure Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | mm[Hg] | mmHg
UCUM | kPa | kPa

## CID 3502 Hemodynamic Resistance Units

Type:

  Non-Extensible

Version:

  20120327

UID:

  1.2.840.10008.6.1.133

Table CID 3502. Hemodynamic Resistance Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | [PRU] | P.R.U.
UCUM | [wood'U] | Wood U
UCUM | dyn.s.cm-5 | dyn.s.cm-5

### Note

P.R.U. is in units of mm[Hg].s/ml; Wood Units is in mm[Hg].min/l

## CID 3503 Indexed Hemodynamic Resistance Units

Type:

  Non-Extensible

Version:

  20120327

UID:

  1.2.840.10008.6.1.134

Table CID 3503. Indexed Hemodynamic Resistance Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | [PRU]/m2 | P.R.U./m2
UCUM | [wood'U]/m2 | Wood U/m2
UCUM | dyn.s.cm-5/m2 | dyn.s.cm-5/m2

## CID 3510 Catheter Size Units

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.135

Table CID 3510. Catheter Size Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | [Ch] | french
UCUM | mm | mm

## CID 3515 Specimen Collection

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.136

Table CID 3515. Specimen Collection

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P3-02000 | specimen collection | 17636008 | C0200345
SRT | PA-20110 | collection of blood specimen for laboratory | 82078001 | C0005834
SRT | PA-2011E | blood sampling from extracorporeal blood circuit | 243776001 | C0419352

## CID 3520 Blood Source Type

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.137

Table CID 3520. Blood Source Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-00376 | Systemic Artery Blood | 371952000 | C1299266
SRT | T-C2007 | Mixed Venous Blood | 116176007 | C0440739
SRT | R-0035B | Pulmonary Artery Blood | 371953005 | C1299267
SRT | R-0035E | Pulmonary Vein Blood | 371954004 | C1299268

## CID 3524 Blood Gas Pressures

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.138

Table CID 3524. Blood Gas Pressures

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID
--- | --- | --- | ---
LN | 11557-6 | Blood Carbon dioxide partial pressure | C0550246
LN | 2019-8 | Arterial Blood Carbon dioxide partial pressure | C0364151
LN | 2021-4 | Venous Blood Carbon dioxide partial pressure | C0364153
LN | 11556-8 | Blood Oxygen partial pressure | C0550440
LN | 2703-7 | Arterial Oxygen partial pressure | C1145645
LN | 2705-2 | Venous Oxygen partial pressure | C1145647
LN | 19217-9 | Oxygen partial pressure at 50% saturation (P50) | C0802130
LN | 19214-6 | Arterial Oxygen partial pressure at 50% saturation | C1153749
LN | 19216-1 | Venous Oxygen partial pressure at 50% saturation | C1153751

## CID 3525 Blood Gas Content

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.139

Table CID 3525. Blood Gas Content

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID
--- | --- | --- | ---
LN | 20565-8 | Blood Carbon dioxide content | C0803374
LN | 2026-3 | Arterial Blood Carbon dioxide content | C0364158
LN | 2027-1 | Venous Blood Carbon dioxide content | C0364159
DCM | 122185 | Blood Oxygen content |
LN | 19218-7 | Arterial Oxygen content | C0802131
LN | 19220-3 | Venous Oxygen content | C0802133
LN | 10232-7 | Aortic Root Oxygen content | C0488752
LN | 10245-9 | Pulmonary Artery Main Oxygen content | C0488765
LN | 10247-5 | Pulmonary Wedge Oxygen content | C0488767

## CID 3526 Blood Gas Saturation

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.140

Table CID 3526. Blood Gas Saturation

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID
--- | --- | --- | ---
DCM | 122187 | Blood Carbon dioxide saturation |
LN | 20564-1 | Blood Oxygen saturation | C0803373
LN | 2708-6 | Arterial Oxygen saturation | C0364851
LN | 2711-0 | Venous Oxygen saturation | C0364854
LN | 2709-4 | Capillary Blood Oxygen Saturation | C0364852
LN | 2710-2 | Capillary Blood Oxygen Saturation, by Oximetry | C0364853

## CID 3527 Blood Base Excess

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.141

Table CID 3527. Blood Base Excess

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID
--- | --- | --- | ---
LN | 11555-0 | Blood Base Excess | C0550221
LN | 1925-7 | Arterial Blood Base Excess | C0364060
LN | 1927-3 | Venous Blood Base Excess | C0364062

## CID 3528 Blood pH

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.142

Table CID 3528. Blood pH

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID
--- | --- | --- | ---
LN | 11558-4 | Blood pH | C0550447
LN | 2744-1 | Arterial Blood pH | C0364887
LN | 2746-6 | Venous Blood pH | C0364889

## CID 3529 Arterial / Venous Content

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.143

Table CID 3529. Arterial / Venous Content

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID
--- | --- | --- | ---
LN | 19218-7 | Arterial Content (FCa) | C0802131
LN | 19220-3 | Venous Content (FCv) | C0802133
DCM | 122188 | Pulmonary Arterial Content (FCpa) |
DCM | 122189 | Pulmonary Venous Content (FCpv) |

## CID 3530 Oxygen Administration Actions

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.144

Table CID 3530. Oxygen Administration Actions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121161 | Begin oxygen administration
DCM | 121162 | End oxygen administration

## CID 3531 Oxygen Administration

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.145

Table CID 3531. Oxygen Administration

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-0034A | Oxygen Administration by nasal cannula | 371907003 | C1299376
SRT | R-00349 | Oxygen Administration by mask | 371908008 | C1299377
DCM | 121163 | Oxygen Administration by ventilator |  |

## CID 3550 Circulatory Support Actions

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.146

Table CID 3550. Circulatory Support Actions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121157 | Begin Circulatory Support
DCM | 121158 | End Circulatory Support

## CID 3551 Ventilation Actions

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.147

Table CID 3551. Ventilation Actions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121168 | Begin Ventilation
DCM | 121169 | End Ventilation

## CID 3552 Pacing Actions

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.148

Table CID 3552. Pacing Actions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121166 | Begin Pacing
DCM | 121167 | End Pacing

## CID 3553 Circulatory Support

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.149

Table CID 3553. Circulatory Support

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-28051 | Intra-Aortic Balloon Pump | 129113006 | C0702122
SRT | R-00303 | External Counter-Pulsation | 371790000 | C1299423
SRT | A-11FCD | Left Ventricular Assist Device | 360066001 | C0181598
SRT | P2-77110 | Extra-corporeal circulation | 182744004 | C0015354
SRT | P1-36858 | Cardiopulmonary bypass | 63697000 | C0007202

## CID 3554 Ventilation

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.150

Table CID 3554. Ventilation

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-002CC | Ambu Bag | 371785003 | C0221812
SRT | R-00359 | Pressure Support Ventilator | 371786002 | C1299420
SRT | R-0038C | Volume Support Ventilator | 371787006 | C1299421

## CID 3555 Pacing

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.151

Table CID 3555. Pacing

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P2-35000 | Pacing | 18590009 | C0199640
SRT | R-00315 | pacing with magnet | 371909000 | C1299378
SRT | P2-35200 | atrial pacing | 69158002 | C0199647
SRT | P2-35002 | ventricular pacing | 344994008 | C0199648
SRT | R-002D9 | A-V sequential pacing | 371910005 | C1299379
SRT | P2-35440 | temporary transcutaneous pacing | 59218006 | C0199657

## CID 3560 Blood Pressure Methods

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.152

Table CID 3560. Blood Pressure Methods

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-00318 | Blood pressure cuff method | 371911009 | C1299380

## CID 3600 Relative Times

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.153

Table CID 3600. Relative Times

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-407E0 | Before | 272113006 | C0740175
SRT | R-407E1 | During | 272114000 | C0347985
SRT | R-42517 | After | 288563008 | C0687676

## CID 3602 Hemodynamic Patient State

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.154

Table CID 3602. Hemodynamic Patient State

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01602 | Baseline state | 128974000 | C1290922
SRT | F-10340 | Supine body position | 40199007 | C0038846
SRT | F-01604 | Resting state | 128975004 | C0679218
SRT | F-01606 | Exercise state | 128976003 | C1290923
SRT | F-01608 | Post-exercise state | 128977007 | C1290924

## CID 3604 Arterial Lesion Locations

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.155

Table CID 3604. Arterial Lesion Locations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3015 “Coronary Arteries”
Include CID 3606 “Arterial Source Locations”

## CID 3606 Arterial Source Locations

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.156

Table CID 3606. Arterial Source Locations

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-42500 | Abdominal aorta | 7832008 | C0003484
SRT | T-45530 | anterior communicating artery | 8012006 | C0149562
SRT | T-45730 | anterior spinal artery | 17388009 | C0149603
SRT | T-42000 | Aorta | 15825003 | C0003483
SRT | T-42300 | Aortic Arch | 57034009 | C0003489
SRT | D3-81922 | Aortic fistula | 128551005 | C1290392
SRT | T-41000 | Artery | 51114001 | C0003842
SRT | T-42100 | Ascending aorta | 54247002 | C0003956
SRT | T-47100 | Axillary Artery | 67937003 | C0004455
SRT | A-00203 | Baffle | 128981007 | C1289790
SRT | T-45800 | basilar artery | 59011009 | C0004811
SRT | T-47160 | Brachial artery | 17137000 | C0006087
SRT | T-46010 | brachiocephalic trunk | 12691009 | C0006094
SRT | T-45010 | Carotid Artery | 69105007 | C0007272
SRT | T-45510 | cerebral artery | 88556005 | C0007770
SRT | T-45100 | Common carotid artery | 32062004 | C0162859
SRT | T-43000 | Coronary Artery | 41801008 | C0205042
SRT | T-42400 | Descending aorta | 32672002 | C3163626
SRT | T-45240 | facial artery | 23074001 | C0226109
SRT | T-47400 | Femoral artery | 7657000 | C0015801
SRT | D4-32504 | Fistula coronary to left atrium | 128555001 | C1290487
SRT | D4-32506 | Fistula coronary to left ventricle | 128556000 | C1290488
SRT | R-002ED | Fistula coronary to right atrium | 373095005 | C1298791
SRT | D4-32510 | Fistula coronary to right ventricle | 128558004 | C1290490
SRT | T-47490 | geniculate artery | 128559007 | C0447108
SRT | T-46420 | Hepatic artery | 76015000 | C0019145
SRT | T-46700 | Iliac artery | 10293006 | C0020887
SRT | T-46010 | Innominate artery | 12691009 | C0006094
SRT | T-45300 | internal carotid artery | 86117002 | C0007276
SRT | T-46200 | Internal mammary artery | 69327007 | C0226276
SRT | T-45410 | lacrimal artery | 59749000 | C0226171
SRT | T-47650 | lateral plantar artery | 44830000 | C0226478
SRT | T-44400 | Left pulmonary artery | 50408007 | C0226069
SRT | T-45230 | lingual artery | 113264009 | C0226104
SRT | T-46960 | lumbar artery | 34635009 | C0226408
SRT | T-46500 | mesenteric artery | 86570000 | C0025465
SRT | T-47660 | medial plantar artery | 74156002 | C0226479
SRT | T-F7001 | Neo-aorta (primitive aorta) | 14944004 | C0231136
SRT | T-F7040 | Neonatal pulmonary artery (primitive PA) | 91707000 | C0231157
SRT | T-45250 | occipital artery | 31145008 | C0226117
SRT | T-45400 | ophthalmic artery | 53549008 | C0029078
SRT | D4-32012 | patent ductus arteriosus | 83330001 | C0013274
SRT | T-47630 | peroneal artery | 8821006 | C0226476
SRT | T-47500 | popliteal artery | 43899006 | C0032649
SRT | T-45320 | posterior communicating artery | 43119007 | C0149559
SRT | D3-4020B | Pulmonary arteriovenous fistula | 253639004 | C0345042
SRT | T-44000 | Pulmonary artery | 81040000 | C0034052
SRT | D4-33142 | Pulmonary artery conduit | 128584005 | C1290491
SRT | R-00360 | Pulmonary vein wedge | 371829003 | C1299456
SRT | T-47300 | radial artery | 45631007 | C0162857
SRT | T-46600 | Renal artery | 2841007 | C0035065
SRT | T-47410 | Right femoral artery | 69833005 | C0226447
SRT | T-44200 | Right pulmonary artery | 78480002 | C0226054
SRT | T-46100 | Subclavian Artery | 36765005 | C0038530
SRT | T-45270 | superficial temporal artery | 15672000 | C0226130
SRT | T-45210 | superior thyroid artery | 72021004 | C0226093
SRT | T-44007 | Systemic collateral Artery to lung | 128589000 | C0345096
SRT | T-42070 | Thoracic aorta | 113262008 | C1522460
SRT | T-4704C | tibial artery | 181351007 | C0085427
SRT | D4-31400 | Truncus Arteriosus Communis | 61959006 | C0041207
SRT | T-F1810 | Umbilical artery | 50536004 | C0041632
SRT | T-45700 | Vertebral artery | 85234005 | C0042559

## CID 3607 Venous Source Locations

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.157

Table CID 3607. Venous Source Locations

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-48503 | Anomalous pulmonary vein | 128585006 | C0265914
SRT | T-49215 | Antecubital Vein | 128553008 | C1276271
SRT | T-49110 | Axillary vein | 68705008 | C0004456
SRT | T-48340 | Azygos vein | 72107004 | C0004526
SRT | T-49230 | Basilic vein | 19715009 | C0226801
SRT | T-49424 | Boyd perforating vein | 128548003 | C1267522
SRT | T-49350 | Brachial vein | 20115005 | C0226812
SRT | T-48003 | Central venous system | 34340008 | C0226503
SRT | T-49240 | cephalic vein | 20699002 | C0226802
SRT | T-49429 | Dodd perforating vein | 128554002 | C1267525
SRT | T-49410 | Femoral vein | 83419000 | C0015809
SRT | T-48820 | gastric vein | 110568007 | C0750610
SRT | T-48720 | hepatic vein | 8993003 | C0019155
SRT | T-4942A | Hunterian perforating vein | 128560002 | C1267526
SRT | T-48710 | Inferior Vena cava | 64131007 | C0042458
SRT | T-48620 | Innominate vein | 8887007 | C0006095
SRT | T-48170 | Internal jugular vein | 12123001 | C0226550
SRT | T-4884A | mesenteric vein | 128583004 | C0025473
SRT | T-48810 | portal vein | 32764006 | C0032718
SRT | T-49535 | posterior medial tributary | 128569001 | C1267527
SRT | T-48581 | Pulmonary vein | 122972007 | C0034090
SRT | D4-33512 | Pulmonary vein confluence | 128566008 | C1290492
SRT | T-48740 | Renal vein | 56400007 | C0035092
SRT | T-D930A | Saphenofemoral junction | 128587003 | C0447132
SRT | T-49530 | Saphenous vein | 60734001 | C0036186
SRT | T-48890 | splenic vein | 35819009 | C0038001
SRT | T-48330 | Subclavian vein | 9454009 | C0038532
SRT | T-48610 | Superior vena cava | 48345005 | C0042459
SRT | T-48832 | Umbilical vein | 284639000 | C0226734
SRT | T-48000 | Vein | 29092000 | C0042449
SRT | R-003AA | Vena anonyma | 371951007 | C1299265

### Note

In a prior version of this Context Group the code T-48500 rather than T-48581 was defined for the concept Pulmonary Vein; this was inconsistent with the DICOM approach of selecting the "structure of" rather than "entire" concept. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 3608 Atrial Source Locations

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.158

Table CID 3608. Atrial Source Locations

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-00203 | Baffle | 128981007 | C1289790
SRT | D4-31005 | Common atrium | 253276007 | C0392482
SRT | T-32330 | Coronary sinus | 31162003 | C0225863
SRT | D4-31052 | Juxtaposed appendage | 128563000 | C1290478
SRT | T-32300 | Left atrium | 82471001 | C0225860
SRT | G-DB27 | Pulmonary artery wedge | 128449009 | C1264742
SRT | G-DB26 | Pulmonary capillary wedge | 128448001 | C1264741
SRT | D4-33514 | Pulmonary venous atrium | 128567004 | C1290493
SRT | T-32190 | Pulmonary chamber in cor triatriatum | 128586007 | C0225841
SRT | T-32200 | Right Atrium | 73829009 | C0225844
SRT | D4-33516 | Systemic venous atrium | 128568009 | C1290494

## CID 3609 Ventricular Source Locations

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.159

Table CID 3609. Ventricular Source Locations

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D4-31120 | Common ventricle | 45503006 | C0152424
SRT | T-32600 | Left ventricle | 87878005 | C0225897
SRT | T-32602 | Left ventricle apex | 128564006 | C0580781
SRT | T-32640 | Left ventricle inflow | 70238003 | C0225911
SRT | T-32650 | Left ventricle outflow tract | 13418002 | C0225912
SRT | T-32500 | Right ventricle | 53085002 | C0225883
SRT | T-32502 | Right ventricle apex | 128565007 | C0445242
SRT | T-32540 | Right ventricle inflow | 8017000 | C0225891
SRT | T-32550 | Right ventricle outflow tract | 44627009 | C0225892

## CID 3610 Gradient Source Locations

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.160

Table CID 3610. Gradient Source Locations

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-35300 | Mitral Valve | 91134007 | C0026264
SRT | T-35400 | Aortic Valve | 34202007 | C0003501
SRT | T-35100 | Tricuspid valve | 46030003 | C0040960
SRT | T-35200 | Pulmonary valve | 39057004 | C0034086
SRT | T-44000 | Pulmonary artery | 81040000 | C0034052
SRT | T-32650 | Left ventricle outflow tract | 13418002 | C0225912
SRT | T-32550 | Right ventricle outflow tract | 44627009 | C0225892
SRT | D4-31150 | Ventricular Septal defect | 30288003 | C0018818
SRT | D4-31220 | Atrial Septal defect | 70142008 | C0018817
SRT | D4-32014 | Coarctation of aorta | 7305005 | C0003492

## CID 3611 Pressure Measurements

Type:

  Extensible

Version:

  20050322

UID:

  1.2.840.10008.6.1.161

Table CID 3611. Pressure Measurements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 109016 | A wave peak pressure |  |
DCM | 122196 | C wave pressure |  |
LN | 8462-4 | Intravascular diastolic blood pressure |  | C0488052
SRT | F-00E22 | Average diastolic blood pressure | 314453003 | C1282163
SRT | F-00E1F | Minimum diastolic blood pressure | 314451001 | C1282161
DCM | 122191 | Ventricular End Diastolic pressure |  |
DCM | 122197 | Gradient pressure, average |  |
DCM | 122198 | Gradient pressure, peak |  |
SRT | F-31150 | Mean blood pressure | 6797001 | C0428886
DCM | 122199 | Pressure at dp/dt max |  |
LN | 8480-6 | Intravascular Systolic Blood pressure |  | C0488055
SRT | F-00E14 | Average systolic blood pressure | 314440001 | C1282151
SRT | F-00E11 | Maximum systolic blood pressure | 314439003 | C1282150
DCM | 109034 | V wave peak pressure |  |
DCM | 122208 | x-descent pressure |  |
DCM | 122209 | y-descent pressure |  |
DCM | 122210 | z-point pressure |  |

## CID 3612 Blood Velocity Measurements

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.162

Table CID 3612. Blood Velocity Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122201 | Diastolic blood velocity, mean
DCM | 122202 | Diastolic blood velocity, peak
DCM | 122203 | Systolic blood velocity, mean
DCM | 122204 | Systolic blood velocity, peak
DCM | 122205 | Blood velocity, mean
DCM | 122206 | Blood velocity, minimum
DCM | 122207 | Blood velocity, peak

## CID 3613 Hemodynamic Time Measurements

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.163

Table CID 3613. Hemodynamic Time Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122182 | R-R interval
DCM | 109072 | Tau
DCM | 122211 | Left Ventricular ejection time
DCM | 122212 | Left Ventricular filling time
DCM | 122213 | Right Ventricular ejection time
DCM | 122214 | Right Ventricular filling time
DCM | 109071 | Indicator mean transit time

## CID 3614 Valve Areas, Non-mitral

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.164

Table CID 3614. Valve Areas, Non-mitral

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-0231F | Aortic Valve Area | 251011009 | C0428817
SRT | F-02321 | Pulmonic Valve Area | 251013007 | C0428819
SRT | F-02322 | Tricuspid Valve Area | 251014001 | C0428820
DCM | 122160 | Derived Non-Valve Area |  |

## CID 3615 Valve Areas

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.165

Table CID 3615. Valve Areas

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 3614 “Valve Areas, Non-mitral” |  |
SRT | F-02320 | Mitral Valve Area | 251012002 | C0221099

## CID 3616 Hemodynamic Period Measurements

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.166

Table CID 3616. Hemodynamic Period Measurements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-002D2 | Aortic Systolic Ejection Period (SEPa) | 371850007 | C1269855
SRT | R-0035C | Pulmonary Systolic Ejection Period (SEPp) | 371848004 | C1269854
SRT | R-0032C | Mitral Diastolic Filling Period (DFPm) | 371849007 | C1299325
SRT | R-003A9 | Tricuspid Diastolic Filling Period (DFPt) | 371847009 | C1299324
SRT | R-002F5 | Derived Period, Non-Valve | 371853009 | C1299328

## CID 3617 Valve Flows

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.167

Table CID 3617. Valve Flows

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-002D3 | Aortic Valve Flow | 371845001 | C1299322
SRT | R-0032D | Mitral Valve Flow | 371837006 | C1299464
SRT | R-0035D | Pulmonary Valve Flow | 371846000 | C1299323
SRT | R-00385 | Tricuspid Valve Flow | 371840006 | C1299467
SRT | R-00394 | Derived Flow, Non-Valve | 371839009 | C1299466

## CID 3618 Hemodynamic Flows

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.168

Table CID 3618. Hemodynamic Flows

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122161 | Pulmonary Flow
DCM | 122162 | Systemic Flow

## CID 3619 Hemodynamic Resistance Measurements

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.169

Table CID 3619. Hemodynamic Resistance Measurements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-03E86 | Pulmonary Vascular Resistance | 276901002 | C0456261
SRT | F-02B35 | Systemic Vascular Resistance | 386530009 | C1258192
DCM | 122215 | Total Pulmonary Resistance |  |
DCM | 122216 | Total Vascular Resistance |  |

## CID 3620 Hemodynamic Ratios

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.170

Table CID 3620. Hemodynamic Ratios

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
LN | 8581-1 | Tibial/brachial index |  | C0488220
SRT | F-0238B | Pulmonary/Systemic Flow Ratio | 251050008 | C0428854
DCM | 122217 | Coronary Flow reserve |  |
DCM | 122218 | Diastolic/Systolic velocity ratio |  |
DCM | 122219 | Hyperemic ratio |  |
SRT | F-031A2 | Pulsatility Index | 252068008 | C0429863
DCM | 122220 | Hemodynamic Resistance Index |  |
Include CID 3621 “Fractional Flow Reserve” |  |

## CID 3621 Fractional Flow Reserve

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.171

Table CID 3621. Fractional Flow Reserve

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-00307 | Fractional flow reserve | 371842003 | C1299469
SRT | R-00308 | Fractional Flow Reserve using intracoronary bolus | 371835003 | C1299462
SRT | R-00309 | Fractional Flow Reserve using intravenous infusion | 371841005 | C1299468

## CID 3627 Measurement Type

Type:

  Extensible

Version:

  20060613

UID:

  1.2.840.10008.6.1.172

Table CID 3627. Measurement Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-002E1 | Best value | 371912002 | C1299381
SRT | R-00317 | Mean | 373098007 | C1298794
SRT | R-00319 | Median | 373099004 | C1298795
SRT | R-0032E | Mode | 373100007 | C1298796
SRT | R-00355 | Point source measurement | 371913007 | C1299382
SRT | R-00353 | Peak to peak | 371914001 | C1299383
SRT | R-41D27 | Visual estimation | 258083009 | C0444684
SRT | R-10260 | Estimated | 414135002 | C0750572
SRT | R-41D2D | Calculated | 258090004 | C0444686
SRT | R-41D41 | Measured | 258104002 | C0444706

## CID 3628 Cardiac Output Methods

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.173

Table CID 3628. Cardiac Output Methods

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-002E5 | Thermal Bath | 371838001 | C1299465
SRT | R-002E7 | Thermal Inline | 371843008 | C1299320
SRT | R-002E6 | Dye Dilution | 373104003 | C1298799

## CID 3629 Procedure Intent

This Context Group specifies the intent for a procedure or a procedure step, depending on the context of invocation. The intent for a procedure step may be different than that of the procedure in which it occurs.

### Note

1. For example, a surgical biopsy procedure may have "Diagnostic Intent", while the imaging procedure step within that procedure may have "Guidance Intent".

2. Collection of specimens is generally "Diagnostic Intent"; "Forensic Intent" is typically used for autopsies; "Palliative Intent" and "Adjuvant Intent" may apply to certain radiotherapy procedures.

3. In SNOMED-CT, "Staging Intent" is a subsidiary concept (refinement) of "Diagnostic Intent". The following are subsidiary concepts of "Therapeutic Intent": Adjunct, Adjuvant, Curative, Neo-adjuvant, Prophylactic, and Supportive. Prophylactic is also a subsidiary concept of Preventive intent.

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.174

Table CID 3629. Procedure Intent

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-408C3 | Diagnostic Intent | 261004008 | C0348026
SRT | R-41531 | Therapeutic Intent | 262202000 | C0302350
SRT | R-002E9 | Combined Diagnostic and Therapeutic Procedure | 371931008 | C1299398
DCM | 113680 | Quality Control Intent |  |
SRT | R-408F2 | Staging intent | 373825000 | C1276306
SRT | R-40641 | Guidance Intent | 363675004 | C1285529
SRT | R-40642 | Forensic Intent | 363676003 | C1285530
SRT | R-42453 | Screening Intent | 360156006 | C1305399
SRT | R-40644 | Palliative Intent | 447295008 | C2960804
SRT | R-41564 | Adjunct intent | 421974008 | C1719882
SRT | R-41561 | Adjuvant intent | 373846009 | C1298675
SRT | R-41560 | Curative intent | 373808002 | C1276305
SRT | R-41562 | Neo-adjuvant intent | 373847000 | C1298676
SRT | R-41563 | Supportive intent | 399707004 | C1302630
SRT | P0-02179 | Preventive intent | 129428001 | C1456501
SRT | P0-021FD | Prophylactic intent | 360271000 | C0199176

## CID 3630 Cardiovascular Anatomic Locations

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.175

Table CID 3630. Cardiovascular Anatomic Locations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3606 “Arterial Source Locations”
Include CID 3607 “Venous Source Locations”
Include CID 3608 “Atrial Source Locations”
Include CID 3609 “Ventricular Source Locations”
Include CID 3610 “Gradient Source Locations”

## CID 3640 Hypertension

Type:

  Extensible

Version:

  20040920

UID:

  1.2.840.10008.6.1.176

Table CID 3640. Hypertension

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D3-40300 | Pulmonary hypertension | 70995007 | C0020542
SRT | D3-02000 | Systemic arterial hypertension | 38341003 | C0020538

## CID 3641 Hemodynamic Assessments

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.177

Table CID 3641. Hemodynamic Assessments

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-03E0D | Left Ventricular Systolic Pressure | 276780008 | C0456189
SRT | F-03E0E | Left Ventricular End-Diastolic Pressure | 276781007 | C0456190
SRT | F-0212C | Pulmonary Artery Pressure | 250767002 | C0428642
SRT | F-03E86 | Pulmonary Vascular Resistance | 276901002 | C0456261
SRT | F-31146 | Pulmonary Capillary Wedge Pressure | 118433006 | C0086879
SRT | F-03DFE | Right Ventricular Systolic Pressure | 276772001 | C0456181
SRT | F-03E02 | Right Ventricular End-Diastolic Pressure | 276774000 | C0456183
SRT | F-03DE9 | Right Atrial Pressure | 276755008 | C0456165
SRT | F-39790 | Vascular Resistance | 88619007 | C0042380
SRT | F-008ED | Diastolic Pressure | 271650006 | C0428883

## CID 3642 Degree Findings

Type:

  Extensible

Version:

  20040920

UID:

  1.2.840.10008.6.1.178

Table CID 3642. Degree Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A316 | Decreased | 1250004 | C0205216
SRT | G-A373 | Elevated | 75540009 | C3163633
SRT | G-A37A | Severely Elevated | 260360000 | C0442804
SRT | R-40765 | Normal Range | 260395002 | C0086715

## CID 3651 Hemodynamic Measurement Phase

This context group is a subset of CID 3250 “Catheterization Procedure Phase”.

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.179

Table CID 3651. Hemodynamic Measurement Phase

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-7293 | Cardiac catheterization baseline phase | 128955008 | C1292432
SRT | G-729B | Cardiac catheterization post contrast phase | 129083002 | C1292440
SRT | G-7298 | Cardiac catheterization post-intervention phase | 128960007 | C1292437
SRT | R-002E4 | Cardiac catheterization test/challenge phase | 373105002 | C1300063
SRT | R-002E3 | Cardiac catheterization gradient assessment phase | 371874005 | C1300078
SRT | P2-71317 | Drug Infusion Challenge | 133882006 | C1297891
SRT | P2-71310 | Exercise challenge | 128967005 | C1293901
SRT | F-01604 | Resting State | 128975004 | C0679218

## CID 3663 Body Surface Area Equations

Type:

  Extensible

Version:

  20100609

UID:

  1.2.840.10008.6.1.180

Table CID 3663. Body Surface Area Equations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122240 | BSA = 0.003207*WT^(0.7285-0.0188 log (WT)) *HT^0.3
DCM | 122241 | BSA = 0.007184*WT^ 0.425*HT^0.725
DCM | 122242 | BSA = 0.0235*WT^0.51456*HTcm^ 0.42246
DCM | 122243 | BSA = 0.024265*WT^0.5378*HTcm^0.3964
DCM | 122244 | BSA = (HT * WT/36) ^0.5
DCM | 122245 | BSA = 1321+0.3433*WT
DCM | 122246 | BSA = 0.0004688*WT^(0.8168-0.0154*log(WT))
DCM | 122266 | BSA = 0.007358*WT^ 0.425*HT^0.725
DCM | 122267 | BSA = 0.010265*WT^ 0.423*HT^0.651
DCM | 122268 | BSA = 0.008883*WT^ 0.444*HT^0.663
DCM | 122269 | BSA = 0.038189*WT^ 0.423*HT^0.362
DCM | 122270 | BSA = 0.009568*WT^ 0.473*HT^0.655

## CID 3664 Oxygen Consumption Equations and Tables

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.181

Table CID 3664. Oxygen Consumption Equations and Tables

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122247 | VO2male = BSA (138.1 - 11.49 * loge(age) + 0.378*HRf)
DCM | 122248 | VO2female = BSA (138.1 - 17.04 * loge(age) + 0.378*HRf)
DCM | 122249 | VO2 = VeSTPD * 10 * (FIO2 - FE02)
DCM | 122250 | VO2 = 152 * BSA
DCM | 122251 | VO2 = 175 * BSA
DCM | 122252 | VO2 = 176 * BSA
DCM | 122253 | Robertson & Reid table
DCM | 122254 | Fleisch table
DCM | 122255 | Boothby table

## CID 3666 P50 Equations

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.182

Table CID 3666. P50 Equations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122256 | if (prem age< 3days) P50 = 19.9
DCM | 122257 | if (age < 1day) P50 = 21.6
DCM | 122258 | if (age < 30day) P50 = 24.6
DCM | 122259 | if (age < 18y) P50 = 27.2
DCM | 122260 | if (age < 40y) P50 = 27.4
DCM | 122261 | if (age > 60y) P50 = 29.3

## CID 3667 Framingham Scores

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.183

Table CID 3667. Framingham Scores

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122230 | 10 Year CHD Risk
DCM | 122231 | Comparative Average10 Year CHD Risk
DCM | 122232 | Comparative Low10 Year CHD Risk

## CID 3668 Framingham Tables

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.184

Table CID 3668. Framingham Tables

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122233 | LDL Cholesterol Score Sheet for Men
DCM | 122234 | LDL Cholesterol Score Sheet for Women
DCM | 122235 | Total Cholesterol Score Sheet for Men
DCM | 122236 | Total Cholesterol Score Sheet for Women

## CID 3670 ECG Procedure Types

Type:

  Extensible

Version:

  20110330

UID:

  1.2.840.10008.6.1.185

Table CID 3670. ECG Procedure Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P2-3120A | 12-Lead ECG | 268400002 | C0430456
SRT | P2-3120E | 15-Lead ECG | 429163003 | C1998169
SRT | P2-3120C | 18-Lead ECG | 425808002 | C1961003

## CID 3671 Reason for ECG Exam

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.186

Table CID 3671. Reason for ECG Exam

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-00300 | Emergency procedure | 373110003 | C1298802
SRT | P1-00410 | Pre-Surgery testing | 110467000 | C1293092
SRT | R-00348 | Outpatient procedure | 371883000 | C1299353
SRT | R-0035A | Procedure in Cardiac Care Unit | 373111004 | C1298803
SRT | P2-10700 | Emergency Department patient visit | 4525004 | C0586082
SRT | R-00302 | Evaluation of murmur | 373112006 | C1298804
SRT | R-0036E | Routine procedure | 373113001 | C1298805

## CID 3672 Pacemakers

This Context Group includes the full set of codes for types of pacemakers specified in the NASPE/BPEG Generic Pacemaker Code (NBG). The Coding Scheme Designator (0008,0102) shall be NBG.

### Note

1. A prior version of this context group used codes from the SCP-ECG vocabulary.

2. Further information at http://www.hrsonline.org/Practice-Guidance/Clinical-Guidelines-Documents/2002-The-Revised-NASPE-BPEG-Generic-Code-for-Antibradycardia-AdaptiveRate-and-Multisite-Pacing. For reference, the scheme is reproduced here:

   Code Position | 1 - Chamber(s) paced | 2 - Chamber(s) sensed | 3 - Response to sensing | 4 - Rate modulation | 5- Multisite pacing
   Code values | O = None A = Atrium V = Ventricle D = Dual (A+V) S = Single(A or V - Mfr designation only) | O = None A = Atrium V = Ventricle D = Dual (A+V) S = Single(A or V - Mfr designation only) | O = None T = Triggered I = Inhibited D = Dual (T+I) | O = None R = Rate modulation | O = None A = Atrium V = Ventricle D = Dual (A+V)

## CID 3673 Diagnosis (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3675 Other Filters (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3676 Lead Measurement Technique

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.190

Table CID 3676. Lead Measurement Technique

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-002DA | Averaged | 371921001 | C1299390
SRT | R-0036D | Routine | 373115008 | C1298806
SRT | R-00319 | Median | 373099004 | C1298795
SRT | R-0036A | Representative | 371916004 | C1299385
SRT | R-00373 | Single Beats | 371871002 | C1299345

## CID 3677 Summary Codes ECG

Type:

  Extensible

Version:

  20111028

UID:

  1.2.840.10008.6.1.191

Table CID 3677. Summary Codes ECG

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-000B7 | Normal ECG | 164854000 | C0522054
SRT | F-38002 | Abnormal ECG | 102594003 | C0522055
SRT | F-38056 | Borderline Normal ECG | 251135002 | C0428951
SRT | F-38095 | ECG Equivocal | 370359005 | C0438155
DCM | 122753 | Non-diagnostic ECG |  |

### Note

A prior version of this context group used codes from the SCP-ECG vocabulary.

## CID 3678 QT Correction Algorithms

Type:

  Extensible

Version:

  20081029

UID:

  1.2.840.10008.6.1.192

Table CID 3678. QT Correction Algorithms

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122730 | Bazett QT Correction Algorithm
DCM | 122731 | Hodges QT Correction Algorithm
DCM | 122732 | Fridericia QTc Algorithm
DCM | 122733 | Framingham QTc Algorithm

### Note

A prior version of this context group used codes from the SCP-ECG vocabulary.

## CID 3679 ECG Morphology Descriptions (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3680 ECG Lead Noise Descriptions

This Context Group comprises the ECG noise annotations of ISO/IEEE 11073-10102. The terms included in the table below may not constitute the complete list; see the ISO/IEEE Standard.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

UID:

  1.2.840.10008.6.1.194

Table CID 3680. ECG Lead Noise Descriptions

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 10:11200 | No noise | MDC_ECG_NOISE_CLEAN
MDC | 10:11216 | Moderate noise | MDC_ECG_NOISE_MODERATE
MDC | 10:11232 | Severe noise | MDC_ECG_NOISE_SEVERE
MDC | 10:11248 | No signal | MDC_ECG_NOISE_NOSIGNAL

A prior version of this context group used codes from the SCP-ECG vocabulary.

## CID 3681 ECG Lead Noise Modifiers (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3682 Probability (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3683 Modifiers (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3684 Trend (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3685 Conjunctive Terms (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3686 ECG Interpretive Statements (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3687 Electrophysiology Waveform Durations

This Context Group consists of the per-lead terms under the hierarchy of Reference ID MDC_ECG_TIME_PD in the ISO/IEEE 11073-10102 nomenclature.

The base terms from that hierarchy are included in the table below for reference. The per-lead base terms are pre-coordinated with concept discriminators for specific leads, and the code values for those pre-coordinated terms are arithmetically derived from the code values of the base terms. For the complete current list of terms and discriminator values, see the ISO/IEEE Standard. All pre-coordinated terms (measurements plus discriminators) within the identified hierarchy are part of this Context Group.

### Note

1. A prior version of this context group used codes from the SCP-ECG coding system.

2. Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

UID:

  1.2.840.10008.6.1.201

Table CID 3687. Electrophysiology Waveform Durations

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:6656 | P duration, per lead | MDC_ECG_TIME_PD_P
MDC | 2:4608 | P onset to P1 duration, per lead | MDC_ECG_TIME_PD_P1
MDC | 2:4864 | P onset to P2 duration, per lead | MDC_ECG_TIME_PD_P2
MDC | 2:5120 | P onset to P3 duration, per lead | MDC_ECG_TIME_PD_P3
MDC | 2:7168 | P offset to QRS onset duration, per lead | MDC_ECG_TIME_PD_PR
MDC | 2:7680 | Q duration, per lead | MDC_ECG_TIME_PD_Q
MDC | 2:7936 | QRS duration, per lead | MDC_ECG_TIME_PD_QRS
MDC | 2:8192 | QT duration, per lead | MDC_ECG_TIME_PD_QT
MDC | 2:11264 | R1 duration, per lead | MDC_ECG_TIME_PD_R_1
MDC | 2:11520 | R2 duration, per lead | MDC_ECG_TIME_PD_R_2
MDC | 2:11776 | R3 duration, per lead | MDC_ECG_TIME_PD_R_3
MDC | 2:12032 | S1 duration, per lead | MDC_ECG_TIME_PD_S_1
MDC | 2:12288 | S2 duration, per lead | MDC_ECG_TIME_PD_S_2
MDC | 2:12544 | S3 duration, per lead | MDC_ECG_TIME_PD_S_3
MDC | 2:11008 | Ventricular activation time, per lead | MDC_ECG_TIME_PD_VENT_ACTIV
MDC | 2:32768 | PP time period, per lead | MDC_ECG_TIME_PD_PP
MDC | 2:33024 | RR time period, per lead | MDC_ECG_TIME_PD_RR
MDC | 2:33280 | PQ time period, per lead | MDC_ECG_TIME_PD_PQ
MDC | 2:33536 | PQ segment time period, per lead | MDC_ECG_TIME_PD_PQ_SEG
MDC | 2:34560 | QTU time period, per lead | MDC_ECG_TIME_PD_QTU

## CID 3688 Electrophysiology Waveform Voltages

This Context Group consists of the codes of the hierarchies under Reference IDs MDC_ECG_ELEC_POTL and MDC_ECG_AMPL of the ISO/IEEE 11073-10102 nomenclature.

The base terms from those hierarchies are included in the table below for reference. The per lead base terms are pre-coordinated with concept discriminators for specific leads, and the code values for those pre-coordinated terms are arithmetically derived from the code values of the base terms. For the complete current list of terms and discriminator values, see the ISO/IEEE Standard. All pre-coordinated terms (measurements plus discriminators) within the identified hierarchies are part of this Context Group.

### Note

1. A prior version of this context group used codes from the SCP-ECG coding system.

2. Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

UID:

  1.2.840.10008.6.1.202

Table CID 3688. Electrophysiology Waveform Voltages

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:1024 | J point Amplitude, per lead | MDC_ECG_AMPL_J
MDC | 2:14848 | Amplitude at 20 ms into ST segment, per lead | MDC_ECG_ELEC_POTL_ST_20
MDC | 2:15104 | Amplitude at 40 ms into ST segment, per lead | MDC_ECG_ELEC_POTL_ST_40
MDC | 2:14336 | Amplitude at 60 ms into ST segment, per lead | MDC_ECG_ELEC_POTL_ST_60
MDC | 2:14592 | Amplitude at 80 ms into ST segment, per lead | MDC_ECG_ELEC_POTL_ST_80
MDC | 2:1280 | P maximum amplitude, per lead | MDC_ECG_AMPL_P_MAX
MDC | 2:1536 | P minimum amplitude, per lead | MDC_ECG_AMPL_P_MIN
MDC | 2:3072 | P3 amplitude, per lead | MDC_ECG_AMPL_P3
MDC | 2:1792 | Q amplitude, per lead | MDC_ECG_AMPL_Q
MDC | 2:2048 | R amplitude, per lead | MDC_ECG_AMPL_R
MDC | 2:12800 | R1 amplitude, per lead | MDC_ECG_ELEC_POTL_R_1
MDC | 2:13056 | R2 amplitude, per lead | MDC_ECG_ELEC_POTL_R_2
MDC | 2:13312 | R3 amplitude, per lead | MDC_ECG_ELEC_POTL_R_3
MDC | 2:2304 | S amplitude, per lead | MDC_ECG_AMPL_S
MDC | 2:13568 | S1 amplitude, per lead | MDC_ECG_ELEC_POTL_S_1
MDC | 2:13824 | S2 amplitude, per lead | MDC_ECG_ELEC_POTL_S_2
MDC | 2:14080 | S3 amplitude, per lead | MDC_ECG_ELEC_POTL_S_3
MDC | 2:2560 | T maximum amplitude, per lead | MDC_ECG_AMPL_T_MAX
MDC | 2:2816 | T minimum amplitude, per lead | MDC_ECG_AMPL_T_MIN
MDC | 2:768 | ST amplitude, per lead | MDC_ECG_AMPL_ST

## CID 3689 ECG Global Waveform Durations

This Context Group consists of the global terms under the hierarchy of Reference ID MDC_ECG_TIME_PD in the ISO/IEEE 11073-10102 nomenclature.

The base terms from that hierarchy are included in the table below for reference. The base terms may be pre-coordinated with concept discriminators, and the code values for those pre-coordinated terms are arithmetically derived from the code values of the base terms. For the complete current list of terms and discriminator values, see the ISO/IEEE Standard. All pre-coordinated terms (measurements plus discriminators) within the identified hierarchy are part of this Context Group.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

UID:

  1.2.840.10008.6.1.927

Table CID 3689. ECG Global Waveform Durations

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:16184 | P duration, global | MDC_ECG_TIME_PD_P_GL
MDC | 2:16140 | PP time period, global | MDC_ECG_TIME_PD_PP_GL
MDC | 2:16144 | PQ time period, global | MDC_ECG_TIME_PD_PQ_GL
MDC | 2:15872 | PR time period, global | MDC_ECG_TIME_PD_PR_GL
MDC | 2:16148 | PQ segment time period, global | MDC_ECG_TIME_PD_PQ_SEG_GL
MDC | 2:16156 | QRS duration, global | MDC_ECG_TIME_PD_QRS_GL
MDC | 2:16160 | QT duration, global | MDC_ECG_TIME_PD_QT_GL
MDC | 2:16000 | RR time period, global | MDC_ECG_TIME_PD_RR_GL
MDC | 2:16004 | QTU time period, global | MDC_ECG_TIME_PD_QTU_GL

## CID 3690 ECG Control Variables Numeric

This Context Group includes the ECG control variables specified in the ISO/IEEE 11073-10102 nomenclature that take numeric values. The terms are included in the table below for reference; these may not constitute the complete current list (see the ISO/IEEE Standard).

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

UID:

  1.2.840.10008.6.1.895

Table CID 3690. ECG Control Variables Numeric

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 10:11393 | Sample rate | MDC_ECG_CTL_VBL_SAMPLE_RATE
MDC | 10:11394 | Sensitivity | MDC_ECG_CTL_VBL_SENSITIVITY
MDC | 10:11395 | Zero offset | MDC_ECG_CTL_VBL_ZERO_OFFSET
MDC | 10:11397 | Pad value | MDC_ECG_CTL_VBL_PAD_VALUE
MDC | 10:11398 | Time skew | MDC_ECG_CTL_VBL_TIME_SKEW
MDC | 10:11399 | Sample skew | MDC_ECG_CTL_VBL_SAMPLE_SKEW
MDC | 10:11400 | Time offset | MDC_ECG_CTL_VBL_TIME_OFFSET
MDC | 10:11403 | Low pass filter cutoff freq | MDC_ECG_CTL_VBL_ATTR_FILTER_CUTOFF_FREQ
MDC | 10:11408 | Notch filter frequency | MDC_ECG_CTL_VBL_ATTR_FILTER_NOTCH_FREQ
MDC | 10:11409 | Notch filter bandwidth | MDC_ECG_CTL_VBL_ATTR_FILTER_NOTCH_BANDWIDTH
MDC | 10:11418 | Interpolator SNR | MDC_ECG_CTL_VBL_INTERPOLATOR_SNR

## CID 3691 ECG Control Variables Text

This Context Group includes the ECG control variables specified in the ISO/IEEE 11073-10102 nomenclature that take text or coded values. The terms are included in the table below for reference; these may not constitute the complete current list (see the ISO/IEEE Standard).

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

UID:

  1.2.840.10008.6.1.896

Table CID 3691. ECG Control Variables Text

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 10:11402 | Low pass filter | MDC_ECG_CTL_VBL_ATTR_FILTER_LOW_PASS
MDC | 10:11404 | High pass filter | MDC_ECG_CTL_VBL_ATTR_FILTER_HIGH_PASS
MDC | 10:11406 | High pass filter description | MDC_ECG_CTL_VBL_ATTR_FILTER_DESCRIPTION
MDC | 10:11407 | Notch filter | MDC_ECG_CTL_VBL_ATTR_FILTER_NOTCH
MDC | 10:11410 | Notch filter description | MDC_ECG_CTL_VBL_ATTR_FILTER_NOTCH _DESCRIPTION
MDC | 10:11412 | Baseline description | MDC_ECG_CTL_VBL_BASELINE_DESC
MDC | 10:11414 | Interpolator | MDC_ECG_CTL_VBL_INTERPOLATOR
MDC | 10:11416 | Interpolator description | MDC_ECG_CTL_VBL_INTERPOLATOR_DESC

## CID 3692 ICDs

This Context Group includes the full set of codes for types of implanted cardioverter/defibrillators (ICDs) specified in the NASPE/BPEG Defibrillator Code (NBD). The Coding Scheme Designator (0008,0102) shall be NBD.

### Note

Further information at http://www.hrsonline.org/News/ep-history/topics-in-depth/modecodehistory.cfm. For reference, the scheme is reproduced here:

Code Position | Shock chamber | Antitachycardia pacing chamber | Tachycardia detection | Antibradycardia pacing chamber
Code values | O = None A = Atrium V = Ventricle D = Dual (A+V) | O = None A = Atrium V = Ventricle D = Dual (A+V) | E = Electrogram H = Hemodynamic | O = None A = Atrium V = Ventricle D = Dual (A+V)

Short Form
ICD-S = ICD with shock capability only ICD-B = ICD with bradycardia pacing as well as shock ICD-T = ICD with tachycardia (and bradycardia) pacing as well as shock

## CID 3700 Cath Diagnosis

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.203

Table CID 3700. Cath Diagnosis

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D3-13040 | Coronary artery disease | 53741008 | C1956346
SRT | D3-15100 | Acute myocardial infarction | 57054005 | C0155626
SRT | F-37012 | Atypical chest pain | 102589003 | C0262384
SRT | D3-13020 | Stable Angina | 233819005 | C0340288
SRT | D3-12400 | Atypical Angina, Variant Angina | 87343002 | C0002963
SRT | D3-12700 | Unstable Angina, Progressive Angina | 4557003 | C0002965
SRT | D3-13014 | Post-infarction angina | 314116003 | C1278535
SRT | R-00368 | Recurrent angina Post-PTCA | 371808007 | C1299436
SRT | R-00367 | Recurrent angina Post-DCA | 371812001 | C1299440
SRT | R-00369 | Recurrent angina Post-Rotational Atherectomy | 371811008 | C1299439
SRT | R-00366 | Recurrent angina Post-Stent | 371809004 | C1299437
SRT | R-00365 | Recurrent angina Post-CABG | 371810009 | C1299438
SRT | D3-16010 | Congestive heart failure | 42343007 | C0018802
SRT | D2-61100 | Pulmonary edema | 19242006 | C0034063
SRT | D3-00200 | cardiogenic shock | 89138009 | C0036980
SRT | R-002CB | Acute ventricular septal rupture | 371817007 | C1299445
SRT | D3-29010 | Mitral valve disease | 11851006 | C0026265
SRT | D3-29011 | Mitral stenosis | 79619009 | C0026269
SRT | D3-29012 | Mitral regurgitation | 48724000 | C0026266
SRT | D3-29096 | Acute mitral regurgitation | 373116009 | C1298807
SRT | D3-13021 | Silent ischemia | 233823002 | C0340291
SRT | R-00336 | s/p MI positive stress for ischemia | 371824008 | C1300077
SRT | D3-26000 | Myocarditis | 50920009 | C0027059
SRT | D3-28102 | Subacute bacterial endocarditis | 73774007 | C0014122
SRT | D3-2906A | Idiopathic hypertrophic subaortic stenosis | 360465008 | C0700053
SRT | D3-40300 | Pulmonary hypertension | 70995007 | C0020542
SRT | D3-29040 | Tricuspid valve disease | 20721001 | C0264882
SRT | D3-29042 | Tricuspid regurgitation | 111287006 | C0040961
SRT | D3-1081C | Mitral valve prolapse | 409712001 | C0026267
SRT | D3-31700 | Ventricular tachycardia | 25569003 | C0042514
SRT | D3-31720 | Ventricular fibrillation | 71908006 | C0042510
SRT | D3-20021 | Congestive cardiomyopathy | 399020009 | C0007193
SRT | D3-02500 | Hypertensive heart disease | 64715009 | C0152105
SRT | D3-22100 | Restrictive cardiomyopathy | 90828009 | C0007196
SRT | D3-90000 | Pericardial disease | 55855009 | C0265122
SRT | D3-90100 | Pericardial tamponade | 35304003 | C0007177
SRT | D3-29020 | Aortic valve disease | 8722008 | C1260873
SRT | D3-29021 | Aortic stenosis | 60573004 | C0003507
SRT | D3-29025 | Aortic insufficiency | 194983005 | C0340377
SRT | D4-31220 | Atrial septal defect | 70142008 | C0018817
SRT | D3-80016 | Aortic dissection | 308546005 | C0340643
SRT | D3-29050 | Pulmonic valve disease | 76267008 | C0034087
SRT | D4-31150 | Ventricular septal defect | 30288003 | C0018818
SRT | D3-83300 | Aortic aneurysm | 67362008 | C0003486
SRT | R-FAE6C | Arrhythmia | 698247007 | C0003811
SRT | D3-31520 | Atrial fibrillation | 49436004 | C0004238
SRT | D4-31000 | heart disease, congenital | 13213009 | C0152021
SRT | D3-91030 | Constrictive pericarditis | 85598007 | C0031048

## CID 3701 Cardiac Valves and Tracts

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.204

Table CID 3701. Cardiac Valves and Tracts

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-35300 | Mitral Valve | 91134007 | C0026264
SRT | T-35400 | Aortic Valve | 34202007 | C0003501
SRT | T-35100 | Tricuspid valve | 46030003 | C0040960
SRT | T-35200 | Pulmonary valve | 39057004 | C0034086
SRT | T-32650 | Left ventricle outflow tract | 13418002 | C0225912

## CID 3703 Wall Motion

Type:

  Extensible

Version:

  20111028

UID:

  1.2.840.10008.6.1.205

Table CID 3703. Wall Motion

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-00378 | Not Evaluated | 373121007 | C1298808
SRT | R-41198 | Unknown | 261665006 | C0439673
DCM | 122288 | Not visualized |  |
SRT | R-00344 | Normal wall motion | 373122000 | C1298809
SRT | R-0030D | Hyperkinetic region | 373123005 | C1298810
SRT | F-32056 | Hypokinesis | 37706002 | C0232172
SRT | R-00327 | Mild Hypokinesis | 371868005 | C1299342
SRT | R-0032F | Moderate Hypokinesis | 371869002 | C1299343
SRT | R-00370 | Severe Hypokinesis | 371870001 | C1299344
SRT | F-30004 | Akinesis | 195675009 | C0232171
SRT | F-32052 | Dyskinesis | 25437005 | C0232168

### Note

In prior editions, this Context Group included incorrect codes for "Hypokinesis" and "Mild Hypokinesis" (see PS3.16-2011).

## CID 3704 Myocardium Wall Morphology Findings

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.206

Table CID 3704. Myocardium Wall Morphology Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 122112 | Normal Myocardium |  |
SRT | D3-10510 | Ventricular Aneurysm | 90539001 | C0392464
DCM | 122113 | Scarred Myocardium |  |
DCM | 122114 | Thinning Myocardium |  |

## CID 3705 Chamber Size

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.207

Table CID 3705. Chamber Size

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-00343 | Normal size cardiac chamber | 373124004 | C1298811
SRT | R-002C6 | Abnormally small cardiac chamber | 373125003 | C1298812
SRT | R-0032A | Mildly Enlarged cardiac chamber | 373126002 | C1298813
SRT | R-00331 | Moderately Enlarged cardiac chamber | 373127006 | C1298814
SRT | R-00316 | Markedly Enlarged cardiac chamber | 373128001 | C1298815

## CID 3706 Overall Contractility

Type:

  Extensible

Version:

  20111028

UID:

  1.2.840.10008.6.1.208

Table CID 3706. Overall Contractility

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-00341 | Normal wall contractility | 373129009 | C1298816
SRT | R-00398 | Hyperkinesis | 371855002 | C1299330
SRT | F-32056 | Hypokinesis | 37706002 | C0232172
SRT | F-30004 | Akinesis | 195675009 | C0232171

## CID 3707 VSD Description

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.209

Table CID 3707. VSD Description

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D4-31154 | Membranous | 94150003 | C0685706
SRT | R-0033B | Non-restrictive | 373131000 | C1298817
SRT | D4-31166 | Restrictive | 253551005 | C0344924
SRT | R-40775 | None | 260413007 | C0549184

## CID 3709 Aortic Root Description

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.210

Table CID 3709. Aortic Root Description

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-0033C | Normal Aortic Root | 373132007 | C1298818
SRT | R-00301 | Enlarged Aortic Root | 373133002 | C1298819
SRT | R-002CD | Aneurysm of Aortic Root | 373134008 | C1298820
SRT | R-002D1 | Annular Abscess of Aortic Root | 373135009 | C1298821
SRT | R-003A1 | Post Stenotic Dilation | 371872009 | C1299346
SRT | D3-83660 | Ruptured Sinus of Valsalva | 21379009 | C0265019

## CID 3710 Coronary Dominance

Type:

  Extensible

Version:

  20111028

UID:

  1.2.840.10008.6.1.211

Table CID 3710. Coronary Dominance

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D4-3252C | Left Coronary Dominance | 253729004 | C0345136
SRT | D4-3252B | Right Coronary Dominance | 253728007 | C0345135
SRT | D4-3252D | Balanced Coronary Dominance | 253730009 | C0345137

### Note

In prior editions, this Context Group included NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes (see PS3.16-2011).

## CID 3711 Valvular Abnormalities

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.212

Table CID 3711. Valvular Abnormalities

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D3-29001 | Stenosis | 44241007 | C0264878
SRT | F-32400 | Regurgitation | 10337008 | C0042300
SRT | R-0030B | Calcified Heart Valve | 373136005 | C1142152
SRT | R-0030F | Immobile Heart Valve | 373137001 | C1298822
DCM | 127856 | Heart Valve Flail |  |
SRT | D3-28005 | Valvular endocarditis | 89736004 | C0264865

## CID 3712 Vessel Descriptors

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.213

Table CID 3712. Vessel Descriptors

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-00389 | Ulcerated | 373138006 | C1298823
SRT | R-0036B | Restenotic | 371893007 | C1299362
SRT | R-002E2 | Bifurcation | 371894001 | C1299363
SRT | R-002EF | Culprit | 371895000 | C1299364
SRT | R-40411 | Aneurysmal | 255378009 | C0439651
SRT | R-002FC | Diffuse Disease | 371915000 | C1299384
SRT | R-00314 | Luminal Irregularities | 371873004 | C1299347
SRT | D4-31B68 | Muscle Bridge | 424045003 | C1827939
SRT | R-10050 | Stenotic | 386139002 | C1272588
SRT | R-10051 | Ectatic | 386140000 | C1272589
SRT | D6-34737 | Calcified | 237897009 | C0342649
SRT | M-35001 | Thrombus | 396339007 | C0087086
SRT | R-10048 | Tortuous | 386137000 | C1272586
SRT | R-10049 | Stented | 386138005 | C1272587

## CID 3713 TIMI Flow Characteristics

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.214

Table CID 3713. TIMI Flow Characteristics

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR Equivalent
--- | --- | --- | --- | --- | ---
SRT | R-0037E | 0: No Perfusion | 371867000 | C1299341 | 106-0, 107-0
SRT | R-0037F | 1: Penetration without Perfusion | 371866009 | C1299340 | 106-1, 107-1
SRT | R-00381 | 2: Partial Perfusion | 371864007 | C1299338 | 106-2, 107-2
SRT | R-00382 | 3: Complete Perfusion | 371865008 | C1299339 | 106-3, 107-3

## CID 3714 Thrombus

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.215

Table CID 3714. Thrombus

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-0033A | No Thrombus | 373140001 | C1276764
SRT | R-00356 | Possible Thrombus | 373141002 | C1298825
SRT | R-002F1 | Definite Thrombus | 373142009 | C1298826
SRT | R-00371 | Severe Thrombus | 373143004 | C1298827

## CID 3715 Lesion Margin

Type:

  Extensible

Version:

  20050614

UID:

  1.2.840.10008.6.1.216

Table CID 3715. Lesion Margin

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A545 | Smooth | 82280004 | C0205357
SRT | G-A402 | Irregular | 49608001 | C0205271
SRT | R-00335 | Multiple Irregularities | 371922008 | C1299391
SRT | R-403CC | Ulcerative | 255321001 | C0041582

## CID 3716 Severity

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.217

Table CID 3716. Severity

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-40775 | None | 260413007 | C0549184
SRT | R-404FA | Mild | 255604002 | C2945599
SRT | R-00329 | Mild to Moderate | 371923003 | C1299392
SRT | G-A002 | Moderate | 6736007 | C0205081
SRT | R-00330 | Moderate to Severe | 371924009 | C1299393
SRT | G-A003 | Severe | 24484000 | C0205082
SRT | R-4099D | Fatal | 399166001 | C1302234

## CID 3717 Myocardial Wall Segments

This 17-segment model of left ventricular myocardial wall segments uses the terminology specified in " AHA Scientific Statement: Standardized Myocardial Segmentation and Nomenclature for Tomographic Imaging of the Heart"(see Section 2).

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.218

Table CID 3717. Myocardial Wall Segments

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-32619 | left ventricle basal anterior segment | 264850008 | C0555926
SRT | R-10075 | left ventricle basal anteroseptal segment | 396482007 | C1300766
SRT | R-10076 | left ventricle basal inferoseptal segment | 396646008 | C1300903
SRT | T-32615 | left ventricle basal inferior segment | 264846001 | C0555929
SRT | R-10079 | left ventricle basal inferolateral segment | 396652009 | C1300909
SRT | R-1007A | left ventricle basal anterolateral segment | 396654005 | C1300911
SRT | T-32617 | left ventricle mid anterior segment | 264848000 | C0555925
SRT | R-10077 | left ventricle mid anteroseptal segment | 396647004 | C1300904
SRT | R-10078 | left ventricle mid inferoseptal segment | 396649001 | C1300906
SRT | T-32616 | left ventricle mid inferior segment | 264847005 | C0555924
SRT | R-1007B | left ventricle mid inferolateral segment | 396655006 | C1300912
SRT | R-1007C | left ventricle mid anterolateral segment | 396656007 | C1300913
SRT | T-32613 | left ventricle apical anterior segment | 264844003 | C0555922
SRT | T-32614 | left ventricle apical septal segment | 264845002 | C0555923
SRT | T-32618 | left ventricle apical inferior segment | 264849008 | C0555930
SRT | T-3261C | left ventricle apical lateral segment | 264853005 | C0555928
SRT | T-32602 | apex of left ventricle | 128564006 | C0580781

## CID 3718 Myocardial Wall Segments in Projection

This context group specifies the left ventricular myocardial wall segments as seen in typical right anterior oblique (RAO) and left anterior oblique (LAO) angiographic projections.

Type:

  Extensible

Version:

  20030614

UID:

  1.2.840.10008.6.1.219

Table CID 3718. Myocardial Wall Segments in Projection

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-32619 | left ventricle basal anterior segment | 264850008 | C0555926
SRT | T-32634 | myocardium of anterolateral region | 73050001 | C0225907
SRT | T-32636 | myocardium of apex of heart | 47962008 | C0225909
SRT | T-32632 | myocardium of diaphragmatic region | 72542009 | C0225905
SRT | T-32615 | left ventricle basal inferior segment | 264846001 | C0555929
SRT | T-32603 | left ventricle basal lateral segment | 277631004 | C0559192
SRT | T-32633 | myocardium of posterolateral region | 33272004 | C0225906
SRT | T-32637 | myocardium of inferolateral region | 16239001 | C0225910
SRT | T-32614 | left ventricle apical septal segment | 264845002 | C0555923
SRT | T-32601 | left ventricular basal septal segment | 277630003 | C0559191
SRT | R-101C0 | left ventricular posterobasal segment | 408720008 | C1443269

## CID 3719 Canadian Clinical Classification

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.220

Table CID 3719. Canadian Clinical Classification

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR [2.0b] Equivalent
--- | --- | --- | --- | --- | ---
SRT | F-A265A | Chest pain not present | 161971004 | C0423635 | 50-0
SRT | D3-12001 | Angina Class I | 61490001 | C0264675 | 50-I
SRT | D3-12002 | Angina Class II | 41334000 | C0264676 | 50-II
SRT | D3-12003 | Angina Class III | 85284003 | C0264677 | 50-IIII
SRT | D3-12004 | Angina Class IV | 89323001 | C0264678 | 50-IV

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3720 Cardiac History Dates (Retired)

This Context Group is retired. See PS3.16-2007.

## CID 3721 Cardiovascular Surgeries

Type:

  Extensible

Version:

  20081027

UID:

  1.2.840.10008.6.1.222

Table CID 3721. Cardiovascular Surgeries

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR [2.0b] Equivalent
--- | --- | --- | --- | --- | ---
SRT | R-102B4 | Percutaneous coronary intervention | 415070008 | C1532338 | 40
SRT | P1-3301A | Coronary artery bypass graft | 232717009 | C0010055 | 42
SRT | P1-32000 | Operation on heart valve | 73544002 | C0190065 | 44
SRT | P1-31C03 | Ablation operation for arrhythmia | 233159005 | C0397403 |
SRT | P0-004BA | Implantation of cardiac pacemaker | 307280005 | C0189842 |
SRT | P1-3157D | Implantation of automatic cardiac defibrillator | 233170003 | C0397417 |
SRT | P1-0555A | Abdominal aortic aneurysm stenting | 307701005 | C0585569 |
SRT | P1-31D00 | Heart transplant | 32413006 | C0018823 |
SRT | P1-080B4 | Correction of congenital cardiovascular deformity | 428613004 | C1997888 |

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3722 Diabetic Therapy

Type:

  Extensible

Version:

  20141103

UID:

  1.2.840.10008.6.1.223

Table CID 3722. Diabetic Therapy

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-02F14 | Diabetic on Dietary Treatment | 170745003 | C0421246
SRT | F-02F15 | Diabetic on Oral Treatment | 170746002 | C0421247
SRT | F-02F16 | Diabetic on Insulin | 170747006 | C0421248

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3723 MI Types

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.224

Table CID 3723. MI Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR [2.0b] Equivalent
--- | --- | --- | --- | --- | ---
SRT | D3-1511A | Non ST Elevation Myocardial Infarction | 401314000 | C1276061 | 94-1
SRT | D3-15119 | ST Elevation Myocardial Infarction | 401303003 | C1303258 | 94-2

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3724 Smoking History

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.225

Table CID 3724. Smoking History

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR [2.0b] Equivalent
--- | --- | --- | --- | --- | ---
SRT | F-9321F | No History of Smoking | 266919005 | C0425293 | 38-0
SRT | S-32000 | Current Smoker | 77176002 | C3241966 | 38-1
SRT | S-32070 | Former Smoker | 8517006 | C0337671 | 38-2

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3726 Indications for Coronary Intervention

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.226

Table CID 3726. Indications for Coronary Intervention

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR Equivalent
--- | --- | --- | --- | --- | ---
DCM | 122171 | Coronary lesion > = 50% stenosis |  |  |
SRT | D3-00200 | Cardiogenic Shock | 89138009 | C0036980 | 123

## CID 3727 Indications for Catheterization

Type:

  Extensible

Version:

  20111028

UID:

  1.2.840.10008.6.1.227

Table CID 3727. Indications for Catheterization

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D3-00200 | cardiogenic shock | 89138009 | C0036980
SRT | D3-10800 | valvular heart disease | 368009 | C0018824
SRT | D3-30000 | Arrhythmia | 44808001 | C0264886
SRT | D3-10030 | ischemic heart disease | 414545008 | C0151744
SRT | F-000FF | cardiac function test abnormal | 165076002 | C0438177
SRT | P1-31D00 | heart transplant | 32413006 | C0018823
SRT | D4-31000 | heart disease - congenital | 13213009 | C0152021
SRT | D3-20000 | cardiomyopathy | 85898001 | C0878544
SRT | D3-10000 | heart disease | 56265001 | C0018799

### Note

1. (D3-10000, SRT, "heart disease") should be used only when a more specific characterization of the disease is not applicable.

2. In prior editions, this Context Group included NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes (see PS3.16-2011).

## CID 3728 Cath Findings

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.228

Table CID 3728. Cath Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-0033F | Normal left heart hemodynamics | 371856001 | C1299331
SRT | R-00342 | Normal right heart hemodynamics | 371859008 | C1299334
SRT | R-0033E | Normal left and right heart hemodynamics | 371858000 | C1299333
SRT | R-00340 | Normal left ventricular systolic function and wall motion | 371857005 | C1299332
SRT | R-0033D | Normal coronary arteries | 371860003 | C1299335
SRT | R-00328 | Mild intimal coronary irregularities, no significant stenoses | 371861004 | C1299336
SRT | D3-13001 | Single vessel coronary artery disease. | 194842008 | C0581374
SRT | D3-13013 | Double vessel coronary artery disease. | 194843003 | C0581375
SRT | D3-1301F | Triple vessel coronary artery disease. | 233817007 | C0340285
SRT | R-00334 | Multi vessel coronary artery disease. | 371803003 | C1299432
SRT | R-00313 | Left main coronary artery disease | 371804009 | C1299433
SRT | R-00372 | Significant coronary bypass graft disease | 371805005 | C1299434
SRT | D3-29021 | Aortic stenosis | 60573004 | C0003507
SRT | D3-29025 | Aortic insufficiency | 194983005 | C0340377
SRT | D3-29011 | Mitral stenosis | 79619009 | C0026269
SRT | D3-29012 | Mitral regurgitation | 48724000 | C0026266
SRT | R-002F3 | Depression of left ventricular systolic function | 371862006 | C1299337
SRT | R-002C8 | Acute mitral regurgitation from chordal rupture | 371813006 | C1299441
SRT | R-002C7 | Acute mitral regurgitation from chordal dysfunction | 371814000 | C1299442
SRT | R-002CA | Acute mitral regurgitation from papillary muscle rupture | 371816003 | C1299444
SRT | R-002C9 | Acute mitral regurgitation from papillary muscle dysfunction | 371815004 | C1299443
SRT | D3-1081C | Mitral valve prolapse | 409712001 | C0026267
SRT | D3-20021 | Congestive cardiomyopathy | 399020009 | C0007193
SRT | D3-23000 | Hypertrophic cardiomyopathy with obstruction | 45227007 | C0007194
SRT | D3-20003 | Hypertrophic cardiomyopathy without obstruction | 195020003 | C0340425
SRT | D3-02500 | Hypertensive heart disease | 64715009 | C0152105
SRT | D3-22100 | Restrictive cardiomyopathy | 90828009 | C0007196
SRT | D3-90100 | Pericardial tamponade | 35304003 | C0007177
SRT | D3-91030 | Constrictive pericarditis | 85598007 | C0031048
SRT | D3-40300 | Pulmonary hypertension | 70995007 | C0020542
SRT | D4-31220 | Atrial septal defect | 70142008 | C0018817
SRT | D4-31150 | Ventricular septal defect | 30288003 | C0018818
SRT | R-002CB | Acute ventricular septal rupture | 371817007 | C1299445
SRT | D4-31000 | heart disease - congenital | 13213009 | C0152021

## CID 3729 Admission Status

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.229

Table CID 3729. Admission Status

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR [2.0b] Equivalent
--- | --- | --- | --- | --- | ---
SRT | P0-10010 | Elective | 8715000 | C0184667 | 17-1
SRT | P0-10800 | Emergency Department | 50849002 | C0583237 | 17-2
SRT | P0-10210 | Transfer | 4563007 | C0184681 | 17-3

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3730 Insurance Payor

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.230

Table CID 3730. Insurance Payor

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 18-1 | Government
NCDR | 2.0b | 18-2 | Commercial
NCDR | 2.0b | 18-3 | Health Maintenance Organization
NCDR | 2.0b | 18-4 | None

## CID 3733 Primary Cause of Death

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.231

Table CID 3733. Primary Cause of Death

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 25-1 | Cardiac
NCDR | 2.0b | 25-2 | Neurologic
NCDR | 2.0b | 25-3 | Renal
NCDR | 2.0b | 25-4 | Vascular
NCDR | 2.0b | 25-5 | Infection
NCDR | 2.0b | 25-6 | Pulmonary
NCDR | 2.0b | 25-7 | Valvular
NCDR | 2.0b | 25-8 | Other

## CID 3735 Acute Coronary Syndrome Time Period

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.232

Table CID 3735. Acute Coronary Syndrome Time Period

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 51-1 | < = 6 hours
NCDR | 2.0b | 51-2 | between 6 hours and 24 hours
NCDR | 2.0b | 51-3 | between 24 hours and 7 days

## CID 3736 NYHA Classification

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.233

Table CID 3736. NYHA Classification

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR [2.0b] Equivalent
--- | --- | --- | --- | --- | ---
SRT | F-3018B | NYHA Class I | 420300004 | C1319793 | 47-I
SRT | F-3018C | NYHA Class II | 421704003 | C1319794 | 47-II
SRT | F-3018D | NYHA Class III | 420913000 | C1319795 | 47-III
SRT | F-3018E | NYHA Class IV | 422293003 | C1319796 | 47-IV

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3737 Non-invasive Test - Ischemia

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.234

Table CID 3737. Non-invasive Test - Ischemia

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 48-1 | Not Done
NCDR | 2.0b | 48-2 | Positive
NCDR | 2.0b | 48-3 | Negative
NCDR | 2.0b | 48-4 | Equivocal
NCDR | 2.0b | 48-5 | Arrhythmia

## CID 3738 Pre-Cath Angina Type

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.235

Table CID 3738. Pre-Cath Angina Type

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 49-I | Atypical Chest Pain
NCDR | 2.0b | 49-II | Stable Angina
NCDR | 2.0b | 49-IIIa | Acute Coronary Syndrome: Unstable Angina
NCDR | 2.0b | 49-IIIb | Acute Coronary Syndrome: Non ST-Elevation Myocardial Infarction
NCDR | 2.0b | 49-IIIc | Acute Coronary Syndrome: ST-Elevation Myocardial Infarction

## CID 3739 Cath Procedure Type

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.236

Table CID 3739. Cath Procedure Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR Equivalent
--- | --- | --- | --- | --- | ---
SRT | P1-31602 | Catheterization of right heart | 40403005 | C0189896 | 54-1
SRT | P1-31604 | Catheterization of left heart | 67629009 | C0189897 | 54-2
SRT | P1-3160A | Catheterization of both left and right heart with graft | 128952006 | C1293383 |
SRT | P1-3160B | Catheterization of both left and right heart without graft | 128953001 | C1293384 |
DCM | 122061 | Percutaneous Coronary Intervention |  |  | 54-3

## CID 3740 Thrombolytic Administration

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.237

Table CID 3740. Thrombolytic Administration

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR Equivalent
--- | --- | --- | --- | --- | ---
SRT | R-0037D | Contraindicated | 373148008 | C1276287 | 57-1
SRT | R-0037C | Administered less than 3 hours before PCI | 371896004 | C1299365 | 57-2
SRT | R-0037A | Administered between 3 and 6 hours before PCI | 371897008 | C1299366 | 57-3
SRT | R-0037B | Administered between 6 hours and 7 days before PCI | 371906007 | C1299375 | 57-4

## CID 3741 Medication Administration, Lab Visit

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.238

Table CID 3741. Medication Administration, Lab Visit

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR Equivalent
--- | --- | --- | --- | --- | ---
SRT | R-00321 | Contraindicated | 373147003 | C1298831 | 58-1
SRT | R-0031B | Administered before lab visit | 371898003 | C1299367 | 58-2
SRT | R-0031C | Administered during lab visit | 371905006 | C1299374 | 58-3
SRT | R-0031A | Administered after lab visit | 371899006 | C1299368 | 58-4

## CID 3742 Medication Administration, PCI

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.239

Table CID 3742. Medication Administration, PCI

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR Equivalent
--- | --- | --- | --- | --- | ---
SRT | R-00320 | Not Administered | 371900001 | C1299369 |
SRT | R-00321 | Contraindicated | 373147003 | C1298831 | 59-1
SRT | R-0031F | Administered Prior to Percutaneous Coronary Intervention | 371904005 | C1299373 | 59-2
SRT | R-0039A | Administered During Percutaneous Coronary Intervention | 371903004 | C1299372 | 59-3
SRT | R-00399 | Administered After Percutaneous Coronary Intervention | 371902009 | C1299371 | 59-4

## CID 3743 Clopidogrel/Ticlopidine Administration

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.240

Table CID 3743. Clopidogrel/Ticlopidine Administration

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR Equivalent
--- | --- | --- | --- | --- | ---
SRT | R-00320 | Not Administered | 371900001 | C1299369 | 60-1
SRT | R-00321 | Contraindicated | 373147003 | C1298831 | 60-2
SRT | R-0031E | Administered Less than 72 Hours before PCI | 371901002 | C1299370 | 60-3
SRT | R-00399 | Administered After Percutaneous Coronary Intervention | 371902009 | C1299371 | 60-4

## CID 3744 EF Testing Method

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.241

Table CID 3744. EF Testing Method

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P5-3003A | Cardiac ventriculography | 252426003 | C0596683
SRT | P5-D3300 | Radionuclide ventriculography | 85606007 | C0034610
SRT | P5-B3000 | Echocardiography | 40701008 | C0013516

### Note

Previously, a non-existent SNOMED code, (P5-B3081, SRT, "Adult echocardiography"), was used in this context group. It has been replaced with the more generic (P5-B3000, SRT, "Echocardiography") (rather than replacement with (P5-B3004, SRT, "Transthoracic echocardiography"); see Table J-1 SNOMED Codes Retired from DICOM Use.

## CID 3745 Calculation Method

Type:

  Extensible

Version:

  20050110

UID:

  1.2.840.10008.6.1.242

Table CID 3745. Calculation Method

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-10260 | Estimated | 414135002 | C0750572
SRT | R-41D2D | Calculated | 258090004 | C0444686

## CID 3746 Percutaneous Entry Site

This Context Group includes concepts for Percutaneous entry that are the most relevant children of SNOMED concept 297211001 "transvascular approach". Other concepts from that hierarchy may be used as local extensions to this Context Group.

Type:

  Extensible

Version:

  20110609

UID:

  1.2.840.10008.6.1.243

Table CID 3746. Percutaneous Entry Site

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-D067 | Via femoral artery | 260590008 | C0442441
SRT | G-D1E4 | Via radial artery | 444850002 | C2919368
SRT | G-D05F | Via brachial artery | 260585005 | C0442436
SRT | G-D054 | Via artery | 103387006 | C0522522
SRT | G-D0C6 | Via arm vein | 261459001 | C0442444
SRT | G-D071 | Via femoral vein | 260601006 | C0442455
SRT | G-D052 | Via vein | 103386002 | C0522521

### Note

In prior editions, this Context Group included NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes (see PS3.16-2009).

## CID 3747 Percutaneous Closure

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.244

Table CID 3747. Percutaneous Closure

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 75-0 | No closure device used at percutaneous entry
NCDR | 2.0b | 75-1 | Percutaneous entry closed by suture
NCDR | 2.0b | 75-2 | Percutaneous entry closed by sealant
NCDR | 2.0b | 75-3 | Percutaneous entry closed by other mechanism

## CID 3748 Angiographic EF Testing Method

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.245

Table CID 3748. Angiographic EF Testing Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122059 | Single plane Angiography
DCM | 122060 | Bi-plane Angiography

## CID 3749 PCI Procedure Result

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.246

Table CID 3749. PCI Procedure Result

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 100-1 | Successful
NCDR | 2.0b | 100-2 | Partially successful
NCDR | 2.0b | 100-3 | Unsuccessful

## CID 3750 Previously Dilated Lesion

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.247

Table CID 3750. Previously Dilated Lesion

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 108-0 | not previously treated
NCDR | 2.0b | 108-1 | balloon only
NCDR | 2.0b | 108-2 | stent only
NCDR | 2.0b | 108-3 | other/any combination

## CID 3752 Guidewire Crossing

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.248

Table CID 3752. Guidewire Crossing

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122301 | Guidewire crossing lesion unsuccessful
DCM | 122302 | Guidewire crossing lesion successful

## CID 3754 Vascular Complications

Type:

  Extensible

Version:

  20050110

UID:

  1.2.840.10008.6.1.249

Table CID 3754. Vascular Complications

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR 2.0b Equivalent Code Value
--- | --- | --- | --- | --- | ---
SRT | M-37000 | Bleeding | 50960005 | C0019080 | 127
SRT | D3-89100 | Occlusion of artery | 1386000 | C0151699 | 128
SRT | R-102B2 | Loss of distal pulse | 414617007 | C1532146 | 129
SRT | D3-81310 | Arterial dissection | 9406001 | C0002949 | 130
SRT | M-32390 | Pseudoaneurysm | 22036004 | C1510412 | 131
SRT | M-39390 | AV Fistula | 128617001 | C0003855 | 132

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3755 Cath Complications

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.250

Table CID 3755. Cath Complications

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR 2.0b Equivalent Code Value
--- | --- | --- | --- | --- | ---
SRT | D3-00200 | Cardiogenic shock | 89138009 | C0036980 | 123
SRT | D3-30000 | Arrhythmia | 44808001 | C0264886 | 124
SRT | D3-8900D | Cerebrovascular Accident or Stroke | 230690007 | C0038454 | 125
SRT | D3-90100 | Cardiac tamponade | 35304003 | C0007177 | 126
SRT | DF-10781 | Contrast media adverse reaction | 292095005 | C0569413 | 133
SRT | D3-16010 | Congestive heart failure | 42343007 | C0018802 | 134
SRT | D7-11010 | Renal failure | 42399005 | C0035078 | 135
SRT | R-102B5 | Emergency Percutaneous Coronary Intervention | 414089002 | C1532297 | 136
SRT | R-102B3 | Emergency Coronary Artery Bypass | 414088005 | C1532296 | 137
SRT | D3-3002F | Cardiac arrest | 410429000 | C0018790 |

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3756 Cardiac Patient Risk Factors

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.251

Table CID 3756. Cardiac Patient Risk Factors

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | NCDR 2.0b Equivalent Code Value
--- | --- | --- | --- | --- | ---
SRT | G-026D | History of congestive heart failure | 161505003 | C0455531 | 30
SRT | G-023F | History of Diabetes | 161445009 | C0455488 | 31
SRT | R-102B6 | History of renal failure | 414417004 | C1533077 | 32
SRT | R-102B7 | History of chronic lung disease | 414415007 | C1533075 | 33
SRT | G-0102 | History of cerebrovascular disease | 308064009 | C0585890 | 34
SRT | D3-8005B | Peripheral vascular disease | 400047006 | C0085096 | 35
SRT | G-03AA | History of myocardial infarction | 399211009 | C1275835 | 36
SRT | G-0269 | History of Hypertension | 161501007 | C0455527 | 37
SRT | R-102B8 | History of hypercholesterolemia | 414416008 | C1533076 | 39
SRT | D3-30000 | Arrhythmia | 44808001 | C0264886 |
SRT | F-0331B | HIV Positive | 165816005 | C0019699 |
UMLS | C0456029 | Infant of mother with gestational diabetes |  | C0456029 |
SRT | G-0586 | Insulin dependent mother (IDM) | 444161008 | C2732238 |

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3757 Cardiac Diagnostic Procedures

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.252

Table CID 3757. Cardiac Diagnostic Procedures

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P5-D3304 | Cardiac blood pool imaging (nuclear) | 35621002 | C0203725
SRT | P5-00A25 | Cardiac cath coronary angiogram and left ventriculogram | 418903008 | C1690980
SRT | P1-31600 | Cardiac catheterization | 41976001 | C0018795
SRT | P5-00A34 | Cardiac catheterization coronary angiogram | 419416005 | C1633729
SRT | P5-08025 | Cardiac CT | 241547009 | C0412618
SRT | P5-080C2 | Cardiac CT for calcium scoring | 426005005 | C1960839
SRT | P5-09011 | Cardiac MRI | 241620005 | C0412692
SRT | P0-00CA7 | Cardiac MRI stress | 431609005 | C2314961
SRT | P5-00A5C | CT angiography of coronary arteries | 419545005 | C1634617
SRT | P5-B3000 | Echocardiography | 40701008 | C0013516
SRT | P5-B3050 | Exercise Stress echocardiography | 433233004 | C0430466
SRT | P0-006E4 | Exercise Tolerance Test | 165079009 | C0015260
SRT | P5-0903A | Magnetic resonance angiography | 241663008 | C0243032
SRT | P5-D30F8 | Nuclear medicine cardiovascular study | 108294005 | C0581579
SRT | P5-D0050 | Perfusion imaging (nuclear) | 35202002 | C0412366
SRT | P5-0A006 | PET heart study | 241439007 | C0412498
SRT | P2-31011 | Pharmacologic and exercise stress test | 428813002 | C1998158
SRT | P2-31107 | Pharmacological stress test | 424064009 | C1827946
SRT | P5-30045 | Radionuclide angiocardiography | 426940008 | C1960212
SRT | P5-D3008 | Radionuclide myocardial perfusion study | 252432008 | C0430471
SRT | P5-0A100 | SPECT | 105371005 | C0040399
SRT | P2-3110B | Stress test using cardiac pacing | 428685003 | C1997441
SRT | P5-B3002 | Transesophageal echocardiography | 105376000 | C0206054
SRT | P5-B3012 | Transthoracic echocardiography | 433236007 | C0430462

### Note

In a prior version of this Context Group, the code P5-B3009 was specified for Exercise stress echocardiography. That code has been retired by SNOMED, and replaced by P5-B3050. Although there is minimal possibility of misinterpretation with SOP Instances that may include the retired code, receiving applications should be aware of this change; see Annex J.

## CID 3758 Cardiovascular Family History

Type:

  Extensible

Version:

  20111028

UID:

  1.2.840.10008.6.1.253

Table CID 3758. Cardiovascular Family History

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-032F | Family history of cardiovascular disease | 266894000 | C0455404
SRT | G-0157 | Family history of diabetes mellitus | 160303001 | C1313937
SRT | G-011E | Family history of myocardial infarction | 266897007 | C0455406
SRT | G-04E3 | Family history of coronary arteriosclerosis | 430091005 | C2317524
SRT | R-2087E | No family history of diabetes | 160274005 | C0455678
SRT | R-20773 | No family history of cardiovascular disease | 160270001 | C0455346
SRT | F-03F6E | Family history unknown | 407559004 | C1319897

## CID 3760 Hypertension Therapy

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.254

Table CID 3760. Hypertension Therapy

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-80135 | Beta blocker | 33252009 | C0001645
SRT | C-80160 | Calcium channel blocker | 48698004 | C0006684
SRT | C-81520 | Nitrate vasodilator | 31970009 | C0360716
SRT | C-80150 | ACE inhibitor | 41549009 | C0003015
SRT | C-81300 | Angiotensin II receptor antagonist | 96308008 | C0521942
SRT | C-72000 | Diuretic | 30492008 | C0012798

## CID 3761 Antilipemic Agents

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.255

Table CID 3761. Antilipemic Agents

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-80609 | Anion exchange resin | 346322006 | C0003072
SRT | C-80610 | Bile acid sequestrant | 83750004 | C0304522
SRT | C-80680 | Fibrate | 108602006 | C0358700
SRT | C-8060A | Fish oils | 346441008 | C0016157
SRT | C-80800 | Statins | 96302009 | C0360714

## CID 3762 Antiarrhythmic Agents

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.256

Table CID 3762. Antiarrhythmic Agents

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-6181B | class I antiarrhythmic agent | 373260001 | C0360692
SRT | F-61861 | class II antiarrhythmic agent | 373278006 | C0360701
SRT | F-61995 | class III antiarrhythmic agent | 372855004 | C0360703
SRT | F-618AE | class IV antiarrhythmic agent | 372693007 | C0360706

## CID 3764 Myocardial Infarction Therapies

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.257

Table CID 3764. Myocardial Infarction Therapies

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-102B4 | Percutaneous coronary intervention | 415070008 | C1532338
SRT | P1-33530 | Insertion of coronary artery stent | 36969009 | C0521232
SRT | P1-3301A | Coronary artery bypass graft | 232717009 | C0010055
SRT | P0-00C29 | Thrombolytic therapy | 426347000 | C0040044

## CID 3769 Concern Types

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.258

Table CID 3769. Concern Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-04BA9 | Complaint | 409586006 | C0277786
SRT | DF-00000 | Disease | 64572001 | C0012634
SRT | R-005AE | Finding | 404684003 | C0037088
SRT | R-005E0 | Finding reported by patient/informant | 418799008 | C1689949
SRT | F-03E55 | Functional performance and activity | 248536006 | C0424866
SRT | F-01000 | Problem | 55607006 | C0033213

## CID 3770 Problem Status

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.259

Table CID 3770. Problem Status

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-42501 | Active problem | 394774009 | C1273826
SRT | G-A270 | Chronic | 90734009 | C0205191
SRT | G-A397 | Intermittent | 7087005 | C0205267
SRT | G-A39A | Recurrent | 255227004 | C2945760
SRT | G-A47B | Suspected | 415684004 | C0750491
SRT | R-42502 | Inactive problem | 394775005 | C1273827
SRT | F-04B88 | Problem resolved | 413322009 | C1446392
SRT | G-A46B | Known absent | 410516002 | C1444640
SRT | P0-30450 | Well controlled | 1194003 | C0184778

## CID 3772 Health Status

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.260

Table CID 3772. Health Status

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-00001 | Alive and well | 81323004 | C0231162
SRT | F-029D4 | In remission | 313386006 | C1277626
SRT | R-209F6 | Symptom free | 162467007 | C0436342
SRT | F-0600C | Chronically ill | 161901003 | C0581862
SRT | F-06001 | Severely ill | 271593001 | C0424547
SRT | F-00100 | Disabled | 21134002 | C0231170
SRT | F-0351E | Severely disabled | 161045001 | C0424990
SRT | F-04DA1 | Deceased | 419099009 | C1546956

## CID 3773 Use Status

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.261

Table CID 3773. Use Status

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-D316 | Ended | 385656004 | C1272693
SRT | G-D30F | Suspended | 385655000 | C1705537
SRT | G-D30B | In progress | 385651009 | C1272688

## CID 3774 Social History

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.262

Table CID 3774. Social History

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-93109 | Tobacco Smoking Behavior | 365981007 | C0453996
SRT | F-931D4 | Drug misuse behavior | 228366006 | C0556386
SRT | R-40C16 | Exercise | 256235009 | C0015259
SRT | F-045CE | Nutrition | 364393001 | C1286103
SRT | F-02573 | Alcohol consumption | 160573003 | C0001948

## CID 3777 Implanted Devices

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.263

Table CID 3777. Implanted Devices

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-11100 | Cardiac pacemaker | 14106009 | C0030163
SRT | A-11206 | Implantable defibrillator | 72506001 | C0162589
SRT | A-11FCD | Left ventricular assist device | 360066001 | C0181598
SRT | A-28040 | Insulin pump | 69805005 | C1140609

## CID 3778 Stages

Type:

  Extensible

Version:

  20081027

UID:

  1.2.840.10008.6.1.638

Table CID 3778. Stages

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-41177 | Stage 0 | 261613009 | C0441763
SRT | R-41DA8 | Stage 1 | 258215001 | C0441766
SRT | R-41DAC | Stage 2 | 258219007 | C0441767
SRT | R-41DB0 | Stage 3 | 258224005 | C0441771
SRT | R-41DB4 | Stage 4 | 258228008 | C0441772
SRT | R-4117B | Stage 5 | 261617005 | C0441777

## CID 3802 Plaque Structures

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.264

Table CID 3802. Plaque Structures

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-40448 | fibrous | 255423002 | C0439709
SRT | M-50080 | fatty degeneration | 29185008 | C0152254
SRT | M-55420 | pathologic calcification | 18115005 | C0006663
SRT | M-72000 | hyperplasia | 76197007 | C0020507
SRT | G-A265 | non-calcified | 17589002 | C0332209
SRT | G-A660 | mixed | 26242008 | C0205430

## CID 3804 Stenosis Measurement Methods

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.265

Table CID 3804. Stenosis Measurement Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122655 | NASCET
DCM | 122656 | ECST
DCM | 122650 | Area Based Method
DCM | 122651 | Diameter Based Method

## CID 3805 Stenosis Types

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.266

Table CID 3805. Stenosis Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D3-81100 | arteriosclerotic vascular disease | 72092001 | C0003850
SRT | M-01460 | compression | 71173004 | C0332459
SRT | R-40448 | fibrous | 255423002 | C0439709
SRT | D3-80505 | Raynaud's disease | 195295006 | C0034734
SRT | M-300F2 | entrapment | 363563002 | C1285497
SRT | D3-80650 | vasculitis | 31996006 | C0042384
SRT | R-423C3 | thrombosis | 264579008 | C0040053
SRT | M-35300 | embolism | 55584005 | C1704212
SRT | D3-80033 | cystic adventitial disease | 234021009 | C1306656

## CID 3806 Stenosis Shape

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.267

Table CID 3806. Stenosis Shape

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-4047B | concentric | 255465008 | C0439744
SRT | R-40416 | eccentric | 255380003 | C0439740

## CID 3807 Volume Measurement Methods

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.268

Table CID 3807. Volume Measurement Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122650 | Area Based Method
DCM | 122651 | Diameter Based Method
DCM | 122652 | Volume Based Method

## CID 3808 Aneurysm Types

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.269

Table CID 3808. Aneurysm Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-32270 | dissecting aneurysm | 26845001 | C0020449
SRT | D3-80017 | inflammatory aneurysm | 261850007 | C0002950
SRT | M-32201 | ruptured aneurysm | 22039006 | C0162869
SRT | M-24614 | berry aneurysm | 125236003 | C0005136
SRT | M-32240 | mixed aneurysm | 85726003 | C0333093
SRT | M-32410 | racemose aneurysm | 14156004 | C0334533
SRT | D3-80002 | cirsoid aneurysm | 233982006 | C0334533
SRT | M-32320 | mycotic aneurysm | 51668007 | C0085808
SRT | M-32310 | miliary aneurysm | 43299000 | C0333097
SRT | M-32340 | saccular aneurysm | 54002007 | C2713497
SRT | M-32221 | varicose aneurysm | 57754000 | C0333091
SRT | M-32350 | fusiform aneurysm | 85431000 | C0333099
SRT | M-32210 | traumatic aneurysm | 110421000 | C1527161
SRT | M-32202 | thrombosed aneurysm | 125271003 | C1265766
SRT | M-32203 | expanding aneurysm | 125272005 | C1265767
SRT | M-32204 | calcified aneurysm | 125273000 | C1265768
SRT | M-32208 | multiple aneurysm | 125274006 | C1265769
SRT | M-32360 | cylindroid aneurysm | 52856002 | C0333100
SRT | M-32260 | serpentine aneurysm | 70984001 | C0333095

## CID 3809 Associated Conditions

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.270

Table CID 3809. Associated Conditions

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D6-90800 | Marfan's Syndrome | 19346006 | C0024796
SRT | M-10000 | Traumatic Abnormality | 19130008 | C0221206

## CID 3810 Vascular Morphology

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.271

Table CID 3810. Vascular Morphology

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-01470 | plaque | 1522000 | C0332461
SRT | M-3400A | stenosis | 415582006 | C0009814
SRT | M-32200 | aneurysm | 85659009 | C0002940
SRT | D3-81310 | arterial dissection | 9406001 | C0002949
SRT | A-25500 | stent | 65818007 | C0038257
SRT | M-34000 | occlusion | 26036001 | C0028778
SRT | M-39390 | arteriovenous fistula | 128617001 | C0003855
SRT | M-91200 | angioma | 2099007 | C0018916
SRT | M-32000 | dilatation | 25322007 | C0012359
SRT | R-FAB5E | vascular coiling | 416061003 | C1562399
SRT | M-31790 | tortuosity | 15690004 | C0333076
SRT | M-32700 | diverticulum | 31113003 | C0012817
SRT | M-520F8 | vascular sclerosis | 107671003 | C0003850
SRT | M-35001 | thrombus | 396339007 | C0087086
SRT | M-32390 | pseudoaneurysm | 22036004 | C1510412
SRT | M-35300 | embolism | 55584005 | C1704212
SRT | M-74880 | fibromuscular dysplasia | 31653004 | C0016052

## CID 3813 Stent Findings

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.272

Table CID 3813. Stent Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-75300 | hypoplasia | 55199003 | C0243069
SRT | M-3400A | stenosis | 415582006 | C0009814
DCM | 122680 | endoleak |  |
SRT | DD-661D2 | migration of implant or internal device | 370512004 | C1299914
DCM | 122684 | stent disintegration |  |
DCM | 122683 | stent fracture |  |

## CID 3814 Stent Composition

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.273

Table CID 3814. Stent Composition

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-25502 | metal stent | 257363003 | C0441290
SRT | A-25501 | plastic stent | 257362008 | C0441289

## CID 3815 Source of Vascular Finding

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.274

Table CID 3815. Source of Vascular Finding

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D3-80515 | thrombosis | 118927008 | C0040053
SRT | M-35300 | embolism | 55584005 | C1704212
SRT | M-72000 | hyperplasia | 76197007 | C0020507
SRT | D3-80650 | vasculitis | 31996006 | C0042384
SRT | M-8FFFF | tumor | 108369006 | C0027651
SRT | DF-00777 | trauma | 417746004 | C3263723
SRT | G-B102 | surgical | 83578000 | C0543467
SRT | R-422A4 | after procedure | 303110006 | C0580203

## CID 3817 Vascular Sclerosis Types

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.275

Table CID 3817. Vascular Sclerosis Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-52450 | adventitial degeneration | 47631006 | C0333493
SRT | M-52210 | arteriosclerosis with fibrinoid necrosis | 32651000 | C0333487
SRT | M-52200 | arteriolosclerosis | 17941002 | C0333486
SRT | M-52000 | arteriosclerosis | 28960008 | C0003850
SRT | M-52100 | atheroma | 48434008 | C0264956
SRT | M-52120 | atherosclerotic fibrous plaque | 20717008 | C0333483
SRT | M-52101 | calcified atheromatous plaque | 29483008 | C0333479
SRT | M-52102 | complicated atheromatous plaque | 74937006 | C0333480
SRT | M-52470 | cystic medical necrosis | 42182000 | C0392775
SRT | M-52240 | elastic vascular sclerosis | 19952003 | C0333488
SRT | M-52130 | fatty streaks | 53151000 | C0333484
SRT | M-52300 | fibroelastosis | 72166006 | C0016038
SRT | M-52302 | diffuse fibroelastosis | 125358004 | C1265866
SRT | M-52301 | focal fibroelastosis | 125357009 | C1265865
SRT | M-52500 | phlebosclerosis | 18016009 | C0333494
SRT | M-52103 | ulcerated atheromatous plaque | 62189002 | C0333481
SRT | M-52400 | vascular wall degeneration | 33593002 | C0333489

## CID 3820 Non-invasive Vascular Procedures

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.276

Table CID 3820. Non-invasive Vascular Procedures

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P5-0903A | vascular MRI | 241663008 | C0243032
SRT | P5-09011 | cardiac MRI | 241620005 | C0412692
SRT | P5-0807F | cardiovascular CT | 303680000 | C0581427
SRT | P5-0802B | CT of abdominal aorta | 241553009 | C0412626
SRT | P5-00A0D | trunk angiography | 303827001 | C0565173
SRT | P5-009BF | peripheral angiography | 271993009 | C0412290

## CID 3821 Papillary Muscle Included/Excluded

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.277

Table CID 3821. Papillary Muscle Included/Excluded

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122620 | Papillary Muscle Excluded
DCM | 122621 | Papillary Muscle Included

## CID 3823 Respiratory Status

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.278

Table CID 3823. Respiratory Status

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-20010 | inspiration | 14910006 | C0004048
SRT | F-20020 | expiration | 58322009 | C0231800
SRT | F-20030 | autonomous breathing | 45804006 | C0231802
SRT | R-40928 | Valsalva maneuver | 261039008 | C0042293
DCM | 122612 | central breathing position |  |
SRT | F-201BD | shallow breathing | 386616007 | C0221161

## CID 3826 Heart Rhythm

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.279

Table CID 3826. Heart Rhythm

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-33300 | normal sinus rhythm | 64730000 | C0232202
SRT | D3-31500 | atrial arrhythmia | 17366009 | C0085611
SRT | D3-31715 | ventricular arrhythmia | 44103008 | C0085612

## CID 3827 Vessel Segments

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.280

Table CID 3827. Vessel Segments

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12105 “Intracranial Cerebral Vessels”
Include CID 12106 “Intracranial Cerebral Vessels (Unilateral)”
Include CID 12104 “Extracranial Arteries”
Include CID 12109 “Lower Extremity Arteries”
Include CID 12110 “Lower Extremity Veins”
Include CID 12107 “Upper Extremity Arteries”
Include CID 12108 “Upper Extremity Veins”
Include CID 12115 “Renal Vessels”
Include CID 12111 “Abdominal Arteries (Lateral)”
Include CID 12112 “Abdominal Arteries (Unilateral)”
Include CID 12113 “Abdominal Veins (Lateral)”
Include CID 12114 “Abdominal Veins (Unilateral)”
Include CID 3015 “Coronary Arteries”
Include CID 3839 “Coronary Veins”
Include CID 3840 “Pulmonary Veins”

## CID 3829 Pulmonary Arteries

Type:

  Extensible

Version:

  20110818

UID:

  1.2.840.10008.6.1.281

Table CID 3829. Pulmonary Arteries

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-44100 | Trunk of pulmonary artery | 45341000 | C0034052
SRT | T-44010 | Suprapulmonic valve area | 79142001 | C0226052
SRT | T-35250 | pulmonary valve sinuses | 90315007 | C0225946
SRT | T-44400 | Left pulmonary artery | 50408007 | C0226069
SRT | T-44200 | Right pulmonary artery | 78480002 | C0226054

### Note

A previous version of this context group used terms with the SNOMED concept "entire" (see PS3.16-2011). The use of "structure" concepts rather than "entire" is described in Section 8.1.1.

## CID 3831 Stenosis Length

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.282

Table CID 3831. Stenosis Length

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-404AC | long | 255511005 | C0205166
SRT | R-4235F | short | 367450005 | C1806781

## CID 3832 Stenosis Grade

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.283

Table CID 3832. Stenosis Grade

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A003 | severe | 24484000 | C0205082
SRT | G-A002 | moderate | 6736007 | C0205081
SRT | R-404FA | mild | 255604002 | C2945599

## CID 3833 Cardiac Ejection Fraction

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.284

Table CID 3833. Cardiac Ejection Fraction

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID
--- | --- | --- | ---
LN | 8810-4 | Left ventricular ejection fraction by CT | C0488725
LN | 8817-9 | Right ventricular ejection fraction by CT | C0488733
LN | 8811-2 | Left ventricular ejection fraction by MR | C0488726
LN | 8818-7 | Right ventricular ejection fraction by MR | C0488734

## CID 3835 Cardiac Volume Measurements

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.285

Table CID 3835. Cardiac Volume Measurements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Equivalent LOINC Code Value
--- | --- | --- | --- | --- | ---
Include CID 3468 “ED Volume” |  |
Include CID 3469 “ES Volume” |  |
SRT | F-32120 | Stroke Volume | 90096001 | C0038455 | 20562-5

## CID 3836 Time-based Perfusion Measurements

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.286

Table CID 3836. Time-based Perfusion Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122631 | Signal Earliest Peak Time
DCM | 122633 | Signal Increase Start Time
DCM | 122634 | Signal Time to Peak
DCM | 122638 | Signal Baseline Start
DCM | 122639 | Signal Baseline End

## CID 3837 Fiducial Feature

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.287

Table CID 3837. Fiducial Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-4215C | Ostium | 264114003 | C0444567
SRT | T-46600 | Renal Artery | 2841007 | C0035065
SRT | T-42580 | Aortic Bifurcation | 73166001 | C0226027
SRT | R-10258 | Common Iliac Bifurcation | 413896006 | C1531837

## CID 3838 Diameter Derivation

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.288

Table CID 3838. Diameter Derivation

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 3488 “Min/Max/Mean” |  |
SRT | G-A117 | Transverse | 62824007 | C0205106
DCM | 122675 | Anterior-Posterior |  |

## CID 3839 Coronary Veins

Type:

  Extensible

Version:

  20110818

UID:

  1.2.840.10008.6.1.289

Table CID 3839. Coronary Veins

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-48340 | Azygos Vein | 72107004 | C0004526
SRT | T-48410 | Coronary Sinus | 90219004 | C0456944
SRT | T-48420 | Great Cardiac Vein | 5928000 | C0226659
SRT | T-48435 | Small Cardiac Vein | 49082002 | C0226661
SRT | T-48403 | Anterior Cardiac Vein | 194996006 | C0226662
SRT | T-48406 | Atrial Vein | 195164009 | C0226666
SRT | T-48407 | Atrioventricular Vein | 195496005 | C0226668
SRT | T-48430 | Middle Cardiac Vein | 73580002 | C0226660
SRT | T-48404 | Ventricular Vein | 195328002 | C0226667
SRT | T-48405 | Smallest Cardiac Vein | 195073003 | C1279372

### Note

A previous version of this context group used terms with the SNOMED concept "entire" (see PS3.16-2011). The use of "structure" concepts rather than "entire" is described in Section 8.1.1.

## CID 3840 Pulmonary Veins

Type:

  Extensible

Version:

  20110818

UID:

  1.2.840.10008.6.1.290

Table CID 3840. Pulmonary Veins

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-4858F | Pulmonary Vein | 430757002 | C2317442
SRT | T-48502 | Left Pulmonary Vein | 27706005 | C0226670
SRT | T-48540 | Inferior Left Pulmonary Vein | 51249003 | C0226686
SRT | T-48530 | Superior Left Pulmonary Vein | 43863001 | C0226682
SRT | T-48501 | Right Pulmonary Vein | 91539005 | C0226669
SRT | T-48520 | Inferior Right Pulmonary Vein | 113273001 | C0226676
SRT | T-48510 | Superior Right Pulmonary Vein | 8629005 | C0226671

### Note

A previous version of this context group used terms with the SNOMED concept "entire" (see PS3.16-2011). The use of "structure" concepts rather than "entire" is described in Section 8.1.1.

## CID 3843 Myocardial Subsegment

Type:

  Extensible

Version:

  20051103

UID:

  1.2.840.10008.6.1.291

Table CID 3843. Myocardial Subsegment

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-427E6 | endocardial | 304059001 | C0014124
SRT | R-40940 | epicardial | 261073003 | C0442016

## CID 3850 Intravascular OCT Flush Agent

Type:

  Extensible

Version:

  20110609

UID:

  1.2.840.10008.6.1.934

Table CID 3850. Intravascular OCT Flush Agent

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Trade Name (Informative)
--- | --- | --- | --- | --- | ---
SRT | C-A7220 | Dextran | 13132007 | C0086140 |
SRT | C-70841 | Saline | 262003004 | C0445115 |
SRT | C-70434 | Lactated Ringer's | 347379006 | C0073385 |
Include CID 12 “Radiographic Contrast Agent” |  |

### Note

Trade names are from http://www.accessdata.fda.gov/scripts/cder/drugsatfda/index.cfm.

## CID 4005 Partial View Section for Mammography

Type:

  Non-Extensible

Version:

  20050110

UID:

  1.2.840.10008.6.1.292

Table CID 4005. Partial View Section for Mammography

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-404CC | Anterior | 255549009 | C0205094
SRT | R-404CE | Posterior | 255551008 | C0205095
SRT | R-42191 | Superior | 264217000 | C1282910
SRT | R-4094A | Inferior | 261089000 | C0542339
SRT | R-404D5 | Medial | 255561001 | C0205098
SRT | G-A104 | Lateral | 49370004 | C0205093
SRT | G-A110 | Central | 26216008 | C0205099

## CID 4009 DX Anatomy Imaged

Type:

  Extensible

Version:

  20040114

UID:

  1.2.840.10008.6.1.293

Table CID 4009. DX Anatomy Imaged

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 4031 “Common Anatomic Regions”

## CID 4010 DX View

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.294

Table CID 4010. DX View

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-10202 | frontal | 399033003 | C0442223
SRT | R-10204 | frontal oblique | 399160007 | C1302231
SRT | R-10206 | antero-posterior | 399348003 | C0442212
SRT | R-10208 | antero-posterior oblique | 399312000 | C1302318
SRT | R-10210 | right posterior oblique | 399038007 | C1275807
SRT | R-10212 | left posterior oblique | 399006002 | C1275802
SRT | R-40888 | postero-anterior | 272479007 | C0457409
SRT | R-10216 | postero-anterior oblique | 399059000 | C1275812
SRT | R-40985 | right anterior oblique | 399356000 | C1275852
SRT | R-10220 | left anterior oblique | 399135007 | C1275823
SRT | G-A145 | sagittal | 30730003 | C0205129
SRT | R-10224 | medial-lateral | 399260004 | C1302283
SRT | R-40783 | lateral oblique | 260427002 | C0442295
SRT | R-10228 | lateral-medial | 399352003 | C1302336
SRT | R-40782 | medial oblique | 260426006 | C0442294
SRT | R-10232 | right lateral | 399198007 | C0442202
SRT | R-10234 | right oblique | 399236003 | C0442291
SRT | R-10236 | left lateral | 399173006 | C0442198
SRT | R-10238 | left oblique | 399184004 | C0442288
SRT | R-10241 | axial | 399061009 | C0442269
SRT | R-10242 | cranio-caudal | 399162004 | C0442215
SRT | R-10244 | caudo-cranial | 399196006 | C1302249
SRT | R-10246 | oblique axial | 399004004 | C1302164
SRT | R-10248 | oblique cranio-caudal | 399288005 | C1302302
SRT | R-10250 | oblique caudo-cranial | 399225005 | C1302262
SRT | R-10252 | frontal-oblique axial | 399132005 | C1275822
SRT | R-10254 | sagittal-oblique axial | 399325008 | C1275850
SRT | R-102C1 | oblique | 399182000 | C0442287
SRT | R-102CD | lateral | 399067008 | C0442197
SRT | R-102C2 | tangential | 399110001 | C0442227
SRT | R-10256 | submentovertical | 399255003 | C0442244
SRT | R-10257 | verticosubmental | 399360002 | C1302340
SRT | R-102C3 | plantodorsal | 399071006 | C1302192
SRT | R-102C4 | dorsoplantar | 399335002 | C1302328
SRT | R-102C5 | parietoacanthial | 399272005 | C1302290
SRT | R-102C6 | acanthioparietal | 399242004 | C1302273
SRT | R-102C7 | orbitoparietal | 399351005 | C1302335
SRT | R-102C8 | parieto-orbital | 399316002 | C1302320
SRT | R-10230 | latero-medial oblique | 399099002 | C1302201
SRT | R-10226 | medio-lateral oblique | 399368009 | C1302345
SRT | G-8300 | tissue specimen | 119376003 | C1292533
SRT | R-40810 | Occlusal projection | 260499007 | C0442276

### Note

1. In a prior version of this Context Group, Lateral Oblique was assigned the code R-10226, and Medial Oblique was assigned the code R-10230, as synonymous with Medio-Lateral Oblique and Latero-Medial Oblique, respectively. SNOMED currently distinguishes between LO and MLO, and between MO and LMO, although in most radiography contexts there is no practical distinction. Receiving applications should be aware that they may receive SOP Instances with the prior code assignments.

2. In a prior version of this Context Group, "right anterior oblique" was assigned the code R-10218, which in SNOMED is actually "Indirect iris transillumination"; this code has been replaced with the correct code R-40985.

3. In a prior version of this Context Group, a concept of "sagittal" was present with a code of R-10222, which in SNOMED is actually "Trypan blue"; this code has been replaced with the general SNOMED qualifier concept G-A145.

## CID 4011 DX View Modifier

Type:

  Extensible

Version:

  20070524

UID:

  1.2.840.10008.6.1.295

Table CID 4011. DX View Modifier

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-10244 | cephalad | 399196006 | C1302249
SRT | R-10242 | caudad | 399162004 | C0442215
SRT | R-40885 | transthoracic | 272476000 | C0442285
SRT | R-4087B | transforamenal | 272466003 | C0442259
SRT | G-D00B | transoral | 118438002 | C0442366
SRT | R-40554 | transorbital | 278318001 | C0457460
DCM | 111069 | Crosstable |  |
SRT | R-421A4 | Mouth closed | 286866000 | C0564684

### Note

In a prior version of this Context Group, the codes R-102C9, R-102CA, R-102CB, R-102CC, and R-102CE were specified for various concepts. Those codes are not actually in SNOMED, and their use in this context is deprecated. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 4012 Projection Eponymous Name

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.296

Table CID 4012. Projection Eponymous Name

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-10261 | Albers-Schonberg | 399142007 | C1302223
SRT | R-10262 | Alexander | 399237007 | C1302270
SRT | R-40A88 | Apple | 422670003 | C1827705
SRT | R-10263 | Arcelin | 399218003 | C1302258
SRT | R-10264 | Beclere | 399263002 | C1302284
SRT | R-10265 | Bertel | 399362005 | C1302341
SRT | R-10266 | Blackett-Healy | 399246001 | C1302276
SRT | R-40809 | Brewerton projection | 260492003 | C0442271
SRT | R-10267 | Broden | 399344001 | C1302332
SRT | R-40A89 | Burman | 422861003 | C1828171
SRT | R-10268 | Cahoon | 399278009 | C1302294
SRT | R-10269 | Caldwell | 399358004 | C0442264
SRT | R-1026A | Camp-Coventry | 399212002 | C1302254
SRT | R-1026B | Causton | 399065000 | C1302190
SRT | R-1026C | Chamberlain | 399148006 | C1302226
SRT | R-1026D | Chassard-Lapine | 399013002 | C1302168
SRT | R-1026E | Chausse | 399355001 | C1302338
SRT | R-1026F | Cleaves | 399245002 | C1302275
SRT | R-10270 | Clements | 399028002 | C1302177
SRT | R-10271 | Clements-Nakayama | 399320003 | C1302322
SRT | R-40A8A | Colcher-Sussman | 423091003 | C1827227
SRT | R-40A8B | Danelius-Miller | 424811006 | C1828231
SRT | R-10272 | Dunlap | 399303002 | C1302310
SRT | R-40A8F | Eraso Modification | 424655003 | C1827856
SRT | R-10273 | Ferguson | 399372008 | C1302349
SRT | R-40A8C | Fisk | 424962005 | C1827093
SRT | R-10274 | Fleischner | 399281004 | C1302296
SRT | R-40A8D | Folio | 425157002 | C1827491
SRT | R-10275 | Friedman | 399103007 | C1302203
SRT | R-10276 | Fuchs | 399073009 | C1302193
SRT | R-40A8E | Garth | 425188003 | C1827580
SRT | R-10277 | Gaynor-Hart | 399082003 | C1302196
SRT | R-10278 | Grandy | 399311007 | C1302317
SRT | R-10279 | Grashey | 399146005 | C1302225
SRT | R-1027A | Haas | 399341009 | C1302330
SRT | R-4080A | Harris Beath axial projection | 260493008 | C0442308
SRT | R-1027B | Henschen | 399199004 | C1302250
SRT | R-1027C | Hickey | 399277004 | C1302293
SRT | R-40A90 | Hirtz Modification | 424086005 | C1828045
SRT | R-1027D | Holly | 399129007 | C1302216
SRT | R-1027E | Holmblad | 399285008 | C1302300
SRT | R-1027F | Hough | 399168000 | C1302236
SRT | R-10280 | Hsieh | 399083008 | C1302197
SRT | R-10281 | Hughston | 399003005 | C1302163
SRT | R-10282 | Isherwood | 399025004 | C0456593
SRT | R-10283 | Judd | 399201002 | C1302252
SRT | R-4080D | Judet projection | 260496000 | C0442309
SRT | R-10284 | Kandel | 399152006 | C1302227
SRT | R-10285 | Kasabach | 399280003 | C1302295
SRT | R-10286 | Kemp Harper | 399227002 | C1302263
SRT | R-40A91 | Kite | 425030002 | C1827203
SRT | R-10287 | Kovacs | 399318001 | C1302321
SRT | R-10288 | Kuchendorf | 399080006 | C1302195
SRT | R-10289 | Kurzbauer | 399332004 | C1302327
SRT | R-1028A | Laquerriere-Pierquin | 399156009 | C1302230
SRT | R-1028B | Lauenstein | 399169008 | C1302237
SRT | R-1028C | Law | 399206007 | C1302253
SRT | R-1028D | Lawrence | 399179005 | C1302241
SRT | R-1028E | Leonard-George | 398996004 | C1302159
SRT | R-1028F | Lewis | 399037002 | C1302179
SRT | R-10290 | Lilienfeld | 399342002 | C1302331
SRT | R-10291 | Lindblom | 399308006 | C1302314
SRT | R-10292 | Lorenz | 399251007 | C1302279
SRT | R-10293 | Low-Beer | 399327000 | C1302324
SRT | R-10294 | Lysholm | 399370000 | C1302347
SRT | R-10295 | May | 399024000 | C1302174
SRT | R-10296 | Mayer | 399000008 | C1302161
SRT | R-10297 | Merchant | 399284007 | C1302299
SRT | R-10298 | Miller | 399005003 | C1302165
SRT | R-40A92 | Moore | 422568001 | C1827499
SRT | R-4080E | Mortice projection | 260497009 | C0442274
SRT | R-40A93 | Neer | 422795009 | C1828002
SRT | R-10299 | Nolke | 399002000 | C1302162
SRT | R-1029A | Norgaard | 399157000 | C0442275
SRT | R-1029B | Ottonello | 399171008 | C1302238
SRT | R-1029C | Pawlow | 399181007 | C1302242
SRT | R-1029D | Pearson | 399365007 | C1302342
SRT | R-1029E | Penner | 399138009 | C1302221
SRT | R-1029F | Pirie | 399022001 | C1302172
SRT | R-40A94 | Rafert | 423720000 | C1827152
SRT | R-40A95 | Rafert-Long | 422534007 | C1827402
SRT | R-102A0 | Rhese | 399234000 | C1302268
SRT | R-40A96 | Robert | 425035007 | C1827274
SRT | R-40A97 | Rosenberg | 425042007 | C1827277
SRT | R-102A1 | Schuller | 399290006 | C1302303
SRT | R-102A2 | Settegast | 399243009 | C1302274
SRT | R-102A3 | Staunig | 399098005 | C1302200
SRT | R-102A4 | Stecher | 399292003 | C1302304
SRT | R-102A5 | Stenvers | 399349006 | C0442232
SRT | R-40A98 | Stryker | 422954003 | C1828322
SRT | R-102A6 | Swanson | 399313005 | C1302319
SRT | R-102A7 | Tarrant | 399247005 | C1302277
SRT | R-102A8 | Taylor | 399296000 | C1302307
SRT | R-102A9 | Teufel | 399127009 | C1302215
SRT | R-102AA | Titterington | 399241006 | C1302272
SRT | R-102AB | Towne | 399270002 | C0442265
SRT | R-102AC | Twining | 399125001 | C1302214
SRT | R-102AD | Valdini | 399330007 | C1302326
SRT | R-40816 | Van Rosen projection | 260506009 | C0442286
SRT | R-407B0 | Waters | 260473000 | C0442243
SRT | R-102AF | West Point | 399130002 | C1302217
SRT | R-102B0 | Wigby-Taylor | 399215000 | C1302257
SRT | R-40A99 | Wolf | 422996004 | C1828400
SRT | R-102B1 | Zanelli | 399026003 | C1302175

## CID 4013 Anatomic Region for Mammography

Type:

  Non-Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.297

Table CID 4013. Anatomic Region for Mammography

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-04000 | Breast | 76752008 | C0006141

## CID 4014 View for Mammography

Type:

  Non-Extensible

Version:

  20090717

UID:

  1.2.840.10008.6.1.298

Table CID 4014. View for Mammography

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | ACR MQCM 1999 Equivalent
--- | --- | --- | --- | --- | ---
SRT | R-10224 | medio-lateral | 399260004 | C1302283 | ML
SRT | R-10226 | medio-lateral oblique | 399368009 | C1302345 | MLO
SRT | R-10228 | latero-medial | 399352003 | C1302336 | LM
SRT | R-10230 | latero-medial oblique | 399099002 | C1302201 | LMO
SRT | R-10242 | cranio-caudal | 399162004 | C0442215 | CC
SRT | R-10244 | caudo-cranial (from below) | 399196006 | C1302249 | FB
SRT | R-102D0 | superolateral to inferomedial oblique | 399188001 | C1302245 | SIO
SRT | R-40AAA | inferomedial to superolateral oblique | 441555000 | C2711617 | ISO
SRT | R-1024A | cranio-caudal exaggerated laterally | 399192008 | C1302247 | XCCL
SRT | R-1024B | cranio-caudal exaggerated medially | 399101009 | C1302202 | XCCM
SRT | G-8310 | tissue specimen from breast | 127457009 | C0444070 |

### Note

1. In a prior version of this Context Group, Cranio-Caudal Exaggerated Laterally was assigned the code Y-X1770, and Cranio-Caudal Exaggerated Medially was assigned the code Y-X1771. Those codes are deprecated, as they are not valid SNOMED codes. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated codes, receiving applications should be aware of this change; see Annex J.

2. While SRT is the preferred Coding Scheme Designator for SNOMED, regulatory approval of mammography systems makes changes to such systems problematic. Implementers should be aware that many systems will continue to use the deprecated designator SNM3 for certain terms in this context group. It is recommended that implementations receiving Mammography Image or CAD SOP Instances support both SNM3 and SRT as equivalent Coding Scheme Designators for Attributes or Content Items that use this Context Group.

3. In a prior version of this Context Group, (R-102CF, SRT, "cranio-caudal exaggerated") was included. This is not a clinically applied view. Use of this term is deprecated, but receiving applications should be aware of its prior existence.

## CID 4015 View Modifier for Mammography

Type:

  Non-Extensible

Version:

  20090717

UID:

  1.2.840.10008.6.1.299

Table CID 4015. View Modifier for Mammography

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Applies only when View ACR MQCM 1999 Equivalent is: | ACR MQCM 1999 Equivalent
--- | --- | --- | --- | --- | --- | ---
SRT | R-102D2 | Cleavage | 399161006 | C1302232 | CC or FB | CV
SRT | R-102D1 | Axillary Tail | 399011000 | C1302167 | MLO | AT
SRT | R-102D3 | Rolled Lateral | 399197002 | C1275832 | any | ...RL
SRT | R-102D4 | Rolled Medial | 399226006 | C1275838 | any | ...RM
SRT | R-102CA | Rolled Inferior | 414493004 | C1532323 | any | ...RI
SRT | R-102C9 | Rolled Superior | 415670009 | C1531911 | any | ...RS
SRT | R-102D5 | Implant Displaced | 399209000 | C1275834 | any | ...ID
SRT | R-102D6 | Magnification | 399163009 | C1302233 | any | M...
SRT | R-102D7 | Spot Compression | 399055006 | C1302185 | any | S...
SRT | R-102C2 | Tangential | 399110001 | C0442227 | any | TAN
SRT | R-40AB3 | Nipple in profile | 442581004 | C2711408 | any | ...NP
SRT | P2-00161 | Anterior compression | 441752004 | C2711933 | any | ...AC
SRT | R-40ABE | Infra-mammary fold | 442593008 | C2711136 | any | ...IMF
SRT | R-40AB2 | Axillary tissue | 442580003 | C2711122 | any | ...AX

### Note

1. The View ACR MQCM 1999 Equivalent is defined in CID 4014 “View for Mammography”.

2. Some applications and View Modifier ACR MQCM 1999 equivalents have been extended by DICOM to incorporate additional known clinical use cases. The View Modifier ACR MQCM 1999 equivalent indicates its use as a prefix (shown by trailing "…") or suffix (shown by preceding "…") to the View ACR MQCM 1999 equivalent, or replacement for the View ACR MQCM 1999 equivalent.

3. While SRT is the preferred Coding Scheme Designator for SNOMED, regulatory approval of mammography systems makes changes to such systems problematic. Implementers should be aware that many systems will continue to use the deprecated designator SNM3 for certain terms in this context group. It is recommended that implementations receiving Mammography Image or CAD SOP Instances support both SNM3 and SRT as equivalent Coding Scheme Designators for Attributes or Content Items that use this Context Group.

## CID 4016 Anatomic Region for Intra-oral Radiography

Type:

  Non-Extensible

Version:

  20150318

UID:

  1.2.840.10008.6.1.300

Table CID 4016. Anatomic Region for Intra-oral Radiography

Coding Scheme Designator | Code Value | Code Meaning | SNODENT Code | SNOMED-CT Concept ID
--- | --- | --- | --- | ---
SRT | T-D1213 | Jaw region | 100108D | 661005
SRT | T-11170 | Maxilla | 108042D | 70925003
SRT | T-11180 | Mandible | 144511D | 91609006
SRT | T-54000 | Teeth, gums and supporting structures | 124191D | 28035005

### Note

In a prior version of this table, the code T-D1217 was specified for the concept "Maxilla and mandible". The use of this code conflicts with its assignment to another concept in SNOMED, and its use in this context is deprecated. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 4017 Anatomic Region Modifier for Intra-oral Radiography

Type:

  Non-Extensible

Version:

  20150318

UID:

  1.2.840.10008.6.1.301

Table CID 4017. Anatomic Region Modifier for Intra-oral Radiography

Coding Scheme Designator | Code Value | Code Meaning | SNODENT Code | SNOMED-CT Concept ID
--- | --- | --- | --- | ---
SRT | R-FB322 | Central incisor region | 178934D | 699453001
SRT | R-FB35C | Lateral incisor region | 178947D | 699511000
SRT | R-FB35B | Canine region | 178952D | 699510004
SRT | R-FB35A | First premolar region | 178968D | 699509009
SRT | R-FB359 | Second premolar region | 178975D | 699508001
SRT | R-FB358 | First molar region | 178981D | 699507006
SRT | R-FB356 | Second molar region | 178999D | 699505003
SRT | R-FB354 | Third molar region | 179005D | 699503005

### Note

In a prior version of this table, SNOMED codes T-51005 through T-5100C were specified for various concepts. The use of these codes conflicts with their assignment to other concepts in SNOMED, and the set of concepts has been replaced. Also, SNOMED code T-5100D was specified for an Occlusal view; this code does not exist in SNOMED, and the concept is more properly considered as a view rather than an anatomic region, hence has been moved to CID 4010, and assigned the correct SNOMED code R-40810. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; the deprecated codes are included in Annex J.

## CID 4018 Primary Anatomic Structure for Intra-oral Radiography (Permanent Dentition - Designation of Teeth)

Type:

  Non-Extensible

Version:

  20150318

UID:

  1.2.840.10008.6.1.302

Table CID 4018. Primary Anatomic Structure for Intra-oral Radiography (Permanent Dentition - Designation of Teeth)

Coding Scheme Designator | Code Value | Code Meaning | ISO 3950 Designation of Quadrant | ISO 3950 Designation of Tooth | SNODENT Code | SNOMED-CT Concept ID
--- | --- | --- | --- | --- | --- | ---
SRT | T-54210 | Maxillary right third molar tooth | 1 | 8 | 133248D | 68085002
SRT | T-54220 | Maxillary right second molar tooth | 1 | 7 | 109449D | 7121006
SRT | T-54230 | Maxillary right first molar tooth | 1 | 6 | 104587D | 5140004
SRT | T-54240 | Maxillary right second premolar tooth | 1 | 5 | 128425D | 36492000
SRT | T-54250 | Maxillary right first premolar tooth | 1 | 4 | 138890D | 57826002
SRT | T-54260 | Maxillary right canine tooth | 1 | 3 | 145111D | 80647007
SRT | T-54270 | Maxillary right lateral incisor tooth | 1 | 2 | 116770D | 11712009
SRT | T-54280 | Maxillary right central incisor tooth | 1 | 1 | 106397D | 22120004
SRT | T-54290 | Maxillary left central incisor tooth | 2 | 1 | 125190D | 31982000
SRT | T-54300 | Maxillary left lateral incisor tooth | 2 | 2 | 103484D | 25748002
SRT | T-54310 | Maxillary left canine tooth | 2 | 3 | 108821D | 72876007
SRT | T-54320 | Maxillary left first premolar tooth | 2 | 4 | 119834D | 61897005
SRT | T-54330 | Maxillary left second premolar tooth | 2 | 5 | 126921D | 23226009
SRT | T-54340 | Maxillary left first molar tooth | 2 | 6 | 135665D | 23427002
SRT | T-54350 | Maxillary left second molar tooth | 2 | 7 | 130330D | 66303006
SRT | T-54360 | Maxillary left third molar tooth | 2 | 8 | 136609D | 87704003
SRT | T-54370 | Mandibular left third molar tooth | 3 | 8 | 129534D | 74344005
SRT | T-54380 | Mandibular left second molar tooth | 3 | 7 | 101391D | 48402004
SRT | T-54390 | Mandibular left first molar tooth | 3 | 6 | 109790D | 89625000
SRT | T-54400 | Mandibular left second premolar tooth | 3 | 5 | 117536D | 24573005
SRT | T-54410 | Mandibular left first premolar tooth | 3 | 4 | 138336D | 2400006
SRT | T-54420 | Mandibular left canine tooth | 3 | 3 | 119269D | 39844006
SRT | T-54430 | Mandibular left lateral tooth | 3 | 2 | 119276D | 77130001
SRT | T-54440 | Mandibular left central incisor tooth | 3 | 1 | 116581D | 113278005
SRT | T-54450 | Mandibular right central incisor tooth | 4 | 1 | 139525D | 15422005
SRT | T-54460 | Mandibular right lateral incisor tooth | 4 | 2 | 113091D | 82628004
SRT | T-54470 | Mandibular right canine tooth | 4 | 3 | 107357D | 47055002
SRT | T-54480 | Mandibular right first premolar tooth | 4 | 4 | 144507D | 80140008
SRT | T-54490 | Mandibular right second premolar tooth | 4 | 5 | 110784D | 8873007
SRT | T-54500 | Mandibular right first molar tooth | 4 | 6 | 143324D | 28480000
SRT | T-54510 | Mandibular right second molar tooth | 4 | 7 | 145772D | 40005008
SRT | T-54520 | Mandibular right third molar tooth | 4 | 8 | 100566D | 38994002

## CID 4019 Primary Anatomic Structure for Intra-oral Radiography (Deciduous Dentition - Designation of Teeth)

Type:

  Non-Extensible

Version:

  20150318

UID:

  1.2.840.10008.6.1.303

Table CID 4019. Primary Anatomic Structure for Intra-oral Radiography (Deciduous Dentition - Designation of Teeth)

Coding Scheme Designator | Code Value | Code Meaning | ISO 3950 Designation of Quadrant | ISO 3950 Designation of Tooth | SNODENT Code | SNOMED-CT Concept ID
--- | --- | --- | --- | --- | --- | ---
SRT | T-54610 | Deciduous maxillary right central incisor tooth | 5 | 1 | 162619D | 245620002
SRT | T-54620 | Deciduous maxillary right lateral incisor tooth | 5 | 2 | 162494D | 245619008
SRT | T-54630 | Deciduous maxillary right canine tooth | 5 | 3 | 124018D | 30618001
SRT | T-54640 | Deciduous maxillary right first molar tooth | 5 | 4 | 162234D | 245616001
SRT | T-54650 | Deciduous maxillary right second molar tooth | 5 | 5 | 130574D | 27855007
SRT | T-54660 | Deciduous maxillary left central incisor tooth | 6 | 1 | 108911D | 51678005
SRT | T-54670 | Deciduous maxillary left lateral incisor tooth | 6 | 2 | 123818D | 43622005
SRT | T-54680 | Deciduous maxillary left canine tooth | 6 | 3 | 140711D | 73937000
SRT | T-54690 | Deciduous maxillary left first molar tooth | 6 | 4 | 141712D | 45234009
SRT | T-54700 | Deciduous maxillary left second molar tooth | 6 | 5 | 112992D | 51943008
SRT | T-54760 | Deciduous mandibular left central incisor tooth | 7 | 1 | 150298D | 89552004
SRT | T-54770 | Deciduous mandibular left lateral incisor tooth | 7 | 2 | 134816D | 14770005
SRT | T-54780 | Deciduous mandibular left canine tooth | 7 | 3 | 162441D | 245639007
SRT | T-54790 | Deciduous mandibular left first molar tooth | 7 | 4 | 118147D | 38896004
SRT | T-54800 | Deciduous mandibular left second molar tooth | 7 | 5 | 144621D | 49330006
SRT | T-54710 | Deciduous mandibular right central incisor tooth | 8 | 1 | 120236D | 67834006
SRT | T-54720 | Deciduous mandibular right lateral incisor tooth | 8 | 2 | 113281D | 22445006
SRT | T-54730 | Deciduous mandibular right canine tooth | 8 | 3 | 105720D | 6062009
SRT | T-54740 | Deciduous mandibular right first molar tooth | 8 | 4 | 162206D | 245631005
SRT | T-54750 | Deciduous mandibular right second molar tooth | 8 | 5 | 107031D | 61868007

## CID 4020 PET Radionuclide

Type:

  Extensible

Version:

  20160119

UID:

  1.2.840.10008.6.1.304

Table CID 4020. PET Radionuclide

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-105A1 | ^11^Carbon | 40565003 | C0302944
SRT | C-107A1 | ^13^Nitrogen | 21576001 | C0302959
SRT | C-1018C | ^14^Oxygen | 424875009 | C1828369
SRT | C-B1038 | ^15^Oxygen | 129504001 | C1268556
SRT | C-111A1 | ^18^Fluorine | 77004003 | C0302995
SRT | C-155A1 | ^22^Sodium | 71633006 | C0303511
SRT | C-135A4 | ^38^Potassium | 423764008 | C1827255
DCM | 126605 | ^43^Scandium |  |
DCM | 126600 | ^44^Scandium |  |
SRT | C-166A2 | ^45^Titanium | 75696008 | C0303635
DCM | 126601 | ^51^Manganese |  |
SRT | C-130A1 | ^52^Iron | 69089000 | C0303218
SRT | C-149A1 | ^52^Manganese | 37225000 | C0303448
DCM | 126607 | ^52m^Manganese |  |
SRT | C-127A4 | ^60^Copper | 425364008 | C1827982
SRT | C-127A1 | ^61^Copper | 71425003 | C0303189
SRT | C-127A5 | ^62^Copper | 422934004 | C1828311
SRT | C-141A1 | ^62^Zinc | 65054007 | C0303361
SRT | C-127A2 | ^64^Copper | 3932008 | C0303190
SRT | C-131A1 | ^66^Gallium | 79477007 | C0303224
SRT | C-131A3 | ^68^Gallium | 35337001 | C0303226
SRT | C-128A2 | ^68^Germanium | 53315004 | C0303198
DCM | 126602 | ^70^Arsenic |  |
SRT | C-115A2 | ^72^Arsenic | 2705002 | C0303037
SRT | C-116A2 | ^73^Selenium | 87437000 | C0303047
SRT | C-113A1 | ^75^Bromine | 17910003 | C0303008
SRT | C-113A2 | ^76^Bromine | 79523006 | C1304532
SRT | C-113A3 | ^77^Bromine | 86521004 | C0303010
SRT | C-159A2 | ^82^Rubidium | 79197006 | C0303554
SRT | C-162A3 | ^86^Yttrium | 10738001 | C0303592
SRT | C-168A4 | ^89^Zirconium | 63360001 | C0303661
DCM | 126603 | ^90^Niobium |  |
SRT | C-162A7 | ^90^Yttrium | 14691008 | C0303596
SRT | C-163AA | ^94m^Technetium | 424079002 | C1828040
SRT | C-114A5 | ^124^Iodine | 40937006 | C0303024
DCM | 126606 | ^152^Terbium |  |

## CID 4021 PET Radiopharmaceutical

Type:

  Extensible

Version:

  20170413

UID:

  1.2.840.10008.6.1.305

Table CID 4021. PET Radiopharmaceutical

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Other Names
--- | --- | --- | --- | --- | ---
DCM | 126752 | 28H1 ^89^Zr |  |  |
DCM | 126713 | 2FA F^18^ |  |  | FA-85380
DCM | 126751 | 7D12 ^89^Zr |  |  |
DCM | 126750 | 7E11 ^89^Zr |  |  |
SRT | C-B1043 | Acetate C^11^ | 129513004 | C1098488 |
DCM | 126729 | AGN-150998 ^89^Zr |  |  | MP0112
SRT | C-B103C | Ammonia N^13^ | 129508003 | C1268560 |
DCM | 126754 | Anti-B220 ^89^Zr |  |  | Anti-CD45R
DCM | 126700 | ATSM Cu^60^ |  |  |
DCM | 126701 | ATSM Cu^61^ |  |  |
DCM | 126702 | ATSM Cu^62^ |  |  |
SRT | C-B07DB | ATSM Cu^64^ | 422855001 | C1828021 |
DCM | 126722 | Benralizumab ^89^Zr |  |  | MEDI-563, KHK4563
DCM | 126516 | Bevacizumab ^89^Zr |  |  | Avastin™ ^89^Zr
DCM | 126727 | Blinatumomab ^89^Zr |  |  | AMG103, MT103
DCM | 126735 | Brentuximab ^89^Zr |  |  | Adcetris™
SRT | C-B07DC | Butanol O^15^ | 422540000 | C1827030 |
SRT | C-B103B | Carbon dioxide O^15^ | 129507008 | C1268559 |
SRT | C-B1045 | Carbon monoxide C^11^ | 129515006 | C1268564 |
SRT | C-B103A | Carbon monoxide O^15^ | 129506004 | C1268558 |
SRT | C-B103F | Carfentanil C^11^ | 129511002 | C1268562 |
DCM | 126513 | Cetuximab ^89^Zr |  |  | Erbitux™ ^89^Zr
DCM | 126517 | cG250-F(ab')(2) ^89^Zr |  |  |
DCM | 126703 | Choline C^11^ |  |  |
DCM | 126715 | CLR1404 I^124^ |  |  |
DCM | 126716 | CLR1404 I^131^ |  |  |
DCM | 126746 | cMAb U36 ^89^Zr |  |  |
DCM | 126515 | cU36 ^89^Zr |  |  |
DCM | 126762 | Df-[FK](2) ^89^Zr |  |  |
DCM | 126763 | Df-[FK](2)-3PEG(4) ^89^Zr |  |  |
DCM | 126520 | Df-CD45 ^89^Zr |  |  |
DCM | 126760 | Df-FK ^89^Zr |  |  |
DCM | 126761 | Df-FK-PEG(3) ^89^Zr |  |  |
DCM | 126747 | DN30 ^89^Zr |  |  |
DCM | 126519 | E4G10 ^89^Zr |  |  |
DCM | 126732 | Ecromeximab ^89^Zr |  |  | KW-2871
UMLS | C2713594 | Edotreotide Ga^68^ |  | C2713594 | DOTATOC, SMT487
SRT | C-B07DD | EDTA Ga^68^ | 423498000 | C1828067 |
DCM | 126704 | Fallypride C^11^ |  |  |
DCM | 126705 | Fallypride F^18^ |  |  |
DCM | 126706 | FLB 457 C^11^ |  |  |
SRT | C-D6858 | Florbetaben F^18^ | 712736002 | C3818757 | NeuroCeq™
SRT | C-E0269 | Florbetapir F^18^ | 456995000 | C3475363 | AV-45, Amyvid™
DCM | 126503 | Flubatine F^18^ |  |  | NCFHEB
SRT | C-E0265 | Fluciclatide F^18^ | 456999006 | C2987729 |
SRT | C-E026A | Fluciclovine F^18^ | 457000009 | C1311253 |
SRT | C-B07DE | Flumazenil C^11^ | 423543007 | C1827653 |
SRT | C-B07DF | Flumazenil F^18^ | 422975006 | C1828330 |
SRT | C-B07E0 | Fluorethyltyrosin F^18^ | 424708001 | C1827913 |
SRT | C-B07E4 | Fluorobenzothiazole F^18^ | 423546004 | C1827131 |
SRT | C-E0273 | Fluorocholine F^18^ | 456992002 | C3531803 |
SRT | C-B1031 | Fluorodeoxyglucose F^18^ | 35321007 | C0046056 |
UMLS | C1831937 | Fluoroestradiol (FES) F^18^ |  | C1831937 |
UMLS | C1541539 | Fluoroetanidazole F^18^ |  | C1541539 | EF5
SRT | C-B1034 | Fluoro-L-dopa F^18^ | 129500005 | C1268553 |
SRT | C-B07E2 | Fluoromethane F^18^ | 422763008 | C1827137 |
SRT | C-B07E1 | Fluoromisonidazole F^18^ | 422598008 | C1827349 | FMISO
UMLS | C2934038 | Fluoropropyl-dihydrotetrabenazine (DTBZ) F^18^ |  | C2934038 | AV-133
DCM | 126707 | Fluorotriopride F^18^ |  |  |
SRT | C-B07E3 | Fluorouracil F^18^ | 425236000 | C1827690 |
DCM | 126718 | Flurpiridaz F^18^ |  |  | BMS-747158-02
SRT | C-E0267 | Flutemetamol F^18^ | 456997008 | C2983948 | Vizamyl™
DCM | 126748 | Fresolimumab ^89^Zr |  |  | GC1008
DCM | 126731 | GA201 ^89^Zr |  |  | RG1760, RO5083945
SRT | C-B1046 | Germanium Ge^68^ | 129516007 | C1268565 |
DCM | 126724 | Glembatumumab vedotin ^89^Zr |  |  | CDX-011, CR011-vcMMAE
SRT | C-B103D | Glutamate N^13^ | 129509006 | C1268561 |
DCM | 126709 | Glutamine C^11^ |  |  |
DCM | 126710 | Glutamine C^14^ |  |  |
DCM | 126711 | Glutamine F^18^ |  |  |
UMLS | C2981788 | ISO-1 F^18^ |  | C2981788 |
DCM | 126514 | J591 ^89^Zr |  |  |
DCM | 126740 | Margetuximab ^89^Zr |  |  | MGAH22
DCM | 126730 | MEDI-551 ^89^Zr |  |  |
SRT | C-B07E5 | Mespiperone C^11^ | 424789007 | C1828032 |
SRT | C-B103E | Methionine C^11^ | 129510001 | C0252667 |
DCM | 126738 | Mogamulizumab ^89^Zr |  |  | AMG761, KW-0761, Poteligeo™
DCM | 126510 | Monoclonal Antibody (mAb) ^64^Cu |  |  |
DCM | 126511 | Monoclonal Antibody (mAb) ^89^Zr |  |  |
SRT | C-B07E6 | Monoclonal antibody I^124^ | 424874008 | C1827605 |
DCM | 126753 | Nanocolloidal albumin ^89^Zr |  |  | Nanocoll
DCM | 126714 | Nifene F^18^ |  |  |
DCM | 126721 | Obinituzimab ^89^Zr |  |  | Afutuzumab, Gazyva™
DCM | 126723 | Ocaratuzumab ^89^Zr |  |  | AME-133v, LY2469298
SRT | C-B1038 | Oxygen O^15^ | 129504001 | C1268556 |
SRT | C-B1039 | Oxygen-water O^15^ | 129505000 | C1268557 |
SRT | C-B1044 | Palmitate C^11^ | 129514005 | C1268563 |
DCM | 126736 | Panitumumab ^89^Zr |  |  | ABX-EGF, Vectibix™
DCM | 126728 | Pegdinetanib ^89^Zr |  |  | BMS-844203, CT-322, Angiocept™
DCM | 126725 | Pinatuzumab vedotin ^89^Zr |  |  | RG7593, DCDT2980S
DCM | 126500 | Pittsburgh compound B C^11^ |  |  | PIB
DCM | 126726 | Polatuzumab vedotin ^89^Zr |  |  | RG7596, DCDS4501A
SRT | C-B07E7 | PTSM Cu^62^ | 422789008 | C1827357 |
DCM | 126518 | R1507 ^89^Zr |  |  |
SRT | C-B1042 | Raclopride C^11^ | 129512009 | C0752264 |
DCM | 126742 | Ranibizumab ^89^Zr |  |  | Lucentis™
DCM | 126737 | Rituximab ^89^Zr |  |  | IDEC-C2B8, Rituxan™
DCM | 126755 | RO5323441 ^89^Zr |  |  |
DCM | 126756 | RO542908 ^89^Zr |  |  |
DCM | 126733 | Roledumab ^89^Zr |  |  | LFB-R593
SRT | C-B1037 | Rubidium chloride Rb^82^ | 129503007 | C1268555 |
DCM | 126741 | SAR3419 ^89^Zr |  |  |
SRT | C-B1032 | Sodium fluoride F^18^ | 129501009 | C0304965 |
SRT | C-B07E8 | Sodium iodide I^124^ | 422980002 | C1828393 |
SRT | C-B1047 | Sodium Na^22^ | 129517003 | C1268566 |
SRT | C-B1033 | Spiperone F^18^ | 129499001 | C1268552 |
DCM | 126502 | T807 F^18^ |  |  | AV-1451
DCM | 126717 | THK5351 F^18^ |  |  |
SRT | C-B1036 | Thymidine (FLT) F^18^ | 129502002 | C1268554 |
DCM | 126512 | Trastuzumab ^89^Zr |  |  | Herceptin™ ^89^Zr
DCM | 126749 | TRC105 ^89^Zr |  |  |
UMLS | C1742831 | tyrosine-3-octreotate Ga^68^ |  | C1742831 | DOTATATE
DCM | 126739 | Ublituximab ^89^Zr |  |  | LFB-R603, TG-1101
DCM | 126734 | XmAb5574 ^89^Zr |  |  | MOR208

## CID 4025 Primary Anatomic Structure for Intra-oral Radiography (Supernumerary Dentition - Designation of Teeth)

Type:

  Non-Extensible

Version:

  20150318

UID:

  1.2.840.10008.6.1.1021

Table CID 4025. Primary Anatomic Structure for Intra-oral Radiography (Supernumerary Dentition - Designation of Teeth)

Coding Scheme Designator | Code Value | Code Meaning | SNODENT Code | SNOMED-CT Concept ID
--- | --- | --- | --- | ---
SRT | R-FC4E0 | Supernumerary deciduous mandibular left canine tooth | 177552D | 707029006
SRT | R-FC4DD | Supernumerary deciduous mandibular left central incisor tooth | 177292D | 707026004
SRT | R-FC4E1 | Supernumerary deciduous mandibular left first molar tooth | 177421D | 707030001
SRT | R-FC4DF | Supernumerary deciduous mandibular left lateral incisor tooth | 177318D | 707028003
SRT | R-FC4E2 | Supernumerary deciduous mandibular left second molar tooth | 177704D | 707031002
SRT | R-FC4DA | Supernumerary deciduous mandibular right canine tooth | 177387D | 707023007
SRT | R-FC4DC | Supernumerary deciduous mandibular right central incisor tooth | 177450D | 707025000
SRT | R-FC4D9 | Supernumerary deciduous mandibular right first molar tooth | 177758D | 707022002
SRT | R-FC4DB | Supernumerary deciduous mandibular right lateral incisor tooth | 177466D | 707024001
SRT | R-FC4D8 | Supernumerary deciduous mandibular right second molar tooth | 177302D | 707021009
SRT | R-FC4D3 | Supernumerary deciduous maxillary left canine tooth | 177497D | 707016006
SRT | R-FC4D1 | Supernumerary deciduous maxillary left central incisor tooth | 177736D | 707014009
SRT | R-FC4D4 | Supernumerary deciduous maxillary left first molar tooth | 177715D | 707017002
SRT | R-FC4D2 | Supernumerary deciduous maxillary left lateral incisor tooth | 177263D | 707015005
SRT | R-FC4D5 | Supernumerary deciduous maxillary left second molar tooth | 177581D | 707018007
SRT | R-FC4CE | Supernumerary deciduous maxillary right canine tooth | 177575D | 707011001
SRT | R-FC4D0 | Supernumerary deciduous maxillary right central incisor tooth | 177696D | 707013003
SRT | R-FC4CD | Supernumerary deciduous maxillary right first molar tooth | 177360D | 707010000
SRT | R-FC4CF | Supernumerary deciduous maxillary right lateral incisor tooth | 177620D | 707012008
SRT | R-FC4CC | Supernumerary deciduous maxillary right second molar tooth | 177665D | 707009005
SRT | R-FC4FD | Supernumerary permanent mandibular left canine tooth | 177523D | 707058009
SRT | R-FC4FF | Supernumerary permanent mandibular left central incisor tooth | 177510D | 707060006
SRT | R-FC4FA | Supernumerary permanent mandibular left first molar tooth | 177478D | 707055007
SRT | R-FC4FC | Supernumerary permanent mandibular left first premolar tooth | 177631D | 707057004
SRT | R-FC4FE | Supernumerary permanent mandibular left lateral incisor tooth | 177271D | 707059001
SRT | R-FC4F9 | Supernumerary permanent mandibular left second molar tooth | 177677D | 707054006
SRT | R-FC4FB | Supernumerary permanent mandibular left second premolar tooth | 177727D | 707056008
SRT | R-FC4F7 | Supernumerary permanent mandibular left third molar tooth | 177743D | 707052005
SRT | R-FC502 | Supernumerary permanent mandibular right canine tooth | 177341D | 707063008
SRT | R-FC500 | Supernumerary permanent mandibular right central incisor tooth | 177285D | 707061005
SRT | R-FC505 | Supernumerary permanent mandibular right first molar tooth | 177413D | 707066000
SRT | R-FC503 | Supernumerary permanent mandibular right first premolar tooth | 177599D | 707064002
SRT | R-FC501 | Supernumerary permanent mandibular right lateral incisor tooth | 177506D | 707062003
SRT | R-FC506 | Supernumerary permanent mandibular right second molar tooth | 177432D | 707067009
SRT | R-FC504 | Supernumerary permanent mandibular right second premolar tooth | 177409D | 707065001
SRT | R-FC507 | Supernumerary permanent mandibular right third molar tooth | 177608D | 707068004
SRT | R-FC4EF | Supernumerary permanent maxillary left canine tooth | 177356D | 707044007
SRT | R-FC4ED | Supernumerary permanent maxillary left central incisor tooth | 177762D | 707042006
SRT | R-FC4F2 | Supernumerary permanent maxillary left first molar tooth | 177654D | 707047000
SRT | R-FC4F0 | Supernumerary permanent maxillary left first premolar tooth | 177445D | 707045008
SRT | R-FC4EE | Supernumerary permanent maxillary left lateral incisor tooth | 177683D | 707043001
SRT | R-FC4F3 | Supernumerary permanent maxillary left second molar tooth | 177373D | 707048005
SRT | R-FC4F1 | Supernumerary permanent maxillary left second premolar tooth | 177325D | 707046009
SRT | R-FC4F4 | Supernumerary permanent maxillary left third molar tooth | 177568D | 707049002
SRT | R-FC4E9 | Supernumerary permanent maxillary right canine tooth | 177339D | 707038008
SRT | R-FC4EC | Supernumerary permanent maxillary right central incisor tooth | 177259D | 707041004
SRT | R-FC4E6 | Supernumerary permanent maxillary right first molar tooth | 177534D | 707035006
SRT | R-FC4E8 | Supernumerary permanent maxillary right first premolar tooth | 177612D | 707037003
SRT | R-FC4EA | Supernumerary permanent maxillary right lateral incisor tooth | 177484D | 707039000
SRT | R-FC4E4 | Supernumerary permanent maxillary right second molar tooth | 177649D | 707033004
SRT | R-FC4E7 | Supernumerary permanent maxillary right second premolar tooth | 177547D | 707036007
SRT | R-FC4E3 | Supernumerary permanent maxillary right third molar tooth | 177394D | 707032009

## CID 4026 Primary Anatomic Structure for Intra-oral and Craniofacial Radiography - Teeth

Type:

  Non-Extensible

Version:

  20150318

UID:

  1.2.840.10008.6.1.1022

Table CID 4026. Primary Anatomic Structure for Intra-oral and Craniofacial Radiography - Teeth

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 4018 “Primary Anatomic Structure for Intra-oral Radiography (Permanent Dentition - Designation of Teeth)”
Include CID 4019 “Primary Anatomic Structure for Intra-oral Radiography (Deciduous Dentition - Designation of Teeth)”
Include CID 4025 “Primary Anatomic Structure for Intra-oral Radiography (Supernumerary Dentition - Designation of Teeth)”

## CID 4028 Craniofacial Anatomic Regions

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.306

Table CID 4028. Craniofacial Anatomic Regions

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-D006D | Bone structure of head and/or neck | 312779009 | C0730130
SRT | T-11501 | Cervical spine | 122494005 | C0728985
SRT | T-11156 | Ethmoid bone | 52374004 | C0015027
SRT | T-AB100 | External ear | 28347008 | C0013453
SRT | T-AA770 | Eyeball | 79652003 | C0229242
SRT | T-D0801 | Eye region | 371398005 | C0015392
SRT | T-11196 | Facial bones | 91397008 | C0015455
SRT | T-11110 | Frontal bone | 74872008 | C0016732
SRT | T-D1100 | Head | 69536005 | C0018670
SRT | T-D1000 | Head and Neck | 774007 | C0460004
SRT | T-11190 | Hyoid bone | 21387005 | C0020417
SRT | T-AB700 | Inner ear | 22945000 | C0022889
SRT | T-AB959 | Internal Auditory Canal | 361078006 | C1283773
SRT | T-D1213 | Jaw region | 661005 | C0022359
SRT | T-1115A | Lacrimal bone | 6229007 | C0222733
SRT | T-24100 | Larynx | 4596009 | C0023078
SRT | T-52000 | Lip | 48477009 | C0023759
SRT | T-11180 | Mandible | 91609006 | C0024687
SRT | T-11133 | Mastoid bone | 59066005 | C0446908
SRT | T-11170 | Maxilla | 70925003 | C0024947
SRT | T-AB300 | Middle ear | 25342003 | C0013455
SRT | T-13100 | Muscle of head | 22688005 | C0224097
SRT | T-13300 | Muscle of neck | 81727001 | C0027532
SRT | T-11149 | Nasal bone | 74386004 | C0027422
SRT | T-D1600 | Neck | 45048000 | C0027530
SRT | T-11140 | Occipital bone | 31640002 | C0028784
SRT | T-11102 | Optic canal | 55024004 | C0450102
SRT | T-D14AE | Orbital structure | 363654007 | C0029180
SRT | T-11160 | Palatine bone | 51283005 | C0222734
SRT | T-22000 | Paranasal sinus | 2095001 | C0030471
SRT | T-11120 | Parietal bone | 24924006 | C0030558
SRT | T-61007 | Salivary gland | 385294005 | C0036098
SRT | T-11100 | Skull | 89546000 | C0037303
SRT | T-51120 | Soft palate | 49460000 | C0030219
SRT | T-11150 | Sphenoid bone | 73117003 | C0037884
SRT | T-61300 | Submandibular gland | 54019009 | C0038556
SRT | T-11130 | Temporal bone | 60911003 | C0039484
SRT | T-15290 | Temporomandibular joint | 53620006 | C0039493
SRT | T-53000 | Tongue | 21974007 | C0040408
SRT | T-54010 | Tooth | 38199008 | C0040426
SRT | T-25000 | Trachea | 44567001 | C0040578
SRT | T-11011 | Vertebral column and cranium | 110517009 | C1266914
SRT | T-21342 | Vomer bone | 87166008 | C0242403
SRT | T-11166 | Zygoma | 13881006 | C0043539

## CID 4030 CT, MR and PET Anatomy Imaged

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.307

Table CID 4030. CT, MR and PET Anatomy Imaged

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 4031 “Common Anatomic Regions”
SRT | T-42500 | Abdominal aorta | 7832008 | C0003484
SRT | T-B3000 | Adrenal gland | 23451007 | C0001625
SRT | T-42300 | Aortic arch | 57034009 | C0003489
SRT | T-A0100 | Brain | 12738006 | C0006104
SRT | T-45010 | Carotid Artery | 69105007 | C0007272
SRT | T-A6000 | Cerebellum | 113305005 | C0007765
SRT | T-45520 | Circle of Willis | 11279006 | C0008812
SRT | T-43000 | Coronary artery | 41801008 | C0205042
SRT | T-A0191 | Cranial venous system | 128320002 | C0447118
SRT | T-41068 | Iliac and/or femoral artery | 299716001 | C0576469
SRT | T-71000 | Kidney | 64033007 | C0022646
SRT | T-62000 | Liver | 10200004 | C0023884
SRT | T-65000 | Pancreas | 15776009 | C0030274
SRT | T-B7000 | Parathyroid | 111002 | C0030518
SRT | T-44000 | Pulmonary artery | 81040000 | C0034052
SRT | T-46600 | Renal artery | 2841007 | C0035065
SRT | T-C3000 | Spleen | 78961009 | C0037993
SRT | T-94000 | Testis | 40689003 | C0039597
SRT | T-42070 | Thoracic aorta | 113262008 | C1522460
SRT | T-C8000 | Thymus | 9875009 | C0040113
SRT | T-B6000 | Thyroid | 69748006 | C0040132
SRT | T-83000 | Uterus | 35039007 | C0042149

## CID 4031 Common Anatomic Regions

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.308

Table CID 4031. Common Anatomic Regions

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-D4000 | Abdomen | 113345001 | C0000726
SRT | R-FAB57 | Abdomen and Pelvis | 416949008 | C1508499
SRT | T-15420 | Acromioclavicular joint | 85856004 | C0001208
SRT | T-15750 | Ankle joint | 70258002 | C0003087
SRT | T-59900 | Anus | 53505006 | C0003461
SRT | T-280A0 | Apex of Lung | 86598002 | C0225703
SRT | T-60610 | Bile duct | 28273000 | C0005400
SRT | T-74000 | Bladder | 89837001 | C0005682
SRT | T-12700 | Bone of lower limb | 72001000 | C0448188
SRT | T-D0821 | Bone of upper limb | 371195002 | C0003793
SRT | T-04000 | Breast | 76752008 | C0006141
SRT | T-26000 | Bronchus | 955009 | C0006255
SRT | T-12770 | Calcaneus | 80144004 | C0006655
SRT | T-11501 | Cervical spine | 122494005 | C0728985
SRT | T-D00F7 | Cervico-thoracic spine | 297171002 | C0729373
SRT | T-D3000 | Chest | 51185008 | C0817096
SRT | R-FAB55 | Chest and Abdomen | 416550000 | C1442171
SRT | R-FAB56 | Chest, Abdomen and Pelvis | 416775004 | C1562547
SRT | T-12310 | Clavicle | 51299004 | C0008913
SRT | T-11BF0 | Coccyx | 64688005 | C0009194
SRT | T-59300 | Colon | 71854001 | C0009368
SRT | T-58200 | Duodenum | 38848004 | C0013303
SRT | T-15430 | Elbow joint | 16953009 | C0013770
SRT | T-D0010 | Entire body | 38266002 | C0229960
SRT | T-56000 | Esophagus | 32849002 | C0014876
SRT | T-DD163 | Esophagus, stomach and duodenum | 110861005 | C1268410
SRT | T-D0300 | Extremity | 66019005 | C0015385
SRT | T-AA000 | Eye | 81745001 | C0015392
SRT | T-D0801 | Eye region | 371398005 | C0700042
SRT | T-11196 | Facial bones | 91397008 | C0015455
SRT | T-12710 | Femur | 71341001 | C0015811
SRT | T-12750 | Fibula | 87342007 | C0016068
SRT | T-D8800 | Finger | 7569003 | C0016129
SRT | T-D9700 | Foot | 56459004 | C0016504
SRT | T-D8500 | Forearm | 14975008 | C0016536
SRT | T-63000 | Gallbladder | 28231008 | C0016976
SRT | T-D8700 | Hand | 85562004 | C0018563
SRT | T-D1100 | Head | 69536005 | C0018670
SRT | T-D1000 | Head and Neck | 774007 | C0460004
SRT | T-32000 | Heart | 80891009 | C0018787
SRT | T-15710 | Hip joint | 29836001 | C0019558
SRT | T-12410 | Humerus | 85050009 | C0020164
SRT | T-58600 | Ileum | 34516001 | C0020885
SRT | T-12340 | Ilium | 22356005 | C0020889
SRT | T-AB959 | Internal Auditory Canal | 361078006 | C1283773
SRT | T-D1213 | Jaw region | 661005 | C0022359
SRT | T-58400 | Jejunum | 21306003 | C0022378
SRT | T-D9200 | Knee | 72696002 | C0022742
SRT | T-59000 | Large intestine | 14742008 | C0021851
SRT | T-24100 | Larynx | 4596009 | C0023078
SRT | T-D9400 | Lower leg | 30021000 | C1140621
SRT | T-D9000 | Lower limb | 61685007 | C0023216
SRT | T-11503 | Lumbar spine | 122496007 | C0024091
SRT | T-D00F9 | Lumbo-sacral spine | 297173004 | C0574025
SRT | T-11180 | Mandible | 91609006 | C0024687
SRT | T-11133 | Mastoid bone | 59066005 | C0446908
SRT | T-11170 | Maxilla | 70925003 | C0024947
SRT | T-D3300 | Mediastinum | 72410000 | C0025066
SRT | T-14668 | Muscle of lower limb | 102292000 | C0584890
SRT | T-13600 | Muscle of upper limb | 30608006 | C0559498
SRT | T-11149 | Nasal bone | 74386004 | C0027422
SRT | T-D1600 | Neck | 45048000 | C0027530
SRT | R-FAB52 | Neck and Chest | 417437006 | C1562459
SRT | R-FAB53 | Neck, Chest and Abdomen | 416152001 | C1562378
SRT | R-FAB54 | Neck, Chest, Abdomen and Pelvis | 416319003 | C1562776
SRT | T-11102 | Optic canal | 55024004 | C0450102
SRT | T-D14AE | Orbital structure | 363654007 | C0029180
SRT | T-65600 | Pancreatic duct and bile duct systems | 110621006 | C1267614
SRT | T-22000 | Paranasal sinus | 2095001 | C0030471
SRT | T-61100 | Parotid gland | 45289007 | C0030580
SRT | T-12730 | Patella | 64234005 | C0030647
SRT | T-D6000 | Pelvis | 12921003 | C0030797
SRT | R-FAB58 | Pelvis and lower extremities | 416631005 | C1562943
DCM | 113681 | Phantom |  | C0282611
SRT | T-92000 | Prostate | 41216001 | C0033572
SRT | T-59600 | Rectum | 34402009 | C0034896
SRT | T-11300 | Rib | 113197003 | C0035561
SRT | T-15680 | Sacroiliac joint | 39723000 | C0036036
SRT | T-11AD0 | Sacrum | 54735007 | C0036037
SRT | T-12280 | Scapula | 79601000 | C0036277
SRT | T-D1460 | Sella turcica | 42575006 | C0036609
SRT | T-12980 | Sesamoid bones of foot | 58742003 | C0278418
SRT | T-D2220 | Shoulder | 16982005 | C0037004
SRT | T-11100 | Skull | 89546000 | C0037303
SRT | T-58000 | Small intestine | 30315005 | C0021852
SRT | T-D04FF | Spine | 421060004 | C0037949
SRT | T-15610 | Sternoclavicular joint | 7844006 | C0038291
SRT | T-11210 | Sternum | 56873002 | C0038293
SRT | T-57000 | Stomach | 69695003 | C0038351
SRT | T-61300 | Submandibular gland | 54019009 | C0038556
SRT | T-15770 | Tarsal joint | 27949001 | C0039318
SRT | T-15290 | Temporomandibular joint | 53620006 | C0039493
SRT | T-D9100 | Thigh | 68367000 | C0039866
SRT | T-11502 | Thoracic spine | 122495006 | C0581269
SRT | T-D00F8 | Thoraco-lumbar spine | 297172009 | C0729374
SRT | T-D8810 | Thumb | 76505004 | C0040067
SRT | T-D9800 | Toe | 29707007 | C0040357
SRT | T-25000 | Trachea | 44567001 | C0040578
SRT | T-D8200 | Upper arm | 40983000 | C0446516
SRT | T-D8000 | Upper limb | 53120007 | C1140618
SRT | T-7000B | Upper urinary tract | 431491007 | C2317509
SRT | T-73000 | Ureter | 87953007 | C0041951
SRT | T-75000 | Urethra | 13648007 | C0041967
SRT | T-88920 | Uterus and fallopian tubes | 110639002 | C1267676
SRT | T-11011 | Vertebral column and cranium | 110517009 | C1266914
SRT | T-15460 | Wrist joint | 74670003 | C1322271
SRT | T-11166 | Zygoma | 13881006 | C0043539

### Note

1. In a prior version of this table, the code T-D1217 was specified for the concept "Maxilla and mandible". The use of this code conflicts with its assignment to another concept in SNOMED, and its use in this context is deprecated. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

2. In a prior version of this table, the code T-D8300 was used for T-15430, T-12402 for T-D8500, T-15710 for T-D2500, T-73800 for T-73000, and T-11167 for T-11166. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 4032 MR Spectroscopy Metabolites

Type:

  Extensible

Version:

  20040322

UID:

  1.2.840.10008.6.1.309

Table CID 4032. MR Spectroscopy Metabolites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 4033 “MR Proton Spectroscopy Metabolites”

## CID 4033 MR Proton Spectroscopy Metabolites

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.310

Table CID 4033. MR Proton Spectroscopy Metabolites

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-65C50 | N-acetylaspartate | 115391007 | C0067684
SRT | F-61080 | Citrate | 59351004 | C0376259
SRT | F-61620 | Choline | 65123005 | C0008405
SRT | F-61380 | Creatine | 14804005 | C0010286
DCM | 113094 | Creatine and Choline |  |
SRT | F-61760 | Lactate | 83036002 | C0376261
SRT | F-63600 | Lipid | 70106000 | C0023779
DCM | 113095 | Lipid and Lactate |  |
DCM | 113080 | Glutamate and glutamine |  |
SRT | F-64210 | Glutamine | 25761002 | C0017797
SRT | F-64460 | Tuarine | 10944007 | C0039350
SRT | F-61A90 | Inositol | 72164009 | C0021547
DCM | 113081 | Choline/Creatine Ratio |  |
DCM | 113082 | N-acetylaspartate/Creatine Ratio |  |
DCM | 113083 | N-acetylaspartate/Choline Ratio |  |
DCM | 113096 | Creatine+Choline/Citrate Ratio |  |

### Note

For the purpose of this context group, where possible, the resonance peak in the spectrum corresponding to a particular metabolite is described using the concept from SNOMED for the substance corresponding to the metabolite. E.g., the code used for "lipid" is the code for "lipid (substance) ", as this concept is effectively post-coordinated by its use in the Metabolite Map Code Sequence (0018,9083) to mean "lipid resonance peaks in MR spectroscopy".

## CID 4040 Endoscopy Anatomic Regions

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.311

Table CID 4040. Endoscopy Anatomic Regions

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-D4000 | Abdomen | 113345001 | C0000726
SRT | T-59490 | Anus, rectum and sigmoid colon | 110612005 | C1267595
SRT | T-60610 | Bile duct | 28273000 | C0005400
SRT | T-74000 | Bladder | 89837001 | C0005682
SRT | T-DD123 | Bladder and urethra | 110837003 | C1268386
SRT | T-26000 | Bronchus | 955009 | C0006255
SRT | T-83200 | Cervix | 71252005 | C0007874
SRT | T-D3000 | Chest | 51185008 | C0817096
SRT | T-DD163 | Esophagus, stomach and duodenum | 110861005 | C1268410
SRT | T-AB200 | External auditory canal | 84301002 | C0013444
SRT | T-63000 | Gallbladder | 28231008 | C0016976
SRT | T-D7000 | Inguinal region | 26893007 | C0018246
SRT | T-15001 | Joint | 39352004 | C0022417
SRT | T-71000 | Kidney | 64033007 | C0022646
SRT | T-D9200 | Knee | 72696002 | C0022742
SRT | T-59000 | Large intestine | 14742008 | C0021851
SRT | T-24100 | Larynx | 4596009 | C0023078
SRT | T-40230 | Lumen of blood vessel | 91747007 | C0524424
SRT | T-D3300 | Mediastinum | 72410000 | C0025066
SRT | T-2300C | Nasopharynx | 360955006 | C1283682
SRT | T-22000 | Paranasal sinus | 2095001 | C0030471
SRT | T-55000 | Pharynx | 54066008 | C0031354
SRT | T-20101 | Pharynx and larynx | 312535008 | C0729889
SRT | T-59600 | Rectum | 34402009 | C0034896
SRT | T-D2220 | Shoulder | 16982005 | C0037004
SRT | T-59470 | Sigmoid colon | 60184004 | C0227391
SRT | T-D04FF | Spine | 421060004 | C0037949
SRT | T-DD006 | Trachea and bronchus | 110726009 | C1268276
SRT | T-7000B | Upper urinary tract | 431491007 | C2317509
SRT | T-73000 | Ureter | 87953007 | C0041951
SRT | T-88920 | Uterus and fallopian tubes | 110639002 | C1267676

### Note

1. See Annex I for examples of the relationship between anatomic regions and type of endoscopy performed.

2. In a prior version of this table, the code T-55002 was used for T-55000, and T-73800 for T-73000. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 4042 XA/XRF Anatomy Imaged

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.312

Table CID 4042. XA/XRF Anatomy Imaged

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3010 “Cardiovascular Anatomic Locations”
Include CID 4031 “Common Anatomic Regions”

## CID 4050 Drug or Contrast Agent Characteristics

Type:

  Extensible

Version:

  20070124

UID:

  1.2.840.10008.6.1.313

Table CID 4050. Drug or Contrast Agent Characteristics

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-C52F | Active Ingredient | 127489000 | C1292749
DCM | 121380 | Active Ingredient Undiluted Concentration |  |
DCM | 121381 | Contrast/Bolus Ingredient Opaque |  |
SRT | G-D705 | Volume | 118565006 | C0449468

## CID 4051 General Devices

Type:

  Extensible

Version:

  20160525

UID:

  1.2.840.10008.6.1.314

Table CID 4051. General Devices

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 8 “Angiographic Interventional Devices”
Include CID 3451 “Calibration Objects”
Include CID 4052 “Phantom Devices”
SRT | A-10150 | Syringe | 61968008 | C0039142

## CID 4052 Phantom Devices

Type:

  Extensible

Version:

  20061023

UID:

  1.2.840.10008.6.1.315

Table CID 4052. Phantom Devices

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113681 | Phantom
DCM | 113682 | ACR Accreditation Phantom - CT
DCM | 113683 | ACR Accreditation Phantom - MR
DCM | 113684 | ACR Accreditation Phantom - Mammography
DCM | 113685 | ACR Accreditation Phantom - Stereotactic Breast Biopsy
DCM | 113686 | ACR Accreditation Phantom - ECT
DCM | 113687 | ACR Accreditation Phantom - PET
DCM | 113688 | ACR Accreditation Phantom - ECT/PET
DCM | 113689 | ACR Accreditation Phantom - PET Faceplate
DCM | 113690 | IEC Head Dosimetry Phantom
DCM | 113691 | IEC Body Dosimetry Phantom
DCM | 113692 | NEMA XR21-2000 Phantom

## CID 4100 T1 Measurement Methods

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.985

Table CID 4100. T1 Measurement Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126350 | T1 by Multiple Flip Angles
DCM | 126351 | T1 by Inversion Recovery
DCM | 126352 | T1 by Fixed Value

### Note

## CID 4101 Tracer Kinetic Models

Type:

  Extensible

Version:

  20160316

UID:

  1.2.840.10008.6.1.986

Table CID 4101. Tracer Kinetic Models

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126340 | Standard Tofts Model
DCM | 126341 | Extended Tofts Model
DCM | 126343 | First Pass Leakage Profile (FPLP) Model
DCM | 126344 | Shutter-Speed Model (SSM)
DCM | 126345 | Gamma Capillary Transit Time (GCTT) Model
DCM | 126346 | Adiabatic Tissue Homogeneity (ATH) Model
DCM | 126347 | Two Compartment Exchange (2CX) Model

### Note

## CID 4102 Perfusion Measurement Methods

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.987

Table CID 4102. Perfusion Measurement Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126300 | Perfusion analysis by Stable Xenon CT technique
DCM | 126301 | Perfusion analysis by IV Iodinated Contrast CT technique
DCM | 126302 | Perfusion analysis by Arterial Spin Labeling MR technique
DCM | 126303 | Perfusion analysis by Susceptibility MR technique

### Note

## CID 4103 Arterial Input Function Measurement Methods

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.988

Table CID 4103. Arterial Input Function Measurement Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126360 | AIF Ignored
DCM | 126361 | Population Averaged AIF
DCM | 126362 | User-defined AIF ROI
DCM | 126363 | Automatically Detected AIF ROI
DCM | 126364 | Blind Estimation of AIF

### Note

The anatomic location relevant to the application of any AIF method is not pre-coordinated in concepts in this Context Group. Typically these would be described by the Finding Site of any related measurements in the appropriate Template.

## CID 4104 Bolus Arrival Time Derivation Methods

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.989

Table CID 4104. Bolus Arrival Time Derivation Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126373 | Temporal Derivative Exceeds Threshold
DCM | 126370 | Time of Peak Concentration
DCM | 126372 | Time of Leading Half-Peak Concentration

### Note

## CID 4105 Perfusion Analysis Methods

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.990

Table CID 4105. Perfusion Analysis Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126310 | Least Mean Square (LMS) deconvolution
DCM | 126311 | Singular Value Decomposition (SVD) deconvolution

### Note

## CID 4106 Quantitative Methods used for Perfusion And Tracer Kinetic Models

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.991

Table CID 4106. Quantitative Methods used for Perfusion And Tracer Kinetic Models

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 4100 “T1 Measurement Methods”
Include CID 4101 “Tracer Kinetic Models”
Include CID 4102 “Perfusion Measurement Methods”
Include CID 4103 “Arterial Input Function Measurement Methods”
Include CID 4104 “Bolus Arrival Time Derivation Methods”
Include CID 4105 “Perfusion Analysis Methods”
DCM | 126342 | Model-free concentration-time quantitification

### Note

1. Concepts from this context group may be used in measurement Templates to describe the measurement method of measurement on an ROI.

   E.g., NUM (126312, DCM, "Ktrans") = 0.0185 /min; (G-C036, SRT, "Measurement Method") = (126341, DCM, "Extended Tofts Model")

## CID 4107 Tracer Kinetic Model Parameters

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.992

Table CID 4107. Tracer Kinetic Model Parameters

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126312 | Ktrans
DCM | 126313 | kep
DCM | 126314 | ve
DCM | 126330 | tau_m
DCM | 126331 | vp

### Note

## CID 4108 Perfusion Model Parameters

Type:

  Extensible

Version:

  20161106

UID:

  1.2.840.10008.6.1.993

Table CID 4108. Perfusion Model Parameters

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113055 | Regional Cerebral Blood Flow
DCM | 126390 | Regional Blood Flow
DCM | 113056 | Regional Cerebral Blood Volume
DCM | 126391 | Regional Blood Volume
DCM | 113052 | Mean Transit Time
DCM | 113069 | Time To Peak
DCM | 126392 | Oxygen Extraction Fraction
DCM | 113084 | Tmax

### Note

## CID 4109 Model-Independent Dynamic Contrast Analysis Parameters

Type:

  Extensible

Version:

  20150916

UID:

  1.2.840.10008.6.1.994

Table CID 4109. Model-Independent Dynamic Contrast Analysis Parameters

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126320 | IAUC
DCM | 126321 | IAUC60
DCM | 126322 | IAUC90
DCM | 126323 | IAUC180
DCM | 126324 | IAUCBN
DCM | 126325 | IAUC60BN
DCM | 126326 | IAUC90BN
DCM | 126327 | IAUC180BN
DCM | 126370 | Time of Peak Concentration
DCM | 126372 | Time of Leading Half-Peak Concentration
DCM | 126371 | Bolus Arrival Time
DCM | 113069 | Time To Peak
DCM | 126374 | Temporal Derivative Threshold
DCM | 126375 | Maximum Slope
DCM | 126376 | Maximum Difference
DCM | 126377 | Tracer Concentration

### Note

(126326, DCM, "IAUC90BN") can be used for DCE-MRI using a Gd-based contrast agent to represent the IAUGCBN measurement in the claim of the QIBA DCE MRI Quantification Profile, though the concept itself is not specific to the modality or the contast agent used. See https://www.rsna.org/QIBA_Protocols_and_Profiles.aspx. See also Ng, CS., et al. "Reproducibility of Perfusion Parameters in Dynamic Contrast-Enhanced MRI of Lung and Liver Tumors: Effect on Estimates of Patient Sample Size in Clinical Trials and on Individual Patient Responses." AJR 194, no. 2 (February 1, 2010): W134–40. http://dx.doi.org/10.2214/AJR.09.3116.

The type of contrast agent and the AIF used for blood normalization may or may not be post-coordinated.

E.g., voxel-wise IAUCBN measurements encoded as a parametric map with the quantity defined by the Quantity Definition Sequence (0040,9220) in a Real World Value Map might be encoded as:

(G-C1C6, SRT, "Quantity") = (126326, DCM, "IAUC90BN")

(G-C036, SRT, "Measurement Method") = (126362, DCM, "User-defined AIF ROI")

(123011, DCM, "Contrast Bolus/Agent") = (C-17800, SRT, "Gadolinium")

E.g., an IAUCBN measurement for an ROI encoded in a structured report might be encoded as:

NUM (126326, DCM, "IAUC90BN") = 0.230 (UNITS = ({normalized}, UCUM, "normalized")

>HAS CONCEPT MOD: CODE (G-C036, SRT, "Measurement Method") = (126364, DCM, "Blind Estimation of AIF")

Note that the generic ROI measurement templates do not have the contrast/bolus agent as a parameter; this may be implicit from context, or inherited from the (121058, DCM,"Procedure reported") in the parent template.

## CID 4110 Tracer Kinetic Modeling Covariates

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.995

Table CID 4110. Tracer Kinetic Modeling Covariates

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID
--- | --- | --- | ---
LN | 20570-8 | Hematocrit | C0803379

## CID 4111 Contrast Characteristics

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.996

Table CID 4111. Contrast Characteristics

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126380 | Contrast Longitudinal Relaxivity

## CID 4200 Ophthalmic Imaging Agent

Type:

  Extensible

Version:

  20040921

UID:

  1.2.840.10008.6.1.316

Table CID 4200. Ophthalmic Imaging Agent

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-B02CC | Fluorescein | 350086004 | C0060520
SRT | C-B0156 | Indocyanine green | 7292004 | C0021234
SRT | C-B0295 | Rose Bengal | 330888007 | C0035857
SRT | C-22853 | Trypan blue | 60441008 | C0041213
SRT | C-B02C5 | Methylene blue | 354064008 | C0025746

## CID 4201 Patient Eye Movement Command

Type:

  Extensible

Version:

  20040921

UID:

  1.2.840.10008.6.1.317

Table CID 4201. Patient Eye Movement Command

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-1022D | Primary gaze | 408744005 | C1443287
SRT | R-404BF | Upward gaze | 255533007 | C0439774
SRT | R-404B9 | Left upgaze | 255525006 | C0439769
SRT | R-404BC | Left gaze | 255530005 | C0439773
SRT | R-404B7 | Left downgaze | 255523004 | C0439772
SRT | R-404B6 | Downgaze | 255521002 | C0439777
SRT | R-404B8 | Right downgaze | 255524005 | C0439763
SRT | R-404BD | Right gaze | 255531009 | C0439765
SRT | R-404BA | Right upgaze | 255526007 | C0439760
SRT | R-10227 | Convergent gaze | 408745006 | C1446614

## CID 4202 Ophthalmic Photography Acquisition Device

Type:

  Extensible

Version:

  20100607

UID:

  1.2.840.10008.6.1.318

Table CID 4202. Ophthalmic Photography Acquisition Device

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-1021A | Fundus Camera | 409898007 | C0179536
SRT | A-2B201 | Slit Lamp Biomicroscope | 397247004 | C0183355
SRT | R-1021B | External Camera | 409903006 | C1444146
SRT | R-1021C | Specular Microscope | 409899004 | C1444145
SRT | A-2B210 | Operating Microscope | 102321001 | C0181849
SRT | A-00E8A | Scanning Laser Ophthalmoscope | 392001008 | C0392288
SRT | R-1021D | Indirect Ophthalmoscope | 409901008 | C0182048
SRT | R-1021E | Direct Ophthalmoscope | 409900009 | C0182047
SRT | R-1021F | Ophthalmic Endoscope | 409902001 | C0493036
SRT | A-00FCA | Keratoscope | 397522002 | C0181448
SRT | A-00FF4 | Pupillograph | 420827006 | C0182567

## CID 4203 Ophthalmic Photography Illumination

Type:

  Extensible

Version:

  20100607

UID:

  1.2.840.10008.6.1.319

Table CID 4203. Ophthalmic Photography Illumination

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-1020E | Dual diffuse direct illumination | 410461001 | C1444589
SRT | R-1020F | Fine slit beam direct illumination | 410462008 | C1444590
SRT | R-10211 | Broad tangential direct illumination | 410463003 | C1444591
SRT | R-10213 | Indirect sclerotic scatter illumination | 410464009 | C1444592
SRT | R-10215 | Indirect retroillumination from the iris | 410465005 | C1444593
SRT | R-10217 | Indirect retroillumination from the retina | 410466006 | C1444594
SRT | R-10218 | Indirect iris transillumination | 410467002 | C1444595
DCM | 111625 | Diffuse direct illumination |  |
DCM | 111627 | Scotopic light |  |
DCM | 111628 | Mesopic light |  |
DCM | 111629 | Photopic light |  |
DCM | 111630 | Dynamic light |  |

Reference: From the OPS web site: http://www.opsweb.org/Op-Photo/SlitLamp/SL/SlitLamp.htm

## CID 4204 Ophthalmic Filter

Type:

  Extensible

Version:

  20110112

UID:

  1.2.840.10008.6.1.320

Table CID 4204. Ophthalmic Filter

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-010E2 | Green optical filter | 445465004 | C2919396
SRT | A-010DF | Red optical filter | 445279009 | C2919397
SRT | A-010DA | Blue optical filter | 445084008 | C2919751
SRT | A-010E0 | Yellow-green optical filter | 445340000 | C2919190
SRT | A-010D8 | Blue-green optical filter | 422915004 | C1828251
SRT | A-010DC | Infrared optical filter | 445169002 | C2919637
SRT | A-010E1 | Polarizing optical filter | 445391002 | C2919554
DCM | 111609 | No filter |  |

## CID 4205 Ophthalmic Lens

Type:

  Extensible

Version:

  20040921

UID:

  1.2.840.10008.6.1.321

Table CID 4205. Ophthalmic Lens

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-10219 | Indirect ophthalmoscopy lens | 409897002 | C1444144
SRT | R-10239 | Concave contact fundus lens | 409783000 | C1444081
SRT | R-1023A | Concave noncontact fundus lens | 410688004 | C1444761
SRT | R-1023B | Contact fundus lens | 410686000 | C1444759
SRT | A-00FAD | Goniolens | 389156006 | C1300255
SRT | R-1023D | Convex noncontact fundus lens | 410687009 | C1444760
SRT | R-1023E | Noncontact fundus lens | 410685001 | C1444758
SRT | R-1023C | Convex contact fundus lens | 410689007 | C1444762

## CID 4206 Ophthalmic Channel Description

Type:

  Extensible

Version:

  20040921

UID:

  1.2.840.10008.6.1.322

Table CID 4206. Ophthalmic Channel Description

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A12F | Blue | 405738005 | C1260957
SRT | R-102C0 | Full Spectrum | 414298005 | C1532530
SRT | G-A11E | Green | 371246006 | C0332583
SRT | R-102BE | Infrared | 414497003 | C1532326
SRT | G-A11A | Red | 371240000 | C1260956
SRT | G-A132 | Red free | 405739002 | C1319009
SRT | R-102BF | Ultraviolet | 415770004 | C1532472

## CID 4207 Ophthalmic Image Position

Type:

  Extensible

Version:

  20110825

UID:

  1.2.840.10008.6.1.323

Table CID 4207. Ophthalmic Image Position

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-10229 | Diabetic Retinopathy Study field 1 | 408734008 | C1443282
SRT | R-1022A | Diabetic Retinopathy Study field 2 | 410434001 | C1444567
SRT | R-1022B | Diabetic Retinopathy Study field 3 | 410435000 | C1444568
SRT | R-1022C | Diabetic Retinopathy Study field 4 | 410436004 | C1444569
SRT | R-1022E | Diabetic Retinopathy Study field 5 | 410437008 | C1444570
SRT | R-1022F | Diabetic Retinopathy Study field 6 | 410438003 | C1444571
SRT | R-10231 | Diabetic Retinopathy Study field 7 | 410439006 | C1444572
DCM | 111621 | Field 1 for Joslin3 field |  |
DCM | 111622 | Field 2 for Joslin 3 field |  |
DCM | 111623 | Field 3 for Joslin 3 field |  |
DCM | 111900 | Macula centered |  |
DCM | 111901 | Disc centered |  |
DCM | 111902 | Lesion centered |  |
DCM | 111903 | Disc-macula centered |  |
DCM | 111904 | Mid-peripheral-superior |  |
DCM | 111905 | Mid-peripheral-superior temporal |  |
DCM | 111906 | Mid-peripheral-temporal |  |
DCM | 111907 | Mid-peripheral-inferior temporal |  |
DCM | 111908 | Mid-peripheral-inferior |  |
DCM | 111909 | Mid-peripheral-inferior nasal |  |
DCM | 111910 | Mid-peripheral-nasal |  |
DCM | 111911 | Mid-peripheral-superior nasal |  |
DCM | 111912 | Peripheral-superior |  |
DCM | 111913 | Peripheral-superior temporal |  |
DCM | 111914 | Peripheral-temporal |  |
DCM | 111915 | Peripheral-inferior temporal |  |
DCM | 111916 | Peripheral-inferior |  |
DCM | 111917 | Peripheral-inferior nasal |  |
DCM | 111918 | Peripheral-nasal |  |
DCM | 111919 | Peripheral-superior nasal |  |

## CID 4208 Mydriatic Agent

Type:

  Extensible

Version:

  20040921

UID:

  1.2.840.10008.6.1.324

Table CID 4208. Mydriatic Agent

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-677B9 | Atropine | 349947003 | C0360182
SRT | C-677C0 | Homatropine | 82264009 | C0062922
SRT | C-97520 | Cyclopentolate | 8348002 | C0010582
SRT | C-68165 | Phenylephrine | 386693003 | C0717985
SRT | C-97580 | Tropicamide | 9190005 | C0041190

## CID 4209 Ophthalmic Anatomic Structure Imaged

Type:

  Extensible

Version:

  20040921

UID:

  1.2.840.10008.6.1.325

Table CID 4209. Ophthalmic Anatomic Structure Imaged

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-AA050 | Anterior chamber of eye | 31636006 | C0003151
SRT | T-AA180 | Both eyes | 40638003 | C0229118
SRT | T-AA310 | Choroid of eye | 68703001 | C0008520
SRT | T-AA400 | Ciliary body | 29534007 | C0008779
SRT | T-AA860 | Conjunctiva | 29445007 | C0009758
SRT | T-AA200 | Cornea | 28726007 | C0010031
SRT | T-AA000 | Eye | 81745001 | C0015392
SRT | T-AA810 | Eyelid | 80243003 | C0015426
SRT | T-AA621 | Fovea centralis | 67046006 | C0016622
SRT | T-AA500 | Iris | 41296002 | C0022077
SRT | T-AA862 | Lacrimal caruncle | 43045000 | C0446860
SRT | T-AA910 | Lacrimal gland | 13561001 | C0022907
SRT | T-AA940 | Lacrimal sac | 3954005 | C0229289
SRT | T-AA700 | Lens | 78076003 | C0023317
SRT | T-AA830 | Lower Eyelid | 62736007 | C0229258
SRT | T-45400 | Ophthalmic artery | 53549008 | C0029078
SRT | T-AA630 | Optic nerve head | 81016008 | C0029127
SRT | T-AA610 | Retina | 5665001 | C0035298
SRT | T-AA110 | Sclera | 18619003 | C0036410
SRT | T-AA820 | Upper Eyelid | 38934000 | C0585636

## CID 4210 Ophthalmic Tomography Acquisition Device

Type:

  Extensible

Version:

  20131014

UID:

  1.2.840.10008.6.1.326

Table CID 4210. Ophthalmic Tomography Acquisition Device

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-00FBE | Optical Coherence Tomography Scanner | 392012008 | C1271441
SRT | R-FAB5A | Retinal Thickness Analyzer | 416567007 | C1562933
SRT | A-00E8B | Confocal Scanning Laser Ophthalmoscope | 392004000 | C1271438
DCM | 111626 | Scheimpflug Camera |  |
SRT | A-00E8C | Scanning Laser Polarimeter | 392007007 | C1271440
DCM | 111945 | Elevation-based corneal tomographer |  |
DCM | 111946 | Reflection-based corneal topographer |  |
DCM | 111947 | Interferometry-based corneal tomographer |  |

## CID 4211 Ophthalmic OCT Anatomic Structure Imaged

Type:

  Extensible

Version:

  20071016

UID:

  1.2.840.10008.6.1.327

Table CID 4211. Ophthalmic OCT Anatomic Structure Imaged

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-AA050 | Anterior chamber of eye | 31636006 | C0003151
SRT | T-AA310 | Choroid of eye | 68703001 | C0008520
SRT | T-AA400 | Ciliary body | 29534007 | C0008779
SRT | T-AA860 | Conjunctiva | 29445007 | C0009758
SRT | T-AA200 | Cornea | 28726007 | C0010031
SRT | T-AA500 | Iris | 41296002 | C0022077
SRT | T-AA700 | Lens | 78076003 | C0023317
SRT | T-AA630 | Optic nerve head | 81016008 | C0029127
SRT | T-AA610 | Retina | 5665001 | C0035298
SRT | T-AA110 | Sclera | 18619003 | C0036410
SRT | T-AA079 | Vitreous | 26386000 | C0229095
SRT | T-AA220 | Corneal epithelium | 15775008 | C0459875
SRT | T-AA260 | Corneal endothelium | 65431007 | C0014258

## CID 4214 Ophthalmic Horizontal Directions

Type:

  Extensible

Version:

  20080124

UID:

  1.2.840.10008.6.1.800

Table CID 4214. Ophthalmic Horizontal Directions

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-C028 | Inward | 255460003 | C0439786
SRT | R-404C7 | Outward | 255543005 | C0439788

## CID 4215 Ophthalmic Vertical Directions

Type:

  Extensible

Version:

  20080124

UID:

  1.2.840.10008.6.1.801

Table CID 4215. Ophthalmic Vertical Directions

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-404BE | Up | 255532002 | C0547043
SRT | R-404B3 | Down | 255518004 | C0205104

## CID 4216 Ophthalmic Visual Acuity Type

Type:

  Extensible

Version:

  20080124

UID:

  1.2.840.10008.6.1.802

Table CID 4216. Ophthalmic Visual Acuity Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111685 | Autorefraction Visual Acuity |  |
DCM | 111686 | Habitual Visual Acuity |  |
DCM | 111687 | Prescription Visual Acuity |  |
SRT | F-04ECE | Potential Acuity Meter Visual Acuity | 424622008 | C1827765
SRT | F-04D54 | Best Corrected Visual Acuity | 419775003 | C1690532
SRT | F-04D53 | Uncorrected Visual Acuity | 420050001 | C1637380
SRT | F-04D55 | Pinhole Visual Acuity | 419475002 | C1642831
SRT | F-04ECF | Brightness Acuity Testing Visual Acuity | 425141002 | C1827482

## CID 4220 Visual Fixation Quality During Acquisition

Type:

  Extensible

Version:

  20090917

UID:

  1.2.840.10008.6.1.819

Table CID 4220. Visual Fixation Quality During Acquisition

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A555 | Steady | 55011004 | C0205361
SRT | G-A556 | Not Steady | 103361006 | C0439829
SRT | G-A385 | Indeterminate | 82334004 | C0205258

## CID 4221 Visual Fixation Quality Problem

Type:

  Extensible

Version:

  20090917

UID:

  1.2.840.10008.6.1.820

Table CID 4221. Visual Fixation Quality Problem

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 110518 | Patient Movement |  |
SRT | F-02FA4 | Eccentric Fixation | 251786004 | C0429578
DCM | 110519 | Operator Error |  |
DCM | 110501 | Equipment failure |  |

## CID 4222 Ophthalmic Macular Grid Problem

Type:

  Extensible

Version:

  20090917

UID:

  1.2.840.10008.6.1.821

Table CID 4222. Ophthalmic Macular Grid Problem

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 4221 “Visual Fixation Quality Problem” |  |
SRT | F-0123A | Constricted Pupil | 301939004 | C0728710
SRT | DA-73402 | Lens Opacity | 193570009 | C0086543
SRT | DA-75300 | Corneal Opacity | 64634000 | C0010038
SRT | DA-7931D | Vitreous Opacity | 422061002 | C0152006
SRT | R-20839 | Poor Visual Fixation | 314348007 | C1277657
SRT | DA-76000 | Eyelid Disease | 60113004 | C0015423
DCM | 111695 | Interfering Tears or Drops |  |
SRT | DA-74100 | Refractive Error | 39021009 | C0034951
DCM | 111209 | Patient Positioning Problem |  |
SRT | F-F1722 | Dry Eyes Problem | 162290004 | C0314719

## CID 4230 Ophthalmic Ultrasound Axial Measurements Type

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.876

Table CID 4230. Ophthalmic Ultrasound Axial Measurements Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111750 | Ultrasound Contact
DCM | 111751 | Ultrasound Immersion

## CID 4231 Lens Status

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.877

Table CID 4231. Lens Status

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | DA-73410 | Aphakic | 24010005 | C0003534
SRT | R-2073F | Phakic | 309649001 | C0587139
SRT | A-040F7 | Phakic IOL | 397559001 | C1301524
SRT | F-02087 | Piggyback IOL | 370951003 | C1299686
SRT | DA-73460 | Pseudophakia | 95217000 | C0684343

## CID 4232 Vitreous Status

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.878

Table CID 4232. Vitreous Status

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-035F3 | Gas in vitreous cavity | 247094004 | C0423372
SRT | DA-7930D | Post-Vitrectomy | 232077005 | C0339563
SRT | F-035FD | Silicone Oil | 247095003 | C0423373
SRT | T-AA092 | Vitreous Only | 372242005 | C1299205

## CID 4233 Ophthalmic Axial Length Measurements Segment Names

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.879

Table CID 4233. Ophthalmic Axial Length Measurements Segment Names

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-AA200 | Cornea | 28726007 | C0010031
SRT | T-AA050 | Anterior Chamber | 31636006 | C0003151
DCM | 111778 | Single or Anterior Lens |  |
DCM | 111779 | Posterior Lens |  |
SRT | T-AA079 | Vitreous Cavity | 26386000 | C0229095

## CID 4234 Refractive Surgery Types

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.880

Table CID 4234. Refractive Surgery Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P1-A3102 | RK | 51683002 | C0022607
SRT | P1-A3835 | PRK | 397516006 | C0395416
SRT | P0-0526F | LASIK | 312965008 | C0752094
SRT | P1-A3846 | LASEK | 414582004 | C1449939

## CID 4235 Keratometry Descriptors

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.881

Table CID 4235. Keratometry Descriptors

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111753 | Manual Keratometry
DCM | 111754 | Auto Keratometry
DCM | 111755 | Simulated Keratometry
DCM | 111756 | Equivalent K-reading

## CID 4236 IOL Calculation Formula

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.882

Table CID 4236. IOL Calculation Formula

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111760 | Haigis
DCM | 111761 | Haigis-L
DCM | 111762 | Holladay 1
DCM | 111763 | Holladay 2
DCM | 111764 | Hoffer Q
DCM | 111765 | Olsen
DCM | 111766 | SRKII
DCM | 111767 | SRK-T

## CID 4237 Lens Constant Type

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.883

Table CID 4237. Lens Constant Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-048FA | A-Constant | 397263007 | C1301307
DCM | 111768 | ACD Constant |  |
DCM | 111769 | Haigis a0 |  |
DCM | 111770 | Haigis a1 |  |
DCM | 111771 | Haigis a2 |  |
DCM | 111772 | Hoffer pACD Constant |  |
DCM | 111773 | Surgeon Factor |  |

## CID 4238 Refractive Error Types

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.884

Table CID 4238. Refractive Error Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | DA-74120 | Myopia | 57190000 | C0027092
SRT | DA-74110 | Hyperopia | 38101003 | C0020490

## CID 4239 Anterior Chamber Depth Definition

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.885

Table CID 4239. Anterior Chamber Depth Definition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111776 | Front Of Cornea To Front Of Lens
DCM | 111777 | Back Of Cornea To Front Of Lens

## CID 4240 Ophthalmic Measurement or Calculation Data Source

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.886

Table CID 4240. Ophthalmic Measurement or Calculation Data Source

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111780 | Measurement From This Device
DCM | 113857 | Manual Entry
DCM | 111781 | External Data Source
DCM | 111782 | Axial Measurements SOP Instance
DCM | 111783 | Refractive Measurements SOP Instance

## CID 4241 Ophthalmic Axial Length Selection Method

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.887

Table CID 4241. Ophthalmic Axial Length Selection Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121412 | Mean value chosen
DCM | 121410 | User chosen value

## CID 4243 Ophthalmic Quality Metric Type

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.889

Table CID 4243. Ophthalmic Quality Metric Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111786 | Standard Deviation of measurements used
DCM | 111787 | Signal to Noise Ratio

## CID 4244 Ophthalmic Agent Concentration Units

Type:

  Extensible

Version:

  20100623

UID:

  1.2.840.10008.6.1.890

Table CID 4244. Ophthalmic Agent Concentration Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | % | Percent
UCUM | mg/ml | mg/ml

## CID 4245 Wide Field Ophthalmic Photography Transformation Method

Type:

  Extensible

Version:

  20150326

UID:

  1.2.840.10008.6.1.1029

Table CID 4245. Wide Field Ophthalmic Photography Transformation Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111791 | Spherical projection
DCM | 111792 | Surface contour mapping

## CID 4250 Visual Field Static Perimetry Test Patterns

Type:

  Extensible

Version:

  20100827

UID:

  1.2.840.10008.6.1.909

Table CID 4250. Visual Field Static Perimetry Test Patterns

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111800 | Visual Field 24-2 Test Pattern
DCM | 111801 | Visual Field 10-2 Test Pattern
DCM | 111802 | Visual Field 30-2 Test Pattern
DCM | 111803 | Visual Field 60-4 Test Pattern
DCM | 111804 | Visual Field Macula Test Pattern
DCM | 111805 | Visual Field Central 40 Point Test Pattern
DCM | 111806 | Visual Field Central 76 Point Test Pattern
DCM | 111807 | Visual Field Peripheral 60 Point Test Pattern
DCM | 111808 | Visual Field Full Field 81 Point Test Pattern
DCM | 111809 | Visual Field Full Field 120 Point Test Pattern
DCM | 111810 | Visual Field G Test Pattern
DCM | 111811 | Visual Field M Test Pattern
DCM | 111812 | Visual Field 07 Test Pattern
DCM | 111813 | Visual Field LVC Test Pattern
DCM | 111814 | Visual Field Central Test Pattern

## CID 4251 Visual Field Static Perimetry Test Strategies

Type:

  Extensible

Version:

  20100827

UID:

  1.2.840.10008.6.1.910

Table CID 4251. Visual Field Static Perimetry Test Strategies

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111815 | Visual Field SITA-Standard Test Strategy
DCM | 111816 | Visual Field SITA-SWAP Test Strategy
DCM | 111817 | Visual Field SITA-Fast Test Strategy
DCM | 111818 | Visual Field Full Threshold Test Strategy
DCM | 111819 | Visual Field FastPac Test Strategy
DCM | 111820 | Visual Field Full From Prior Test Strategy
DCM | 111821 | Visual Field Optima Test Strategy
DCM | 111822 | Visual Field Two-Zone Test Strategy
DCM | 111823 | Visual Field Three-Zone Test Strategy
DCM | 111824 | Visual Field Quantify-Defects Test Strategy
DCM | 111825 | Visual Field TOP Test Strategy
DCM | 111826 | Visual Field Dynamic Test Strategy
DCM | 111827 | Visual Field Normal Test Strategy
DCM | 111828 | Visual Field 1-LT Test Strategy
DCM | 111829 | Visual Field 2-LT Test Strategy
DCM | 111830 | Visual Field LVS Test Strategy
DCM | 111831 | Visual Field GATE Test Strategy
DCM | 111832 | Visual Field GATEi Test Strategy
DCM | 111833 | Visual Field 2LT-Dynamic Test Strategy
DCM | 111834 | Visual Field 2LT-Normal Test Strategy
DCM | 111835 | Visual Field Fast Threshold Test Strategy
DCM | 111836 | Visual Field CLIP Test Strategy
DCM | 111837 | Visual Field CLASS Strategy

## CID 4252 Visual Field Static Perimetry Screening Test Modes

Type:

  Extensible

Version:

  20100827

UID:

  1.2.840.10008.6.1.911

Table CID 4252. Visual Field Static Perimetry Screening Test Modes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111838 | Age corrected
DCM | 111839 | Threshold related
DCM | 111840 | Single luminance
DCM | 111841 | Foveal sensitivity related
DCM | 111842 | Related to non macular sensitivity
DCM | 121410 | User chosen value

## CID 4253 Visual Field Static Perimetry Fixation Strategy

Type:

  Extensible

Version:

  20100827

UID:

  1.2.840.10008.6.1.912

Table CID 4253. Visual Field Static Perimetry Fixation Strategy

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111843 | Automated Optical |  |
DCM | 111844 | Blind Spot Monitoring |  |
DCM | 111845 | Macular Fixation Testing |  |
DCM | 111846 | Observation by Examiner |  |
SRT | R-40775 | None | 260413007 | C0549184

## CID 4254 Visual Field Static Perimetry Test Analysis Results

Type:

  Extensible

Version:

  20100827

UID:

  1.2.840.10008.6.1.913

Table CID 4254. Visual Field Static Perimetry Test Analysis Results

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111847 | Outside normal limits |  |
DCM | 111848 | Borderline |  |
DCM | 111849 | Abnormally high sensitivity |  |
DCM | 111850 | General reduction in sensitivity |  |
DCM | 111851 | Borderline and general reduction in sensitivity |  |
SRT | M-00101 | Within normal limits | 125112009 | C1265570

## CID 4255 Visual Field Illumination Color

Type:

  Extensible

Version:

  20100827

UID:

  1.2.840.10008.6.1.914

Table CID 4255. Visual Field Illumination Color

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A11D | Yellow | 371244009 | C0221205
SRT | G-A12B | White | 371251000 | C0220938
SRT | G-A11A | Red | 371240000 | C1260956
SRT | G-A12F | Blue | 405738005 | C1260957
SRT | G-A11E | Green | 371246006 | C0332583

## CID 4256 Visual Field Procedure Modifier

Type:

  Extensible

Version:

  20100827

UID:

  1.2.840.10008.6.1.915

Table CID 4256. Visual Field Procedure Modifier

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-42453 | Screening | 360156006 | C1305399
SRT | R-408C3 | Diagnostic | 261004008 | C0348026

## CID 4257 Visual Field Global Index Name

Type:

  Extensible

Version:

  20100827

UID:

  1.2.840.10008.6.1.916

Table CID 4257. Visual Field Global Index Name

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111852 | Visual Field Index
DCM | 111853 | Visual Field Loss Due to Diffuse Defect
DCM | 111854 | Visual Field Loss Due to Local Defect
DCM | 111855 | Glaucoma Hemifield Test Analysis
DCM | 111856 | Optical Fixation Measurements

## CID 4260 Ophthalmic Mapping Units for Real World Value Mapping

Type:

  Extensible

Version:

  20110825

UID:

  1.2.840.10008.6.1.936

Table CID 4260. Ophthalmic Mapping Units for Real World Value Mapping

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | um | micrometer

## CID 4261 Ophthalmic Mapping Acquisition Method

Type:

  Extensible

Version:

  20110825

UID:

  1.2.840.10008.6.1.937

Table CID 4261. Ophthalmic Mapping Acquisition Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111920 | Time domain
DCM | 111921 | Spectral domain
DCM | 111922 | No corneal compensation
DCM | 111923 | Corneal birefringence compensation
DCM | 111924 | Retinal topography

## CID 4262 Retinal Thickness Definition

Type:

  Extensible

Version:

  20110825

UID:

  1.2.840.10008.6.1.938

Table CID 4262. Retinal Thickness Definition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111925 | Retinal nerve fiber layer thickness
DCM | 111926 | Ganglion cell complex thickness
DCM | 111927 | Total retinal thickness (ILM to IS-OS)
DCM | 111928 | Total retinal thickness (ILM to RPE)
DCM | 111929 | Total retinal thickness (ILM to BM)

## CID 4263 Ophthalmic Thickness Map Value Type

Type:

  Extensible

Version:

  20110825

UID:

  1.2.840.10008.6.1.939

Table CID 4263. Ophthalmic Thickness Map Value Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111930 | Absolute ophthalmic thickness
DCM | 111931 | Thickness deviation category from normative data
DCM | 111932 | Thickness deviation from normative data

## CID 4264 Ophthalmic Map Purposes of Reference

Type:

  Extensible

Version:

  20110825

UID:

  1.2.840.10008.6.1.940

Table CID 4264. Ophthalmic Map Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121311 | Localizer
DCM | 121322 | Source image for image processing operation
DCM | 111933 | Related ophthalmic thickness map

## CID 4265 Ophthalmic Thickness Deviation Categories

Type:

  Extensible

Version:

  20110825

UID:

  1.2.840.10008.6.1.941

Table CID 4265. Ophthalmic Thickness Deviation Categories

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111935 | p>5%
DCM | 111936 | p<5%
DCM | 111937 | p<2%
DCM | 111938 | p<1%
DCM | 111939 | p<0.5%

## CID 4266 Ophthalmic Anatomic Structure Reference Point

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.942

Table CID 4266. Ophthalmic Anatomic Structure Reference Point

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-01000 | Morphologically Abnormal Structure | 49755003 | C0332447
SRT | M-01100 | Lesion | 52988006 | C0221198
SRT | T-AA621 | Fovea centralis | 67046006 | C0016622
SRT | T-AA630 | Optic nerve head | 81016008 | C0029127
DCM | 111934 | Disc-Fovea |  |
SRT | T-AA200 | Cornea | 28726007 | C0010031

### Note

(M-01000, SRT, "Morphologically Abnormal Structure") was previously described with a Code Meaning of "Lesion", but that synonym has been retired as "inappropriate" in SNOMED. The Code Meaning has been replaced with the preferred SNOMED term, and the separate concept (M-01100, SRT, "Lesion") added.

## CID 4267 Corneal Topography Mapping Units for Real World Value Mapping

Type:

  Extensible

Version:

  20131014

UID:

  1.2.840.10008.6.1.965

Table CID 4267. Corneal Topography Mapping Units for Real World Value Mapping

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | um | micrometer
UCUM | diop | diopters
UCUM | mm | mm

## CID 4268 Corneal Topography Map Value Type

Type:

  Extensible

Version:

  20131014

UID:

  1.2.840.10008.6.1.966

Table CID 4268. Corneal Topography Map Value Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111940 | Corneal axial power map
DCM | 111941 | Corneal instantaneous power map
DCM | 111942 | Corneal refractive power map
DCM | 111943 | Corneal elevation map
DCM | 111944 | Corneal wavefront map

## CID 4270 OCT-A Processing Algorithm Families

Type:

  Extensible

Version:

  20170405

UID:

  1.2.840.10008.6.1.1150

Table CID 4270. OCT-A Processing Algorithm Families

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128252 | OCT-A amplitude decorrelation
DCM | 128253 | OCT-Acomplex variance
DCM | 128254 | OCT-Aspeckle variance
DCM | 128255 | OCT-Acorrelation mapping
DCM | 128256 | Doppler OCT-A

## CID 4271 En Face Image Types

Type:

  Extensible

Version:

  20170405

UID:

  1.2.840.10008.6.1.1151

Table CID 4271. En Face Image Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128257 | Retina depth encoded vasculature flow
DCM | 128258 | Retina depth encoded structural reflectance map
DCM | 128259 | Retina vasculature flow
DCM | 128260 | Retina structural reflectance map
DCM | 128261 | Vitreous vasculature flow
DCM | 128262 | Vitreous structural reflectance map
DCM | 128263 | Radial peripapillary vasculature flow
DCM | 128264 | Radial peripapillary structural reflectance map
DCM | 128265 | Superficial retina vasculature flow
DCM | 128266 | Superficial retina structural reflectance map
DCM | 128267 | Middle inner retina vasculature flow
DCM | 128268 | Middle inner structural reflectance map
DCM | 128269 | Deep retina vasculature flow
DCM | 128270 | Deep retina structural reflectance map
DCM | 128271 | Outer retina vasculature flow
DCM | 128272 | Outer retina structural reflectance map
DCM | 128273 | Choriocapillaris vasculature flow
DCM | 128274 | Choriocapillaris structural reflectance map
DCM | 128275 | Choroid vasculature flow
DCM | 128276 | Choroid structural reflectance map
DCM | 128277 | Whole eye vasculature flow
DCM | 128278 | Whole eye structural reflectance map

## CID 4272 Opt Scan Pattern Types

Type:

  Extensible

Version:

  20170405

UID:

  1.2.840.10008.6.1.1152

Table CID 4272. OPT Scan Pattern Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128279 | Cube B-scan pattern
DCM | 128280 | Raster B-scan pattern
DCM | 128281 | Line B-scan pattern
DCM | 128282 | Radial B-scan pattern
DCM | 128283 | Cross B-scan pattern
DCM | 128284 | Circle B-scan pattern
DCM | 128285 | Concentric circle B-scan pattern
DCM | 128286 | Circle-raster B-scan pattern
DCM | 128287 | Circle-radial B-scan pattern
DCM | 128288 | Grid B-scan pattern

## CID 4273 Retinal Segmentation Surfaces

Type:

  Extensible

Version:

  20170405

UID:

  1.2.840.10008.6.1.1153

Table CID 4273. Retinal Segmentation Surfaces

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-AA62D | ILM - Internal limiting membrane | 280677004 | C0459664
DCM | 128289 | Outer surface of RNFL |  |
DCM | 128290 | Outer surface of GCL |  |
DCM | 128291 | Outer surface of IPL |  |
DCM | 128292 | Outer surface of INL |  |
DCM | 128293 | Outer surface of OPL |  |
DCM | 128294 | Outer surface of HFL |  |
SRT | T-AA650 | ELM - External limiting membrane | 76710003 | C0229209
DCM | 128295 | Surface between Inner and Outer Segments of the photoreceptors |  |
DCM | 128296 | Surface of the interdigitating zone between retina and RPE |  |
DCM | 128297 | Anterior surface of the RPE |  |
DCM | 128298 | Surface of the center of the RPE |  |
DCM | 128299 | Posterior surface of the RPE |  |
DCM | 128300 | Outer surface of the BM |  |
DCM | 128301 | Surface of the choroid-sclera interface |  |
DCM | 128302 | Outer surface of the CC |  |

## CID 5000 Languages

Context Group ID 5000 comprises the language tag coding scheme of [RFC 5646]. The Coding Scheme Designator (0008,0102) shall be RFC5646.

### Note

1. The [RFC 5646] coding scheme is constructed from a primary subtag component encoded using the shortest language codes of [ISO 639], plus codes for extensions for languages not represented in [ISO 639]. The code optionally includes additional subtag components, for scripts encoded using the four letter codes of [ISO 15924], and for regions encoded using the two letter country codes of [ISO 3166].

2. [RFC 5646] may be obtained at http://www.ietf.org/rfc/rfc5646.txt. [RFC 5646] obsoletes [RFC 4646], [RFC 3066] and [RFC 1766], but is forward compatible with those specifications. In previous editions of the Standard, [RFC 4646] codes were used with a Coding Scheme Designator of IETF4646. [RFC 5646] is a superset of [RFC 4646], which formalizes support for [ISO 639-3].

3. [ISO 639] codes may be obtained at http://www.loc.gov/standards/iso639-2/langhome.html.

4. The two letter country codes of [ISO 3166] may be obtained at https://www.iso.org/obp/ui/#search/code/

5. IANA language tag registrations may be obtained at http://www.iana.org/assignments/language-subtag-registry/language-subtag-registry

6. In previous editions of the Standard, this Context Group formerly included the three letter language codes of [ISO 639-2]/B, using Coding Scheme Designator ISO639_2, or the language codes of [RFC 3066], using Coding Scheme Designator RFC3066, and several IANA-registered language code extensions, using Coding Scheme Designator IANARFC1766.

7. In previous editions of the Standard, this Context Group provided only language identifiers, with national or regional variant identified in a separate attribute or Content Item.

## CID 5001 Countries

Context Group ID 5001 comprises the two letter country code scheme of ISO 3166. The Coding Scheme Designator (0008,0102) shall be ISO3166_1.

### Note

The two letter country codes of ISO 3166 may be obtained at https://www.iso.org/obp/ui/#search/code/

## CID 5002 Organizations

Context Group ID 5002 comprises the object identifier scheme of ISO 8824-1 and ISO 9834-1, when applied to organizational identifiers (see Section 8.2). The Coding Scheme Designator (0008,0102) shall be ISO_OID.

## CID 6000 Overall Breast Composition

### Note

In future extensions, Overall Breast Composition terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.330

Table CID 6000. Overall Breast Composition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6001 “Overall Breast Composition from BI-RADS®”

## CID 6001 Overall Breast Composition from BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E77)

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.331

Table CID 6001. Overall Breast Composition from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01711 | Almost entirely fat | 129716005 | C0231248
SRT | F-01712 | Scattered fibroglandular densities | 129717001 | C0544447
SRT | F-01713 | Heterogeneously dense | 129718006 | C0231249
SRT | F-01714 | Extremely dense | 129719003 | C1268647

## CID 6002 Change Since Last Mammogram or Prior Surgery

### Note

In future extensions, Change Since Last Mammogram or Prior Surgery terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.332

Table CID 6002. Change Since Last Mammogram or Prior Surgery

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6003 “Change Since Last Mammogram or Prior Surgery from BI-RADS®”

## CID 6003 Change Since Last Mammogram or Prior Surgery from BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E79)

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.333

Table CID 6003. Change Since Last Mammogram or Prior Surgery from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01721 | New finding | 129721008 | C1268649
SRT | F-01722 | Finding partially removed | 129722001 | C1268650
SRT | F-01723 | No significant changes in the finding | 129723006 | C1268651
SRT | M-02520 | Increase in size | 15454001 | C0332509
SRT | M-02530 | Decrease in size | 19776001 | C0332511
SRT | F-01726 | Increase in number of calcifications | 129726003 | C1268654
SRT | F-01727 | Decrease in number of calcifications | 129727007 | C1268655
SRT | F-01728 | Less defined | 129728002 | C1268656
SRT | F-01729 | More defined | 129729005 | C1268657
SRT | F-0172A | Removal of implant since previous mammogram | 129730000 | C1268658
SRT | F-0172B | Implant revised since previous mammogram | 129731001 | C1268659

## CID 6004 Mammography Characteristics of Shape

### Note

In future extensions, Mammography Characteristics of Shape terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.334

Table CID 6004. Mammography Characteristics of Shape

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6005 “Characteristics of Shape from BI-RADS®”

## CID 6005 Characteristics of Shape from BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E80)

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.335

Table CID 6005. Characteristics of Shape from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-02100 | Round shape | 42700002 | C0332490
SRT | M-02120 | Ovoid shape (Oval) | 84360004 | C0332492
SRT | G-A640 | Lobular | 40266001 | C0205417
SRT | G-A402 | Irregular | 49608001 | C0205271

## CID 6006 Mammography Characteristics of Margin

### Note

In future extensions, Mammography Characteristics of Margin terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.336

Table CID 6006. Mammography Characteristics of Margin

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6007 “Characteristics of Margin from BI-RADS®”

## CID 6007 Characteristics of Margin from BI-RADS®

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.337

Table CID 6007. Characteristics of Margin from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01741 | Circumscribed lesion | 129738007 | C1268666
SRT | F-01742 | Microlobulated lesion | 129739004 | C1268667
SRT | F-01743 | Obscured lesion | 129740002 | C1268668
SRT | F-01744 | Indistinct lesion | 129741003 | C1268669
SRT | F-01745 | Spiculated lesion | 129742005 | C1268670
DCM | 111343 | Angular margins |  |

## CID 6008 Density Modifier

### Note

In future extensions, Density Modifier terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.338

Table CID 6008. Density Modifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6009 “Density Modifier from BI-RADS®”

## CID 6009 Density Modifier from BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E82)

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.339

Table CID 6009. Density Modifier from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01751 | High density lesion | 129744006 | C1268672
SRT | F-01752 | Equal density (isodense) lesion | 129745007 | C1268673
SRT | F-01753 | Low density (not containing fat) lesion | 129746008 | C1268674
SRT | F-01754 | Fat containing (radiolucent) lesion | 129747004 | C1268675

## CID 6010 Mammography Calcification Types

### Note

In future extensions, Mammography Calcification Types terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.340

Table CID 6010. Mammography Calcification Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6011 “Calcification Types from BI-RADS®”

## CID 6011 Calcification Types from BI-RADS®

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.341

Table CID 6011. Calcification Types from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01761 | Coarse (popcorn-like) calcification | 129749001 | C1268677
SRT | F-01762 | Dystrophic calcification | 129750001 | C0333582
SRT | F-01763 | Eggshell calcification | 129751002 | C1313950
SRT | F-01764 | Large rod-like calcification | 129752009 | C1268678
SRT | F-01765 | Milk of calcium calcification | 129753004 | C1268679
SRT | F-01766 | Lucent-centered calcification | 129754005 | C1268680
SRT | F-01767 | Punctate calcification | 129755006 | C1265883
SRT | F-01768 | Round shaped calcification | 129756007 | C1268681
SRT | F-01769 | Calcified skin of breast | 129757003 | C1268682
SRT | F-0176A | Calcified suture material | 129758008 | C1268683
SRT | F-0176B | Vascular calcification | 129759000 | C1268684
SRT | F-0176C | Amorphous calcification | 129760005 | C1268685
SRT | F-0176D | Fine, linear (casting) calcification | 129761009 | C1268686
SRT | F-0176E | Fine linear, branching (casting) calcification | 129762002 | C1268687
SRT | F-0176F | Heterogeneous calcification | 129763007 | C1268688
DCM | 111344 | Fine pleomorphic calcification |  |
SRT | D7-90435 | Microcalcifications of the breast | 44771000 | C0520594
DCM | 111345 | Macrocalcifications |  |

## CID 6012 Calcification Distribution Modifier

### Note

In future extensions, Calcification Distribution Modifier terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.342

Table CID 6012. Calcification Distribution Modifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6013 “Calcification Distribution Modifier from BI-RADS®”

## CID 6013 Calcification Distribution Modifier from BI-RADS®

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.343

Table CID 6013. Calcification Distribution Modifier from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01770 | Diffuse calcification distribution | 129764001 | C1268689
SRT | F-01771 | Linear calcification distribution | 129765000 | C1268690
SRT | F-01772 | Grouped calcification distribution | 129766004 | C1268691
SRT | F-01773 | Regional calcification distribution | 129767008 | C1268692
SRT | F-01774 | Segmental calcification distribution | 129768003 | C1268693
DCM | 111346 | Calcifications within a mass |  |
DCM | 111347 | Calcifications outside of a mass |  |

## CID 6014 Mammography Single Image Finding

### Note

In future extensions, Mammography Single Image Finding terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.344

Table CID 6014. Mammography Single Image Finding

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 6015 “Single Image Finding from BI-RADS®” |  |
DCM | 111099 | Selected region |  |
DCM | 111100 | Breast geometry |  |
DCM | 111101 | Image Quality |  |
DCM | 111102 | Non-lesion |  |
SRT | T-04100 | Nipple | 24142002 | C0028109

## CID 6015 Single Image Finding from BI-RADS®

### Note

Collected from BI-RADS®

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.345

Table CID 6015. Single Image Finding from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01796 | Mammography breast density | 129793001 | C1268717
SRT | F-01776 | Individual Calcification | 129770007 | C1268695
SRT | F-01775 | Calcification Cluster | 129769006 | C1268694
SRT | F-01795 | Architectural distortion of breast | 129792006 | C1268716
SRT | F-01797 | Tubular density | 129794007 | C1268718
SRT | T-C430B | Intramammary lymph node | 443808008 | C2733350
SRT | F-01798 | Trabecular thickening of breast | 129795008 | C1268719
SRT | F-01710 | Breast composition | 129715009 | C0005890
SRT | F-01799 | Skin retraction of breast | 129796009 | C0238832
SRT | F-0179A | Skin thickening of breast | 129797000 | C1268720
SRT | DC-721C4 | Axillary adenopathy | 127189005 | C0578735
SRT | D0-00050 | Skin lesion | 95324001 | C0037284
DCM | 111111 | Cooper's ligament changes |  |
SRT | M-36300 | Edema | 79654002 | C0013604
DCM | 111112 | Mass in the skin |  |
DCM | 111113 | Mass on the skin |  |
SRT | T-C4710 | Axillary lymph node | 68171009 | C0729594

## CID 6016 Mammography Composite Feature

### Note

In future extensions, Mammography Composite Feature terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20050110

UID:

  1.2.840.10008.6.1.346

Table CID 6016. Mammography Composite Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111459 | Mass with calcifications
Include CID 6014 “Mammography Single Image Finding”
Include CID 6017 “Composite Feature from BI-RADS®”

## CID 6017 Composite Feature from BI-RADS®

### Note

Collected from BI-RADS® Third Edition

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.347

Table CID 6017. Composite Feature from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01791 | Mammographic breast mass | 129788004 | C1268712
SRT | F-01792 | Focal asymmetric breast tissue | 129789007 | C1268713
SRT | F-01793 | Asymmetric breast tissue | 129790003 | C1268714

## CID 6018 Clockface Location or Region

### Note

In future extensions, Clockface Location or Region terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.348

Table CID 6018. Clockface Location or Region

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 6019 “Clockface Location or Region from BI-RADS®” |  |
SRT | T-D3050 | Chest wall | 78904004 | C0205076

## CID 6019 Clockface Location or Region from BI-RADS®

### Note

From BI-RADS® 3.1, with Addendum 3.1 (National Mammography Database, E96)

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.349

Table CID 6019. Clockface Location or Region from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01781 | 1 o'clock position | 129772004 | C1268696
SRT | F-01782 | 2 o'clock position | 129773009 | C1268697
SRT | F-01783 | 3 o'clock position | 129774003 | C1268698
SRT | F-01784 | 4 o'clock position | 129775002 | C1268699
SRT | F-01785 | 5 o'clock position | 129776001 | C1268700
SRT | F-01786 | 6 o'clock position | 129777005 | C1268701
SRT | F-01787 | 7 o'clock position | 129778000 | C1268702
SRT | F-01788 | 8 o'clock position | 129779008 | C1268703
SRT | F-01789 | 9 o'clock position | 129780006 | C1268704
SRT | F-0178A | 10 o'clock position | 129781005 | C1268705
SRT | F-0178B | 11 o'clock position | 129782003 | C1268706
SRT | F-0178C | 12 o'clock position | 129783008 | C1268707
SRT | F-0178D | Subareolar region | 129784002 | C1268708
SRT | F-0178E | Axillary tail region | 129785001 | C1268709
SRT | F-0178F | Central region of breast | 129786000 | C1268710
SRT | F-01794 | Axilla region | 129791004 | C1268715

## CID 6020 Quadrant Location

### Note

In future extensions, Quadrant Location terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.350

Table CID 6020. Quadrant Location

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6021 “Quadrant Location from BI-RADS®”

## CID 6021 Quadrant Location from BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E97)

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.351

Table CID 6021. Quadrant Location from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-04004 | Upper outer quadrant of breast | 76365002 | C0222598
SRT | T-04002 | Upper inner quadrant of breast | 77831004 | C0222596
SRT | T-04005 | Lower outer quadrant of breast | 33564002 | C0222599
SRT | T-04003 | Lower inner quadrant of breast | 19100000 | C0222597

## CID 6022 Side

### Note

In future extensions, Side terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Non-Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.352

Table CID 6022. Side

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6023 “Side from BI-RADS®”

## CID 6023 Side from BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E98)

Type:

  Non-Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.353

Table CID 6023. Side from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-04030 | Left breast | 80248007 | C0222601
SRT | T-04020 | Right breast | 73056007 | C0222600
SRT | T-04080 | Both breasts | 63762007 | C0222605

## CID 6024 Depth

### Note

In future extensions, Depth terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.354

Table CID 6024. Depth

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6025 “Depth from BI-RADS®”

## CID 6025 Depth from BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E99)

Type:

  Extensible

Version:

  20160405

UID:

  1.2.840.10008.6.1.355

Table CID 6025. Depth from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-404CC | Anterior | 255549009 | C1704448
SRT | R-4081A | Middle | 260528009 | C2939193
SRT | R-404CE | Posterior | 255551008 | C0205095

## CID 6026 Mammography Assessment

### Note

In future extensions, Mammography Assessment terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.356

Table CID 6026. Mammography Assessment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6027 “Assessment from BI-RADS®”
DCM | 111120 | Post Procedure Mammograms for Marker Placement

## CID 6027 Assessment from BI-RADS®

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.357

Table CID 6027. Assessment from BI-RADS®

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | --- | ---
SRT |  | F-037BB | 0 - Incomplete - Need additional imaging evaluation +/- priors | 397138000 | C1301244
SRT |  | F-037BC | 1 - Negative | 397140005 | C1301245
SRT |  | F-037BD | 2 - Benign | 397141009 | C1301246
SRT |  | F-037BF | 3 - Probably Benign | 397143007 | C1301247
SRT |  | F-037C0 | 4 - Suspicious | 397144001 | C1301248
BI | 4.0 | MA.II.A.5.4A | 4A - Low suspicion |  |
BI | 4.0 | MA.II.A.5.4B | 4B - Intermediate suspicion |  |
BI | 4.0 | MA.II.A.5.4C | 4C - Moderate suspicion |  |
SRT |  | F-037C1 | 5 - Highly suggestive of malignancy | 397145000 | C1301249
BI | 4.0 | MA.II.A.5.6 | 6 - Known biopsy proven malignancy |  |

### Note

1. The code meanings are those from BI-RADS Atlas 5th edition, which removed the management recommendation from the assessment category.

2. The code meaning for category 0 is shortened to fit the 64 character limitation of the Value Representation. In BI-RADS 5th edition, the full meaning is "Incomplete - Need additional imaging evaluation and/or prior mammograms for comparison".

## CID 6028 Mammography Recommended Follow-up

### Note

In future extensions, Mammography Recommended Follow-up terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.358

Table CID 6028. Mammography Recommended Follow-up

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 6029 “Recommended Follow-up from BI-RADS®” |  |
DCM | 111121 | Follow-up post biopsy as directed by clinician |  |
SRT | P0-006F1 | Nuclear medicine procedure | 371572003 | C0203634
SRT | P0-009B4 | Evaluation procedure | 386053000 | C1261322
DCM | 111410 | Surgical consult |  |

## CID 6029 Recommended Follow-up from BI-RADS®

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.359

Table CID 6029. Recommended Follow-up from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111135 | Additional projections |  |
SRT | R-102D6 | Magnification views | 399163009 | C1302233
SRT | R-102D7 | Spot compression | 399055006 | C1302185
DCM | 111136 | Spot magnification view(s) |  |
SRT | P5-B0000 | Diagnostic ultrasonography | 16310003 | C0041618
DCM | 111138 | Old films for comparison |  |
SRT | P5-40060 | Mammary ductogram | 18102001 | C0203033
DCM | 111140 | Normal interval follow-up |  |
DCM | 111141 | Any decision to biopsy should be based on clinical assessment |  |
DCM | 111142 | Follow-up at short interval (1-11 months) |  |
DCM | 111143 | Biopsy should be considered |  |
DCM | 111144 | Needle localization and biopsy |  |
DCM | 111145 | Histology using core biopsy |  |
DCM | 111146 | Suggestive of malignancy - take appropriate action |  |
DCM | 111147 | Cytologic analysis |  |
DCM | 111148 | Biopsy should be strongly considered |  |
DCM | 111149 | Highly suggestive of malignancy - take appropriate action |  |
DCM | 111122 | Known biopsy proven malignancy - take appropriate action |  |
SRT | P5-0900D | MRI of breast | 241615005 | C0344104

## CID 6030 Mammography Pathology Codes

### Note

In future extensions, Mammography Pathology Codes terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.360

Table CID 6030. Mammography Pathology Codes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6031 “Benign Pathology Codes from BI-RADS®”
Include CID 6032 “High Risk Lesions Pathology Codes from BI-RADS®”
Include CID 6033 “Malignant Pathology Codes from BI-RADS®”

## CID 6031 Benign Pathology Codes from BI-RADS®

### Note

From BI-RADS® Third Edition, with Addendum 3.1 (National Mammography Database,F110)

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.361

Table CID 6031. Benign Pathology Codes from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-41610 | Abscess | 44132006 | C0000833
SRT | M-74200 | Adenosis | 57597008 | C0334050
SRT | M-81400 | Adenoma | 32048006 | C0001430
SRT | M-83240 | Adenolipoma | 22024005 | C0334325
SRT | M-73310 | Apocrine Metaplasia | 81274009 | C0269252
SRT | M-89830 | Adenomyoepithelioma | 128765009 | C1266146
SRT | M-55160 | Amyloid (tumor) | 37279009 | C0333572
DCM | 111251 | Normal axillary node |  |
SRT | M-88610 | Angiolipoma | 73219006 | C0206632
DCM | 111252 | Axillary node with calcifications |  |
SRT | M-76100 | Angiomatosis | 14350002 | C0002992
DCM | 111253 | Axillary node hyperplasia |  |
SRT | F-8A063 | Asynchronous involution of breast | 130963002 | C1295577
SRT | D7-90035 | Cyst of breast | 399294002 | C0006144
DCM | 111255 | Benign cyst with blood |  |
DCM | 111256 | Benign Calcifications |  |
SRT | M-92200 | Chondroma | 31186001 | C0936248
SRT | M-85040 | Intracystic papilloma | 47488001 | C0334374
DCM | 111258 | Ductal adenoma |  |
SRT | D7-90370 | Mammary duct ectasia | 22049009 | C0152442
DCM | 111259 | Diabetic fibrous mastopathy |  |
SRT | M-72170 | Ductal hyperplasia, Usual | 67617000 | C0333994
SRT | M-88211 | Extra abdominal desmoid | 47284001 | C0079218
SRT | D4-48014 | Ectopic (accessory) breast tissue | 1896004 | C0266012
SRT | M-33415 | Epidermal inclusion cyst | 419670003 | C0014511
SRT | M-36300 | Edema | 79654002 | C0013604
SRT | M-90100 | Fibroadenoma | 65877006 | C0206650
DCM | 111263 | Fibroadenomatoid hyperplasia |  |
DCM | 111264 | Fibroadenolipoma |  |
SRT | M-44140 | Foreign body (reaction) | 37058002 | C0016549
SRT | D7-90310 | Fibrocystic disease of breast | 27431007 | C0016034
SRT | M-78266 | Focal fibrosis | 45559001 | C0521195
SRT | M-78800 | Fibromatosis | 19928005 | C0016048
SRT | D7-90434 | Fat necrosis of breast | 21381006 | C0156321
SRT | D7-90364 | Galactocele | 42385006 | C0152243
SRT | M-95800 | Granular cell tumor | 12169001 | C0085167
SRT | M-90160 | Giant fibroadenoma | 34882000 | C0334500
SRT | D7-90420 | Gynecomastia | 4754008 | C0018418
SRT | M-75500 | Hamartoma | 51398009 | C0018552
SRT | M-91200 | Hemangioma | 2099007 | C0018916
SRT | D3-F0620 | Hemangioma of subcutaneous tissue | 93473009 | C0685200
SRT | M-91220 | Hemangioma - venous | 56468002 | C0334532
SRT | M-35060 | Hematoma | 35566002 | C0018944
SRT | M-72000 | Hyperplasia, usual | 76197007 | C0020507
SRT | D7-90452 | Infarction of breast | 77296004 | C0269266
SRT | M-40000 | Inflammation | 23583003 | C0021368
SRT | T-C430B | Intramammary lymph node | 443808008 | C2733350
SRT | M-85030 | Intraductal papilloma | 5244003 | C0206713
SRT | M-90300 | Juvenile fibroadenoma | 46212000 | C0346158
DCM | 111277 | Juvenile papillomatosis |  |
SRT | M-82040 | Lactating adenoma | 128651002 | C1266023
SRT | M-88500 | Lipoma | 46720004 | C0023798
DCM | 111279 | Lactational change |  |
SRT | D7-90428 | Breast lobular hyperplasia | 6703006 | C0269263
SRT | M-88900 | Leiomyoma | 44598004 | C0042133
SRT | T-C4000 | Lymph node | 59441001 | C0024204
DCM | 111281 | Large duct papilloma |  |
SRT | D3-87780 | Thrombophlebitis of breast (Mondor's disease) | 69954004 | C0265070
SRT | M-88250 | Myofibroblastoma | 128738002 | C0242404
DCM | 111284 | Microglandular adenosis |  |
DCM | 111285 | Multiple Intraductal Papillomas |  |
DCM | 111286 | No abnormality |  |
DCM | 111287 | Normal breast tissue |  |
SRT | M-95400 | Neurofibroma | 89084002 | C0027830
SRT | M-95401 | Neurofibromatosis | 81669005 | C0162678
SRT | D7-F0810 | Benign neoplasm of nipple of female breast (Nipple adenoma) | 92248004 | C0686290
DCM | 111290 | Oil cyst (fat necrosis cyst) |  |
SRT | M-80500 | Papilloma | 23730008 | C0030354
SRT | M-89400 | Pleomorphic adenoma | 8360001 | C0026277
DCM | 111291 | Post reduction mammoplasty |  |
DCM | 111292 | Pseudoangiomatous stromal hyperplasia |  |
SRT | M-78731 | Radial scar | 133855003 | C1297883
SRT | M-74220 | Sclerosing adenosis | 50916005 | C0235590
SRT | M-36050 | Seroma | 56021002 | C0262627
DCM | 111296 | Silicone granuloma |  |
SRT | M-78060 | Scar tissue | 12402003 | C2004491
SRT | M-82110 | Tubular adenoma | 19665009 | C0334292
DCM | 111298 | Virginal hyperplasia |  |

## CID 6032 High Risk Lesions Pathology Codes from BI-RADS®

### Note

From BI-RADS® Third Edition, with Addendum 3.1 (National Mammography Database,F110)

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.362

Table CID 6032. High Risk Lesions Pathology Codes from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-72175 | Atypical intraductal hyperplasia | 6660000 | C0521187
SRT | M-72105 | Atypical lobular hyperplasia | 33889003 | C0442835
SRT | D7-F0A02 | Lobular carcinoma in situ of breast | 109888004 | C0279563
DCM | 111299 | Peripheral duct papillomas |  |
SRT | M-90201 | Phyllodes tumor | 71232009 | C0010701

## CID 6033 Malignant Pathology Codes from BI-RADS®

### Note

From BI-RADS® Third Edition, with Addendum 3.1 (National Mammography Database,F110)

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.363

Table CID 6033. Malignant Pathology Codes from BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-82003 | Adenoid cystic carcinoma | 11671000 | C0010606
DCM | 111300 | Axillary node with lymphoma |  |
DCM | 111301 | Axillary nodal metastases |  |
SRT | M-84013 | Apocrine adenocarcinoma | 57141000 | C0334346
SRT | M-91203 | Angiosarcoma | 39000009 | C0018923
DCM | 111307 | Basal cell carcinoma of nipple |  |
DCM | 111303 | Blood vessel (vascular) invasion |  |
SRT | M-84803 | Mucinous adenocarcinoma (Colloid carcinoma) | 72495009 | C0007130
DCM | 111304 | Carcinoma in children |  |
SRT | M-92203 | Chondrosarcoma | 14990007 | C0008479
DCM | 111305 | Carcinoma in ectopic breast |  |
DCM | 111306 | Carcinoma with endocrine differentiation |  |
SRT | M-85012 | Comedocarcinoma (intraductal) | 78197004 | C0334369
SRT | D7-F0902 | Carcinoma in situ of male breast | 92652009 | C0686328
SRT | M-85733 | Carcinoma with metaplasia | 22694002 | C0334396
DCM | 111309 | Cartilaginous and osseous change |  |
DCM | 111310 | Carcinoma in pregnancy and lactation |  |
SRT | M-89803 | Carcinosarcoma | 63264007 | C0007140
DCM | 111312 | Intraductal comedocarcinoma with necrosis |  |
DCM | 111341 | Intraductal carcinoma, high grade |  |
DCM | 111313 | Intraductal carcinoma, low grade |  |
SRT | M-85072 | Intraductal carcinoma micro-papillary | 128696009 | C1266080
SRT | M-88103 | Fibrosarcoma | 53654007 | C0016057
SRT | M-83153 | Glycogen-rich carcinoma | 74280008 | C0334319
SRT | M-91501 | Hemangiopericytoma | 36060005 | C0018922
SRT | M-96503 | Hodgkin's disease (lymphoma) | 14537002 | C0019829
SRT | M-82013 | Invasive cribriform carcinoma | 30156004 | C0205643
DCM | 111315 | Intracystic papillary carcinoma |  |
SRT | M-85003 | Infiltrating duct carcinoma | 82711006 | C1412014
DCM | 111316 | Invasive and in-situ carcinoma |  |
SRT | M-85203 | Invasive lobular carcinoma | 89740008 | C0206692
SRT | M-85303 | Inflammatory carcinoma | 32968003 | C0334385
SRT | M-80503 | Papillary carcinoma (invasive) | 25910003 | C0007133
DCM | 111318 | Leukemic infiltration |  |
SRT | M-88903 | Leiomyosarcoma | 51549004 | C0023269
SRT | M-88503 | Liposarcoma | 49430005 | C0023827
SRT | M-83143 | Lipid-rich (lipid-secreting) carcinoma | 3839000 | C0334318
DCM | 111320 | Lympathic vessel invasion |  |
SRT | M-95903 | Lymphoma | 21964009 | C0024299
DCM | 111322 | Occult carcinoma presenting with axillary lymph node metastases |  |
DCM | 111323 | Metastatic cancer to the breast |  |
DCM | 111324 | Metastatic cancer to the breast from the colon |  |
DCM | 111325 | Metastatic cancer to the breast from the lung |  |
DCM | 111326 | Metastatic melanoma to the breast |  |
DCM | 111327 | Metastatic cancer to the breast from the ovary |  |
DCM | 111328 | Metastatic sarcoma to the breast |  |
SRT | M-85103 | Medullary carcinoma | 32913002 | C0206693
DCM | 111329 | Multifocal intraductal carcinoma |  |
DCM | 111330 | Metastatic disease to axillary node |  |
SRT | M-88303 | Malignant fibrous histiocytoma | 34360000 | C0334463
DCM | 111332 | Multifocal invasive ductal carcinoma |  |
DCM | 111333 | Metastasis to an intramammary lymph node |  |
DCM | 111334 | Malignant melanoma of nipple |  |
SRT | M-95913 | Non-Hodgkin's lymphoma | 1929004 | C0024305
SRT | D0-F035F | Neoplasm of the mammary skin | 126510002 | C1290094
SRT | M-91803 | Osteogenic sarcoma | 21708004 | C0029463
SRT | M-80502 | Papillary carcinoma in-situ | 10376009 | C0334242
SRT | M-85403 | Paget's disease, mammary (of the nipple) | 2985005 | C0030185
SRT | M-97313 | Plasmacytoma | 10639003 | C0032131
SRT | M-90203 | Phyllodes tumor, malignant | 87913009 | C0600066
DCM | 111338 | Recurrent malignancy |  |
SRT | M-84903 | Signet ring cell carcinoma | 87737001 | C0206696
DCM | 111340 | Squamous cell carcinoma of the nipple |  |
SRT | M-78190 | Spindle cell nodule (tumor) | 110451006 | C0333821
SRT | M-85023 | Secretory (juvenile) carcinoma of the breast | 41919003 | C0334371
SRT | M-80703 | Squamous cell carcinoma | 28899001 | C0007137
SRT | M-82113 | Tubular adenocarcinoma | 4631006 | C0205645

## CID 6034 Intended Use of CAD Output

Type:

  Non-Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.364

Table CID 6034. Intended Use of CAD Output

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111150 | Presentation Required: Rendering device is expected to present
DCM | 111151 | Presentation Optional: Rendering device may present
DCM | 111152 | Not for Presentation: Rendering device expected not to present

## CID 6035 Composite Feature Relations

Type:

  Non-Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.365

Table CID 6035. Composite Feature Relations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111153 | Target content items are related temporally
DCM | 111154 | Target content items are related spatially
DCM | 111155 | Target content items are related contra-laterally

## CID 6036 Scope of Feature

Type:

  Non-Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.366

Table CID 6036. Scope of Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111156 | Feature detected on the only image
DCM | 111157 | Feature detected on only one of the images
DCM | 111158 | Feature detected on multiple images
DCM | 111159 | Feature detected on images from multiple modalities

## CID 6037 Mammography Quantitative Temporal Difference Type

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.367

Table CID 6037. Mammography Quantitative Temporal Difference Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-017B1 | Difference in size | 129806009 | C1268722
SRT | F-017B2 | Difference in opacity | 129807000 | C1268723
SRT | F-017B3 | Difference in location | 129808005 | C1268724
SRT | F-017B4 | Difference in spatial proximity | 129809002 | C1268725
SRT | F-017B5 | Difference in number of calcifications | 129810007 | C1268726

## CID 6038 Mammography Qualitative Temporal Difference Type

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.368

Table CID 6038. Mammography Qualitative Temporal Difference Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-017B6 | Difference in shape | 129811006 | C1268727
SRT | F-017B7 | Difference in margin | 129812004 | C1268728
SRT | F-017B8 | Difference in symmetry | 129813009 | C1268729

## CID 6039 Nipple Characteristic

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.369

Table CID 6039. Nipple Characteristic

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-02000 | Normal shape | 31842008 | C0332480
SRT | D7-90554 | Nipple retraction | 31845005 | C0221370

## CID 6040 Non-lesion Object Type

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.370

Table CID 6040. Non-lesion Object Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 6401 “Non-lesion Object Type - Physical Objects”
Include CID 6402 “Non-lesion Object Type - Substances”
Include CID 6403 “Non-lesion Object Type - Tissues”
Unspecified |  |  |  |

### Note

The use of (111176, DCM, "Unspecified") is explicitly permitted in this context group to allow for the communication of measurements of an object of unknown type using TID 4012 “Mammography CAD Non-lesion”.

## CID 6041 Mammography Image Quality Finding

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.371

Table CID 6041. Mammography Image Quality Finding

Coding Scheme Designator | Code Value | Code Meaning | Source
--- | --- | --- | ---
DCM | 111177 | View and Laterality Marker is missing | MQSA
DCM | 111178 | View and Laterality Marker does not have both view and laterality | MQCM 1999
DCM | 111179 | View and Laterality Marker does not have approved codes | MQCM 1999
DCM | 111180 | View and Laterality Marker is not near the axilla | MQCM 1999
DCM | 111181 | View and Laterality Marker overlaps breast tissue | MQCM 1999
DCM | 111182 | View and Laterality Marker is partially obscured | MQCM 1999
DCM | 111183 | View and Laterality Marker is incorrect |
DCM | 111184 | View and Laterality Marker is off image |
DCM | 111185 | Flash is not near edge of film | MQCM 1999
DCM | 111186 | Flash is illegible, does not fit, or is lopsided | MQSA
DCM | 111187 | Flash doesn't include patient name and additional patient id | MQCM 1999
DCM | 111188 | Flash doesn't include date of examination | MQCM 1999
DCM | 111189 | Flash doesn't include facility name and location | MQSA
DCM | 111190 | Flash doesn't include technologist identification | MQCM 1999
DCM | 111191 | Flash doesn't include cassette/screen/detector identification | MQCM 1999
DCM | 111192 | Flash doesn't include mammography unit identification | MQCM 1999
DCM | 111193 | Date sticker is missing | MQCM 1999
DCM | 111194 | Technical factors missing | MQCM 1999
DCM | 111195 | Collimation too close to breast | MQCM 1999
DCM | 111196 | Inadequate compression | MQCM 1999
DCM | 111197 | MLO Insufficient pectoral muscle | MQCM 1999
DCM | 111198 | MLO No fat is visualized posterior to fibroglandular tissues | MQCM 1999
DCM | 111199 | MLO Poor separation of deep and superficial breast tissues | MQCM 1999
DCM | 111200 | MLO Evidence of motion blur | MQCM 1999
DCM | 111201 | MLO Inframammary fold is not open | MQCM 1999
DCM | 111202 | CC Not all medial tissue visualized | MQCM 1999
DCM | 111203 | CC Nipple not centered on image | MQCM 1999
DCM | 111204 | CC Posterior nipple line does not measure within 1 cm of MLO | MQCM 1999
DCM | 111205 | Nipple not in profile |
DCM | 111206 | Insufficient implant displacement incorrect | MQCM 1999
DCM | 111208 | Grid artifact(s) |
DCM | 111209 | Positioning |
DCM | 111210 | Motion blur |
DCM | 111211 | Under exposed |
DCM | 111212 | Over exposed |
DCM | 111213 | No image |
DCM | 111214 | Detector artifact(s) |
DCM | 111215 | Artifact(s) other than grid or detector artifact |
DCM | 111216 | Mechanical failure |
DCM | 111217 | Electrical failure |
DCM | 111218 | Software failure |
DCM | 111219 | Inappropriate image processing |
DCM | 111220 | Other failure |
DCM | 111221 | Unknown failure |

## CID 6042 Status of Results

Type:

  Non-Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.372

Table CID 6042. Status of Results

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111222 | Succeeded
DCM | 111223 | Partially Succeeded
DCM | 111224 | Failed
DCM | 111225 | Not Attempted

## CID 6043 Types of Mammography CAD Analysis

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.373

Table CID 6043. Types of Mammography CAD Analysis

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Notes
--- | --- | --- | --- | --- | ---
SRT | P5-B3402 | Spatial collocation analysis | 133884007 | C1297892 | See Note 1
SRT | P5-B3404 | Spatial proximity analysis | 133885008 | C1297893 | See Note 2
SRT | P5-B3406 | Temporal correlation | 133886009 | C1297894 |
SRT | P5-B3408 | Image quality analysis | 133887000 | C1297895 |
SRT | P5-B3410 | Focal asymmetric density analysis | 133888005 | C1297896 |
SRT | P5-B3412 | Asymmetric breast tissue analysis | 133889002 | C1297897 |
SRT | P5-B3414 | Breast composition analysis | 133890006 | C1297898 |
DCM | 111233 | Individual Impression / Recommendation Analysis |  |  |
DCM | 111234 | Overall Impression / Recommendation Analysis |  |  |

### Note

1. Spatial Collocation Analysis is used to identify features that are the same or located in the same place.

2. Spatial Proximity Analysis is used to identify features that are related spatially, such as nipple retraction associated with a spiculated mass.

## CID 6044 Types of Image Quality Assessment

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.374

Table CID 6044. Types of Image Quality Assessment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111235 | Unusable - Quality renders image unusable
DCM | 111236 | Usable - Does not meet the quality control standard
DCM | 111237 | Usable - Meets the quality control standard

## CID 6045 Mammography Types of Quality Control Standard

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.375

Table CID 6045. Mammography Types of Quality Control Standard

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111238 | Mammography Quality Control Manual 1999, ACR
DCM | 111239 | Title 21 CFR Section 900, Subpart B
DCM | 111240 | Institutionally defined quality control standard

## CID 6046 Units of Follow-up Interval

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.376

Table CID 6046. Units of Follow-up Interval

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | d | day
UCUM | wk | week
UCUM | mo | month
UCUM | a | year

## CID 6047 CAD Processing and Findings Summary

Type:

  Non-Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.377

Table CID 6047. CAD Processing and Findings Summary

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111241 | All algorithms succeeded; without findings
DCM | 111242 | All algorithms succeeded; with findings
DCM | 111243 | Not all algorithms succeeded; without findings
DCM | 111244 | Not all algorithms succeeded; with findings
DCM | 111245 | No algorithms succeeded; without findings

## CID 6048 CAD Operating Point Axis Label

Type:

  Extensible

Version:

  20060612

UID:

  1.2.840.10008.6.1.378

Table CID 6048. CAD Operating Point Axis Label

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111012 | Certainty of Finding
DCM | 111047 | Probability of cancer
DCM | 111086 | False Markers per Image
DCM | 111087 | False Markers per Case
DCM | 111088 | Case Sensitivity
DCM | 111089 | Lesion Sensitivity
DCM | 111090 | Case Specificity
DCM | 111091 | Image Specificity

## CID 6050 Breast Procedure Reported

Type:

  Extensible

Version:

  20090819

UID:

  1.2.840.10008.6.1.379

Table CID 6050. Breast Procedure Reported

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111408 | Film Screen Mammography |  |
DCM | 111409 | Digital Mammography |  |
SRT | P5-B8500 | Ultrasonography of breast | 47079000 | C0080264
SRT | P5-0900D | MRI of breast | 241615005 | C0344104
SRT | P1-48011 | Pre-biopsy localization of breast lesion | 237380007 | C0473515
SRT | P1-48145 | Fine needle aspiration of breast | 387736007 | C0542415
SRT | P1-48142 | Diagnostic aspiration of breast cyst | 287572003 | C0565162
SRT | P1-48304 | Core needle biopsy of breast | 44578009 | C0191853
SRT | P1-4830F | Breast - surgical biopsy | 274331003 | C0585992
SRT | P5-40060 | Mammary ductogram | 18102001 | C0203033
SRT | P5-0801C | CT of breast | 241539009 | C0412609
SRT | P5-D0042 | Radionuclide localization of tumor, limited area | 66377006 | C0203652
SRT | P5-40030 | Specimen radiography of breast | 80865008 | C0203031
SRT | P2-4A000 | Examination of breast | 46662001 | C0199850
DCM | 111410 | Surgical consult |  |
DCM | 111411 | Mammography CAD |  |
SRT | P1-65359 | Sentinel lymph node biopsy | 396487001 | C0796693
SRT | P5-D0061 | Radioisotope scan of lymphatic system | 169167001 | C0412375
DCM | 111123 | Marker placement |  |
SRT | P1-05535 | Insertion of catheter | 45211000 | C0007430

## CID 6051 Breast Procedure Reason

### Note

Some of these terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.380

Table CID 6051. Breast Procedure Reason

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 6061 “Breast Imaging Procedure Modifiers” |  |
DCM | 111415 | Additional evaluation requested from prior study |  |
DCM | 111416 | Follow-up at short interval from prior study |  |
DCM | 111417 | History of breast augmentation, asymptomatic |  |
DCM | 111418 | Review of an outside study |  |
DCM | 111402 | Clinical finding |  |
SRT | P1-48830 | Reduction mammoplasty | 59214008 | C0191922
SRT | P5-C0000 | Radiation therapy | 53438000 | C1522449
SRT | P1-48840 | Augmentation mammoplasty | 22890008 | C0191925
DCM | 111419 | Additional evaluation requested from abnormal screening exam |  |
SRT | P5-C018A | Brachytherapy | 384692006 | C0006098
DCM | 111420 | History of benign breast biopsy |  |
DCM | 111421 | Personal history of breast cancer with breast conservation therapy |  |
DCM | 111124 | Personal history of breast cancer with mastectomy |  |
DCM | 111125 | Known biopsy proven malignancy |  |
SRT | G-03D3 | Personal history of breast cancer | 415076002 | C1387407
DCM | 111590 | Recall for technical reasons |  |
DCM | 111591 | Recall for imaging findings |  |
DCM | 111592 | Recall for patient symptoms/ clinical findings |  |

## CID 6052 Breast Imaging Report Section Title

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.381

Table CID 6052. Breast Imaging Report Section Title

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111423 | Physical Examination Results |  |
DCM | 111424 | Comparison to previous exams |  |
DCM | 121070 | Findings |  |
LN | 19005-8 | Impressions |  | C0801998
DCM | 121074 | Recommendations |  |
DCM | 121076 | Conclusions |  |
DCM | 121078 | Addendum |  |
SRT | F-01710 | Breast composition | 129715009 | C0005890
DCM | 111413 | Overall Assessment |  |
DCM | 121058 | Procedure reported |  |
DCM | 111401 | Reason for procedure |  |

## CID 6053 Breast Imaging Report Elements

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.382

Table CID 6053. Breast Imaging Report Elements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111423 | Physical Examination Results |  |
DCM | 111424 | Comparison to previous exams |  |
DCM | 121071 | Finding |  |
DCM | 121073 | Impression |  |
DCM | 121075 | Recommendation |  |
DCM | 121077 | Conclusion |  |
SRT | F-01710 | Breast composition | 129715009 | C0005890
DCM | 111413 | Overall Assessment |  |
DCM | 121058 | Procedure reported |  |
DCM | 111401 | Reason for procedure |  |

## CID 6054 Breast Imaging Findings

Type:

  Extensible

Version:

  20050110

UID:

  1.2.840.10008.6.1.383

Table CID 6054. Breast Imaging Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-8A084 | Breast normal | 290084006 | C0567498
SRT | F-8A057 | Calcification of breast | 309587003 | C0587094
SRT | A-04010 | Implant | 40388003 | C0021102
Include CID 6016 “Mammography Composite Feature” |  |
Include CID 6057 “Ductography Findings for Breast” |  |
Include CID 6064 “Ultrasound Findings for Breast” |  |

## CID 6055 Breast Clinical Finding or Indicated Problem

### Note

Some of these terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.384

Table CID 6055. Breast Clinical Finding or Indicated Problem

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-207D7 | O/E - Breast lump palpated | 268951004 | C0437107
SRT | D7-90565 | Bloody nipple discharge | 290113009 | C0541951
DCM | 111478 | Non-bloody discharge (from nipple) |  |
DCM | 111479 | Difficult physical/clinical examination |  |
SRT | D7-90010 | Disorder of breast implant | 271989003 | C0405486
SRT | F-0179A | Skin thickening of breast | 129797000 | C1268720
SRT | F-01799 | Skin retraction of breast | 129796009 | C0238832
SRT | D7-90560 | Peau d'orange surface of breast | 87386002 | C0425791
SRT | F-8A09C | Nipple problem | 290119008 | C0567530
SRT | R-20099 | O/E - axillary lymphadenopathy | 164150006 | C0437624
SRT | F-8A030 | Breast pain | 53430007 | C0024902
DCM | 111480 | Cancer elsewhere |  |
SRT | D7-90530 | Breast lump | 89164003 | C0024103
SRT | F-8A074 | Discoloration of skin of breast | 290069002 | C0567486
SRT | F-01760 | Radiographic calcification finding | 129748009 | C0015663
DCM | 111126 | Image detected mass |  |
SRT | F-03753 | Nipple discharge symptom | 162164007 | C0149741
SRT | F-4410C | Erythema | 247441003 | C0041834
SRT | R-202A9 | O/E - lymphadenopathy | 274303007 | C0558515
SRT | DF-00577 | Disseminated malignancy of unknown primary | 285645000 | C0563521

## CID 6056 Associated Findings for Breast

### Note

These terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.385

Table CID 6056. Associated Findings for Breast

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | D7-9002A | Breast hematoma | 302924003 | C0342095
SRT | M-78280 | Surgical scar | 63130001 | C0334150
SRT | D7-90554 | Nipple retraction | 31845005 | C0221370
Include CID 6015 “Single Image Finding from BI-RADS®” |  |

## CID 6057 Ductography Findings for Breast

### Note

These terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.386

Table CID 6057. Ductography Findings for Breast

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111287 | Normal breast tissue |  |
DCM | 111425 | Intraluminal filling defect |  |
SRT | D7-90370 | Mammary duct ectasia | 22049009 | C0152442
DCM | 111426 | Multiple filling defect |  |
DCM | 111427 | Abrupt duct termination |  |
DCM | 111428 | Extravasation |  |
DCM | 111429 | Duct narrowing |  |
DCM | 111430 | Cyst fill |  |

## CID 6058 Procedure Modifiers for Breast

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.387

Table CID 6058. Procedure Modifiers for Breast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6059 “Breast Implant Types”
Include CID 6060 “Breast Biopsy Techniques”
Include CID 6061 “Breast Imaging Procedure Modifiers”
Include CID 12224 “Ultrasound Image Modes”

## CID 6059 Breast Implant Types

### Note

Some of these terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.388

Table CID 6059. Breast Implant Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-04830 | Breast implant, type not specified | 2282003 | C0179412
SRT | A-04831 | Silicone gel implant | 257357007 | C0441274
DCM | 111481 | Saline implant |  |
DCM | 111482 | Polyurethane implant |  |
DCM | 111483 | Percutaneous silicone injection |  |
DCM | 111484 | Combination implant |  |
DCM | 111485 | Pre-pectoral implant |  |
DCM | 111486 | Retro-pectoral implant |  |

## CID 6060 Breast Biopsy Techniques

### Note

Some of these terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.389

Table CID 6060. Breast Biopsy Techniques

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P1-030C4 | Lumpectomy | 392021009 | C0851238
UMLS | C0024881 | Mastectomy |  | C0024881
SRT | P1-4834A | Quadrantectomy of breast | 172049005 | C0337354
SRT | P5-00032 | Diagnostic radiography, stereotactic localization | 64318009 | C0202577
SRT | P5-B0700 | Ultrasonic guidance procedure | 61593002 | C0442973
SRT | P5-40010 | Mammography | 71651007 | C0024671
DCM | 111487 | Mammographic (crosshair) |  |
DCM | 111488 | Mammographic (grid) |  |
SRT | P1-03107 | Magnetic resonance imaging guided biopsy | 277592004 | C0456854
SRT | P1-03106 | Computed tomography guided biopsy | 277591006 | C0456853
DCM | 111489 | Palpation guided |  |
DCM | 111490 | Vacuum assisted |  |

### Note

1. In a prior version of this Context Group, the code P1-03115 was specified for the concept "Ultrasound guided biopsy". The use of this code is too restrictive, and its use in this context is deprecated. There is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use.

2. The incorrect code P1-43850 was previously used for mastectomy, presumably a two-character transposition of the actual SNOMED concept (P1-48350, SRT, "Mastectomy"); since the correct SNOMED concept is inactive (has a ConceptStatus in SNOMED of ambiguous) and there is no replacement, the corresponding UMLS concept (which maps to multiple coding schemes) is used instead. Currently SNOMED contains a more generic parent concept "Excision of breast tissue", which includes procedures that are less than a mastectomy, e.g., "excisional biopsy of breast", and only specific types of mastectomy, e.g., "simple mastectomy" or "mastectomy of left breast".

## CID 6061 Breast Imaging Procedure Modifiers

Type:

  Extensible

Version:

  20090819

UID:

  1.2.840.10008.6.1.390

Table CID 6061. Breast Imaging Procedure Modifiers

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-42453 | Screening | 360156006 | C1305399
SRT | R-408C3 | Diagnostic | 261004008 | C0348026
DCM | 111127 | Targeted |  |
DCM | 111128 | Survey |  |
DCM | 122505 | Calibration |  |
DCM | 110002 | Quality Control |  |
DCM | 111144 | Needle localization and biopsy |  |
DCM | 111123 | Marker placement |  |

## CID 6062 Interventional Procedure Complications

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.391

Table CID 6062. Interventional Procedure Complications

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | DD-66A67 | Hemorrhage postprocedure | 110265006 | C0032788
DCM | 111491 | Abnormal discharge |  |
SRT | F-01FBA | Hematoma - postoperative | 213262007 | C0472340
SRT | D0-00165 | Weal | 247472004 | C0221232
SRT | DD-67700 | Infection as complication of medical care | 69698001 | C0274432
SRT | F-A2632 | Persistent pain following procedure | 279047007 | C0458166
SRT | D2-80300 | Pneumothorax | 36118008 | C0032326
SRT | D0-00058 | Rash | 271807003 | C0015230
SRT | M-02570 | Swelling | 65124004 | C0038999
SRT | F-A558A | Vasovagal syncope | 398665005 | C0042420
DCM | 111492 | No complications |  |

## CID 6063 Interventional Procedure Results

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.392

Table CID 6063. Interventional Procedure Results

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A249 | Benign | 30807003 | C0205183
SRT | R-41DDC | High risk tumor | 258270003 | C0475283
SRT | G-A425 | Malignant | 21594007 | C0205282
SRT | M-09024 | Insufficient sample | 281268007 | C0460062
SRT | F-01E06 | Indeterminate result | 280416009 | C0459425

## CID 6064 Ultrasound Findings for Breast

### Note

These terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.393

Table CID 6064. Ultrasound Findings for Breast

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01BF8 | Ultrasound scan normal | 169254007 | C0581117
SRT | D7-90035 | Cyst of breast | 399294002 | C0006144
DCM | 111460 | Complex cyst |  |
DCM | 111461 | Intracystic lesion |  |
SRT | D7-90370 | Mammary duct ectasia | 22049009 | C0152442
DCM | 111462 | Solid mass |  |
SRT | T-C4000 | Lymph node | 59441001 | C0024204
SRT | D7-90382 | Sebaceous cyst of skin of breast | 76649007 | C0342082
DCM | 111129 | Clustered microcysts |  |
DCM | 111130 | Complicated cyst |  |
SRT | M-30400 | Foreign body | 19227008 | C0016542

## CID 6065 Instrument Approach

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.394

Table CID 6065. Instrument Approach

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-404D5 | Medial | 255561001 | C0205098
SRT | G-A104 | Lateral | 49370004 | C0205093
SRT | R-42191 | Superior | 264217000 | C1282910
SRT | R-4094A | Inferior | 261089000 | C0542339
DCM | 111432 | Inferolateral to superomedial |  |
DCM | 111433 | Inferomedial to superolateral |  |
DCM | 111434 | Superolateral to inferomedial |  |
DCM | 111435 | Superomedial to inferolateral |  |

## CID 6066 Target Confirmation

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.395

Table CID 6066. Target Confirmation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111443 | Target contained in the specimen
DCM | 111444 | Target partially obtained in the specimen
DCM | 111445 | Target not in the specimen
DCM | 111446 | Calcifications seen in the core
DCM | 111447 | Lesion completely removed
DCM | 111448 | Lesion partially removed
DCM | 111449 | Fluid obtained

## CID 6067 Fluid Color

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.396

Table CID 6067. Fluid Color

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A12B | White color | 371251000 | C0220938
SRT | G-A11D | Yellow color | 371244009 | C0221205
DCM | 111450 | Light brown color |  |
SRT | G-A11E | Green color | 371246006 | C0332583
SRT | G-A12D | Gray color | 371253002 | C1269776
DCM | 111451 | Dark red color |  |
DCM | 111452 | Dark brown color |  |
SRT | R-4205B | Clear | 263707001 | C2963144
SRT | G-A12E | Brown color | 371254008 | C0678579
DCM | 111453 | Bright red color |  |
DCM | 111454 | Blood tinged color |  |
SRT | G-A12C | Black color | 371252007 | C0439541

## CID 6068 Tumor Stages From AJCC

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.397

Table CID 6068. Tumor Stages From AJCC

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111494 | Stage 0
DCM | 111495 | Stage I
DCM | 111496 | Stage IIA
DCM | 111497 | Stage IIB
DCM | 111498 | Stage IIIA
DCM | 111499 | Stage IIIB
DCM | 111500 | Stage IIIC
DCM | 111501 | Stage IV

## CID 6069 Nottingham Combined Histologic Grade

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.398

Table CID 6069. Nottingham Combined Histologic Grade

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-02B9B | Nottingham Combined Grade cannot be determined | 384668003 | C1273755
SRT | G-F616 | Nottingham Combined Grade I: 3-5 points | 369790002 | C1298194
SRT | G-F617 | Nottingham Combined Grade II: 6-7 points | 369791003 | C1298195
SRT | G-F618 | Nottingham Combined Grade III: 8-9 points | 369792005 | C1298196

## CID 6070 Bloom-Richardson Histologic Grade

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.399

Table CID 6070. Bloom-Richardson Histologic Grade

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-F211 | Grade 1: well differentiated | 54102005 | C0475269
SRT | G-F212 | Grade 2: moderately differentiated | 1663004 | C0475270
SRT | G-F213 | Grade 3: poorly differentiated | 61026006 | C0475271
SRT | R-41DC5 | Grade 4: undifferentiated | 258245003 | C0475272

## CID 6071 Histologic Grading Method

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.400

Table CID 6071. Histologic Grading Method

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111502 | Bloom-Richardson Grade |  |
SRT | R-00288 | Nottingham Combined Grade | 372276001 | C1276778

## CID 6072 Breast Implant Findings

### Note

These terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.401

Table CID 6072. Breast Implant Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111503 | Normal implants |  |
DCM | 111504 | Asymmetric implants |  |
DCM | 111505 | Calcified implant |  |
DCM | 111506 | Distorted implant |  |
DCM | 111507 | Silicone-laden lymph nodes |  |
DCM | 111508 | Free silicone |  |
DCM | 111509 | Herniated implant |  |
SRT | DD-66544 | Rupture of breast implant | 237473006 | C0405491
DCM | 111510 | Explantation |  |

## CID 6080 Gynecological Hormones

### Note

Some of these terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.402

Table CID 6080. Gynecological Hormones

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-B7100 | Contraceptives | 108899006 | C0009871
SRT | C-A0900 | Estrogen product | 61946003 | C0014939
SRT | C-A1204 | Progesterone product | 50318003 | C0033308
SRT | C-781E0 | Tamoxifen | 75959001 | C0039286
DCM | 111542 | Unspecified gynecological hormone |  |
SRT | C-A0005 | Raloxifene | 109029006 | C0244404
SRT | F-61B21 | Anastrozole | 386910003 | C0290883

## CID 6081 Breast Cancer Risk Factors

### Note

Some of these terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.403

Table CID 6081. Breast Cancer Risk Factors

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111550 | Personal breast cancer history |  |
DCM | 111551 | History of endometrial cancer |  |
DCM | 111552 | History of ovarian cancer |  |
DCM | 111553 | History of high risk lesion on previous biopsy |  |
DCM | 111554 | Post menopausal patient |  |
SRT | F-84430 | Nulliparous | 102877006 | C0425979
DCM | 111555 | Late child bearing (after 30) |  |
DCM | 111556 | BRCA1 breast cancer gene |  |
DCM | 111557 | BRCA2 breast cancer gene |  |
DCM | 111558 | BRCA3 breast cancer gene |  |
SRT | G-04C5 | Family history of breast cancer | 429740004 | C1261325
DCM | 111559 | Weak family history of breast cancer |  |
DCM | 111560 | Intermediate family history of breast cancer |  |
DCM | 111561 | Very strong family history of breast cancer |  |
DCM | 111562 | Family history of prostate cancer |  |
DCM | 111563 | Family history unknown |  |
SRT | R-207AD | No family history of breast carcinoma | 313376005 | C1277317

## CID 6082 Gynecological Procedures

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.404

Table CID 6082. Gynecological Procedures

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P0-05CCA | Endometrial biopsy | 386802000 | C1510477
SRT | P1-8330D | Hysterectomy | 236886002 | C0020699
SRT | P1-03151 | Dilation and curettage | 13091001 | C0012358

## CID 6083 Procedures for Breast

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.405

Table CID 6083. Procedures for Breast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111564 | Nipple discharge cytology
Include CID 6050 “Breast Procedure Reported”
Include CID 6084 “Mammoplasty Procedures”
Include CID 6085 “Therapies for Breast”

## CID 6084 Mammoplasty Procedures

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.406

Table CID 6084. Mammoplasty Procedures

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P1-48501 | Breast implantation | 119853006 | C0178391
SRT | P1-48830 | Reduction mammoplasty | 59214008 | C0191922
SRT | P1-48820 | Breast reconstruction | 33496007 | C0085076
SRT | P1-48520 | Removal of breast implant | 27315000 | C0191909

## CID 6085 Therapies for Breast

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.407

Table CID 6085. Therapies for Breast

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P0-0058E | Chemotherapy | 367336001 | C3665472
SRT | P5-C0000 | Radiation therapy | 53438000 | C1522449
SRT | P0-007AC | Hormone therapy | 169413002 | C0279025
SRT | P1-67D40 | Bone marrow transplant | 23719005 | C0005961

## CID 6086 Menopausal Phase

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.408

Table CID 6086. Menopausal Phase

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-41FFF | Before menopause | 309606002 | C0587111
SRT | R-422A5 | During menopause | 303111005 | C0587112
SRT | R-410C3 | After menopause | 307429007 | C0587113
SRT | D7-76202 | Postsurgical menopause | 371036001 | C0740421
SRT | D7-76200 | Artificial menopause state | 31351009 | C0232972

## CID 6087 General Risk Factors

This context group collects risk factor terms from specialized risk factor context groups into one aggregate list for general purpose use.

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.409

Table CID 6087. General Risk Factors

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6081 “Breast Cancer Risk Factors”
Include CID 6088 “OB-GYN Maternal Risk Factors”

## CID 6088 OB-GYN Maternal Risk Factors

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.410

Table CID 6088. OB-GYN Maternal Risk Factors

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-023F | History of - diabetes mellitus | 161445009 | C0455488
SRT | G-0269 | History of - hypertension | 161501007 | C0455527
SRT | G-0244 | History of - obesity | 161453001 | C0455493
SRT | G-02D0 | History of - regular medication | 161656000 | C0455633
SRT | G-0338 | History of substance abuse | 371422002 | C1299544
SRT | G-0335 | History of - cardiovascular disease | 266995000 | C0455539
DCM | 111565 | Uterine malformations |  |
SRT | G-0304 | History of - ectopic pregnancy | 161763005 | C0438096
DCM | 111566 | Spontaneous Abortion |  |
DCM | 111567 | Gynecologic condition |  |
DCM | 111568 | Gynecologic surgery |  |
SRT | G-031E | History of - eclampsia | 161806007 | C0438072
SRT | G-031F | History of - severe pre-eclampsia | 161807003 | C0438073
DCM | 111569 | Previous LBW or IUGR birth |  |
DCM | 111570 | Previous fetal malformation/syndrome |  |
SRT | G-0305 | History of - premature delivery | 161765003 | C0438076
DCM | 111571 | Previous RH negative or blood dyscrasia at birth |  |
SRT | G-0319 | History of infertility | 161798008 | C0438063
DCM | 111572 | History of multiple fetuses |  |
SRT | D8-20100 | Multiple pregnancy | 16356006 | C0032989
DCM | 111573 | Current pregnancy, known or suspected malformations/syndromes |  |
DCM | 111574 | Family history, fetal malformation/syndrome |  |

## CID 6089 Substances

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.411

Table CID 6089. Substances

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-21047 | Ethyl alcohol | 419442005 | C0001962
SRT | R-FBDEA | Amphetamine | 703842006 | C0002658
SRT | F-61D6F | Marijuana | 398705004 | C0678449
SRT | F-61C76 | Cocaine | 387085005 | C0009170
SRT | F-61AC4 | Heroin | 387341002 | C0011892
SRT | C-63A10 | Lysergic acid diethlyamide | 15698006 | C0024334
SRT | F-6169A | Mescaline | 373780001 | C0025460
SRT | C-6A180 | Phencyclidine | 9721008 | C0031381
SRT | F-61A95 | Methadone | 387286002 | C0025605
SRT | F-618D7 | Morphine | 373529000 | C0026549
SRT | F-618FE | Methlyphenidate | 373337007 | C0025810
SRT | C-F3310 | Chewing tobacco | 81911001 | C0008038
SRT | C-F3302 | Cigarette smoking tobacco | 66562002 | C0301612
SRT | F-61117 | Caffeine | 255641001 | C0006644

## CID 6090 Relative Usage, Exposure Amount

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.412

Table CID 6090. Relative Usage, Exposure Amount

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111575 | High
DCM | 111576 | Medium
DCM | 111577 | Low
DCM | 111587 | No known exposure

## CID 6091 Relative Frequency of Event Values

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.413

Table CID 6091. Relative Frequency of Event Values

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-40377 | Continuous | 255238004 | C0549178
SRT | G-7154 | Frequent | 70232002 | C0332183
SRT | R-40365 | Mid-frequency | 255218000 | C0439604
SRT | G-7155 | Infrequent | 27789000 | C0521114
SRT | R-40B16 | As required | 225761000 | C0558288
SRT | R-4112F | Single event | 307486002 | C0585347

## CID 6092 Quantitative Concepts for Usage, Exposure

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.414

Table CID 6092. Quantitative Concepts for Usage, Exposure

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-C0B7 | Dosage | 260911001 | C0178602
DCM | 111578 | Dose frequency |  |
DCM | 111579 | Rate of exposure |  |
DCM | 111580 | Volume of use |  |

## CID 6093 Qualitative Concepts for Usage, Exposure Amount

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.415

Table CID 6093. Qualitative Concepts for Usage, Exposure Amount

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111581 | Relative dose amount
DCM | 111582 | Relative amount of exposure
DCM | 111583 | Relative amount of use

## CID 6094 Qualitative Concepts for Usage, Exposure Frequency

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.416

Table CID 6094. Qualitative Concepts for Usage, Exposure Frequency

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111584 | Relative dose frequency
DCM | 111585 | Relative frequency of exposure
DCM | 111586 | Relative frequency of use

## CID 6095 Numeric Properties of Procedures

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.417

Table CID 6095. Numeric Properties of Procedures

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111465 | Needle Gauge
DCM | 111467 | Needle Length

## CID 6096 Pregnancy Status

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.418

Table CID 6096. Pregnancy Status

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-81890 | not pregnant | 60001007 | C0232973
SRT | F-84094 | possible pregnancy | 102874004 | C0425965
SRT | F-84000 | patient currently pregnant | 77386006 | C0549206
SRT | R-41198 | Unknown | 261665006 | C0439673

## CID 6097 Side of Family

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.419

Table CID 6097. Side of Family

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111541 | Maternal |  |
SRT | R-40333 | Paternal | 224944003 | C0337493

## CID 6100 Chest Component Categories

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.420

Table CID 6100. Chest Component Categories

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-28000 | Lung | 39607008 | C0024109
DCM | 112052 | Bronchovascular |  |
SRT | T-29000 | Pleural structure | 3120008 | C0032225
SRT | T-D3300 | Mediastinum | 72410000 | C0025066
SRT | T-32000 | Heart | 80891009 | C0018787
DCM | 112053 | Osseous |  |
SRT | T-4000E | Systemic vascular structure | 281157001 | C0459962
SRT | R-420AE | Muscular | 263816006 | C0442025

## CID 6101 Chest Finding or Feature

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.421

Table CID 6101. Chest Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112061 | Abnormal lines (1D)
DCM | 112033 | Abnormal opacity
DCM | 112062 | Abnormal lucency
DCM | 112063 | Abnormal calcifications
DCM | 112064 | Abnormal texture
DCM | 112005 | Radiographic anatomy
DCM | 111102 | Non-lesion
DCM | 111101 | Image quality
DCM | 111099 | Selected region

## CID 6102 Chest Finding or Feature Modifier

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.422

Table CID 6102. Chest Finding or Feature Modifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6103 “Abnormal Lines Finding or Feature”
Include CID 6104 “Abnormal Opacity Finding or Feature”
Include CID 6105 “Abnormal Lucency Finding or Feature”
Include CID 6106 “Abnormal Texture Finding or Feature”
Include CID 6109 “Radiographic Anatomy Finding or Feature”
Include CID 6138 “Chest Non-lesion Object Type”

## CID 6103 Abnormal Lines Finding or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.423

Table CID 6103. Abnormal Lines Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 112065 | Reticulonodular pattern |  |
DCM | 112104 | Air-fluid level |  |
DCM | 112105 | Corona radiata |  |
DCM | 112106 | Honeycomb pattern |  |
DCM | 112107 | Fleischner's line(s) |  |
DCM | 112108 | Intralobular lines |  |
DCM | 112109 | Kerley A line |  |
DCM | 112110 | Kerley B line |  |
DCM | 112111 | Kerley C lines |  |
DCM | 112112 | Parenchymal band |  |
SRT | D2-60302 | Plate-like atelectasis | 40779009 | C0264494
DCM | 112113 | Reticular pattern |  |
DCM | 112114 | Septal line(s) |  |
DCM | 112115 | Subpleural line |  |
DCM | 112116 | Tramline shadow |  |
DCM | 112117 | Tubular shadow |  |

## CID 6104 Abnormal Opacity Finding or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.424

Table CID 6104. Abnormal Opacity Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 112066 | Beaded septum sign |  |
DCM | 112067 | Nodular pattern |  |
DCM | 112059 | Primary complex |  |
DCM | 112068 | Pseudoplaque |  |
DCM | 112065 | Reticulonodular pattern |  |
DCM | 112069 | Signet-ring sign |  |
DCM | 112004 | Abnormal interstitial pattern |  |
SRT | F-20172 | Coin lesion | 308689002 | C0009250
DCM | 112118 | Density |  |
DCM | 112119 | Dependent opacity |  |
DCM | 112120 | Ground glass opacity |  |
DCM | 112121 | Infiltrate |  |
SRT | M-03000 | Mass | 4147007 | C0577559
DCM | 112122 | Micronodule |  |
SRT | M-03010 | Nodule | 27925004 | C0028259
DCM | 112001 | Opacity |  |
DCM | 112123 | Phantom tumor (pseudotumor) |  |
DCM | 112124 | Shadow |  |
DCM | 112125 | Small irregular opacities |  |
DCM | 112126 | Small rounded opacities |  |
DCM | 112127 | Tree-in-bud sign |  |
SRT | D3-40230 | Pulmonary embolism | 59282003 | C0034065

## CID 6105 Abnormal Lucency Finding or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.425

Table CID 6105. Abnormal Lucency Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 112070 | Air bronchiologram |  |
DCM | 112071 | Air bronchogram |  |
DCM | 112072 | Air crescent |  |
SRT | F-20240 | Air-trapping | 76171001 | C0231819
DCM | 112073 | Halo sign |  |
SRT | D2-81180 | Pneumomediastinum | 16838000 | C0025062
SRT | D2-80300 | Pneumothorax | 36118008 | C0032326

## CID 6106 Abnormal Texture Finding or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.426

Table CID 6106. Abnormal Texture Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112067 | Nodular pattern
DCM | 112065 | Reticulonodular pattern
DCM | 112004 | Abnormal interstitial pattern
DCM | 112128 | Granular pattern
DCM | 112106 | Honeycomb pattern
DCM | 112129 | Miliary pattern
DCM | 112130 | Mosaic pattern
DCM | 112113 | Reticular pattern
DCM | 112125 | Small irregular opacities

## CID 6107 Width Descriptor

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.427

Table CID 6107. Width Descriptor

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-40750 | Enlarged | 260376009 | C0442800
SRT | R-41727 | Narrow | 134223000 | C0333164
DCM | 112077 | Vasoconstriction |  |
DCM | 112078 | Vasodilation |  |

## CID 6108 Chest Anatomic Structure Abnormal Distribution

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.428

Table CID 6108. Chest Anatomic Structure Abnormal Distribution

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-20240 | Air-trapping | 76171001 | C0231819
DCM | 112079 | Architectural distortion |  |
DCM | 112080 | Mosaic perfusion |  |
DCM | 112060 | Oligemia |  |
DCM | 112081 | Pleonemia |  |

## CID 6109 Radiographic Anatomy Finding or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.429

Table CID 6109. Radiographic Anatomy Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6110 “Lung Anatomy Finding or Feature”
Include CID 6111 “Bronchovascular Anatomy Finding or Feature”
Include CID 6112 “Pleura Anatomy Finding or Feature”
Include CID 6113 “Mediastinum Anatomy Finding or Feature”
Include CID 6114 “Osseous Anatomy Finding or Feature”
Include CID 6116 “Muscular Anatomy”
Include CID 6117 “Vascular Anatomy”
DCM | 112082 | Interface
DCM | 112083 | Line
DCM | 112084 | Lucency

## CID 6110 Lung Anatomy Finding or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.430

Table CID 6110. Lung Anatomy Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-28770 | Lobe of lung | 31094006 | C0225752
DCM | 112085 | Midlung window |  |
DCM | 112054 | Secondary pulmonary lobule |  |
SRT | T-280D0 | Segment of lung | 72674008 | C0225705

## CID 6111 Bronchovascular Anatomy Finding or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.431

Table CID 6111. Bronchovascular Anatomy Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-20001 | Airway structure | 89187006 | C0458827
SRT | T-26000 | Bronchus | 955009 | C0006255
SRT | T-25201 | Carina | 28700002 | C0225594
DCM | 112086 | Carina angle |  |
DCM | 112087 | Centrilobular structures |  |
SRT | T-28080 | Hilum of lung | 46750007 | C0225701

## CID 6112 Pleura Anatomy Finding or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.432

Table CID 6112. Pleura Anatomy Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 112088 | Anterior junction line |  |
SRT | T-D051D | Fissure of lung | 278983006 | C0458078
DCM | 112089 | Posterior junction line |  |

## CID 6113 Mediastinum Anatomy Finding or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20130617

UID:

  1.2.840.10008.6.1.433

Table CID 6113. Mediastinum Anatomy Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-15420 | Acromioclavicular Joint | 85856004 | C0001208
SRT | T-35400 | Aortic Valve | 34202007 | C0003501
SRT | D4-31220 | Atrial Septal Defect | 70142008 | C0018817
SRT | T-32100 | Atrium | 59652004 | C0018792
SRT | T-18774 | Axillary Fascia | 368536000 | C0225236
DCM | 112090 | Azygoesophageal recess interface |  |
SRT | T-25201 | Carina | 28700002 | C0225594
SRT | T-B4000 | Carotid Body | 51345006 | C0007277
SRT | T-11240 | Costal Cartilage | 50016007 | C0222787
SRT | T-D3412 | Esophageal Hiatus | 280062008 | C0230160
SRT | T-56000 | Esophagus | 32849002 | C0014876
SRT | T-D0634 | Fascial layer | 120576005 | C1268198
SRT | T-32000 | Heart | 80891009 | C0018787
DCM | 112095 | Hiatus |  |
SRT | T-26500 | Left main bronchus | 75245000 | C0225630
SRT | T-42370 | Ligamentum arteriosum | 2160002 | C0226023
SRT | T-C4000 | Lymph node | 59441001 | C0024204
SRT | T-35300 | Mitral Valve | 91134007 | C0026264
DCM | 112091 | Paraspinal line |  |
DCM | 112092 | Posterior tracheal stripe |  |
SRT | T-35200 | Pulmonary valve | 39057004 | C0034086
SRT | T-26100 | Right main bronchus | 70074004 | C0225608
DCM | 112093 | Right tracheal stripe |  |
DCM | 112094 | Stripe |  |
SRT | T-C6510 | Thoracic Duct | 1732005 | C0039979
SRT | T-C8000 | Thymus Gland | 9875009 | C0040113
SRT | T-B6000 | Thyroid | 69748006 | C0040132
SRT | T-25000 | Trachea | 44567001 | C0040578
SRT | T-14171 | Trapezius muscle | 31764008 | C0224361
SRT | T-35100 | Tricuspid Valve | 46030003 | C0040960
SRT | T-32400 | Ventricle | 21814001 | C0018827

## CID 6114 Osseous Anatomy Finding or Feature

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.434

Table CID 6114. Osseous Anatomy Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-12310 | Clavicle | 51299004 | C0008913
SRT | T-12410 | Humerus | 85050009 | C0020164
SRT | T-11300 | Rib | 113197003 | C0035561
SRT | T-12280 | Scapula | 79601000 | C0036277
SRT | T-D04FF | Spine | 421060004 | C0037949
SRT | T-11210 | Sternum | 56873002 | C0038293
SRT | T-11510 | Vertebra | 51282000 | C0549207

## CID 6115 Osseous Anatomy Modifiers

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.435

Table CID 6115. Osseous Anatomy Modifiers

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-12281 | Acromion process of scapula | 31934006 | C0001209
SRT | T-11307 | Angle of rib | 14510004 | C0222812
SRT | T-11511 | Arch of vertebra | 40265002 | C0223076
SRT | T-11220 | Body of sternum | 52509009 | C0222771
SRT | T-11219 | Clavicular notch of sternum | 75319007 | C0222770
SRT | T-12282 | Coracoid process of scapula | 8931003 | C0223626
SRT | T-11308 | Costal groove | 17399006 | C0222813
SRT | T-12287 | Dorsal aspect of scapula | 51698000 | C0223631
SRT | T-1228A | Glenoid cavity of scapula | 46385009 | C1261046
SRT | T-11301 | Head of rib | 12872006 | C0222806
SRT | T-116EF | Inferior articular facet of axis | 181901007 | C0223115
SRT | T-1153F | Inferior articular process of vertebra | 317766009 | C0223083
SRT | T-11514 | Lamina of vertebra | 89340005 | C0223079
SRT | T-11211 | Manubrium of sternum | 37285002 | C0024764
SRT | T-11303 | Neck of rib | 72184008 | C0222808
SRT | T-12200 | Pectoral girdle | 26444007 | C0427245
SRT | T-11515 | Pedicle of vertebra | 78972004 | C0223080
DCM | 112096 | Rib Scalene Tubercle |  |
DCM | 112101 | Scapular Infraspinatus Fossa |  |
DCM | 112099 | Scapular Spine |  |
DCM | 112100 | Scapular Supraspinatus Fossa |  |
SRT | T-11309 | Shaft of rib | 41601005 | C0448161
SRT | T-11512 | Spinous process of vertebra | 55678000 | C0223077
SRT | T-11221 | Sternal angle | 44612009 | C0222772
DCM | 112098 | Subscapular Fossa |  |
SRT | T-116EE | Superior articular facet of axis | 181900008 | C0223114
SRT | T-1153E | Superior articular process of vertebra | 317665004 | C0223082
SRT | T-11218 | Suprasternal notch | 26493002 | C0222769
SRT | T-11513 | Transverse process or vertebra | 73400003 | C0223078
SRT | T-11304 | Tubercle of rib | 113198008 | C0222809
SRT | T-1151F | Vertebral canal | 61853006 | C0037922
SRT | T-11531 | Vertebral foramen | 280734009 | C0459720
DCM | 112097 | Vertebral Intervertebral Notch |  |
SRT | T-11227 | Xiphoid process of sternum | 20298003 | C0043356

## CID 6116 Muscular Anatomy

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.436

Table CID 6116. Muscular Anatomy

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-35020 | Chordae tendineae cordis | 102298001 | C0008484
SRT | T-13660 | Deltoid muscle | 35259002 | C0224234
SRT | T-D3400 | Diaphragm | 5798000 | C0011980
SRT | T-14020 | Erector spinae muscle | 44947003 | C0224301
SRT | T-14161 | External intercostal muscle | 53967007 | C1744535
SRT | T-14030 | Iliocostalis muscle | 57651003 | C0224302
SRT | T-13620 | Infraspinatus muscle | 72573008 | C0584882
SRT | T-14165 | Innermost intercostal muscles | 24062007 | C0224357
SRT | T-32150 | Interatrial septum | 58095006 | C0225836
SRT | T-14163 | Internal intercostal muscle | 41313007 | C1744536
SRT | T-32410 | Interventricular septum | 589001 | C0225870
SRT | T-14172 | Latissimus dorsi muscle | 15665001 | C0224362
SRT | T-14150 | Levatores costarum muscles | 73930003 | C1744586
SRT | T-14040 | Longissimus muscle | 88340001 | C0224306
SRT | T-14110 | Pectoralis major muscle | 60005003 | C0585574
SRT | T-14120 | Pectoralis minor muscle | 18686000 | C0224347
SRT | T-13450 | Scalenous anterior muscle | 50755001 | C0224173
SRT | T-14140 | Serratus anterior muscle | 18346003 | C0224349
SRT | T-14050 | Spinalis muscle | 4317002 | C0224310
SRT | T-13310 | Sternocleidomastoid muscle | 22823000 | C0224153
SRT | T-14166 | Subcostal muscle | 64658001 | C0224358
SRT | T-13650 | Subscapularis muscle | 90588001 | C0584884
SRT | T-13610 | Supraspinatus muscle | 6423006 | C0584869
SRT | T-13640 | Teres major muscle | 1193009 | C0224232
SRT | T-13630 | Teres minor muscle | 51159009 | C0224231
SRT | T-32423 | Trabeculae carnae | 118755002 | C0502348
SRT | T-14167 | Transversus thoracis | 88454005 | C1744608
SRT | T-14171 | Trapezius muscle | 31764008 | C0224361

## CID 6117 Vascular Anatomy

Type:

  Extensible

Version:

  20130617

UID:

  1.2.840.10008.6.1.437

Table CID 6117. Vascular Anatomy

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 3015 “Coronary Arteries”
SRT | T-42300 | Aortic arch | 57034009 | C0003489
SRT | T-42310 | Aortic isthmus | 88593004 | C0226019
DCM | 112102 | Aortic knob |  |
DCM | 112103 | Arch of the Azygos vein |  |
SRT | T-42100 | Ascending aorta | 54247002 | C0003956
SRT | T-47100 | Axillary Artery | 67937003 | C0004455
SRT | T-49110 | Axillary vein | 68705008 | C0004456
SRT | T-48340 | Azygos vein | 72107004 | C0004526
SRT | T-47160 | Brachial artery | 17137000 | C0006087
SRT | T-A9090 | Brachial plexus | 36582005 | C0006090
SRT | T-46010 | Brachiocephalic trunk | 12691009 | C0006094
SRT | T-48620 | Brachiocephalic vein | 8887007 | C0006095
SRT | T-46310 | Bronchial artery | 64468002 | C0006257
SRT | T-45100 | Common carotid artery | 32062004 | C0162859
SRT | T-46180 | Costocervical trunk | 3159004 | C0226273
SRT | T-D0765 | Descending aorta | 281130003 | C0011666
SRT | T-461A0 | Dorsal scapular artery | 91732003 | C0500583
SRT | T-4630D | Esophageal artery | 206034008 | C0226294
SRT | T-46940 | Inferior phrenic artery | 29660000 | C0226406
SRT | T-48710 | Inferior vena cava | 64131007 | C0042458
SRT | T-D305A | Intercostal artery | 281134007 | C0459917
SRT | T-48170 | Internal jugular vein | 12123001 | C0226550
SRT | T-46200 | Internal thoracic artery | 69327007 | C0226276
SRT | T-46210 | Pericardiophrenic Artery | 3924000 | C0226287
SRT | T-44000 | Pulmonary artery | 81040000 | C0034052
SRT | T-44100 | Pulmonary trunk | 45341000 | C0034052
SRT | T-48581 | Pulmonary vein | 122972007 | C0034090
SRT | T-46100 | Subclavian artery | 36765005 | C0038530
SRT | T-48330 | Subclavian vein | 9454009 | C0038532
SRT | T-46350 | Superior phrenic artery | 38991005 | C0226295
SRT | T-48610 | Superior vena cava | 48345005 | C0042459
SRT | T-46130 | Thyrocervical trunk | 6538005 | C0226263
SRT | T-45700 | Vertebral artery | 85234005 | C0042559

### Note

In a prior version of this Context Group the code T-48500 rather than T-48581 was defined for the concept Pulmonary Vein; this was inconsistent with the DICOM approach of selecting the "structure of" rather than "entire" concept. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 6118 Size Descriptor

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.438

Table CID 6118. Size Descriptor

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 112131 | Extremely small |  |
DCM | 112132 | Very small |  |
SRT | R-404A8 | Small | 255507004 | C0700321
SRT | R-404A9 | Medium | 255508009 | C0439536
SRT | R-404AA | Large | 255509001 | C0549177
SRT | R-40750 | Enlarged | 260376009 | C0442800
DCM | 112133 | Too small |  |

## CID 6119 Chest Border Shape

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.439

Table CID 6119. Chest Border Shape

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-02100 | Round shape | 42700002 | C0332490
DCM | 112134 | Elliptic |  |
SRT | G-A402 | Irregular | 49608001 | C0205271
DCM | 112135 | Lobulated |  |
DCM | 112136 | Spiculated |  |

## CID 6120 Chest Border Definition

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.440

Table CID 6120. Chest Border Definition

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-40771 | Well defined | 260409000 | C0442825
DCM | 112137 | Sharply defined |  |
SRT | R-428E7 | Poorly defined | 300841009 | C0577553
DCM | 112138 | Distinctly defined |  |
DCM | 112139 | Well demarcated |  |
DCM | 112140 | Sharply demarcated |  |
DCM | 112141 | Poorly demarcated |  |
DCM | 112142 | Circumscribed |  |

## CID 6121 Chest Orientation Descriptor

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.441

Table CID 6121. Chest Orientation Descriptor

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A142 | Horizontal | 24020000 | C0205126
SRT | G-A144 | Vertical | 33096000 | C0205128
SRT | G-A472 | Oblique | 21114003 | C0205315

## CID 6122 Chest Content Descriptor

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.442

Table CID 6122. Chest Content Descriptor

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 112143 | Air |  |
SRT | T-D008A | Fat | 256674009 | C0015677
DCM | 112144 | Soft tissue |  |
DCM | 112145 | Calcium |  |
SRT | M-30400 | Foreign material (iodized oil, mercury,talc) | 19227008 | C0016542

## CID 6123 Chest Opacity Descriptor

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.443

Table CID 6123. Chest Opacity Descriptor

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112146 | Acinar
DCM | 112147 | Air space
DCM | 112148 | Fibronodular
DCM | 112149 | Fluffy
DCM | 112150 | Linear
DCM | 112151 | Profusion
DCM | 112152 | Silhouette sign

## CID 6124 Location in Chest

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.444

Table CID 6124. Location in Chest

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6125 “General Chest Location”
Include CID 6126 “Location in Lung”
Include CID 6127 “Segment Location in Lung”

## CID 6125 General Chest Location

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.445

Table CID 6125. General Chest Location

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A110 | Central | 26216008 | C0205099
SRT | G-A111 | Peripheral | 14414005 | C0205100
SRT | G-A122 | Apical | 43674008 | C0205111
SRT | G-A123 | Basal | 57195005 | C0205112

## CID 6126 Location in Lung

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.446

Table CID 6126. Location in Lung

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-D3208 | Upper zone of lung | 281392002 | C0559286
SRT | T-D3209 | Middle zone of lung | 281393007 | C0559287
SRT | T-D320A | Lower zone of lung | 281394001 | C0559288
SRT | T-28820 | Upper lobe of lung | 45653009 | C0225756
SRT | T-28825 | Middle lobe of lung | 40020002 | C0225757
SRT | T-28830 | Lower lobe of lung | 90572001 | C0225758
DCM | 112153 | Subpleural |  |

## CID 6127 Segment Location in Lung

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.447

Table CID 6127. Segment Location in Lung

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-28230 | Anterior segment of right upper lobe | 39743006 | C0225718
SRT | T-28630 | Anterior segment of left upper lobe | 22270008 | C0225742
SRT | T-28220 | Posterior segment of right upper lobe | 3236000 | C0225717

## CID 6128 Chest Distribution Descriptor

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.448

Table CID 6128. Chest Distribution Descriptor

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 112154 | Bat's wing distribution |  |
DCM | 112155 | Butterfly distribution |  |
DCM | 112156 | Centrilobular |  |
DCM | 112157 | Coalescent |  |
SRT | G-A321 | Diffuse | 19648000 | C0205219
SRT | M-020FA | Discoid | 255282008 | C0439641
SRT | G-A324 | Disseminated | 65709003 | C0205221
SRT | G-A351 | Focal | 87017008 | C0205234
SRT | G-A366 | Generalized | 60132005 | C0205246
DCM | 112158 | Lobar |  |
SRT | G-A443 | Multifocal | 524008 | C0205292
SRT | G-A137 | Segmental | 62372003 | C0205122
SRT | G-A572 | Systemic | 31099001 | C0205373

## CID 6129 Chest Site Involvement

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.449

Table CID 6129. Chest Site Involvement

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-28000 | Lung | 39607008 | C0024109
SRT | T-D3300 | Mediastinum | 72410000 | C0025066
DCM | 112158 | Lobar |  |
SRT | T-1A007 | Interstitial tissue | 85293002 | C0225318
SRT | R-40939 | Bronchial | 261061003 | C0205039
SRT | T-28080 | Hilum of lung | 46750007 | C0225701
SRT | T-42000 | Aorta | 15825003 | C0003483
SRT | T-29000 | Pleural structure | 3120008 | C0032225
SRT | T-D3050 | Chest wall | 78904004 | C0205076
SRT | T-D4001 | Upper abdomen | 80581009 | C2937240

## CID 6130 Severity Descriptor

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.450

Table CID 6130. Severity Descriptor

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-404FA | Mild | 255604002 | C2945599
SRT | G-A002 | Moderate | 6736007 | C0205081
SRT | G-A003 | Severe | 24484000 | C0205082
SRT | R-424BE | Acute onset | 373933003 | C1276802
SRT | G-A270 | Chronic | 90734009 | C0205191
DCM | 112159 | Hyper-acute |  |
SRT | G-A561 | Subacute | 19939008 | C0205365

## CID 6131 Chest Texture Descriptor

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.451

Table CID 6131. Chest Texture Descriptor

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112160 | Homogeneous (uniform opacity)
DCM | 112161 | Inhomogeneous

## CID 6132 Chest Calcification Descriptor

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.452

Table CID 6132. Chest Calcification Descriptor

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01763 | Eggshell calcification | 129751002 | C1313950
SRT | F-01761 | Coarse (popcorn-like) calcification | 129749001 | C1268677
DCM | 112162 | Target |  |
SRT | G-A405 | Laminated | 88446008 | C0205274
DCM | 112163 | Fibrocalcific |  |
DCM | 112164 | Flocculent |  |
SRT | R-403A7 | Nodular | 255288007 | C0205297
SRT | F-12100 | Ossification | 83323007 | 83323007

## CID 6133 Chest Quantitative Temporal Difference Type

Type:

  Extensible

Version:

  20100604

UID:

  1.2.840.10008.6.1.453

Table CID 6133. Chest Quantitative Temporal Difference Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-05173 | Difference in size | 442714003 | C2711955
SRT | F-05179 | Difference in location | 442726008 | C2711109

## CID 6134 Chest Qualitative Temporal Difference Type

Type:

  Extensible

Version:

  20090717

UID:

  1.2.840.10008.6.1.454

Table CID 6134. Chest Qualitative Temporal Difference Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-0517E | Difference in border shape | 442755000 | C2711283
SRT | F-05166 | Difference in border definition | 442688001 | C2711343
SRT | F-0516C | Difference in distribution | 442704007 | C2711851
SRT | F-05170 | Difference in site involvement | 442711006 | C2711937
SRT | F-05167 | Difference in substance | 442691001 | C2711644
SRT | F-0516A | Difference in Texture | 442700003 | C2711323
SRT | F-01722 | Finding partially removed | 129722001 | C1268650
SRT | F-01723 | No significant changes in the finding | 129723006 | C1268651
SRT | M-02520 | Increase in size | 15454001 | C0332509
SRT | M-02530 | Decrease in size | 19776001 | C0332511
SRT | F-01728 | Less defined | 129728002 | C1268656
SRT | F-01729 | More defined | 129729005 | C1268657

## CID 6135 Image Quality Finding

Type:

  Extensible

Version:

  20090402

UID:

  1.2.840.10008.6.1.455

Table CID 6135. Image Quality Finding

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111208 | Grid artifact(s)
DCM | 111209 | Positioning
DCM | 111210 | Motion blur
DCM | 111211 | Under exposed
DCM | 111212 | Over exposed
DCM | 111213 | No image
DCM | 111214 | Detector artifact(s)
DCM | 111215 | Artifact(s) other than grid or detector artifact
DCM | 111216 | Mechanical failure
DCM | 111217 | Electrical failure
DCM | 111218 | Software failure
DCM | 111219 | Inappropriate image processing
DCM | 111220 | Other failure
DCM | 111221 | Unknown failure
RADLEX | RID11327 | Beam-hardening artifact

## CID 6136 Chest Types of Quality Control Standard

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.456

Table CID 6136. Chest Types of Quality Control Standard

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112035 | Performance of Pediatric and Adult Chest Radiography, ACR
DCM | 112036 | ACR Position Statement
DCM | 111240 | Institutionally defined quality control standard
DCM | 112184 | Performance of Pediatric and Adult Thoracic CT
DCM | 112185 | Performance of CT for Detection of Pulmonary Embolism in Adults
DCM | 112186 | Performance of High-Resolution CT of the Lungs in Adults

## CID 6137 Types of CAD Analysis

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.457

Table CID 6137. Types of CAD Analysis

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Notes
--- | --- | --- | --- | --- | ---
SRT | P5-B3402 | Spatial collocation analysis | 133884007 | C1297892 | See Note 1
SRT | P5-B3404 | Spatial proximity analysis | 133885008 | C1297893 | See Note 2
SRT | P5-B3406 | Temporal correlation | 133886009 | C1297894 |
SRT | P5-B3408 | Image quality analysis | 133887000 | C1297895 |

### Note

1. Spatial Co-location Analysis is used to identify features that are the same or located in the same place.

2. Spatial Proximity Analysis is used to identify different features that are related spatially.

## CID 6138 Chest Non-lesion Object Type

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.458

Table CID 6138. Chest Non-lesion Object Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 6404 “Chest Non-lesion Object Type - Physical Objects”
Include CID 6405 “Chest Non-lesion Object Type - Tissues”

### Note

The use of (111176, DCM, "Unspecified") was previously included in this context group but was removed since it does not make sense to have Chest CAD detections of an unknown type (was using TID 4015 “CAD Detections Performed” invoked by TID 4100 “Chest CAD Document Root”, TID 4102 “Chest CAD Composite Feature” and TID 4104 “Chest CAD Single Image Finding” of an unknown type).

## CID 6139 Non-lesion Modifiers

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.459

Table CID 6139. Non-lesion Modifiers

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-40819 | Internal | 260521003 | C0205102
SRT | R-40941 | External | 261074009 | C0205101

## CID 6140 Calculation Methods

Type:

  Extensible

Version:

  20070625

UID:

  1.2.840.10008.6.1.460

Table CID 6140. Calculation Methods

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-10260 | Estimated | 414135002 | C0750572
DCM | 112187 | Unspecified method of calculation |  |
DCM | 112055 | Agatston scoring method |  |
DCM | 112056 | Volume scoring method |  |
DCM | 112057 | Mass scoring method |  |
DCM | 112188 | Two-dimensional method |  |
DCM | 112189 | Three-dimensional method |  |

## CID 6141 Attenuation Coefficient Measurements

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.461

Table CID 6141. Attenuation Coefficient Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112031 | Attenuation Coefficient
DCM | 112179 | Minimum Attenuation Coefficient
DCM | 112180 | Maximum Attenuation Coefficient
DCM | 112181 | Mean Attenuation Coefficient
DCM | 112182 | Median Attenuation Coefficient
DCM | 112183 | Standard Deviation of Attenuation Coefficient

## CID 6142 Calculated Value

Type:

  Extensible

Version:

  20070625

UID:

  1.2.840.10008.6.1.462

Table CID 6142. Calculated Value

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112017 | Cavity extent as percent of volume
DCM | 112018 | Calcification extent as percent of surface
DCM | 112019 | Calcification extent as percent of volume
DCM | 112058 | Calcium score
DCM | 112191 | Breast tissue density
DCM | 112192 | Volume of parenchymal tissue
DCM | 112193 | Volume of breast
DCM | 112194 | Mass of parenchymal tissue
DCM | 112195 | Mass of breast
DCM | 112196 | Area of Vascular Calcification
DCM | 112197 | Volume of Vascular Calcification
DCM | 112198 | Percentage of Vascular Calcification
DCM | 112199 | Mass of Vascular Calcification
DCM | 112200 | Average calcification distance in a calcification cluster
DCM | 112201 | Standard deviation distance of calcifications in a cluster

## CID 6143 Lesion Response

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.463

Table CID 6143. Lesion Response

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6144 “RECIST Defined Lesion Response”

## CID 6144 RECIST Defined Lesion Response

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.464

Table CID 6144. RECIST Defined Lesion Response

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112041 | Target Lesion Complete Response
DCM | 112042 | Target Lesion Partial Response
DCM | 112043 | Target Lesion Progressive Disease
DCM | 112044 | Target Lesion Stable Disease
DCM | 112045 | Non-Target Lesion Complete Response
DCM | 112046 | Non-Target Lesion Incomplete Response or Stable Disease
DCM | 112047 | Non-Target Lesion Progressive Disease

## CID 6145 Baseline Category

### Note

From RECIST

Type:

  Extensible

Version:

  20030108

UID:

  1.2.840.10008.6.1.465

Table CID 6145. Baseline Category

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112074 | Target Lesion at Baseline
DCM | 112075 | Non-Target Lesion at Baseline
DCM | 112076 | Non-Lesion at Baseline

## CID 6146 Time Point Types

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.1002

Table CID 6146. Time Point Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UMLS | C1442488 | Baseline
UMLS | C3539075 | Pretreatment
DCM | 126074 | Posttreatment
DCM | 126075 | Eligibility
UMLS | C1699701 | Unscheduled
UMLS | C1708760 | Nadir

### Note

1. (C1442488, UMLS, "Baseline") is (C25213, NCIt, "Baseline"). The undefined (121079, DCM, "Baseline") that is used in CID 7003 Diagnostic Imaging Report Purposes of Reference is not used in this context.

2. (C3539075, UMLS, "Pretreatment") is (C103341, NCIt, "Pretreatment").

3. (C1708760, UMLS, "Nadir") is (C43517, NCIt, "Nadir"), and is a synonymn for "lowest", though "nadir" is more commnly used in the context of therapeutic response criteria.

## CID 6147 Response Criteria

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.1004

Table CID 6147. Response Criteria

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112029 | WHO
DCM | 126080 | RECIST 1.0
DCM | 126081 | RECIST 1.1
NCIt | C114879 | RANO

## CID 6151 Background Echotexture

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.466

Table CID 6151. Background Echotexture

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111351 | Homogeneous fat echotexture
DCM | 111352 | Homogeneous fibroglandular echotexture
DCM | 111353 | Heterogeneous echotexture

## CID 6152 Orientation

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.467

Table CID 6152. Orientation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111355 | Parallel
DCM | 111356 | Not parallel

## CID 6153 Lesion Boundary

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.468

Table CID 6153. Lesion Boundary

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111358 | Abrupt interface
DCM | 111359 | Echogenic halo

## CID 6154 Echo Pattern

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.469

Table CID 6154. Echo Pattern

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111361 | Anechoic
DCM | 111362 | Hyperechoic
DCM | 111363 | Complex
DCM | 111364 | Hypoechoic
DCM | 111365 | Isoechoic

## CID 6155 Posterior Acoustic Features

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20060622

UID:

  1.2.840.10008.6.1.470

Table CID 6155. Posterior Acoustic Features

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111367 | No posterior acoustic features
DCM | 111368 | Posterior enhancement
DCM | 111369 | Posterior shadowing
DCM | 111370 | Combined posterior enhancement and shadowing

## CID 6157 Vascularity

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.471

Table CID 6157. Vascularity

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111373 | Vascularity not present
DCM | 111374 | Vascularity not assessed
DCM | 111375 | Vascularity present in lesion
DCM | 111376 | Vascularity present immediately adjacent to lesion
DCM | 111377 | Diffusely increased vascularity in surrounding tissue

## CID 6158 Correlation to Other Findings

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.472

Table CID 6158. Correlation to Other Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111381 | Correlates to physical exam findings
DCM | 111382 | Correlates to mammography findings
DCM | 111383 | Correlates to MRI findings
DCM | 111384 | Correlates to ultrasound findings
DCM | 111385 | Correlates to other imaging findings
DCM | 111386 | No correlation to other imaging findings
DCM | 111387 | No correlation to clinical findings

## CID 6159 Malignancy Type

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.473

Table CID 6159. Malignancy Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111389 | Invasive breast carcinoma |  |
SRT | M-85002 | Intraductal carcinoma, non-infiltrating | 86616005 | C0007124
DCM | 111390 | Other malignancy type |  |

## CID 6160 Breast Primary Tumor Assessment From AJCC

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.474

Table CID 6160. Breast Primary Tumor Assessment From AJCC

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-003B8 | TX: Primary tumor cannot be assessed (breast) | 373173008 | C1276754
SRT | R-003B9 | T0: No evidence of primary tumor (breast) | 373174002 | C1276755
SRT | R-003BB | Tis: Carcinoma in situ (breast) | 373175001 | C1276756
SRT | R-003BC | Tis: Ductal carcinoma in situ (breast) | 373176000 | C1276757
SRT | R-003BD | Tis: Lobular carcinoma in situ (breast) | 373177009 | C1276758
SRT | R-003BE | Tis: Paget's disease of the nipple with no tumor | 373178004 | C1269975
SRT | R-003BA | T1: Tumor 2 cm or less in greatest dimension (breast) | 373172003 | C1272784
SRT | R-003BF | T1mic: Microinvasion 0.1 cm or less in greatest dimension... | 373179007 | C1269976
SRT | R-003C0 | T1a: Tumor more than 0.1 cm but not more than 0.5 cm... | 373180005 | C1269977
SRT | R-003C1 | T1b: Tumor more than 0.5 cm but not more than 1 cm... | 373204007 | C1269981
SRT | R-003C2 | T1c: Tumor more than 1 cm but not more than 2 cm... | 373183007 | C1272785
SRT | R-003C3 | T2: Tumor more than 2 cm but not more than 5 cm... | 373182002 | C1269978
SRT | R-003C4 | T3: Tumor more than 5 cm in greatest dimension (breast) | 373184001 | C1269979
SRT | R-003C5 | T4: Tumor of any size with direct extension to chest wall... | 373185000 | C1276759
SRT | R-003C6 | T4a: Tumor of any size with extension to chest wall, not incl... | 373186004 | C1276760
SRT | R-003C7 | T4b: Tumor of any size with edema (including peau d'orange) ... | 373187008 | C1276761
SRT | R-003C8 | T4c: Tumor of any size with direct extension to chest wall... | 373189006 | C1268960
SRT | R-003C9 | T4: Inflammatory carcinoma (breast) | 373190002 | C1276762

## CID 6161 Clinical Regional Lymph Node Assessment for Breast

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.475

Table CID 6161. Clinical Regional Lymph Node Assessment for Breast

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-003CA | NX: Regional lymph nodes cannot be assessed... | 373150000 | C1276765
SRT | R-003CB | N0: No regional lymph node metastasis histologically... | 373151001 | C1272783
SRT | R-003D0 | N1: Metastasis in 1 to 3 axillary lymph nodes... | 373156006 | C1276766
SRT | R-003D6 | N2: Metastasis in 4 to 9 axillary lymph nodes... | 373162001 | C1276749
SRT | R-003D7 | N2a: Metastasis in 4 to 9 axillary lymph nodes (...2.0 mm)... | 373163006 | C1276750
SRT | R-003D8 | N2b: Metastasis in clinically apparent internal... nodes... | 373164000 | C1276751
SRT | G-F749 | N3: Metastasis to ipsilateral internal mammary lymph node(s) | 369991007 | C1276711
SRT | R-003D9 | N3a: Metastasis in 10 or more axillary lymph nodes... | 373165004 | C1276752
SRT | R-003DA | N3b: Metastasis in clinically apparent ipsilateral internal... | 373167007 | C1274009
SRT | R-003DB | N3c: Metastasis in ipsilateral supraclavicular lymph nodes... | 373166003 | C1276753

## CID 6162 Assessment of Metastasis for Breast

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.476

Table CID 6162. Assessment of Metastasis for Breast

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-003DC | MX: Distant metastasis cannot be assessed (breast) | 373170006 | C1268958
SRT | R-003DD | M0: No distant metastasis (breast) | 373169005 | C1268957
SRT | R-003DE | M1: Distant metastasis (breast) | 373171005 | C1268959

## CID 6163 Menstrual Cycle Phase

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.477

Table CID 6163. Menstrual Cycle Phase

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111392 | 1st week |  |
DCM | 111393 | 2nd week |  |
DCM | 111394 | 3rd week |  |
SRT | F-840B3 | Menstruation present | 289894009 | C0567306

## CID 6164 Time Intervals

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.478

Table CID 6164. Time Intervals

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111396 | < 3 months ago
DCM | 111397 | 4 months to 1 year ago
DCM | 111398 | > 1 year ago
DCM | 111399 | Not sure

## CID 6165 Breast Linear Measurements

Type:

  Extensible

Version:

  20050822

UID:

  1.2.840.10008.6.1.479

Table CID 6165. Breast Linear Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7470 “Linear Measurements”
DCM | 121242 | Distance from nipple
DCM | 121243 | Distance from skin
DCM | 121244 | Distance from chest wall

## CID 6166 CAD Geometry Secondary Graphical Representation

Type:

  Extensible

Version:

  20060822

UID:

  1.2.840.10008.6.1.480

Table CID 6166. CAD Geometry Secondary Graphical Representation

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 113661 | Outline of lobulations |  |
DCM | 113662 | Inner limits of fuzzy margin |  |
DCM | 113663 | Outer limits of fuzzy margin |  |
DCM | 113664 | Outline of spiculations |  |
DCM | 113665 | Linear spiculation |  |
DCM | 113666 | Pixelated spiculations |  |
SRT | G-A185 | Long axis | 103339001 | C0522487
SRT | G-A186 | Short axis | 103340004 | C0522488
DCM | 113669 | Orthogonal location arc |  |
DCM | 113670 | Orthogonal location arc inner margin |  |
DCM | 113671 | Orthogonal location arc outer margin |  |

## CID 6200 Colon Overall Assessment

Type:

  Extensible

Version:

  20090402

UID:

  1.2.840.10008.6.1.787

Table CID 6200. Colon Overall Assessment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112240 | C0 - Inadequate Study/Awaiting Prior Comparisons
DCM | 112241 | C1 - Normal Colon or Benign Lesion
DCM | 112242 | C2 - Intermediate Polyp or Indeterminate Finding
DCM | 112243 | C3 - Polyp, Possibly Advanced Adenoma
DCM | 112244 | C4 - Colonic Mass, Likely Malignant

## CID 6201 Colon Finding or Feature

Type:

  Extensible

Version:

  20090402

UID:

  1.2.840.10008.6.1.788

Table CID 6201. Colon Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 111101 | Image quality |  |
DCM | 111099 | Selected region |  |
SRT | D5-41170 | Polyp of colon | 68496003 | C0009376
SRT | D5-F131F | Tumor of colon | 126838000 | C0009375
SRT | F-54005 | Rectal mass | 248523006 | C0240873
SRT | M-32700 | Diverticulum | 31113003 | C0012817
SRT | T-59345 | Colonic haustra | 6533001 | C0227361
SRT | T-59666 | Feces | 39477002 | C0015733
SRT | M-88500 | Lipoma | 46720004 | C0023798
SRT | T-50153 | Intraluminal fluid | 442170005 | C2711278
SRT | F-61D54 | Contrast media | 385420005 | C0009924
SRT | T-58650 | Ileocecal valve | 23153004 | C0020880
SRT | M-32704 | Inverted diverticulum | 441901008 | C2711356
SRT | M-18000 | Operative Site | 43526002 | C0332850
DCM | 111102 | Non-lesion |  |
DCM | 112238 | Anatomic non-colon |  |

## CID 6202 Colon Finding or Feature Modifier

Type:

  Extensible

Version:

  20090402

UID:

  1.2.840.10008.6.1.789

Table CID 6202. Colon Finding or Feature Modifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6203 “Colon Non-lesion Object Type”
Include CID 6204 “Anatomic Non-colon Findings”

## CID 6203 Colon Non-lesion Object Type

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.790

Table CID 6203. Colon Non-lesion Object Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-32110 | Bullet | 86122002 | C0336699
SRT | A-13600 | Staple | 56353002 | C0524724
SRT | A-13500 | Suture | 27065002 | C0038969
SRT | M-78060 | Scar tissue | 12402003 | C2004491
SRT | A-26800 | Catheter | 19923001 | C0085590
DCM | 112173 | Chest tube |  |
SRT | A-14611 | Vena cava filter | 257409000 | C0080306
SRT | A-04000 | Prosthesis | 53350007 | C0175649
SRT | A-26434 | Jejunostomy tube | 126065006 | C0879216
DCM | 112175 | Kidney stent |  |
SRT | A-11C08 | Ureteral stent | 286558002 | C0183518
DCM | 112176 | Pancreatic stent |  |
SRT | A-61000 | Jewelry | 80919006 | C0336902
DCM | 112178 | Coin |  |
SRT | A-12024 | Pin | 77444004 | C0175718
SRT | A-30360 | Needle | 79068005 | C0027551
DCM | 112171 | Fiducial mark |  |
SRT | A-120DD | Colostomy set | 341036005 | C0180028
SRT | A-10DBC | Colostomy bag | 339648008 | C0180026
SRT | A-1009E | Ileostomy set | 342706005 | C0181271
SRT | A-10029 | Ileostomy bag | 417136005 | C1563151
SRT | A-10703 | Urostomy set | 344575009 | C0467978
SRT | A-105E3 | Urostomy bag | 344088002 | C0467658
SRT | A-26440 | Rectal tube | 67966000 | C0175752
SRT | A-26864 | Urethral catheter | 34759008 | C0179800

### Note

The use of (111176, DCM, "Unspecified") was previously included in this context group but was removed since it does not make sense to have Colon CAD composite feature modifiers (TID 4125 “Colon CAD Composite Feature”) and single image finding modifiers (TID 4127 “Colon CAD Single Image Finding”) of an unknown type.

## CID 6204 Anatomic Non-colon Findings

Type:

  Extensible

Version:

  20090402

UID:

  1.2.840.10008.6.1.791

Table CID 6204. Anatomic Non-colon Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-62000 | Liver | 10200004 | C0023884
SRT | T-C3000 | Spleen | 78961009 | C0037993
SRT | T-71000 | Kidney | 64033007 | C0022646
SRT | T-B3000 | Adrenal gland | 23451007 | C0001625
SRT | T-42000 | Aorta | 15825003 | C0003483
SRT | T-48710 | Inferior vena cava | 64131007 | C0042458
SRT | T-28000 | Lung | 39607008 | C0024109
SRT | T-D016E | Bone | 272673000 | C0262950
SRT | T-94000 | Testis | 40689003 | C0039597
SRT | T-83000 | Uterus | 35039007 | C0042149
SRT | T-87000 | Ovary | 15497006 | C0029939
SRT | T-83200 | Cervix | 71252005 | C0007874
SRT | T-92000 | Prostate | 41216001 | C0033572
SRT | T-93000 | Seminal Vesicle | 64739004 | C0036628
SRT | T-59600 | Rectum | 34402009 | C0034896
SRT | T-74000 | Bladder | 89837001 | C0005682
SRT | T-13001 | Muscle | 71616004 | C0026845
SRT | T-40000 | Blood Vessel | 59820001 | C0005847
SRT | T-59200 | Appendix | 66754008 | C0003617
SRT | T-D0874 | Appendiceal stump | 441850003 | C2711602

## CID 6205 Clockface Location for Colon

Type:

  Extensible

Version:

  20090402

UID:

  1.2.840.10008.6.1.792

Table CID 6205. Clockface Location for Colon

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-01781 | 1 o'clock position | 129772004 | C1268696
SRT | F-01782 | 2 o'clock position | 129773009 | C1268697
SRT | F-01783 | 3 o'clock position | 129774003 | C1268698
SRT | F-01784 | 4 o'clock position | 129775002 | C1268699
SRT | F-01785 | 5 o'clock position | 129776001 | C1268700
SRT | F-01786 | 6 o'clock position | 129777005 | C1268701
SRT | F-01787 | 7 o'clock position | 129778000 | C1268702
SRT | F-01788 | 8 o'clock position | 129779008 | C1268703
SRT | F-01789 | 9 o'clock position | 129780006 | C1268704
SRT | F-0178A | 10 o'clock position | 129781005 | C1268705
SRT | F-0178B | 11 o'clock position | 129782003 | C1268706
SRT | F-0178C | 12 o'clock position | 129783008 | C1268707

## CID 6206 Recumbent Patient Orientation for Colon

Type:

  Extensible

Version:

  20090402

UID:

  1.2.840.10008.6.1.793

Table CID 6206. Recumbent Patient Orientation for Colon

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-10310 | Prone | 1240000 | C0033422
SRT | F-10340 | Supine | 40199007 | C0038846
SRT | F-10317 | right lateral decubitus | 102535000 | C0559228
SRT | F-10319 | left lateral decubitus | 102536004 | C0559227

## CID 6207 Colon Quantitative Temporal Difference Type

Type:

  Extensible

Version:

  20090717

UID:

  1.2.840.10008.6.1.794

Table CID 6207. Colon Quantitative Temporal Difference Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-05173 | Difference in size | 442714003 | C2711955
SRT | F-05179 | Difference in location | 442726008 | C2711109
SRT | F-0516E | Difference in attenuation | 442707000 | C2711926

## CID 6208 Colon Types of Quality Control Standard

Type:

  Extensible

Version:

  20090402

UID:

  1.2.840.10008.6.1.795

Table CID 6208. Colon Types of Quality Control Standard

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112036 | ACR Position Statement
DCM | 111240 | Institutionally defined quality control standard
DCM | 112248 | ACR Guideline, Performance of Adult CT Colonography
DCM | 112249 | ACR Standard, CT medical physics performance monitoring

## CID 6209 Colon Morphology Descriptor

Type:

  Extensible

Version:

  20090402

UID:

  1.2.840.10008.6.1.796

Table CID 6209. Colon Morphology Descriptor

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A530 | Sessile | 5712003 | C0205348
SRT | G-A477 | Pedunculated | 25126001 | C0205320
SRT | G-A485 | Flat | 6041008 | C0205324
SRT | R-404F0 | Circumferential | 255593009 | C0205113
SRT | M-38000 | Ulcer | 56208002 | C0041582

## CID 6210 Location in Intestinal Tract

Type:

  Extensible

Version:

  20090402

UID:

  1.2.840.10008.6.1.797

Table CID 6210. Location in Intestinal Tract

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-59600 | Rectum | 34402009 | C0034896
SRT | T-59470 | Sigmoid colon | 60184004 | C0227391
SRT | T-59460 | Descending colon | 32622004 | C0227389
SRT | T-59440 | Transverse colon | 485005 | C0227386
SRT | T-59420 | Ascending colon | 9040008 | C0227375
SRT | T-59100 | Cecum | 32713005 | C0007531
SRT | T-59442 | Splenic flexure of colon | 72592005 | C0227387
SRT | T-59438 | Hepatic flexure of colon | 48338005 | C0227385

## CID 6211 Colon CAD Material Description

Type:

  Extensible

Version:

  20090402

UID:

  1.2.840.10008.6.1.798

Table CID 6211. Colon CAD Material Description

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 112144 | Soft tissue |  |
SRT | T-D008A | Fat | 256674009 | C0015677
SRT | A-80230 | Air | 15158005 | C0001861
SRT | T-11034 | Bone matrix | 45001002 | C0005962

## CID 6212 Calculated Value for Colon Findings

Type:

  Extensible

Version:

  20090402

UID:

  1.2.840.10008.6.1.799

Table CID 6212. Calculated Value for Colon Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-0045B | Polyp stalk length | 395511002 | C1273121
SRT | R-00286 | Polyp size, largest dimension | 373197004 | C1272618
DCM | 112232 | Polyp stalk width |  |
DCM | 112233 | Distance from anus |  |

## CID 6300 Prostate Sector Anatomy

### Note

In future extensions, Prostate Sector Anatomy terms that are not derived from PI-RADS v2 should be added to this context group.

Type:

  Extensible

Version:

  20161106

UID:

  1.2.840.10008.6.1.1138

Table CID 6300. Prostate Sector Anatomy

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6301 “Prostate Sector Anatomy from PI-RADS v2”
Include CID 6302 “Prostate Sector Anatomy from European Concensus 16 Sector (Minimal) Model”
Include CID 6303 “Prostate Sector Anatomy from European Concensus 27 Sector (Optimal) Model”

## CID 6301 Prostate Sector Anatomy from PI-RADS v2

### Note

From [PI-RADS v2].

Type:

  Extensible

Version:

  20161106

UID:

  1.2.840.10008.6.1.1139

Table CID 6301. Prostate Sector Anatomy from PI-RADS v2

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | FMA ID | PI-RADS v2 Abbreviation | NCI Thesaurus
--- | --- | --- | --- | --- | --- | --- | ---
SRT | R-FFFFC | Central zone of left half prostate | 716901006 | C4273550 | 302475 | Base L CZ | C128587
SRT | R-FFFD0 | Central zone of right half prostate | 716900007 | C4274157 | 302473 | Base R CZ | C128593
SRT | R0-00025 | Left anterior apical peripheral zone of prostate | 716937001 | C4274170 | 328760 | Apex L PZa | C128575
SRT | R-FFFE5 | Left anterior apical transition zone of prostate | 716931000 | C4274174 | 328795 | Apex L TZa | C128578
SRT | R0-00001 | Left anterior basal peripheral zone of prostate | 716905002 | C4273857 | 328753 | Base L PZa | C128588
SRT | R-FFFD6 | Left anterior basal transition zone of prostate | 716897000 | C4274207 | 328785 | Base L TZa | C128589
SRT | R-FFFB0 | Left anterior middle peripheral zone of prostate | 716920008 | C4274185 | 328768 | Mid L PZa | C128600
SRT | R0-00013 | Left anterior middle transition zone of prostate | 716914007 | C4274190 | 328784 | Mid L TZa | C128603
SRT | R-FFFDF | Left apical anterior fibromuscular stroma of prostate | 716927006 | C4274178 | 328772 | Apex L AS | C128574
SRT | R-FFFDC | Left basal anterior fibromuscular stroma of prostate | 716893001 | C4274482 | 328758 | Base L AS | C128586
SRT | R0-00027 | Left middle anterior fibromuscular stroma of prostate | 716910003 | C4274479 | 328781 | Mid L AS | C128599
SRT | R0-00014 | Left posterior apical transition zone of prostate | 716933002 | C4274173 | 328775 | Apex L TZp | C128579
SRT | R-FFFB7 | Left posterior basal transition zone of prostate | 716899002 | C4274204 | 328789 | Base L TZp | C128590
SRT | R-FFFAB | Left posterior middle transition zone of prostate | 716916009 | C4274189 | 328786 | Mid L TZp | C128604
SRT | R-FFFDD | Left posterolateral apical peripheral zone of prostate | 716939003 | C4274168 | 328752 | Apex L PZpl | C128576
SRT | R-FFFC2 | Left posterolateral basal peripheral zone of prostate | 716907005 | C4274197 | 328759 | Base L PZpl | C128591
SRT | R-FFFE9 | Left posterolateral middle peripheral zone of prostate | 716922000 | C4274180 | 328791 | Mid L PZpl | C128601
SRT | R0-0000B | Left posteromedial apical peripheral zone of prostate | 716941002 | C4274166 | 328792 | Apex L PZpm | C128577
SRT | R-FFFB5 | Left posteromedial middle peripheral zone of prostate | 716924004 | C4274183 | 328777 | Mid L PZpm | C128602
SRT | T-93020 | Left seminal vesicle | 42320003 | C0227980 | 19388 | L SV | C128598
SRT | R-FFFD9 | Male external urethral sphincter | 717027004 | C0815353 | 19733 | US | C128612
SRT | R0-00003 | Right anterior apical peripheral zone of prostate | 716936005 | C4274125 | 328779 | Apex R PZa | C128581
SRT | R0-00006 | Right anterior apical transition zone of prostate | 716930004 | C4274131 | 328761 | Apex R TZa | C128584
SRT | R-FFFE2 | Right anterior basal peripheral zone of prostate | 716904003 | C4274200 | 328798 | Base R PZa | C128594
SRT | R0-00000 | Right anterior basal transition zone of prostate | 716896009 | C4273547 | 328793 | Base R TZa | C128596
SRT | R-FFFCD | Right anterior middle peripheral zone of prostate | 716919002 | C4274141 | 328796 | Mid R PZa | C128606
SRT | R-FFFF4 | Right anterior middle transition zone of prostate | 716913001 | C4274147 | 328800 | Mid R TZa | C128609
SRT | R-FFFFD | Right apical anterior fibromuscular stroma of prostate | 716926002 | C4273870 | 328801 | Apex R AS | C128580
SRT | R-FFFF2 | Right basal anterior fibromuscular stroma of prostate | 716892006 | C4273849 | 328778 | Base R AS | C128592
SRT | R0-00004 | Right middle anterior fibromuscular stroma of prostate | 716909008 | C4273544 | 328783 | Mid R AS | C128605
SRT | R-FFFF3 | Right posterior apical transition zone of prostate | 716932007 | C4274099 | 328763 | Apex R TZp | C128585
SRT | R-FFFB1 | Right posterior basal transition zone of prostate | 716898005 | C4274205 | 328799 | Base R TZp | C128597
SRT | R-FFFC9 | Right posterior middle transition zone of prostate | 716915008 | C4273542 | 328787 | Mid R TZp | C128610
SRT | R-FFFC0 | Right posterolateral apical peripheral zone of prostate | 716938006 | C4273861 | 328782 | Apex R PZpl | C128582
SRT | R0-0001E | Right posterolateral basal peripheral zone of prostate | 716906001 | C4274198 | 328797 | Base R PZpl | C128595
SRT | R0-0000C | Right posterolateral middle peripheral zone of prostate | 716921007 | C4274184 | 328771 | Mid R PZpl | C128607
SRT | R-FFFEA | Right posteromedial apical peripheral zone of prostate | 716940001 | C4274167 | 328764 | Apex R PZpm | C128583
SRT | R-FFFD4 | Right posteromedial middle peripheral zone of prostate | 716923005 | C4274181 | 328766 | Mid R PZpm | C128608
SRT | T-93010 | Right seminal vesicle | 74308000 | C0227979 | 19387 | R SV | C128611

## CID 6302 Prostate Sector Anatomy from European Concensus 16 Sector (Minimal) Model

### Note

From [Prostate Eu Concensus].

Type:

  Extensible

Version:

  20161106

UID:

  1.2.840.10008.6.1.1140

Table CID 6302. Prostate Sector Anatomy from European Concensus 16 Sector (Minimal) Model

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | FMA ID | 16 Sector Code
--- | --- | --- | --- | --- | --- | ---
SRT | R-FFFFC | Central zone of left half prostate | 716901006 | C4273550 | 302475 | 6p
SRT | R-FFFD0 | Central zone of right half prostate | 716900007 | C4274157 | 302473 | 1p
SRT | R-FFFD3 | Left apical peripheral zone of prostate | 716935009 | C4274171 | 328790 | 10p
SRT | R-FFFA9 | Left apical transition zone of prostate | 716929009 | C4274176 | 328769 | 6a
SRT | R0-00020 | Left basal part transition zone of prostate | 716895008 | C4274160 | 328755 | 4a
SRT | R-FFFC4 | Left basal peripheral zone of prostate | 716903009 | C4274120 | 328765 | 7p
SRT | R-FFFE6 | Left lateral middle peripheral zone of prostate | 716918005 | C4274142 | 328767 | 9p
SRT | R-FFFD5 | Left middle transition zone of prostate | 716912006 | C4274192 | 328762 | 5a
SRT | R-FFFB5 | Left posteromedial middle peripheral zone of prostate | 716924004 | C4274183 | 328777 | 8p
SRT | T-93020 | Left seminal vesicle | 42320003 | C0227980 | 19388 | L SV
SRT | R-FFFD9 | Male external urethral sphincter | 717027004 | C0815353 | 19733 | US
SRT | R-FFFB3 | Right apical peripheral zone of prostate | 716934008 | C4274128 | 328794 | 5p
SRT | R-FFFC1 | Right apical transition zone of prostate | 716928001 | C4273855 | 328773 | 3a
SRT | R0-00018 | Right basal peripheral zone of prostate | 716902004 | C4274155 | 328802 | 2p
SRT | R-FFFBE | Right basal transition zone of prostate | 716894007 | C4274164 | 328780 | 1a
SRT | R0-0000F | Right lateral middle peripheral zone of prostate | 716917000 | C4274143 | 328803 | 4p
SRT | R-FFFB6 | Right middle transition zone of prostate | 716911004 | C4273545 | 328757 | 2a
SRT | R-FFFD4 | Right posteromedial middle peripheral zone of prostate | 716923005 | C4274181 | 328766 | 3p
SRT | T-93010 | Right seminal vesicle | 74308000 | C0227979 | 19387 | R SV

## CID 6303 Prostate Sector Anatomy from European Concensus 27 Sector (Optimal) Model

### Note

From [Prostate Eu Concensus].

Type:

  Extensible

Version:

  20161106

UID:

  1.2.840.10008.6.1.1141

Table CID 6303. Prostate Sector Anatomy from European Concensus 27 Sector (Optimal) Model

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | FMA ID | 27 Sector Code
--- | --- | --- | --- | --- | --- | ---
SRT | R0-0001B | Apical anterior fibromuscular stroma of prostate | 716925003 | C4274179 | 302546 | 15as
SRT | R0-00017 | Basal anterior fibromuscular stroma of prostate | 716891004 | C4273850 | 302539 | 13as
SRT | R-FFFFC | Central zone of left half prostate | 716901006 | C4273550 | 302475 | 7p
SRT | R-FFFD0 | Central zone of right half prostate | 716900007 | C4274157 | 302473 | 1p
SRT | R0-00025 | Left anterior apical peripheral zone of prostate | 716937001 | C4274170 | 328760 | 12a
SRT | R0-00001 | Left anterior basal peripheral zone of prostate | 716905002 | C4273857 | 328753 | 8a
SRT | R-FFFB0 | Left anterior middle peripheral zone of prostate | 716920008 | C4274185 | 328768 | 10a
SRT | R-FFFA9 | Left apical transition zone of prostate | 716929009 | C4274176 | 328769 | 11a
SRT | R0-00020 | Left basal part transition zone of prostate | 716895008 | C4274160 | 328755 | 7a
SRT | R-FFFD5 | Left middle transition zone of prostate | 716912006 | C4274192 | 328762 | 9a
SRT | R-FFFDD | Left posterolateral apical peripheral zone of prostate | 716939003 | C4274168 | 328752 | 12p
SRT | R-FFFC2 | Left posterolateral basal peripheral zone of prostate | 716907005 | C4274197 | 328759 | 8p
SRT | R-FFFE9 | Left posterolateral middle peripheral zone of prostate | 716922000 | C4274180 | 328791 | 10p
SRT | R0-0000B | Left posteromedial apical peripheral zone of prostate | 716941002 | C4274166 | 328792 | 11p
SRT | R-FFFB5 | Left posteromedial middle peripheral zone of prostate | 716924004 | C4274183 | 328777 | 9p
SRT | T-93020 | Left seminal vesicle | 42320003 | C0227980 | 19388 | L SV
SRT | R-FFFD9 | Male external urethral sphincter | 717027004 | C0815353 | 19733 | US
SRT | R-FFFE0 | Middle anterior fibromuscular stroma of prostate | 716908000 | C4274194 | 302542 | 14as
SRT | R0-00003 | Right anterior apical peripheral zone of prostate | 716936005 | C4274125 | 328779 | 6a
SRT | R-FFFE2 | Right anterior basal peripheral zone of prostate | 716904003 | C4274200 | 328798 | 2a
SRT | R-FFFCD | Right anterior middle peripheral zone of prostate | 716919002 | C4274141 | 328796 | 4a
SRT | R-FFFC1 | Right apical transition zone of prostate | 716928001 | C4273855 | 328773 | 5a
SRT | R-FFFBE | Right basal transition zone of prostate | 716894007 | C4274164 | 328780 | 1a
SRT | R-FFFB6 | Right middle transition zone of prostate | 716911004 | C4273545 | 328757 | 3a
SRT | R-FFFC0 | Right posterolateral apical peripheral zone of prostate | 716938006 | C4273861 | 328782 | 6p
SRT | R0-0001E | Right posterolateral basal peripheral zone of prostate | 716906001 | C4274198 | 328797 | 2p
SRT | R0-0000C | Right posterolateral middle peripheral zone of prostate | 716921007 | C4274184 | 328771 | 4p
SRT | R-FFFEA | Right posteromedial apical peripheral zone of prostate | 716940001 | C4274167 | 328764 | 5p
SRT | R-FFFD4 | Right posteromedial middle peripheral zone of prostate | 716923005 | C4274181 | 328766 | 3p
SRT | T-93010 | Right seminal vesicle | 74308000 | C0227979 | 19387 | R SV

## CID 6401 Non-lesion Object Type - Physical Objects

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1185

Table CID 6401. Non-lesion Object Type - Physical Objects

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-32475 | BB shot (Lead Pellet) | 102378009 | C0522706
SRT | A-32110 | Bullet | 86122002 | C0336699
SRT | A-11101 | Cardiac Pacemaker | 118378005 | C1289799
SRT | A-26800 | Catheter | 19923001 | C0085590
SRT | A-12062 | Clip | 77720000 | C0175722
SRT | A-0110F | Collimator | 228761004 | C0454169
SRT | A-10042 | Compression paddle | 129460009 | C1268544
SRT | A-16016 | ID Plate | 129467007 | C1268548
SRT | A-04010 | Implant | 40388003 | C0021102
SRT | A-1016B | J Wire | 129463006 | C1268545
SRT | A-00D7B | Opaque Marker | 262301009 | C0445402
DCM | 111175 | Other Marker |  |
SRT | A-13600 | Staple | 56353002 | C0524724
SRT | A-13500 | Suture | 27065002 | C0038969

### Note

This Context Group formerly included SNOMED code J-83250, which has been replaced with A-00D7B. See Annex J.

## CID 6402 Non-lesion Object Type - Substances

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1186

Table CID 6402. Non-lesion Object Type - Substances

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | C-B0300 | Contrast agent | 7140000 | C0009924

## CID 6403 Non-lesion Object Type - Tissues

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1187

Table CID 6403. Non-lesion Object Type - Tissues

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-78060 | Scar tissue | 12402003 | C2004491

## CID 6404 Chest Non-lesion Object Type - Physical Objects

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1188

Table CID 6404. Chest Non-lesion Object Type - Physical Objects

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-32110 | Bullet | 86122002 | C0336699
SRT | A-11100 | Cardiac Pacemaker | 14106009 | C0030163
SRT | A-040CB | Cardiac pacemaker lead | 360129009 | C1283151
SRT | A-26800 | Catheter | 19923001 | C0085590
DCM | 112174 | Central line |  |
SRT | A-12210 | Cervical collar | 63562005 | C0175751
DCM | 112173 | Chest tube |  |
DCM | 112178 | Coin |  |
SRT | A-25350 | Endotracheal tube | 26412008 | C0336630
SRT | A-26430 | Feeding tube | 25062003 | C2945625
DCM | 112171 | Fiducial mark |  |
SRT | A-04110 | Heart valve prosthesis | 25510005 | C0018825
SRT | A-26434 | Jejunostomy tube | 126065006 | C0879216
SRT | A-61000 | Jewelry | 80919006 | C0336902
DCM | 112175 | Kidney stent |  |
SRT | A-30360 | Needle | 79068005 | C0027551
DCM | 112177 | Nipple ring |  |
DCM | 112176 | Pancreatic stent |  |
SRT | A-12024 | Pin | 77444004 | C0175718
DCM | 112172 | Portacath |  |
SRT | A-04000 | Prosthesis | 53350007 | C0175649
SRT | A-13600 | Staple | 56353002 | C0524724
SRT | A-13500 | Suture | 27065002 | C0038969
SRT | P1-26100 | Tracheotomy | 48387007 | C0040590
SRT | A-11C08 | Ureteric stent | 286558002 | C0183518
SRT | A-14611 | Vena cava filter | 257409000 | C0080306

## CID 6405 Chest Non-lesion Object Type - Tissues

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1189

Table CID 6405. Chest Non-lesion Object Type - Tissues

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-78060 | Scar tissue | 12402003 | C2004491

## CID 7000 Diagnostic Imaging Report Document Titles

Type:

  Extensible

Version:

  20140102

UID:

  1.2.840.10008.6.1.481

Table CID 7000. Diagnostic Imaging Report Document Titles

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID
--- | --- | --- | ---
LN | 18745-0 | Cardiac Catheterization Report | C0801759
LN | 11540-2 | CT Abdomen Report | C0551722
LN | 11538-6 | CT Chest Report | C0551723
LN | 11539-4 | CT Head Report | C0551724
LN | 18747-6 | CT Report | C0801761
LN | 18748-4 | Diagnostic Imaging Report | C0801762
LN | 11522-0 | Echocardiography Report | C0551715
LN | 18760-9 | Ultrasound Report | C0801774
LN | 11541-0 | MRI Head Report | C0551725
LN | 18755-9 | MRI Report | C0801769
LN | 18756-7 | MRI Spine Report | C0801770
LN | 18757-5 | Nuclear Medicine Report | C0801771
LN | 17787-3 | Nuclear Medicine Thyroid Scan Report | C0800894
LN | 11525-3 | Ultrasound Obstetric and Gyn Report | C0551717
LN | 18758-3 | PET Scan Report | C0801772
LN | 11528-7 | Radiology Report | C0551720
LN | 18750-0 | Cardiac Electrophysiology Report | C0801764
LN | 11524-0 | ECG Report |
LN | 18752-6 | Exercise Stress Test Report | C0801766
LN | 18754-2 | Holter Study Report | C0801768
LN | 43468-8 | X-Ray Report | C1714805
LN | 38269-7 | DEXA Skeletal System Study Report | C1526358
DCM | 111400 | Breast Imaging Report |
LN | 24606-6 | Mammography Screening Report | C0881841
LN | 49512-7 | Fluoroscopy Study Report | C1977263
LN | 47048-4 | Diagnostic Interventional Radiology Report | C1831148

## CID 7001 Diagnostic Imaging Report Headings

Type:

  Extensible

Version:

  20150324

UID:

  1.2.840.10008.6.1.482

Table CID 7001. Diagnostic Imaging Report Headings

Coding Scheme Designator | Code Value | Code Meaning | UMLS Concept Unique ID | Equivalent DCMR (DCM) Code
--- | --- | --- | --- | ---
LN | 11329-0 | History | C0551569 | 121060
LN | 55115-0 | Request | C2708293 | 121062
LN | 55111-9 | Current Procedure Descriptions | C2708286 | 121064
LN | 55114-3 | Prior Procedure Descriptions | C2708291 | 121066
LN | 18834-2 | Previous Findings | C0801832 | 121068
LN | 59776-5 | Findings | C2926606 | 121070
LN | 19005-8 | Impressions | C0801998 | 121072
LN | 18783-1 | Recommendations | C0801796 | 121074
LN | 55110-1 | Conclusions | C2708285 | 121076
LN | 55107-7 | Addendum | C2708272 | 121078
LN | 18785-6 | Indications for Procedure | C0801797 | 121109
LN | 55108-5 | Patient Presentation | C2708282 | 121110
LN | 55109-3 | Complications | C2708284 | 121113
LN | 55112-7 | Summary | C2708288 | 121111
LN | 55113-5 | Key Images | C2708289 | 121180
LN | 73569-6 | Radiation Exposure and Protection Information | C3654408 | 113923
LN | 55752-0 | Clinical Information | C2708732 |
LN | 29549-3 | Medications Administered | C0945765 |
LN | 73568-8 | Communication of Critical Results | C3654409 |

### Note

- In previous editions of the Standard, this Context Group included codes of DCMR, using Coding Scheme Designator DCM. The preferable encoding of these concepts is using the LOINC codes, however, the support of equivalent DCMR codes is recommended for backward compatibility.

- In a prior version of this Context Group, the code (18782-3, LN, "Study Observation") was specified for report heading "Findings". This has now been replaced by (59776-5, LN, "Procedure Findings").

## CID 7002 Diagnostic Imaging Report Elements

Type:

  Extensible

Version:

  20030327

UID:

  1.2.840.10008.6.1.483

Table CID 7002. Diagnostic Imaging Report Elements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
LN | 11329-0 | History |  | C0551569
LN | 55115-0 | Request |  | C2708293
DCM | 121065 | Procedure Description |  |
DCM | 121069 | Previous Finding |  |
DCM | 121071 | Finding |  |
DCM | 121073 | Impression |  |
DCM | 121075 | Recommendation |  |
DCM | 121077 | Conclusion |  |
SRT | DD-60002 | Complication of Procedure | 116224001 | C0742724
DCM | 121110 | Patient Presentation |  |
DCM | 121111 | Summary |  |

## CID 7003 Diagnostic Imaging Report Purposes of Reference

Type:

  Extensible

Version:

  20100604

UID:

  1.2.840.10008.6.1.484

Table CID 7003. Diagnostic Imaging Report Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121079 | Baseline
DCM | 121080 | Best illustration of finding
DCM | 121112 | Source of Measurement
DCM | 121200 | Illustration of ROI

## CID 7004 Waveform Purposes of Reference

Type:

  Extensible

Version:

  20090409

UID:

  1.2.840.10008.6.1.485

Table CID 7004. Waveform Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121301 | Simultaneous Doppler
DCM | 121302 | Simultaneous Hemodynamic
DCM | 121303 | Simultaneous ECG
DCM | 121304 | Simultaneous Voice Narrative
DCM | 121305 | Simultaneous Respiratory Waveform
DCM | 121306 | Simultaneous Arterial Pulse Waveform
DCM | 121307 | Simultaneous Phonocardiographic Waveform

## CID 7005 Contributing Equipment Purposes of Reference

Type:

  Extensible

Version:

  20130617

UID:

  1.2.840.10008.6.1.486

Table CID 7005. Contributing Equipment Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109101 | Acquisition Equipment
DCM | 109102 | Processing Equipment
DCM | 109103 | Modifying Equipment
DCM | 109104 | De-identifying Equipment
DCM | 109105 | Frame Extracting Equipment
DCM | 109106 | Enhanced Multi-frame Conversion Equipment
DCM | MEDIM | Portable Media Importer Equipment
DCM | FILMD | Film Digitizer
DCM | DOCD | Document Digitizer Equipment
DCM | VIDD | Video Tape Digitizer Equipment

## CID 7006 SR Document Purposes of Reference

Type:

  Extensible

Version:

  20061023

UID:

  1.2.840.10008.6.1.487

Table CID 7006. SR Document Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121331 | Equivalent CDA Document
DCM | 121332 | Complete Rendering for Presentation
DCM | 121333 | Partial Rendering for Presentation
DCM | 121334 | Extended Rendering for Presentation
DCM | 121335 | Source Document

## CID 7007 Signature Purpose

Context Group ID 7007 comprises the signature purposes codes of ASTM E 2084-00. The Coding Scheme Designator (0008,0102) shall be "ASTM-sigpurpose". The ASTM document defines the signature purpose codes as OIDs. For the purposes of this Coding Scheme only the leaf digit is used as the Code Value (0008,0100).

### Note

ASTM E 1762 provides the full definitions for the signature purpose OIDs defined by E 2084. The recommended Code Meanings (0008,0104) are the titles of the definitions for the leaves of the OIDs. For example, the OID 1.2.840.10065.1.12.1 corresponds to the leaf "id-purpose-author", whose meaning could be encoded as "Author Signature" and whose code value is 1.

## CID 7008 Media Import

This Context Group specifies items that may be conveyed in the Billing and Materials Management Module (see PS3.3).

Type:

  Extensible

Version:

  20061024

UID:

  1.2.840.10008.6.1.489

Table CID 7008. Media Import

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110020 | Sheet Film Digitized
DCM | 110021 | Cine Film Digitized
DCM | 110022 | Video Tape Digitized
DCM | 110023 | Paper Digitized
DCM | 110024 | CD Imported
DCM | 110025 | DVD Imported
DCM | 110026 | MOD Imported
DCM | 110027 | Studies Imported
DCM | 110028 | Instances Imported

## CID 7009 Purpose of Reference to Predecessor Report

Type:

  Extensible

Version:

  20090826

UID:

  1.2.840.10008.6.1.818

Table CID 7009. Purpose of Reference to Predecessor Report

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121360 | Replaced report
DCM | 121361 | Addended report
DCM | 121362 | Preliminary report
DCM | 121363 | Partial report
DCM | 122073 | Current procedure evidence

### Note

The concepts of replaced and addended correspond to REPLACEMENT and ADDENDUM in HL7 V2.6 Chapter 9, with the exception that an EDITED value is not supported due to incompatibility with HL7 CDA.

## CID 7010 Key Object Selection Document Title

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.490

Table CID 7010. Key Object Selection Document Title

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113000 | Of Interest
DCM | 113001 | Rejected for Quality Reasons
DCM | 113002 | For Referring Provider
DCM | 113003 | For Surgery
DCM | 113004 | For Teaching
DCM | 113005 | For Conference
DCM | 113006 | For Therapy
DCM | 113007 | For Patient
DCM | 113008 | For Peer Review
DCM | 113009 | For Research
DCM | 113010 | Quality Issue
DCM | 113013 | Best In Set
DCM | 113018 | For Printing
DCM | 113020 | For Report Attachment
DCM | 113021 | For Litigation
DCM | 113030 | Manifest
DCM | 113031 | Signed Manifest
DCM | 113032 | Complete Study Content
DCM | 113033 | Signed Complete Study Content
DCM | 113034 | Complete Acquisition Content
DCM | 113035 | Signed Complete Acquisition Content
DCM | 113036 | Group of Frames for Display
DCM | 113037 | Rejected for Patient Safety Reasons
DCM | 113038 | Incorrect Modality Worklist Entry
DCM | 113039 | Data Retention Policy Expired
DCM | 113022 | Collection of Presentation States
DCM | 128181 | Diagnostic Source Images
DCM | 128182 | Segmentation Result
DCM | 128183 | Registration Result
DCM | 128195 | For Diagnosis
DCM | 128218 | Diagnosis Input Used
DCM | 128196 | For Segmentation
DCM | 128219 | Contouring Input Used
DCM | 128199 | For Plan Comparison
DCM | 128220 | Plan Comparison Input Used
DCM | 128203 | For Tumor Board
DCM | 128221 | Tumor Board Input Used
DCM | 128208 | For Tumor Registry
DCM | 128222 | Tumor Registry Input Used
DCM | 128207 | For Clinical Trial Submission
DCM | 128223 | Clinical Trial Submission Input Used
Include CID 7023 “RT Process Output”
Include CID 7024 “RT Process Input”
Include CID 7025 “RT Process Input Used”
Include CID 7014 “Export Additional Information Document Titles”

## CID 7011 Rejected for Quality Reasons

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.491

Table CID 7011. Rejected for Quality Reasons

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111207 | Image artifact(s)
DCM | 111208 | Grid artifact(s)
DCM | 111209 | Positioning
DCM | 111210 | Motion blur
DCM | 111211 | Under exposed
DCM | 111212 | Over exposed
DCM | 111213 | No image
DCM | 111214 | Detector artifact(s)
DCM | 111215 | Artifact(s) other than grid or detector artifact
DCM | 111216 | Mechanical failure
DCM | 111217 | Electrical failure
DCM | 111218 | Software failure
DCM | 111219 | Inappropriate image processing
DCM | 111220 | Other failure
DCM | 111221 | Unknown failure
DCM | 113026 | Double exposure

## CID 7012 Best in Set

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.492

Table CID 7012. Best in Set

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113014 | Study
DCM | 113015 | Series
DCM | 113016 | Performed Procedure Step
DCM | 113017 | Stage-View

## CID 7013 Non-Image Source Instance Purposes of Reference

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1134

Table CID 7013. Non-Image Source Instance Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128224 | Source measurement
DCM | 128225 | Source report
DCM | 128226 | Source raw data
Include CID 7019 “Segmentation Non-Image Source Purposes of Reference”

### Note

This context group previously contained a code for "source image", which has been removed.

## CID 7014 Export Additional Information Document Titles

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1178

Table CID 7014. Export Additional Information Document Titles

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128710 | For Teaching File Export
DCM | 128711 | For Clinical Trial Export
DCM | 128713 | For Research Collection Export
DCM | 128714 | For Publication Export

## CID 7015 Export Delay Reasons

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1179

Table CID 7015. Export Delay Reasons

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128715 | Delay export until final report is available
DCM | 128716 | Delay export until clinical information is available
DCM | 128717 | Delay export until confirmation of diagnosis is available
DCM | 128718 | Delay export until histopathology is available
DCM | 128719 | Delay export until other laboratory results are available
DCM | 128720 | Delay export until patient is discharged
DCM | 128721 | Delay export until patient dies
DCM | 128722 | Delay export until expert review is available

## CID 7016 Level of Difficulty

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1180

Table CID 7016. Level of Difficulty

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128725 | Primary level
DCM | 128726 | Intermediate level
DCM | 128727 | Advanced level

## CID 7017 Category of Teaching Material - Imaging

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1181

Table CID 7017. Category of Teaching Material - Imaging

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128728 | Musculoskeletal imaging subject matter
DCM | 128729 | Pulmonary imaging subject matter
DCM | 128730 | Cardiovascular imaging subject matter
DCM | 128731 | Gastrointestinal imaging subject matter
DCM | 128732 | Genitourinary imaging subject matter
DCM | 128733 | Neuroimaging subject matter
DCM | 128734 | Vascular and interventional imaging subject matter
DCM | 128735 | Nuclear medicine imaging subject matter
DCM | 128736 | Ultrasound imaging subject matter
DCM | 128737 | Pediatric imaging subject matter
DCM | 128738 | Breast imaging subject matter

### Note

The contents of this context group correspond to the American Board of Radiology categories in use at the time the IHE TCE Profile was developed.

## CID 7018 Miscellaneous Document Titles

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1182

Table CID 7018. Miscellaneous Document Titles

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128712 | Additional Teaching File Information

## CID 7019 Segmentation Non-Image Source Purposes of Reference

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1183

Table CID 7019. Segmentation Non-Image Source Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 128227 | Source real world value map | Real world value map used as the source for derivation. E.g., the map applied to source images before processing them, such as for a threshold based segmentation operation. |

## CID 7020 Document Titles

Context Group ID 7020 comprises all document names (i.e., terms with Scale "DOC") within the HIPAA Attachments class of the LOINC coding scheme. The Coding Scheme Designator shall be LN.

### Note

1. A subset of this Context Group directly applicable to imaging reports is in CID 7000 “Diagnostic Imaging Report Document Titles”.

2. The LOINC coding scheme can be found at http://www.regenstrief.org/loinc.

## CID 7021 Measurement Report Document Titles

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.997

Table CID 7021. Measurement Report Document Titles

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126000 | Imaging Measurement Report
DCM | 126001 | Oncology Measurement Report
DCM | 126002 | Dynamic Contrast MR Measurement Report
DCM | 126003 | PET Measurement Report

## CID 7022 Radiotherapy Purposes of Reference

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.1115

Table CID 7022. Radiotherapy Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 121310 | RT treatment plan for the position being verified |  |

## CID 7023 RT Process Output

Type:

  Extensible

Version:

  20160908

UID:

  1.2.840.10008.6.1.1135

Table CID 7023. RT Process Output

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128184 | Pre-Planning Result
DCM | 128185 | RT Prescription Result
DCM | 128186 | Dose Calculation Image Series
DCM | 128187 | Coordinate Alignment Image Series
DCM | 128188 | RT Treatment Simulation Result
DCM | 128189 | RT Planning Result
DCM | 128190 | Dosimetric Result
DCM | 128191 | Patient Setup Verification Result
DCM | 128192 | RT Treatment Session Result
DCM | 128193 | RT Treatment Course Summary
DCM | 128194 | RT Treatment QA Result

### Note

The concepts in the CID are intended to be a declarative statement to represent the output of an operation, without implying that this operation was part of a particular workflow or that the output will be used in any future operation.

## CID 7024 RT Process Input

Type:

  Extensible

Version:

  20160908

UID:

  1.2.840.10008.6.1.1136

Table CID 7024. RT Process Input

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128180 | For RT Workflow
DCM | 128197 | For RT Prescription
DCM | 128198 | For RT Treatment Planning
DCM | 128200 | For RT Plan Summation
DCM | 128201 | For Physician Review
DCM | 128202 | For Physicist Review
DCM | 128204 | For Plan Quality Assurance
DCM | 128205 | For Machine Quality Assurance
DCM | 128206 | For Patient Setup Verification

### Note

The concepts in the CID are intended to be a declarative statement to represent the potential input of an operation, without implying that this operation is part of a particular workflow, that this input will be used at all in any subsequent operation, that only parts of the referenced instances will be used, or that instances other than those referenced will be used as input.

## CID 7025 RT Process Input Used

Type:

  Extensible

Version:

  20160908

UID:

  1.2.840.10008.6.1.1137

Table CID 7025. RT Process Input Used

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 128209 | RT Workflow Input Used
DCM | 128210 | RT Prescription Input Used
DCM | 128211 | RT Treatment Planning Input Used
DCM | 128212 | RT Plan Summation Input Used
DCM | 128213 | Physician Review Input Used
DCM | 128214 | Physicist Review Input Used
DCM | 128215 | Plan Quality Assurance Input Used
DCM | 128216 | Machine Quality Assurance Input Used
DCM | 128217 | Patient Setup Verification Input Used

### Note

The concepts in the CID are intended to be a declarative statement to represent input that has been used in an operation, without implying that this operation was part of a particular workflow or how this input was collected.

## CID 7026 Radiotherapeutic Dose Measurement Devices

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1177

Table CID 7026. Radiotherapeutic Dose Measurement Devices

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-FCCF2 | Medical x-ray film | 706247001 | C3873821
DCM | 128701 | 3D Gel |  |
DCM | 128702 | Diode Array |  |
DCM | 128703 | Ion Chamber Array |  |
SRT | R-FCE69 | Thermoluminescent radiation dosimeter | 464983000 | C3881975
DCM | 128704 | Diode |  |
DCM | 128705 | Liquid Ion Chamber |  |
SRT | R-FCC16 | MOSFET radiation therapy dosimetry system dosimeter | 701933006 | C3872923
DCM | 128706 | OSLD |  |
DCM | 128707 | Ion Chamber |  |
SRT | R-FD5EB | Digital imager | 468440006 | C3877969
DCM | 128708 | Diamond Detector |  |

## CID 7030 Institutional Departments, Units and Services

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.817

Table CID 7030. Institutional Departments, Units and Services

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 128170 | Abdominal Radiology |  |
SRT | R-300E3 | Accident and Emergency | 225728007 | C0562508
SRT | R-30246 | Allergy and Immunology | 309913004 | C0587451
SRT | R-3023A | Anesthesiology | 309901009 | C0002907
SRT | R-30247 | Audiology | 309914005 | C0587452
DCM | 128171 | Biomedical Engineering |  |
SRT | R-3027F | Breast Surgery | 309968000 | C0587504
SRT | R-3060E | Burns Intensive Care | 426439001 | C1959926
SRT | R-30240 | Cardiac Intensive Care | 309907008 | C0587446
SRT | R-30282 | Cardiac Surgery | 309971008 | C0587507
SRT | R-30248 | Cardiology | 309915006 | C0587453
SRT | R-30280 | Cardiothoracic Surgery | 309969008 | C0587505
DCM | 128172 | Cardiovascular Radiology |  |
SRT | R-30276 | Child and Adolescent Psychiatry | 309959002 | C0587495
SRT | R-421EB | Clinical Biochemistry | 310076001 | C0587609
SRT | R-3023B | Clinical Oncology | 309902002 | C0587443
SRT | R-3028E | Colorectal Surgery | 309983005 | C0587519
SRT | R-4221E | Computerized Tomography Service | 310128004 | C0587659
SRT | R-4225D | Cytology | 310200001 | C0587725
SRT | R-30283 | Dental Surgery | 309972001 | C0587508
SRT | R-30250 | Dermatology | 309923008 | C0587461
SRT | R-3061B | Diagnostic Imaging | 441662001 | C2711258
SRT | R-3028A | Endocrine Surgery | 309979005 | C0587515
SRT | R-30252 | Endocrinology | 309925001 | C0587463
SRT | R-421D4 | Endoscopy | 310030000 | C0587565
SRT | R-3028B | Gastrointestinal Surgery | 309980008 | C0587516
SRT | R-30254 | General Medicine | 309927009 | C0587465
SRT | R-3028F | General Surgery | 309984004 | C0587520
SRT | R-3025A | Geriatric Medicine | 309933000 | C0587471
SRT | R-30264 | Gynecology | 309943002 | C0587481
SRT | R-30290 | Hand Surgery | 309985003 | C0587521
SRT | R-3026F | Hematology | 309954007 | C0587491
SRT | R-4223B | Hepatobiliary Surgery | 310158005 | C0587687
SRT | R-3061D | Histopathology | 441950002 | C2711413
SRT | R-3025B | Infectious Disease | 309934006 | C0587472
DCM | 128173 | Information Technology |  |
SRT | R-3023D | Intensive Care | 309904001 | C0021708
SRT | R-FF0C4 | Interventional Radiology Service | 708174004 | C3872675
SRT | R-4221D | Magnetic Resonance Imaging Service | 310127009 | C0587658
SRT | R-3061E | Medical Intensive Care | 441994008 | C2711734
SRT | R-30270 | Medical Microbiology | 309956009 | C0587492
DCM | 128174 | Medical Physics |  |
DCM | 128175 | Musculoskeletal Radiology |  |
SRT | R-305CE | Neonatal Intensive Care | 405269005 | C0021709
SRT | R-3025D | Nephrology | 309936008 | C0587474
SRT | R-3025E | Neurology | 309937004 | C0587475
UMLS | C2183225 | Neuroradiology |  | C2183225
SRT | R-4223C | Neurosurgery | 310159002 | C0587688
SRT | R-3025F | Nuclear Medicine | 309938009 | C0587476
SRT | R-30265 | Obstetrics | 309944008 | C0028775
SRT | R-30263 | Obstetrics and Gynecology | 309942007 | C0587480
SRT | R-3025C | Ophthalmology | 309935007 | C0587473
SRT | R-42207 | Optometry | 310105000 | C0587638
SRT | R-30285 | Oral Surgery | 309974000 | C0587510
SRT | R-30294 | Orthopedic Surgery | 309989009 | C0587525
SRT | R-30289 | Otorhinolaryngology | 309978002 | C0587514
SRT | R-3026A | Pain Management | 309949003 | C0587486
SRT | R-30260 | Palliative Care | 309939001 | C0587477
SRT | R-3026B | Pathology | 309950003 | C0587487
SRT | R-30243 | Pediatric Intensive Care | 309910001 | C0021710
SRT | R-305EA | Pediatric Medicine | 420223003 | C1628316
SRT | R-30269 | Pediatric Oncology | 309948006 | C0587485
DCM | 128177 | Pediatric Radiology |  |
SRT | R-30296 | Pediatric Surgery | 309991001 | C0587527
SRT | R-302A2 | Physiotherapy | 310464005 | C0587975
SRT | R-30297 | Plastic Surgery | 309992008 | C0587528
SRT | S-8000A | Primary Care Department | 441480003 | C2711449
SRT | R-30275 | Psychiatry | 309958005 | C0587494
SRT | R-42219 | Psychology | 310123008 | C0587654
SRT | R-3024B | Pulmonology | 309918008 | C0587456
SRT | R-3027B | Radiology | 309964003 | C0587500
SRT | R-3023C | Radiotherapy | 309903007 | C0587444
SRT | R-30261 | Rehabilitation | 309940004 | C0587478
SRT | R-30262 | Rheumatology | 309941000 | C0587479
SRT | R-42203 | Speech and Language Therapy | 310101009 | C0587634
SRT | R-3027D | Stroke | 309966001 | C0587502
SRT | R-3027E | Surgery | 309967005 | C0587503
SRT | R-305EB | Surgical Intensive Care | 418433008 | C1690590
DCM | 128179 | Thoracic Radiology |  |
SRT | R-30281 | Thoracic Surgery | 309970009 | C0587506
SRT | R-30298 | Transplant Surgery | 309993003 | C0587529
SRT | R-30299 | Trauma Surgery | 309994009 | C0587530
SRT | R-30616 | Tropical Medicine | 441548002 | C2711407
SRT | R-42246 | Ultrasonography | 310169008 | C0587698
SRT | R-3029A | Urology | 309995005 | C0587531
SRT | R-3029B | Vascular Surgery | 309996006 | C0587532

### Note

In SNOMED, there is often a choice of unit, department or service concepts; in DICOM, the department concept is preferred and used in this context group.

## CID 7035 Actionable Finding Classification

Type:

  Extensible

Version:

  20150324

UID:

  1.2.840.10008.6.1.1026

Table CID 7035. Actionable Finding Classification

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
RADLEX | RID49480 | ACR Category 1 Actionable Finding
RADLEX | RID49481 | ACR Category 2 Actionable Finding
RADLEX | RID49482 | ACR Category 3 Actionable Finding

## CID 7036 Image Quality Assessment

Type:

  Extensible

Version:

  20150324

UID:

  1.2.840.10008.6.1.1027

Table CID 7036. Image Quality Assessment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
RADLEX | RID12 | Diagnostic quality
RADLEX | RID13 | Limited quality
RADLEX | RID14 | Non-diagnostic quality

## CID 7039 Pediatric Size Categories

Type:

  Extensible

Version:

  20170417

UID:

  1.2.840.10008.6.1.1173

Table CID 7039. Pediatric Size Categories

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 7040 “Broselow-Luten Pediatric Size Categories”

## CID 7040 Broselow-Luten Pediatric Size Categories

Type:

  Non-Extensible

Version:

  20100127

UID:

  1.2.840.10008.6.1.824

Table CID 7040. Broselow-Luten Pediatric Size Categories

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-051E3 | Broselow Luten Pink Zone (6-7 kg) | 444488009 | C2733122
SRT | F-051DF | Broselow Luten Red Zone (8-9 kg) | 444471002 | C2732530
SRT | F-051E4 | Broselow Luten Purple Zone (10-11 kg) | 444489001 | C2733258
SRT | F-051E8 | Broselow Luten Yellow Zone (12-14 kg) | 444505007 | C2732308
SRT | F-051E7 | Broselow Luten White Zone (15-18 kg) | 444504006 | C2732835
SRT | F-051E0 | Broselow Luten Blue Zone (19-23 kg) | 444474005 | C2733154
SRT | F-051E5 | Broselow Luten Orange Zone (24-29 kg) | 444496004 | C2732302
SRT | F-051E6 | Broselow Luten Green Zone (30-36 kg) | 444503000 | C2732991

## CID 7041 Calcium Scoring Patient Size Categories

Type:

  Extensible

Version:

  20170417

UID:

  1.2.840.10008.6.1.1174

Table CID 7041. Calcium Scoring Patient Size Categories

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 7042 “CMDCTECC Calcium Scoring Patient Size Categories”

## CID 7042 CMDCTECC Calcium Scoring Patient Size Categories

Patient sizes for calibrating calcium scoring, from the Consortium for Multi-Detector CT Evaluation of Coronary Calcium.

Type:

  Non-Extensible

Version:

  20100127

UID:

  1.2.840.10008.6.1.825

Table CID 7042. CMDCTECC Calcium Scoring Patient Size Categories

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113601 | Small: < 32.0 cm lateral thickness
DCM | 113602 | Medium: 32.0-38.0 cm lateral thickness
DCM | 113603 | Large: > 38.0 cm lateral thickness

## CID 7050 De-identification Method

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.925

Table CID 7050. De-identification Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113100 | Basic Application Confidentiality Profile
DCM | 113101 | Clean Pixel Data Option
DCM | 113102 | Clean Recognizable Visual Features Option
DCM | 113103 | Clean Graphics Option
DCM | 113104 | Clean Structured Content Option
DCM | 113105 | Clean Descriptors Option
DCM | 113106 | Retain Longitudinal Temporal Information Full Dates Option
DCM | 113107 | Retain Longitudinal Temporal Information Modified Dates Option
DCM | 113108 | Retain Patient Characteristics Option
DCM | 113109 | Retain Device Identity Option
DCM | 113110 | Retain UIDs Option
DCM | 113111 | Retain Safe Private Option
DCM | 113112 | Retain Institution Identity Option

## CID 7100 RCS Registration Method Type

Type:

  Extensible

Version:

  20040115

UID:

  1.2.840.10008.6.1.494

Table CID 7100. RCS Registration Method Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125021 | Frame of Reference Identity
DCM | 125023 | Acquisition Equipment Alignment
DCM | 125025 | Visual Alignment
DCM | 125022 | Fiducial Alignment
DCM | 125024 | Image Content-based Alignment

## CID 7101 Brain Atlas Fiducials

Type:

  Extensible

Version:

  20040115

UID:

  1.2.840.10008.6.1.495

Table CID 7101. Brain Atlas Fiducials

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 125030 | Inter-Hemispheric Plane |  |
SRT | T-A2980 | Anterior Commissure | 62872008 | C0152335
SRT | T-A4904 | Posterior Commissure | 279336005 | C0152327
DCM | 125031 | Right Hemisphere Most Anterior |  |
DCM | 125032 | Right Hemisphere Most Posterior |  |
DCM | 125033 | Right Hemisphere Most Superior |  |
DCM | 125034 | Right Hemisphere Most Inferior |  |
DCM | 125035 | Left Hemisphere Most Anterior |  |
DCM | 125036 | Left Hemisphere Most Posterior |  |
DCM | 125037 | Left Hemisphere Most Superior |  |
DCM | 125038 | Left Hemisphere Most Inferior |  |

## CID 7110 Fiducials Categories

Type:

  Extensible

Version:

  20160919

UID:

  1.2.840.10008.6.1.1132

Table CID 7110. Fiducials Categories

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 112171 | Fiducial mark |  |
SRT | R-FF2E7 | Anatomical point | 711101009 | C0504079
SRT | T-D002F | Body surface point | 183973000 | C0567332
SRT | R-FDCFF | Body reference point marker | 706484002 | C3872476

## CID 7111 Fiducials

Type:

  Extensible

Version:

  20160919

UID:

  1.2.840.10008.6.1.1133

Table CID 7111. Fiducials

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 3496 “IVUS Fiducial Points”
Include CID 3837 “Fiducial Feature”
Include CID 7101 “Brain Atlas Fiducials”

## CID 7140 Brain Structures for Volumetric Measurements

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.967

Table CID 7140. Brain Structures for Volumetric Measurements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-A3230 | Amygdala | 4958002 | C0002708
SRT | T-D0558 | Brain Stem | 119238007 | C1268144
SRT | T-A3200 | Caudate Nucleus | 11000004 | C0007461
SRT | T-A6040 | Cerebellar Cortex | 25991003 | C0007759
SRT | T-A6080 | Cerebellar White Matter | 33060004 | C0152381
SRT | T-A2020 | Cerebral Grey Matter | 40146001 | C0007776
SRT | T-A2030 | Cerebral White Matter | 68523003 | C0152295
SRT | T-D1400 | Cranial Cavity | 1101003 | C0230041
SRT | T-A1502 | Cranial Subarachnoid Space | 33930006 | C0228145
SRT | T-A1604 | Fifth Ventricle | 180933005 | C0228158
SRT | T-A1820 | Fourth Ventricle | 35918002 | C0149556
SRT | T-A3500 | Globus Pallidus | 14738005 | C0017651
SRT | T-A2570 | Hippocampus | 5366008 | C0019564
SRT | T-A1509 | Cerebellar Subarachnoid Space | 263972004 | C0446676
SRT | T-A1720 | Inferior Horn of Lateral Ventricle | 53118009 | C0152283
SRT | T-A1650 | Lateral Ventricle | 66720007 | C0152279
SRT | T-A0149 | Nucleus Accumbens | 427667007 | C0028633
SRT | T-A0190 | Intracranial structure | 128319008 | C1267697
SRT | T-A3400 | Putamen | 89278009 | C0034169
SRT | T-D0593 | Thalamus | 119406000 | C0458271
SRT | T-A1740 | Third ventricle | 49841001 | C0149555
DCM | 110700 | Ventral Diencephalon |  |
DCM | 110701 | White Matter T1 Hypointensity |  |
DCM | 110702 | White Matter T2 Hyperintensity |  |

### Note

1. (T-D1400, SRT, "Cranial Cavity") may be used to describe the volume of the entire intra-cranial space (intra-cranial volume or ICV) though the coded concept used is "structure of" rather than "entire" to be consistent with normal DICOM practice.

2. (T-A6040, SRT, "Cerebellar Cortex") is the gray matter of the cerebellum (as distinct from (T-A6080, SRT, "Cerebellar white matter")).

3. (T-A1502, SRT, "Cranial Subarachnoid Space") may be used to describe the volume of the exterior CSF (surrounding the brain, excluding the ventricles).

4. (T-A1509, SRT, "Cerebellar Subarachnoid Space") may be used to describe the volume of the inferior intracranial CSF space (infra-tentorial).

## CID 7150 Segmentation Property Categories

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.496

Table CID 7150. Segmentation Property Categories

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Segmentation Property Type Context Group
--- | --- | --- | --- | --- | ---
SRT | T-D0050 | Tissue | 85756007 | C0040300 | CID 7191 “Tissue Segmentation Property Types”
SRT | T-D000A | Anatomical Structure | 123037004 | C1268086 | CID 7192 “Anatomical Structure Segmentation Property Types”
SRT | A-00004 | Physical object | 260787004 | C0085089 | CID 7193 “Physical Object Segmentation Property Types”
SRT | M-01000 | Morphological Abnormal Structure | 49755003 | C0221198 | CID 7194 “Morphological Abnormal Structure Segmentation Property Types”
SRT | R-42019 | Function | 246464006 | C0542341 | CID 7195 “Function Segmentation Property Types”
SRT | R-42018 | Spatial and Relational Concept | 309825002 | C0587374 | CID 7196 “Spatial and Relational Concept Segmentation Property Types”
SRT | T-D0080 | Body Substance | 91720002 | C0504082 | CID 7197 “Body Substance Segmentation Property Types”
SRT | F-61002 | Substance | 105590001 | C0439861 | CID 7198 “Substance Segmentation Property Types”

## CID 7151 Segmentation Property Types

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.497

Table CID 7151. Segmentation Property Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7191 “Tissue Segmentation Property Types”
Include CID 7192 “Anatomical Structure Segmentation Property Types”
Include CID 7193 “Physical Object Segmentation Property Types”
Include CID 7194 “Morphological Abnormal Structure Segmentation Property Types”
Include CID 7195 “Function Segmentation Property Types”
Include CID 7196 “Spatial and Relational Concept Segmentation Property Types”
Include CID 7197 “Body Substance Segmentation Property Types”
Include CID 4273 “Retinal Segmentation Surfaces”

## CID 7152 Cardiac Structure Segmentation Types

Type:

  Extensible

Version:

  20130617

UID:

  1.2.840.10008.6.1.498

Table CID 7152. Cardiac Structure Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-32000 | Heart | 80891009 | C0018787
SRT | T-42000 | Aorta | 15825003 | C0003483
SRT | T-32600 | Left Ventricle | 87878005 | C0225897
SRT | T-32500 | Right Ventricle | 53085002 | C0225883
SRT | T-39000 | Pericardium | 76848001 | C0031050
SRT | T-39050 | Pericardial cavity | 25489000 | C0225972

## CID 7153 CNS Segmentation Types

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.499

Table CID 7153. CNS Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-B1100 | Adenohypophysis | 62818001 | C0032008
SRT | T-A3230 | Amygdala | 4958002 | C0002708
SRT | T-A1220 | Arachnoid | 75042008 | C0003707
FMA | 276650 | Arcuate Fasciculus |  | C2329633
SRT | T-A0100 | Brain | 12738006 | C0006104
SRT | T-A0109 | Brain cerebrospinal fluid pathway | 280371009 | C0459387
SRT | T-D0558 | Brain stem | 119238007 | C1268144
SRT | T-A1600 | Brain ventricle | 35764002 | C0007799
SRT | T-A3200 | Caudate nucleus | 11000004 | C0007461
SRT | T-A0090 | Central nervous system | 21483005 | C0927232
SRT | T-A6080 | Cerebellar white matter | 33060004 | C0152381
SRT | T-A1800 | Cerebral aqueduct | 80447000 | C0007769
SRT | T-A2020 | Cerebral cortex | 40146001 | C0007776
SRT | T-A2970 | Cerebral fornix | 87463005 | C0152334
SRT | T-A2020 | Cerebral Grey Matter | 40146001 | C0007776
SRT | T-A2030 | Cerebral White Matter | 68523003 | C0152295
SRT | T-A1000 | Cerebrospinal Fluid | 65216001 | C0007806
SRT | T-A2840 | Cingulum | 37035000 | C0228272
SRT | T-A2700 | Corpus callosum | 88442005 | C0010090
SRT | T-A3100 | Corpus striatum | 31428008 | C0010097
SRT | T-A0102 | Diencephalon | 87563008 | C0012144
SRT | T-A1120 | Dura mater | 18545000 | C0013313
SRT | T-A2594 | Entorhinal Cortex | 3937002 | C0175196
SRT | T-A1820 | Fourth ventricle | 35918002 | C0149556
SRT | T-A2200 | Frontal lobe | 83251001 | C0016733
SRT | T-A3500 | Globus pallidus | 14738005 | C0017651
SRT | T-A0096 | Gray Matter | 389081007 | C1300312
SRT | T-A2570 | Hippocampus | 5366008 | C0019564
SRT | T-A6640 | Inferior cerebellar peduncle | 67701001 | C0152393
SRT | T-A2850 | Inferior longitudinal fasciculus | 55233005 | C0228273
SRT | T-A2610 | Insula | 36169008 | C0021640
SRT | T-A7093 | Lateral corticospinal tract | 461002 | C0152402
SRT | T-A1650 | Lateral ventricle | 66720007 | C0152279
SRT | T-A0036 | Limbic lobe | 279215006 | C0458337
SRT | T-A5271 | Medial Lemniscus | 30114003 | C0228420
SRT | T-A1110 | Meninges | 1231004 | C0025285
SRT | T-A5100 | Midbrain | 61962009 | C0025462
SRT | T-A6630 | Middle cerebellar peduncle | 33723005 | C0152392
SRT | T-A0149 | Nucleus accumbens | 427667007 | C0028633
SRT | T-B1200 | Neurohypophysis | 37512009 | C0032009
SRT | T-A2400 | Occipital lobe | 31065004 | C0028785
SRT | T-A800B | Optic chiasm | 244453006 | C0029126
SRT | T-A2880 | Optic radiation | 70105001 | C0228277
SRT | T-A8060 | Optic tract | 53238003 | C0152405
SRT | T-A2300 | Parietal lobe | 16630005 | C0030560
SRT | T-A1280 | Pia mater | 23180006 | C0031869
SRT | T-B2000 | Pineal Gland | 45793000 | C0031939
SRT | T-B1000 | Pituitary | 56329008 | C0032005
SRT | T-A4904 | Posterior cerebral commissure | 279336005 | C0152327
SRT | T-A3400 | Putamen | 89278009 | C0034169
SRT | T-D0721 | Spinal cerebrospinal fluid pathway | 280401006 | C0459413
SRT | T-A7010 | Spinal cord | 2748008 | C0037925
SRT | T-A7020 | Spinal cord gray matter | 12958003 | C0475853
SRT | T-A7070 | Spinal cord white matter | 27088001 | C0458457
SRT | T-A1500 | Subarachnoid space | 35951006 | C0038527
SRT | T-A5160 | Substantia nigra | 70007007 | C0038590
SRT | T-A6620 | Superior cerebellar peduncle | 11089000 | C0152391
SRT | T-A2820 | Superior longitudinal fasciculus | 89202009 | C0228270
SRT | T-A0103 | Telencephalon | 11628009 | C0039452
SRT | T-A2500 | Temporal lobe | 78277001 | C0039485
SRT | T-A4000 | Thalamus | 42695009 | C0039729
SRT | T-A1740 | Third ventricle | 49841001 | C0149555
SRT | T-A2830 | Uncinate fasciculus | 26230003 | C0228271
SRT | T-A0095 | White Matter | 389080008 | C1300311

## CID 7154 Abdominal Segmentation Types

Type:

  Extensible

Version:

  20130617

UID:

  1.2.840.10008.6.1.500

Table CID 7154. Abdominal Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-D4000 | Abdomen | 113345001 | C0000726
SRT | T-D4010 | Abdominal cavity | 52731004 | C0230168
SRT | T-14001 | Abdominal wall muscle | 195879000 | C1279385
SRT | T-B3000 | Adrenal gland | 23451007 | C0001625
SRT | T-42500 | Abdominal aorta | 7832008 | C0003484
SRT | T-60610 | Bile Duct | 28273000 | C0005400
SRT | T-48710 | Inferior vena cava | 64131007 | C0042458
SRT | T-71000 | Kidney | 64033007 | C0022646
SRT | T-62000 | Liver | 10200004 | C0023884
SRT | T-D4600 | Omentum | 27398004 | C0028977
SRT | T-D4425 | Peritoneal cavity | 83670000 | C1704247
SRT | T-D4400 | Peritioneum | 15425007 | C0031153
SRT | T-D4900 | Retroperitoneal space | 82849001 | C0035359
SRT | T-02480 | Skin of abdomen | 75093004 | C0222166
SRT | T-58000 | Small Intestine | 30315005 | C0021852
SRT | T-C3000 | Spleen | 78961009 | C0037993
SRT | T-70001 | Urinary system | 122489005 | C1508753

## CID 7155 Thoracic Segmentation Types

Type:

  Extensible

Version:

  20130617

UID:

  1.2.840.10008.6.1.501

Table CID 7155. Thoracic Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-40000 | Blood Vessel | 59820001 | C0005847
SRT | T-D0170 | Bone of thorax | 272710004 | C0448157
SRT | T-26000 | Bronchus | 955009 | C0006255
SRT | T-14122 | Chest wall muscle | 372074006 | C1269825
SRT | T-12310 | Clavicle | 51299004 | C0008913
SRT | T-D3400 | Diaphragm | 5798000 | C0011980
SRT | T-56000 | Esophagus | 32849002 | C0014876
SRT | T-28830 | Lower lobe of lung | 90572001 | C0225758
SRT | T-28000 | Lung | 39607008 | C0024109
SRT | T-D3300 | Mediastinum | 72410000 | C0025066
SRT | T-28300 | Middle lobe of right lung | 72481006 | C0225757
SRT | T-29000 | Pleura | 3120008 | C0032225
SRT | T-11300 | Rib | 113197003 | C0035561
SRT | T-02424 | Skin of chest | 74160004 | C0222149
SRT | T-11210 | Sternum | 56873002 | C0038293
SRT | T-11502 | Thoracic spine | 122495006 | C0581269
SRT | T-D3000 | Thorax | 51185008 | C0817096
SRT | T-25000 | Trachea | 44567001 | C0040578
SRT | T-C8000 | Thymus | 9875009 | C0040113
SRT | T-28820 | Upper lobe of lung | 45653009 | C0225756

## CID 7156 Vascular Segmentation Types

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.502

Table CID 7156. Vascular Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-1A190 | Adventitia | 30180000 | C0225342
SRT | T-30000 | Cardiovascular system | 113257007 | C0007226
SRT | T-1A170 | Intima | 8361002 | C0162864
SRT | T-40230 | Lumen | 91747007 | C0524424
SRT | T-1A180 | Media | 61695000 | C0162867
SRT | T-4105E | Systemic artery | 281159003 | C0459964
SRT | T-48081 | Systemic vein | 360592004 | C0447117
SRT | M-35001 | Thrombus | 396339007 | C0087086
SRT | M-520F8 | Vascular sclerosis | 107671003 | C0003850

## CID 7157 Device Segmentation Types

Type:

  Extensible

Version:

  20130617

UID:

  1.2.840.10008.6.1.503

Table CID 7157. Device Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-12024 | Bone Pin | 77444004 | C0175718
SRT | A-12030 | Bone Screw | 68183006 | C0005975
SRT | A-11100 | Cardiac Pacemaker | 14106009 | C0030163
SRT | A-11206 | Defibrillator | 72506001 | C0162589
SRT | A-04200 | Dental Prosthesis | 27606000 | C0162686
SRT | A-04036 | Inlay Dental Restoration | 272287005 | C0441351
SRT | A-11FCD | Left ventricular assist device | 360066001 | C0181598
SRT | A-30360 | Needle | 79068005 | C0027551
SRT | A-04034 | Radioactive implant | 19443004 | C0521196
SRT | A-25500 | Stent | 65818007 | C0038257

## CID 7158 Artifact Segmentation Types

Type:

  Extensible

Version:

  20130617

UID:

  1.2.840.10008.6.1.504

Table CID 7158. Artifact Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-00916 | Clothing | 272180002 | C0009072
SRT | M-30400 | Foreign body | 19227008 | C0016542
SRT | A-17350 | Table | 86407004 | C0039224

## CID 7159 Lesion Segmentation Types

Type:

  Extensible

Version:

  20150106

UID:

  1.2.840.10008.6.1.505

Table CID 7159. Lesion Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-41610 | Abscess | 44132006 | C0000833
SRT | M-35000 | Blood clot | 75753009 | C0302148
SRT | M-3340A | Cyst | 367643001 | C0010709
SRT | M-36300 | Edema | 79654002 | C0013604
SRT | M-35300 | Embolus | 55584005 | C1704212
SRT | M-37000 | Hemorrhage | 50960005 | C0019080
SRT | M-40000 | Inflammation | 23583003 | C0021368
SRT | M-03000 | Mass | 4147007 | C0577559
SRT | M-54000 | Necrosis | 6574001 | C0027540
SRT | M-8FFFF | Neoplasm | 108369006 | C0027651
SRT | M-80003 | Neoplasm, Primary | 86049000 | C1306459
SRT | M-80006 | Neoplasm, Secondary | 14799000 | C2939419
SRT | M-03010 | Nodule | 27925004 | C0028259

## CID 7160 Pelvic Organ Segmentation Types

Type:

  Extensible

Version:

  20130617

UID:

  1.2.840.10008.6.1.506

Table CID 7160. Pelvic Organ Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-74000 | Bladder | 89837001 | C0005682
SRT | T-83200 | Cervix | 71252005 | C0007874
SRT | T-88000 | Fallopian tube | 31435000 | C0015560
SRT | T-80010 | Female external genitalia | 86969008 | C0227747
SRT | T-80020 | Female internal genitalia | 87759004 | C0227748
SRT | T-90010 | Male external genitalia | 90418005 | C0227922
SRT | T-90020 | Male internal genitalia | 38242008 | C0227923
SRT | T-87000 | Ovary | 15497006 | C0029939
SRT | T-92000 | Prostate | 41216001 | C0033572
SRT | T-59600 | Rectum | 34402009 | C0034896
SRT | T-93000 | Seminal Vesicle | 64739004 | C0036628
SRT | T-94000 | Testis | 40689003 | C0039597
SRT | T-83000 | Uterus | 35039007 | C0042149
SRT | T-82000 | Vagina | 76784001 | C0042232
SRT | T-96000 | Vas deferens | 57671007 | C0042360

## CID 7161 Physiology Segmentation Types

Type:

  Extensible

Version:

  20060822

UID:

  1.2.840.10008.6.1.507

Table CID 7161. Physiology Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-0039F | Perfusion | 371863001 | C1276288

## CID 7162 Surface Processing Algorithm Families

Type:

  Extensible

Version:

  20080829

UID:

  1.2.840.10008.6.1.636

Table CID 7162. Surface Processing Algorithm Families

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 123101 | Neighborhood Analysis
DCM | 123102 | Adaptive Filtering
DCM | 123103 | Edge Detection
DCM | 123104 | Morphological Operations
DCM | 123105 | Histogram Analysis
DCM | 123106 | Multi-Scale/Resolution Filtering
DCM | 123107 | Cluster Analysis
DCM | 123108 | Multispectral Processing
DCM | 123109 | Manual Processing
DCM | 123110 | Artificial Intelligence
DCM | 123111 | Deformable Models

## CID 7165 Abstract Segmentation Types

Type:

  Extensible

Version:

  20151113

UID:

  1.2.840.10008.6.1.962

Table CID 7165. Abstract Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 125040 | Background |  |
SRT | T-D0050 | Tissue | 85756007 | C0040300
SRT | F-61779 | Waste Material | 289925000 | C0043045
DCM | 125041 | Registration Input |  |
DCM | 113132 | Single subject extracted from group |  |
NCIt | C94970 | Reference Region |  | C2986814

## CID 7166 Common Tissue Segmentation Types

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.963

Table CID 7166. Common Tissue Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-41000 | Artery | 51114001 | C0003842
SRT | T-C2000 | Blood | 87612001 | C0005767
SRT | T-40000 | Blood vessel | 59820001 | C0005847
SRT | F-03D38 | Body fat | 248300009 | C0344335
SRT | T-D016E | Bone | 272673000 | C0262950
SRT | T-40050 | Capillary | 20982000 | C0006901
SRT | T-D021B | Cartilage | 309312004 | C0007301
SRT | T-1A200 | Connective tissue | 21793004 | C0009780
SRT | T-18010 | Ligament | 52082005 | C0023685
SRT | T-C4000 | Lymph node | 59441001 | C0024204
SRT | T-C6000 | Lymphatic system | 89890002 | C0024235
SRT | T-C6010 | Lymphatic vessel | 83555006 | C0229889
SRT | T-15009 | Meniscus | 74135004 | C0224498
SRT | T-13001 | Muscle | 71616004 | C0026845
SRT | T-D0598 | Nerve | 119410002 | C1268169
SRT | T-D0060 | Organ | 113343008 | C0229983
SRT | T-01000 | Skin | 39937001 | C1123023
SRT | T-17010 | Tendon | 13024002 | C0039508
SRT | T-D0050 | Tissue | 85756007 | C0040300
SRT | T-48000 | Vein | 29092000 | C0042449

### Note

Blood and body fat are considered tissues rather than body substances because they are cellular.

## CID 7167 Peripheral Nervous System Segmentation Types

Type:

  Extensible

Version:

  20130617

UID:

  1.2.840.10008.6.1.964

Table CID 7167. Peripheral Nervous System Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-A0140 | Peripheral nervous system | 3058005 | C0206417
SRT | T-A0500 | Peripheral nerve | 84782009 | C0031119
SRT | T-A8000 | Cranial nerve | 25238003 | C0010268
SRT | T-A8640 | Vagus nerve | 88882009 | C0042276
SRT | T-A9605 | Autonomic nerve | 53520000 | C0206250
SRT | T-A9630 | Sympathetic trunk | 44909008 | C0228972

## CID 7180 Abstract Multi-dimensional Image Model Component Semantics

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.917

Table CID 7180. Abstract Multi-dimensional Image Model Component Semantics

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 4033 “MR Proton Spectroscopy Metabolites” |  |
DCM | 113063 | T1 |  |
DCM | 113065 | T2 |  |
DCM | 113064 | T2* |  |
DCM | 113058 | Proton Density |  |
DCM | 110800 | Spin Tagging Perfusion MR Signal Intensity |  |
DCM | 113070 | Velocity encoded |  |
DCM | 113067 | Temperature encoded |  |
DCM | 110801 | Contrast Agent Angio MR Signal Intensity |  |
DCM | 110802 | Time Of Flight Angio MR Signal Intensity |  |
DCM | 110803 | Proton Density Weighted MR Signal Intensity |  |
DCM | 110804 | T1 Weighted MR Signal Intensity |  |
DCM | 110805 | T2 Weighted MR Signal Intensity |  |
DCM | 110806 | T2* Weighted MR Signal Intensity |  |
Include Section CID 7270 “MR Diffusion Component Semantics” |  |
Include Section CID 7271 “MR Diffusion Anisotropy Indices” |  |
Include Section CID 7272 “MR Diffusion Model Parameters” |  |
DCM | 110807 | Field Map MR Signal Intensity |  |
DCM | 110816 | T1 Weighted Dynamic Contrast Enhanced MR Signal Intensity |  |
DCM | 110817 | T2 Weighted Dynamic Contrast Enhanced MR Signal Intensity |  |
DCM | 110818 | T2* Weighted Dynamic Contrast Enhanced MR Signal Intensity |  |
DCM | 110819 | Blood Oxygenation Level |  |
DCM | 110820 | Nuclear Medicine Projection Activity |  |
DCM | 110821 | Nuclear Medicine Tomographic Activity |  |
DCM | 110822 | Spatial Displacement X Component |  |
DCM | 110823 | Spatial Displacement Y Component |  |
DCM | 110824 | Spatial Displacement Z Component |  |
DCM | 110825 | Hemodynamic Resistance |  |
DCM | 110826 | Indexed Hemodynamic Resistance |  |
DCM | 112031 | Attenuation Coefficient |  |
DCM | 110827 | Tissue Velocity |  |
DCM | 110828 | Flow Velocity |  |
SRT | P0-02241 | Power Doppler | 425704008 | C1960437
DCM | 110829 | Flow Variance |  |
DCM | 110830 | Elasticity |  |
DCM | 110831 | Perfusion |  |
DCM | 110832 | Speed of sound |  |
DCM | 110833 | Ultrasound Attenuation |  |
DCM | 113068 | Student's T-test |  |
DCM | 113071 | Z-score |  |
DCM | 113057 | R-Coefficient |  |
DCM | 126220 | R2-Coefficient |  |
DCM | 126221 | Chi-square |  |
DCM | 126222 | D-W |  |
DCM | 126223 | AIC |  |
DCM | 126224 | BIC |  |
DCM | 110834 | RGB R Component |  |
DCM | 110835 | RGB G Component |  |
DCM | 110836 | RGB B Component |  |
DCM | 110837 | YBR FULL Y Component |  |
DCM | 110838 | YBR FULL CB Component |  |
DCM | 110839 | YBR FULL CR Component |  |
DCM | 110840 | YBR PARTIAL Y Component |  |
DCM | 110841 | YBR PARTIAL CB Component |  |
DCM | 110842 | YBR PARTIAL CR Component |  |
DCM | 110843 | YBR ICT Y Component |  |
DCM | 110844 | YBR ICT CB Component |  |
DCM | 110845 | YBR ICT CR Component |  |
DCM | 110846 | YBR RCT Y Component |  |
DCM | 110847 | YBR RCT CB Component |  |
DCM | 110848 | YBR RCT CR Component |  |
DCM | 110849 | Echogenicity |  |
DCM | 110850 | X-Ray Attenuation |  |
DCM | 110852 | MR signal intensity |  |
DCM | 110853 | Binary Segmentation |  |
DCM | 110854 | Fractional Probabilistic Segmentation |  |
DCM | 110855 | Fractional Occupancy Segmentation |  |
DCM | 126393 | R1 |  |
DCM | 126394 | R2 |  |
DCM | 126395 | R2* |  |
DCM | 113098 | Magnetization Transfer Ratio |  |
DCM | 126396 | Magnetic Susceptibility |  |
Include Section CID 4107 “Tracer Kinetic Model Parameters” |  |
Include Section CID 4108 “Perfusion Model Parameters” |  |
Include Section CID 4109 “Model-Independent Dynamic Contrast Analysis Parameters” |  |
DCM | 126400 | Standardized Uptake Value |  |
DCM | 126401 | SUVbw |  |
DCM | 126402 | SUVlbm |  |
DCM | 126406 | SUVlbm(James128) |  |
DCM | 126405 | SUVlbm(Janma) |  |
DCM | 126403 | SUVbsa |  |
DCM | 126404 | SUVibw |  |
Include CID 10070 “Radiation Dose Types” |  |

## CID 7181 Abstract Multi-dimensional Image Model Component Units

Type:

  Extensible

Version:

  20170413

UID:

  1.2.840.10008.6.1.918

Table CID 7181. Abstract Multi-dimensional Image Model Component Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3500 “Pressure Units”
Include CID 3502 “Hemodynamic Resistance Units”
Include CID 3503 “Indexed Hemodynamic Resistance Units”
Include CID 7460 “Units of Linear Measurement”
Include CID 7461 “Units of Area Measurement”
Include CID 7462 “Units of Volume Measurement”
Include CID 84 “PET Units”
Include CID 7277 “Units of Diffusion Rate Area Over Time”
UCUM | 1 | no units
UCUM | {ratio} | ratio
UCUM | [hnsf'U] | Hounsfield Unit
UCUM | {counts} | Counts
UCUM | {counts}/s | Counts per second
UCUM | [arb'U] | arbitrary unit
UCUM | ppm | ppm
UCUM | cm/s | centimeter/second
UCUM | mm/s | millimeter/second
UCUM | dB | decibel
UCUM | Cel | degrees Celsius
UCUM | ml/min | milliliter per minute
UCUM | ml/s | milliliter per second
UCUM | ms | millisecond
UCUM | s | second
UCUM | Hz | Hertz
UCUM | mT | milliTesla
UCUM | {Particles}/[100]g{Tissue} | number particles per 100 gram of tissue
UCUM | s/mm2 | second per square millimeter
UCUM | ml/[100]g/min | milliliter per 100 gram per minute
UCUM | ml/[100]ml | milliliter per 100 milliliter
UCUM | mmol/kg{WetWeight} | millimoles per kg wet weight
UCUM | /min | /min
UCUM | /s | /s

## CID 7182 Abstract Multi-dimensional Image Model Dimension Semantics

Type:

  Extensible

Version:

  20100825

UID:

  1.2.840.10008.6.1.919

Table CID 7182. Abstract Multi-dimensional Image Model Dimension Semantics

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110856 | Linear Displacement
DCM | 110857 | Photon Energy
DCM | 110858 | Time
DCM | 110859 | Angle

## CID 7183 Abstract Multi-dimensional Image Model Dimension Units

Type:

  Extensible

Version:

  20100825

UID:

  1.2.840.10008.6.1.920

Table CID 7183. Abstract Multi-dimensional Image Model Dimension Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7460 “Units of Linear Measurement”
UCUM | ms | Millisecond
UCUM | s | Second
UCUM | deg | Degree
UCUM | rad | Radian

## CID 7184 Abstract Multi-dimensional Image Model Axis Direction

Type:

  Extensible

Version:

  20100825

UID:

  1.2.840.10008.6.1.921

Table CID 7184. Abstract Multi-dimensional Image Model Axis Direction

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110860 | Left-Right Axis
DCM | 110861 | Head-Foot Axis
DCM | 110862 | Anterior-Posterior Axis
DCM | 110863 | Apex-Base Axis
DCM | 110864 | Anterior-Inferior Axis
DCM | 110865 | Septum-Wall Axis

## CID 7185 Abstract Multi-dimensional Image Model Axis Orientation

Type:

  Extensible

Version:

  20100825

UID:

  1.2.840.10008.6.1.922

Table CID 7185. Abstract Multi-dimensional Image Model Axis Orientation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110866 | Right To Left
DCM | 110867 | Left To Right
DCM | 110868 | Head To Foot
DCM | 110869 | Foot To Head
DCM | 110870 | Anterior To Posterior
DCM | 110871 | Posterior To Anterior
DCM | 110872 | Apex To Base
DCM | 110873 | Base To Apex
DCM | 110874 | Anterior To Inferior
DCM | 110875 | Inferior To Anterior
DCM | 110876 | Septum To Wall
DCM | 110877 | Wall To Septum

## CID 7186 Abstract Multi-dimensional Image Model Qualitative Dimension Sample Semantics

Type:

  Extensible

Version:

  20100825

UID:

  1.2.840.10008.6.1.923

Table CID 7186. Abstract Multi-dimensional Image Model Qualitative Dimension Sample Semantics

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 4033 “MR Proton Spectroscopy Metabolites”
DCM | 110810 | Volumetric Diffusion Dxx Component
DCM | 110811 | Volumetric Diffusion Dxy Component
DCM | 110812 | Volumetric Diffusion Dxz Component
DCM | 110813 | Volumetric Diffusion Dyy Component
DCM | 110814 | Volumetric Diffusion Dyz Component
DCM | 110815 | Volumetric Diffusion Dzz Component
DCM | 110834 | RGB R Component
DCM | 110835 | RGB G Component
DCM | 110836 | RGB B Component
DCM | 110837 | YBR FULL Y Component
DCM | 110838 | YBR FULL CB Component
DCM | 110839 | YBR FULL CR Component
DCM | 110840 | YBR PARTIAL Y Component
DCM | 110841 | YBR PARTIAL CB Component
DCM | 110842 | YBR PARTIAL CR Component
DCM | 110843 | YBR ICT Y Component
DCM | 110844 | YBR ICT CB Component
DCM | 110845 | YBR ICT CR Component
DCM | 110846 | YBR RCT Y Component
DCM | 110847 | YBR RCT CB Component
DCM | 110848 | YBR RCT CR Component

## CID 7191 Tissue Segmentation Property Types

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1190

Table CID 7191. Tissue Segmentation Property Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 6403 “Non-lesion Object Type - Tissues”
Include CID 6405 “Chest Non-lesion Object Type - Tissues”
Include CID 7166 “Common Tissue Segmentation Types”

## CID 7192 Anatomical Structure Segmentation Property Types

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1191

Table CID 7192. Anatomical Structure Segmentation Property Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 4 “Anatomic Region”
Include CID 3010 “Cardiovascular Anatomic Locations”
Include CID 3827 “Vessel Segments”
Include CID 3829 “Pulmonary Arteries”
Include CID 4028 “Craniofacial Anatomic Regions”
Include CID 4030 “CT, MR and PET Anatomy Imaged”
Include CID 4040 “Endoscopy Anatomic Regions”
Include CID 6110 “Lung Anatomy Finding or Feature”
Include CID 6111 “Bronchovascular Anatomy Finding or Feature”
Include CID 6112 “Pleura Anatomy Finding or Feature”
Include CID 6113 “Mediastinum Anatomy Finding or Feature”
Include CID 6114 “Osseous Anatomy Finding or Feature”
Include CID 6116 “Muscular Anatomy”
Include CID 6117 “Vascular Anatomy”
Include CID 7152 “Cardiac Structure Segmentation Types”
Include CID 7153 “CNS Segmentation Types”
Include CID 7154 “Abdominal Segmentation Types”
Include CID 7155 “Thoracic Segmentation Types”
Include CID 7156 “Vascular Segmentation Types”
Include CID 7160 “Pelvic Organ Segmentation Types”
Include CID 7167 “Peripheral Nervous System Segmentation Types”
Include CID 4273 “Retinal Segmentation Surfaces”

## CID 7193 Physical Object Segmentation Property Types

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1192

Table CID 7193. Physical Object Segmentation Property Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 8 “Angiographic Interventional Devices”
Include CID 6401 “Non-lesion Object Type - Physical Objects”
Include CID 6404 “Chest Non-lesion Object Type - Physical Objects”
Include CID 7157 “Device Segmentation Types”
Include CID 7158 “Artifact Segmentation Types”

## CID 7194 Morphological Abnormal Structure Segmentation Property Types

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1193

Table CID 7194. Morphological Abnormal Structure Segmentation Property Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 7159 “Lesion Segmentation Types”

## CID 7195 Function Segmentation Property Types

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1194

Table CID 7195. Function Segmentation Property Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 7161 “Physiology Segmentation Types”

## CID 7196 Spatial and Relational Concept Segmentation Property Types

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1195

Table CID 7196. Spatial and Relational Concept Segmentation Property Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 112082 | Interface |  |
DCM | 112083 | Line |  |
DCM | 112084 | Lucency |  |
Include CID 7165 “Abstract Segmentation Types”

### Note

Some of the concepts in this context group are derived from CID 6109 “Radiographic Anatomy Finding or Feature”.

## CID 7197 Body Substance Segmentation Property Types

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1196

Table CID 7197. Body Substance Segmentation Property Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-60650 | Bile | 70150004 | C0005388
SRT | T-D0070 | Body fluid | 32457005 | C0005889
SRT | T-59666 | Feces | 39477002 | C0015733
SRT | C-10080 | Gas | 74947009 | C0017110
SRT | T-70060 | Urine | 78014005 | C0042036

## CID 7198 Substance Segmentation Property Types

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1197

Table CID 7198. Substance Segmentation Property Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 6402 “Non-lesion Object Type - Substances”

## CID 7201 Referenced Image Purposes of Reference

Type:

  Extensible

Version:

  20090409

UID:

  1.2.840.10008.6.1.508

Table CID 7201. Referenced Image Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121311 | Localizer
DCM | 121312 | Biopsy localizer
DCM | 121313 | Other partial views
DCM | 121314 | Other image of biplane pair
DCM | 121315 | Other image of stereoscopic pair
DCM | 121316 | Images related to standalone object
DCM | 121317 | Spectroscopy
DCM | 121338 | Anatomic image
DCM | 121339 | Functional image
DCM | 121340 | Spectral filtered image
DCM | 121341 | Device localizer
DCM | 121346 | Acquisition frames corresponding to volume
DCM | 121347 | Volume corresponding to spatially-related acquisition frames
DCM | 121348 | Temporal Predecessor
DCM | 121349 | Temporal Successor

## CID 7202 Source Image Purposes of Reference

Type:

  Extensible

Version:

  20170405

UID:

  1.2.840.10008.6.1.509

Table CID 7202. Source Image Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121320 | Uncompressed predecessor
DCM | 121321 | Mask image for image processing operation
DCM | 121322 | Source image for image processing operation
DCM | 121329 | Source image for montage
DCM | 121330 | Lossy compressed predecessor
DCM | 121358 | For Processing predecessor
DCM | 113130 | Predecessor containing group of imaging subjects
DCM | 128250 | Structural image for image processing
DCM | 128251 | Flow image for image processing

## CID 7203 Image Derivation

Type:

  Extensible

Version:

  20170405

UID:

  1.2.840.10008.6.1.510

Table CID 7203. Image Derivation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113040 | Lossy Compression
DCM | 113041 | Apparent Diffusion Coefficient
DCM | 113042 | Pixel by pixel addition
DCM | 113043 | Diffusion weighted
DCM | 113044 | Diffusion Anisotropy
DCM | 113045 | Diffusion Attenuated
DCM | 113046 | Pixel by pixel division
DCM | 113047 | Pixel by pixel mask
DCM | 113048 | Pixel by pixel Maximum
DCM | 113049 | Pixel by pixel mean
DCM | 113050 | Metabolite Maps from spectroscopy data
DCM | 113051 | Pixel by pixel Minimum
DCM | 113052 | Mean Transit Time
DCM | 113053 | Pixel by pixel multiplication
DCM | 113054 | Negative Enhancement Integral
DCM | 113055 | Regional Cerebral Blood Flow
DCM | 113056 | Regional Cerebral Blood Volume
DCM | 113057 | R-Coefficient
DCM | 113058 | Proton Density
DCM | 113059 | Signal Change
DCM | 113060 | Signal to Noise
DCM | 113061 | Standard Deviation
DCM | 113062 | Pixel by pixel subtraction
DCM | 113063 | T1
DCM | 113064 | T2*
DCM | 113065 | T2
DCM | 113066 | Time Course of Signal
DCM | 113067 | Temperature encoded
DCM | 113068 | Student's T-Test
DCM | 113069 | Time To Peak
DCM | 113084 | Tmax
DCM | 113070 | Velocity encoded
DCM | 113071 | Z-Score
DCM | 113072 | Multiplanar reformatting
DCM | 113073 | Curved multiplanar reformatting
DCM | 113074 | Volume rendering
DCM | 113075 | Surface rendering
DCM | 113076 | Segmentation
DCM | 113077 | Volume editing
DCM | 113078 | Maximum intensity projection
DCM | 113079 | Minimum intensity projection
DCM | 113085 | Spatial resampling
DCM | 113086 | Edge enhancement
DCM | 113087 | Smoothing
DCM | 113088 | Gaussian blur
DCM | 113089 | Unsharp mask
DCM | 113090 | Image stitching
DCM | 113091 | Spatially-related frames extracted from the volume
DCM | 113092 | Temporally-related frames extracted from the set of volumes
DCM | 113097 | Multi-energy proportional weighting
DCM | 113093 | Polar to Rectangular Scan Conversion
DCM | 113131 | Extraction of individual subject from group
DCM | 128303 | OCT B-scan analysis

## CID 7205 Purpose of Reference to Alternate Representation

Type:

  Extensible

Version:

  20040322

UID:

  1.2.840.10008.6.1.511

Table CID 7205. Purpose of Reference to Alternate Representation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121324 | Source image
DCM | 121325 | Lossy compressed image
DCM | 121326 | Alternate SOP Class instance
DCM | 121327 | Full fidelity image
DCM | 121328 | Alternate Photometric Interpretation image

## CID 7210 Related Series Purposes of Reference

Type:

  Extensible

Version:

  20140627

UID:

  1.2.840.10008.6.1.512

Table CID 7210. Related Series Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122400 | Simultaneously Acquired
DCM | 122401 | Same Anatomy
DCM | 122402 | Same Indication
DCM | 122403 | For Attenuation Correction
DCM | 121323 | Source series for image processing operation

## CID 7215 Spectroscopy Purpose of Reference

Type:

  Extensible

Version:

  20140105

UID:

  1.2.840.10008.6.1.970

Table CID 7215. Spectroscopy Purpose of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121318 | Spectroscopy Data for Water Phase Correction

## CID 7220 RT Dose Derivation

Type:

  Extensible

Version:

  20140106

UID:

  1.2.840.10008.6.1.968

Table CID 7220. RT Dose Derivation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121370 | Composed from prior doses
DCM | 121371 | Composed from prior doses and current plan

## CID 7221 RT Dose Purpose of Reference

Type:

  Extensible

Version:

  20140106

UID:

  1.2.840.10008.6.1.969

Table CID 7221. RT Dose Purpose of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121372 | Source dose for composing current dose

## CID 7222 Parametric Map Derivation Image Purpose of Reference

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.1009

Table CID 7222. Parametric Map Derivation Image Purpose of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121322 | Source Image for Image Processing Operation

## CID 7250 Multi-Frame Subset Type

This Context Group specifies the terms used to identify a subset of frames of a multi-frame image. It is used for encoding an equivalent of the Image SOP Instance Reference Macro (see PS3.3) in an HL7 v3 data structure (see HL7 v3 CMET, COCT_RM830120 "A_DicomCompositeObjectReference minimal").

Type:

  Extensible

Version:

  20070625

UID:

  1.2.840.10008.6.1.513

Table CID 7250. Multi-Frame Subset Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121190 | Referenced Frames
DCM | 121191 | Referenced Segment

## CID 7260 Diffusion Acquisition Value Types

Type:

  Extensible

Version:

  20150918

UID:

  1.2.840.10008.6.1.1059

Table CID 7260. Diffusion Acquisition Value Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113221 | HARDI
DCM | 113222 | DKI
DCM | 113223 | DTI
DCM | 113224 | DSI
DCM | 113225 | LSDI
DCM | 113226 | Single Shot EPI
DCM | 113227 | Multiple Shot EPI
DCM | 113228 | Parallel Imaging

## CID 7261 Diffusion Model Value Types

Type:

  Extensible

Version:

  20150918

UID:

  1.2.840.10008.6.1.1060

Table CID 7261. Diffusion Model Value Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113231 | Single Tensor
DCM | 113232 | Multi Tensor
DCM | 113233 | Model Free
DCM | 113234 | CHARMED
DCM | 113224 | DSI
DCM | 113236 | DOT
DCM | 113237 | PAS
DCM | 113238 | Spherical Deconvolution

## CID 7262 Diffusion Tractography Algorithm Families

Type:

  Extensible

Version:

  20150918

UID:

  1.2.840.10008.6.1.1061

Table CID 7262. Diffusion Tractography Algorithm Families

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113211 | Deterministic
DCM | 113212 | Probabilistic
DCM | 113213 | Global
DCM | 113214 | FACT
DCM | 113215 | Streamline
DCM | 113216 | TEND
DCM | 113217 | Bootstrap
DCM | 113218 | Euler
DCM | 113219 | Runge-Kutta

## CID 7263 Diffusion Tractography Measurement Types

Type:

  Extensible

Version:

  20150918

UID:

  1.2.840.10008.6.1.1062

Table CID 7263. Diffusion Tractography Measurement Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113201 | Trace
DCM | 113202 | Mean Diffusivity
DCM | 113041 | Apparent Diffusion Coefficient
DCM | 110808 | Fractional Anisotropy
DCM | 110809 | Relative Anisotropy
DCM | 113203 | Radial Diffusivity
DCM | 113204 | Axial Diffusivity
DCM | 113205 | Mean Kurtosis
DCM | 113206 | Apparent Kurtosis Coefficient
DCM | 113207 | Radial Kurtosis
DCM | 113208 | Axial Kurtosis
DCM | 113209 | Fractional Kurtosis Anisotropy
DCM | 110810 | Volumetric Diffusion Dxx Component
DCM | 110811 | Volumetric Diffusion Dxy Component
DCM | 110812 | Volumetric Diffusion Dxz Component
DCM | 110813 | Volumetric Diffusion Dyy Component
DCM | 110814 | Volumetric Diffusion Dyz Component
DCM | 110815 | Volumetric Diffusion Dzz Component

## CID 7270 MR Diffusion Component Semantics

Type:

  Extensible

Version:

  20170413

UID:

  1.2.840.10008.6.1.1165

Table CID 7270. MR Diffusion Component Semantics

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Units
--- | --- | --- | --- | --- | ---
DCM | 113043 | Diffusion weighted |  |  | DT (1, UCUM, "no units")
DCM | 110810 | Volumetric Diffusion Dxx Component |  |  | DT (1, UCUM, "no units")
DCM | 110811 | Volumetric Diffusion Dxy Component |  |  | DT (1, UCUM, "no units")
DCM | 110812 | Volumetric Diffusion Dxz Component |  |  | DT (1, UCUM, "no units")
DCM | 110813 | Volumetric Diffusion Dyy Component |  |  | DT (1, UCUM, "no units")
DCM | 110814 | Volumetric Diffusion Dyz Component |  |  | DT (1, UCUM, "no units")
DCM | 110815 | Volumetric Diffusion Dzz Component |  |  | DT (1, UCUM, "no units")

## CID 7271 MR Diffusion Anisotropy Indices

Type:

  Extensible

Version:

  20170413

UID:

  1.2.840.10008.6.1.1166

Table CID 7271. MR Diffusion Anisotropy Indices

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Units
--- | --- | --- | --- | --- | ---
DCM | 110808 | Fractional Anisotropy |  |  | DT ({0:1}, UCUM, "range 0:1")
DCM | 110809 | Relative Anisotropy |  |  | DT ({ratio}, UCUM, "ratio")
DCM | 113288 | Volume Ratio |  |  | DT ({0:1}, UCUM, "range 0:1")

## CID 7272 MR Diffusion Model Parameters

Type:

  Extensible

Version:

  20170413

UID:

  1.2.840.10008.6.1.1167

Table CID 7272. MR Diffusion Model Parameters

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Units
--- | --- | --- | --- | --- | ---
DCM | 113041 | Apparent Diffusion Coefficient |  | C3890194 | DCID 7277 “Units of Diffusion Rate Area Over Time”
DCM | 113289 | Diffusion Coefficient |  |  | DCID 7277 “Units of Diffusion Rate Area Over Time”
DCM | 113290 | Mono-exponential Apparent Diffusion Coefficient |  |  | DCID 7277 “Units of Diffusion Rate Area Over Time”
DCM | 113291 | Slow Diffusion Coefficient |  |  | DCID 7277 “Units of Diffusion Rate Area Over Time”
DCM | 113292 | Fast Diffusion Coefficient |  |  | DCID 7277 “Units of Diffusion Rate Area Over Time”
DCM | 113293 | Fast Diffusion Coefficient Fraction |  |  | DT ({0:1}, UCUM, "range 0:1")
DCM | 113294 | Kurtosis Diffusion Coefficient |  |  | DCID 7277 “Units of Diffusion Rate Area Over Time”
DCM | 113295 | Gamma Distribution Scale Parameter |  |  | DT (1, UCUM, "no units")
DCM | 113296 | Gamma Distribution Shape Parameter |  |  | DT (1, UCUM, "no units")
DCM | 113297 | Gamma Distribution Mode |  |  | DT (1, UCUM, "no units")
DCM | 113298 | Distributed Diffusion Coefficient |  |  | DCID 7277 “Units of Diffusion Rate Area Over Time”
DCM | 113299 | Anomalous Exponent Parameter |  |  | DT ({0:1}, UCUM, "range 0:1")

## CID 7273 MR Diffusion Models

Type:

  Extensible

Version:

  20170413

UID:

  1.2.840.10008.6.1.1168

Table CID 7273. MR Diffusion Models

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113250 | Mono-exponential diffusion model
DCM | 113251 | Bi-exponential (IVIM) diffusion model
DCM | 113252 | Kurtosis diffusion model
DCM | 113253 | Gamma distribution model
DCM | 113254 | Stretched exponential diffusion model
DCM | 113255 | Truncated Gaussian diffusion model

### Note

## CID 7274 MR Diffusion Model Fitting Methods

Type:

  Extensible

Version:

  20170413

UID:

  1.2.840.10008.6.1.1169

Table CID 7274. MR Diffusion Model Fitting Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113260 | Log of ratio of two samples
DCM | 113261 | Least squares fit of multiple samples
DCM | 113265 | Levenberg–Marquardt
DCM | 113266 | Trust-Region
DCM | 113267 | Fixed-Dp
DCM | 113268 | Segmented-Unconstrained
DCM | 113269 | Segmented-Constrained
DCM | 113270 | Bayesian-Probability

### Note

## CID 7275 MR Diffusion Model Specific Methods

Type:

  Extensible

Version:

  20170413

UID:

  1.2.840.10008.6.1.1170

Table CID 7275. MR Diffusion Model Specific Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113285 | Voxelwise selection of b-value

### Note

## CID 7276 MR Diffusion Model Inputs

Type:

  Extensible

Version:

  20170413

UID:

  1.2.840.10008.6.1.1171

Table CID 7276. MR Diffusion Model Inputs

Coding Scheme Designator | Code Value | Code Meaning | Units
--- | --- | --- | ---
DCM | 113240 | Source image diffusion b-value | DT (s/mm2, UCUM, "s/mm2")

### Note

## CID 7277 Units of Diffusion Rate Area Over Time

Type:

  Extensible

Version:

  20170413

UID:

  1.2.840.10008.6.1.1172

Table CID 7277. Units of Diffusion Rate Area Over Time

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | mm2/s | mm2/s
UCUM | um2/ms | um2/ms
UCUM | um2/s | um2/s
UCUM | 10-6.mm2/s | 10-6.mm2/s

### Note

## CID 7300 Implant Materials

Type:

  Extensible

Version:

  20101102

UID:

  1.2.840.10008.6.1.1031

Table CID 7300. Implant Materials

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | F-61166 | Nickel Titanium | 261250004 | C0076736
SRT | F-611FC | Gold Alloy | 256496006 | C0018027
SRT | F-61207 | Stainless Steel Material | 256506002 | C0038126
SRT | F-61DF9 | Polymer | 412155002 | C0032521
SRT | F-61202 | Carbon Fiber | 256501007 | C0108411

## CID 7301 Intervention Types

Type:

  Extensible

Version:

  20101102

UID:

  1.2.840.10008.6.1.1032

Table CID 7301. Intervention Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P1-14810 | Hip joint reconstruction | 119614000 | C1293219
SRT | P1-14505 | Hip joint implantation | 119610009 | C1293213
SRT | P1-103D3 | Resurfacing of the femoral head | 445185007 | C2919830
SRT | P1-189C2 | Resurfacing of the patella | 239503002 | C0408429

## CID 7302 Implant Templates View Orientations

Type:

  Extensible

Version:

  20101102

UID:

  1.2.840.10008.6.1.1033

Table CID 7302. Implant Templates View Orientations

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-10206 | Antero-Posterior | 399348003 | C0442212
SRT | R-10226 | Medio-Lateral | 399368009 | C1302345
SRT | R-10228 | Lateral-Medial | 399352003 | C1302336

## CID 7303 Implant Templates Modified View Orientations

Type:

  Extensible

Version:

  20101102

UID:

  1.2.840.10008.6.1.1034

Table CID 7303. Implant Templates Modified View Orientations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112300 | AP+45
DCM | 112301 | AP-45

## CID 7304 Implant Target Anatomy

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.1035

Table CID 7304. Implant Target Anatomy

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-15750 | Ankle Joint | 70258002 | C0003087
SRT | T-11501 | Cervical Spine | 122494005 | C0728985
SRT | T-D00F7 | Cervico-Thoracic Spine | 297171002 | C0729373
SRT | T-15430 | Elbow Joint | 16953009 | C0013770
SRT | T-11196 | Facial Bones | 91397008 | C0015455
SRT | T-12710 | Femur | 71341001 | C0015811
SRT | T-12711 | Head of Femur | 2812003 | C0015813
SRT | T-D078C | Proximal Femur | 310651003 | C0588192
SRT | T-12717 | Shaft of Femur | 41111004 | C0588193
SRT | T-D078D | Distal Femur | 310652005 | C0588194
SRT | T-15516 | Finger Joint | 125682004 | C0016125
SRT | T-15710 | Hip Joint | 24136001 | C0019558
SRT | T-D1213 | Jaw Region | 661005 | C0022359
SRT | T-D9200 | Knee | 72696002 | C0022742
SRT | T-11503 | Lumbar Spine | 122496007 | C0024091
SRT | T-D0059 | Lumbo-Sacral Spine | 243898001 | C0446379
SRT | T-11180 | Mandible | 91609006 | C0024687
SRT | T-11170 | Maxilla | 70925003 | C0024947
SRT | T-12730 | Patella | 64234005 | C0030647
SRT | T-12310 | Clavicle | 51299004 | C0008913
SRT | T-D2220 | Shoulder | 16982005 | C0037004
SRT | T-12410 | Humerus | 85050009 | C0020164
SRT | T-1240F | Proximal Humerus | 119524001 | C0588209
SRT | T-12412 | Shaft of Humerus | 20760004 | C0588210
SRT | T-1241F | Distal Humerus | 118495001 | C0588211
SRT | T-12420 | Radius | 62413002 | C0034627
SRT | T-1242A | Proximal Radius | 12881000 | C0588205
SRT | T-12423 | Shaft of Radius | 47728000 | C0588208
SRT | T-1242B | Distal Radius | 75129005 | C0588207
SRT | T-12430 | Ulna | 23416004 | C0041600
SRT | T-1243A | Proximal Ulna | 34318004 | C0588201
SRT | T-12435 | Shaft of Ulna | 21133008 | C0588204
SRT | T-1243B | Distal Ulna | 91238003 | C0588203
SRT | T-11100 | Skull | 89546000 | C0037303
SRT | T-12600 | Hand | 24097009 | C0448064
SRT | T-11502 | Thoracic Spine | 122495006 | C0581269
SRT | T-D00FA | Thoraco-Lumbar Spine | 297174005 | C0574026
SRT | T-15460 | Wrist Joint | 74670003 | C1322271
SRT | T-12375 | Pelvis | 118645006 | C0448168
SRT | T-12750 | Fibula | 87342007 | C0016068
SRT | T-12780 | Talus | 67453005 | C0039277
SRT | T-12770 | Calcaneus | 80144004 | C0006655
SRT | T-12740 | Tibia | 12611008 | C0040184
SRT | T-12746 | Shaft of Tibia | 52687003 | C0588199
SRT | T-1274B | Distal Tibia | 64605006 | C0588200

### Note

Consistent with other concepts in this context group that refer to specific bones or joints, the concept for Elbow has been changed from T-D8300 to T-15430 used in a prior version of this table. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 7305 Implant Planning Landmarks

Type:

  Extensible

Version:

  20101102

UID:

  1.2.840.10008.6.1.1036

Table CID 7305. Implant Planning Landmarks

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7306 “Human Hip Implant Planning Landmarks”

## CID 7306 Human Hip Implant Planning Landmarks

Type:

  Extensible

Version:

  20101102

UID:

  1.2.840.10008.6.1.1037

Table CID 7306. Human Hip Implant Planning Landmarks

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112302 | Anatomical axis of femur
DCM | 112303 | Acetabular Center of Rotation
DCM | 112304 | Femur Head Center of Rotation

## CID 7307 Implant Component Types

Type:

  Extensible

Version:

  20101102

UID:

  1.2.840.10008.6.1.1038

Table CID 7307. Implant Component Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7308 “Human Hip Implant Component Types”
Include CID 7309 “Human Trauma Implant Component Types”

## CID 7308 Human Hip Implant Component Types

Type:

  Extensible

Version:

  20101102

UID:

  1.2.840.10008.6.1.1039

Table CID 7308. Human Hip Implant Component Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 112305 | Acetabular Cup Shell |  |
DCM | 112306 | Acetabular Cup Insert |  |
DCM | 112307 | Acetabular Cup Monoblock |  |
SRT | A-04459 | Femoral Head Prosthesis | 304121006 | C0015803
DCM | 112308 | Femoral Head Ball Component |  |
DCM | 112309 | Femoral Head Cone Taper Component |  |
DCM | 112310 | Femoral Stem |  |
DCM | 112311 | Femoral Stem Distal Component |  |
DCM | 112312 | Femoral Stem Proximal Component |  |
DCM | 112313 | Femoral Stem Component |  |
DCM | 112314 | Neck Component |  |
DCM | 112315 | Monoblock Stem |  |
DCM | 112325 | Distal Centralizer |  |
DCM | 112316 | Prosthetic Shaft Augment |  |
DCM | 112317 | Femoral Head Resurfacing Component |  |

## CID 7309 Human Trauma Implant Component Types

Type:

  Extensible

Version:

  20101102

UID:

  1.2.840.10008.6.1.1040

Table CID 7309. Human Trauma Implant Component Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-12030 | Screw | 68183006 | C0005975
SRT | A-12010 | Bone Plate | 271003 | C0005971
SRT | A-12018 | DHS Plate | 257327003 | C0441261
SRT | A-12020 | Bone Nail | 63289001 | C0336579

## CID 7310 Implant Fixation Method

Type:

  Extensible

Version:

  20101102

UID:

  1.2.840.10008.6.1.1041

Table CID 7310. Implant Fixation Method

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | P0-02126 | Anchoring | 129380009 | C1292829
SRT | P0-02125 | Fusion | 129379006 | C1283075
SRT | P0-021D6 | Gluing | 360038009 | C1283084
SRT | P1-1099B | Internal fixation using internal fixator system | 257837004 | C0441561
SRT | P1-10999 | Internal fixation using plate | 257835007 | C0441559
SRT | P1-10998 | Internal fixation using screw | 257834006 | C0441558
SRT | P1-10997 | Internal fixation using staple | 257833000 | C0441557
SRT | R-41C37 | Cemented component fixation | 257771002 | C0441496
SRT | R-42808 | Uncemented component fixation | 304367000 | C0582264
SRT | P1-08080 | Repair by nailing | 35860002 | C0021885
DCM | 112318 | Pinning |  |
DCM | 112319 | Sewing |  |
DCM | 112320 | Bolting |  |
DCM | 112321 | Wedging |  |

## CID 7320 Planning Methods

Type:

  Extensible

Version:

  20101102

UID:

  1.2.840.10008.6.1.924

Table CID 7320. Planning Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112340 | Generic 2D Planning
DCM | 112341 | Generic 3D Planning
DCM | 112342 | Generic Planning for Hip Replacement
DCM | 112343 | Generic Planning for Knee Replacement
DCM | 112344 | Müller Method Planning for Hip Replacement

## CID 7445 Device Participating Roles

Type:

  Extensible

Version:

  20120406

UID:

  1.2.840.10008.6.1.1042

Table CID 7445. Device Participating Roles

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113859 | Irradiating Device
DCM | 121097 | Recording
DCM | 113942 | X-Ray Reading Device

## CID 7449 Reader Specialty

Type:

  Extensible

Version:

  20160601

UID:

  1.2.840.10008.6.1.1119

Table CID 7449. Reader Specialty

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 128006 | Abdominal Imaging Specialty |  |
DCM | 128007 | Cardiac Imaging Specialty |  |
DCM | 128008 | Head and Neck Imaging Specialty |  |
DCM | 128009 | Musculoskeletal Imaging Specialty |  |
DCM | 128010 | Neurology Specialty |  |
DCM | 128011 | Neuroradiology Imaging Specialty |  |
DCM | 128012 | OB/Gyn Imaging Specialty |  |
DCM | 128013 | Oncologic Imaging Specialty |  |
DCM | 128014 | Oncology Specialty |  |
DCM | 128015 | Thoracic Imaging Specialty |  |
DCM | 128016 | Pediatric Imaging Specialty |  |
DCM | 128017 | Vascular Imaging Specialty |  |

## CID 7450 Person Roles

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.514

Table CID 7450. Person Roles

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 121025 | Patient |  |
SRT | J-00552 | Healthcare professional | 223366009 | C1704312
SRT | S-11090 | Friend | 113163005 | C0079382
Include CID 7451 “Family Member” |  |
Include CID 7452 “Organizational Roles” |  |

## CID 7451 Family Member

Type:

  Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.515

Table CID 7451. Family Member

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | S-10121 | Natural mother | 65656005 | C0337486
SRT | S-10131 | Natural father | 9947008 | C0337494
SRT | S-10151 | Natural sister | 73678001 | C0337515
SRT | S-10161 | Natural brother | 60614009 | C0337528
SRT | S-101A1 | Aunt | 25211005 | C0337576
SRT | S-101A2 | Uncle | 38048003 | C0337577
SRT | S-10154 | Half-sister | 2272004 | C0337518
SRT | S-10164 | Half-brother | 45929001 | C0337531
SRT | S-10115 | Natural grand-mother | 17945006 | C0337476
SRT | S-10116 | Natural grand-father | 62296006 | C0337477
SRT | S-10181 | Natural daughter | 83420006 | C0337552
SRT | S-10191 | Natural son | 113160008 | C0337564
SRT | S-101A9 | Female first cousin | 270002 | C0337584
SRT | S-101AA | Male first cousin | 11993008 | C0337585

## CID 7452 Organizational Roles

Type:

  Extensible

Version:

  20170626

UID:

  1.2.840.10008.6.1.516

Table CID 7452. Organizational Roles

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | J-0016E | Medical Practitioner | 158965000 | C1306754
SRT | J-004E8 | Physician | 309343006 | C0031831
DCM | 128670 | Head of Radiology |  |
DCM | 128671 | Chair of Protocol Committee |  |
DCM | 128676 | Representative of Protocol Committee |  |
DCM | 128677 | Representative of Ethics Committee |  |
DCM | 128675 | Head of Cardiology |  |
DCM | 128673 | Administrator of Radiology Department |  |
SRT | J-07100 | Nurse | 106292003 | C0028661
SRT | J-00187 | Radiologic Technologist | 159016003 | C0402007
DCM | 128674 | Lead Radiologic Technologist |  |
SRT | J-06173 | Radiation Therapist | 3430008 |
SRT | J-00187 | Radiographer | 159016003 | C0402007
UMLS | C1144859 | Intern |  | C1144859
SRT | J-005E6 | Resident | 405277009 | C1320928
SRT | J-00172 | Registrar | 158971006 | C0401974
DCM | 121088 | Fellow |  |
SRT | J-005E8 | Attending | 405279007 | C1320929
SRT | J-0050A | Consultant | 309390008 | C0586911
UMLS | C1441532 | Consulting Physician |  | C1441532
SRT | J-0714A | Scrub nurse | 415506007 | C1531952
SRT | J-00556 | Surgeon | 304292004 | C0582175
DCM | 121092 | Sonologist |  |
UMLS | C1954848 | Sonographer |  | C1954848
UMLS | C2985483 | Radiation Physicist |  | C2985483
UMLS | C1708969 | Medical Physicist |  | C1708969

### Note

1. The distinction between a "physician" and a "surgeon" and a "medical practitioner" is subject to regional variation. In the US, "physician" is often equated with "medical practitioner", and a "surgeon" is considered to be a "physician". In the UK, a "surgeon" is a "medical practitioner" but is not a "physician". In SNOMED, "physician" and "surgeon" are distinct siblings with no direct relationship, and both are children of "medical practitioner". It is recommended that "medical practitioner" be used rather than "physician" when there is uncertainty over whether the person is or is not a "surgeon".

2. There is no distinction between a "radiographer" and a "radiologic technologist", hence the same SNOMED concept is used for both, and "radiologic technologist" is provided as a synonym for use in the US.

3. In the US, the medical practitioner not in training responsible for the care of a hospital patient is referred to as an "attending". In the UK they are referred to as a "consultant". Though these two concepts are essentially the same, they are separate concepts in SNOMED, which defines no explicit relationship between them.

4. 4. A distinction is made between a Consultant and a Consulting Physician since these are separate concepts in UMLS. A Consultant is defined as "individuals referred to for expert or professional advice or services" (MSH) whereas a Consulting Physician is defined as "a physician that has expertise in a specific medical discipline that can offer expertise or advice to other physicians and healthcare providers" (from NCI/PT). In UK practice a "consultant" is always a medical practitioner. In SNOMED, (J-0050A, SRT, "Consultant") is actually described as "Hospital Consultant" and is a child of "Medical practitioner grade (occupation)".

## CID 7453 Performing Roles

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.517

Table CID 7453. Performing Roles

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 121094 | Performing |  |
UMLS | C1709880 | Referring |  | C1709880
DCM | 121096 | Requesting |  |
DCM | 121097 | Recording |  |
DCM | 121098 | Verifying |  |
DCM | 121099 | Assisting |  |
SRT | J-0714B | Circulating Nurse | 413854007 | C1531633
DCM | 121101 | Standby |  |
DCM | 113850 | Irradiation Authorizing |  |
DCM | 113851 | Irradiation Administering |  |

## CID 7454 Animal Taxonomic Rank Values

Type:

  Extensible

Version:

  20160211

UID:

  1.2.840.10008.6.1.518

Table CID 7454. Animal Taxonomic Rank Values

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | ITIS TSN
--- | --- | --- | --- | --- | ---
SRT | L-85003 | homo sapiens | 337915000 | C0086418 | 180092
SRT | L-000F9 | Felis | 388626009 | C0524517 | 180586
SRT | L-00376 | Felis catus (domestic cat) | 448169003 | C0007450 | 183798
SRT | L-000A9 | Equus | 388445009 | C1265527 | 180689
SRT | L-8A102 | Equus caballus (domestic horse) | 35354009 | C0019944 | 180691
SRT | L-8C3FD | Ovis | 388254009 | C0036945 | 180709
SRT | L-8C336 | Ovis aries (domestic sheep) | 125099002 | C1123019 | 552475
SRT | L-8B1FB | Sus | 388393002 | C1265533 | 180721
SRT | L-8B100 | Sus scrofa | 78678003 | C1135183 | 180722
SRT | L-8C3FB | Capra | 388249000 | C1265549 | 180714
SRT | L-8C306 | Capra hircus (domestic goat) | 125097000 | C0018019 | 180715
SRT | L-881FC | Canis | 388490000 | C0524516 | 180595
SRT | L-88121 | Canis lupus | 36855005 | C1510418 | 180596
SRT | L-88124 | Canis lupus familiaris (domestic dog) | 448771007 | C0012984 | 726821
SRT | L-8BA18 | Bos | 388168008 | C1265540 | 183837
SRT | L-8B9F9 | Bovinae | 107007004 | C0325235 | 552332
SRT | L-8B941 | Bos taurus (domestic cow) | 34618005 | C1140701 | 183838
SRT | L-87830 | Mus genus | 447482001 | C0026809 | 180365
SRT | L-87831 | Mus musculus (House mouse) | 447612001 | C0025914 | 180366
ITIS_TSN | 180278 | Peromyscus leucopus (American white-footed mouse) |  |  | 180278
ITIS_TSN | 180276 | Peromyscus maniculatus (Deer mouse) |  |  | 180276
SRT | L-877FB | Rattus | 371564000 | C0034721 | 180361
SRT | L-877FC | Rattus norvegicus (common rat) | 371565004 | C0034693 | 180363
ITIS_TSN | 180346 | Sigmodon genus (cotton rat) |  | C0037070 | 180346
SRT | L-87A02 | Cavia porcellus (domestic guinea pig) | 125076001 | C0999699 | 584713
SRT | L-88423 | Mustela putorius furo (ferret) | 449310008 | C0015859 | 727313
SRT | L-86B02 | Oryctolagus cuniculus (European rabbit) | 36571002 | C0324889 | 180129
SRT | L-001DE | Callithrix jacchus (common marmoset) | 406733009 | C0006765 | 572915

### Note

Codes that are now defined in SNOMED as "ambiguous" (conceptstatus = 4) were previously included in this table, but have been retired and replaced with unambiguous alternatives (e.g., (L-80700, SRT, "Canine species") has been replaced with genus (L-881FC, SRT, "Canis"), species (L-88121, SRT, "Canis lupus") and subspecies (L-88124, SRT, "Canis lupus familiaris")). Note that in UMLS, there is a lack of distinction between "Canis familiaris" and "Canis lupus familiaris". The replaced codes are (L-85B00, SRT, "homo sapiens"), (L-80A00, SRT, "Feline species"), (L-80400, SRT, "Equine species), (L-80300, SRT, "Ovine species"), (L-80500, SRT, "Porcine species"), (L-80200, SRT, "Caprine species"), (L-80700, SRT, "Canine species") and (L-80100, SRT, "Bovine species").

## CID 7455 Sex

This Context Group includes terms for the finding of sex of a subject for clinical purposes, such as selection of sex-based growth metrics.

Type:

  Non-Extensible

Version:

  20040112

UID:

  1.2.840.10008.6.1.519

Table CID 7455. Sex

Coding Scheme Designator | Code Value | Code Meaning | Patient's Sex (0010,0040) Equivalent
--- | --- | --- | ---
DCM | M | Male | M
DCM | F | Female | F
DCM | U | Unknown sex |
DCM | MP | Male Pseudohermaphrodite |
DCM | FP | Female Pseudohermaphrodite |
DCM | H | Hermaphrodite |
DCM | MC | Male changed to Female |
DCM | FC | Female changed to Male |
DCM | 121104 | Ambiguous sex |
DCM | 121102 | Other sex |
DCM | 121103 | Undetermined sex | O

### Note

1. These terms are distinct from the gender of a subject for administrative purposes, although the default value for clinical sex is often based on the administrative gender (e.g., see TID 1007 “Subject Context, Patient”). The administrative value "O" from Patient's Sex (0010,0040) maps by default to "undetermined" for clinical purposes.

2. This Context Group in a prior edition of the Standard included codes improperly attributed to ISO 5218.

3. These terms are derived from the terminology and codes for sex in ASTM E1633-02a "Standard Specification for Coded Values Used in the Electronic Health Record."

## CID 7456 Units of Measure for Age

Type:

  Non-Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.520

Table CID 7456. Units of Measure for Age

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | a | year
UCUM | mo | month
UCUM | wk | week
UCUM | d | day
UCUM | h | hour
UCUM | min | minute

## CID 7457 Sex - Male Female or Both

Type:

  Extensible

Version:

  20151110

UID:

  1.2.840.10008.6.1.1067

Table CID 7457. Sex - Male Female or Both

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | M | Male |  |
DCM | F | Female |  |
DCM | 127146 | Mixed sex |  |

## CID 7460 Units of Linear Measurement

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.521

Table CID 7460. Units of Linear Measurement

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | cm | centimeter
UCUM | mm | millimeter
UCUM | um | micrometer

## CID 7461 Units of Area Measurement

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.522

Table CID 7461. Units of Area Measurement

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | cm2 | square centimeter
UCUM | mm2 | square millimeter
UCUM | um2 | square micrometer

## CID 7462 Units of Volume Measurement

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.523

Table CID 7462. Units of Volume Measurement

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | dm3 | cubic decimeter
UCUM | cm3 | cubic centimeter
UCUM | mm3 | cubic millimeter
UCUM | um3 | cubic micrometer

### Note

A "cubic decimeter" is a "liter", just as a "cubic centimeter" is a "milliliter" (of water). Though there are specific units "l" and "ml" in UCUM, only one form is included here, since this context group is intended for use for volume measurements of a physical object derived from one or more images, rather than of fluid volume.

## CID 7464 General Region of Interest Measurement Modifiers

This context group contains modifiers of measurements of ROIs.

Type:

  Extensible

Version:

  20121101

UID:

  1.2.840.10008.6.1.951

Table CID 7464. General Region of Interest Measurement Modifiers

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 3488 “Min/Max/Mean” |  |
SRT | R-10047 | Standard Deviation | 386136009 | C0871420
SRT | R-40507 | Total | 255619001 | C0439810
SRT | R-00319 | Median | 373099004 | C1298795
SRT | R-0032E | Mode | 373100007 | C1298796
DCM | 126031 | Peak Value Within ROI |  |
UMLS | C0681921 | Coefficient of Variance |  | C0681921
DCM | 126051 | Skewness |  |
DCM | 126052 | Kurtosis |  |
UMLS | C1711260 | Variance |  | C1711260
UMLS | C2347976 | Root Mean Square |  | C2347976

## CID 7465 Measurements Derived From Multiple ROI Measurements

Type:

  Extensible

Version:

  20121101

UID:

  1.2.840.10008.6.1.952

Table CID 7465. Measurements Derived From Multiple ROI Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 226 “Population Statistical Descriptors”
Include CID 227 “Sample Statistical Descriptors”

## CID 7466 PET Region of Interest Measurements

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.999

Table CID 7466. PET Region of Interest Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126032 | Metabolic Volume
DCM | 126033 | Total Lesion Glycolysis
DCM | 126034 | Glycolysis
DCM | 126035 | Total Lesion Proliferation
DCM | 126036 | Proliferative Activity
DCM | 126037 | Standardized Added Metabolic Activity
DCM | 126038 | Standardized Added Metabolic Activity Background
DCM | 126039 | Lesion to Background SUV Ratio
DCM | 126040 | Background for Lesion to Background SUV Ratio

## CID 7467 Grey Level Co-occurrence Matrix Measurements

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.1000

Table CID 7467. Grey Level Co-occurrence Matrix Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126060 | Entropy of GLCM
DCM | 126061 | Energy of GLCM
DCM | 126062 | Homogeneity of GLCM
DCM | 126063 | Contrast of GLCM
DCM | 126064 | Dissimilarity of GLCM
DCM | 126065 | ASM of GLCM
DCM | 126066 | Correlation of GLCM

## CID 7468 Texture Measurements

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.1001

Table CID 7468. Texture Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7467 “Grey Level Co-occurrence Matrix Measurements”
DCM | 126050 | Fractal Dimension

## CID 7469 Generic Intensity and Size Measurements

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.1003

Table CID 7469. Generic Intensity and Size Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7180 “Abstract Multi-dimensional Image Model Component Semantics”
Include CID 7470 “Linear Measurements”
Include CID 7471 “Area Measurements”
Include CID 7472 “Volume Measurements”

## CID 7470 Linear Measurements

Type:

  Extensible

Version:

  20160314

UID:

  1.2.840.10008.6.1.524

Table CID 7470. Linear Measurements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-D7FE | Length | 410668003 | C1444754
DCM | 121211 | Path length |  |
DCM | 121206 | Distance |  |
SRT | G-A220 | Width | 103355008 | C0487742
SRT | G-D785 | Depth | 131197000 | C0205125
SRT | M-02550 | Diameter | 81827009 | C1301886
SRT | G-A185 | Long Axis | 103339001 | C0522487
SRT | G-A186 | Short Axis | 103340004 | C0522488
SRT | G-A193 | Major Axis | 131187009 | C1295723
SRT | G-A194 | Minor Axis | 131188004 | C1295724
SRT | G-A195 | Perpendicular Axis | 131189007 | C1295725
SRT | G-A196 | Radius | 131190003 | C1306504
SRT | G-A197 | Perimeter | 131191004 | C1295726
SRT | M-02560 | Circumference | 74551000 | C0332520
SRT | G-A198 | Diameter of circumscribed circle | 131192006 | C1295727
DCM | 121207 | Height |  |

## CID 7471 Area Measurements

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.525

Table CID 7471. Area Measurements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A166 | Area | 42798000 | C0205146
SRT | G-A16A | Area of defined region | 131184002 | C1295720

## CID 7472 Volume Measurements

Type:

  Extensible

Version:

  20020904

UID:

  1.2.840.10008.6.1.526

Table CID 7472. Volume Measurements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-D705 | Volume | 118565006 | C0449468
DCM | 121216 | Volume estimated from single 2D region |  |
DCM | 121218 | Volume estimated from two non-coplanar 2D regions |  |
DCM | 121217 | Volume estimated from three or more non-coplanar 2D regions |  |
DCM | 121222 | Volume of sphere |  |
DCM | 121221 | Volume of ellipsoid |  |
DCM | 121220 | Volume of circumscribed sphere |  |
DCM | 121219 | Volume of bounding three dimensional region |  |

## CID 7473 General Area Calculation Methods

Type:

  Extensible

Version:

  20070827

UID:

  1.2.840.10008.6.1.527

Table CID 7473. General Area Calculation Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122501 | Area of closed irregular polygon
DCM | 122502 | Area of a closed NURBS

## CID 7474 General Volume Calculation Methods

Type:

  Extensible

Version:

  20141110

UID:

  1.2.840.10008.6.1.528

Table CID 7474. General Volume Calculation Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122503 | Integration of sum of closed areas on contiguous slices
DCM | 126030 | Sum of segmented voxel volumes

## CID 7480 Breed

Type:

  Extensible

Version:

  20170914

UID:

  1.2.840.10008.6.1.529

Table CID 7480. Breed

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 7486 “Mixed Breeds” |  |
SRT | L-80139 | Hereford cattle superbreed | 125074003 | C0324066
SRT | L-8C338 | Merino sheep superbreed | 125101009 | C1265459
SRT | L-80121 | Africander cattle breed | 131426006 | C1269178
SRT | L-80122 | Ankole cattle breed | 131427002 | C1295943
SRT | L-80123 | Ankole-Watusi cattle breed | 131428007 | C1295944
SRT | L-80124 | Baladicattle cattle breed | 131429004 | C1295945
SRT | L-80125 | Belmont Red cattle breed | 131430009 | C1295946
SRT | L-80126 | Bonsmara cattle breed | 131431008 | C1295947
SRT | L-80127 | Damietta cattle breed | 131432001 | C1295948
SRT | L-80128 | Horro cattle breed | 131433006 | C1295949
SRT | L-80129 | Kuri cattle breed | 131434000 | C1295950
SRT | L-8012A | Nguni cattle breed | 131435004 | C1295951
SRT | L-8012B | Philippine Native cattle breed | 131436003 | C1269179
SRT | L-8012C | Romagnola cattle breed | 131437007 | C1295952
SRT | L-8012E | Sanhe cattle breed | 131438002 | C1295953
SRT | L-8012F | Tswana cattle breed | 131439005 | C1295954
SRT | L-80138 | Tuli cattle breed | 131440007 | C1295955
SRT | L-8013A | Aliab Dinka cattle breed | 131441006 | C1295956
SRT | L-8013B | Alur cattle breed | 131442004 | C1295957
SRT | L-8013C | Ankina cattle breed | 131443009 | C1295958
SRT | L-8013D | Apulian Podolian cattle breed | 131444003 | C1295959
SRT | L-8013E | Arado cattle breed | 131445002 | C1269180
SRT | L-8013F | Aweil Dinka cattle breed | 131446001 | C1295960
SRT | L-8014C | Bahima cattle breed | 131447005 | C1295961
SRT | L-8014D | Bapedi cattle breed | 131448000 | C1295962
SRT | L-8014E | Baria (Vietnam/Madagascar) cattle breed | 131449008 | C1295963
SRT | L-8014F | Barotse cattle breed | 131450008 | C1295964
SRT | L-8015A | Barra do Cuanzo cattle breed | 131451007 | C1295965
SRT | L-8015B | Bashi cattle breed | 131452000 | C1295966
SRT | L-8015C | Basuto cattle breed | 131453005 | C1295967
SRT | L-8015D | Batangas cattle breed | 131454004 | C1295968
SRT | L-8015E | Bavenda cattle breed | 131455003 | C1295969
SRT | L-8015F | Beja cattle breed | 131456002 | C1295970
SRT | L-80161 | Calabrian cattle breed | 131457006 | C1295971
SRT | L-80162 | Blonde-du Cap Bon cattle breed | 131458001 | C1295972
SRT | L-80163 | Cham-Doc cattle breed | 131459009 | C1295973
SRT | L-80164 | Chernigov cattle breed | 131460004 | C1295974
SRT | L-80165 | Chino Santandereano cattle breed | 131461000 | C1295975
SRT | L-80166 | Cinisara cattle breed | 131462007 | C1295976
SRT | L-80167 | Cuprem Hybrid cattle breed | 131463002 | C1295977
SRT | L-80168 | Dabieshan cattle breed | 131464008 | C1295978
SRT | L-80169 | Damara cattle breed | 131465009 | C1295979
SRT | L-8016A | Danakil cattle breed | 131466005 | C1295980
SRT | L-8016B | Dnieper cattle breed | 131467001 | C1295981
SRT | L-8016C | Doayo cattle breed | 131468006 | C1295982
SRT | L-8016D | Eastern Nuer cattle breed | 131469003 | C1269181
SRT | L-8016E | Egyptian cattle breed | 131470002 | C1295983
SRT | L-8016F | Fogera cattle breed | 131471003 | C1295984
SRT | L-80177 | Garfagnina cattle breed | 131472005 | C1295985
SRT | L-80178 | Grati cattle breed | 131473000 | C1295986
SRT | L-80179 | Gaunling cattle breed | 131474006 | C1295987
SRT | L-8017A | Halhin Gol cattle breed | 131475007 | C1295988
SRT | L-8017B | Holmonger cattle breed | 131476008 | C1295989
SRT | L-8017C | Ilocos cattle breed | 131477004 | C1295990
SRT | L-8017D | Iloilo cattle breed | 131478009 | C1295991
SRT | L-8017E | Inkuku cattle breed | 131479001 | C1295992
SRT | L-8017F | Iskar cattle breed | 131480003 | C1295993
SRT | L-80180 | Istrian cattle breed | 131481004 | C1295994
SRT | L-80181 | Javanese Ongole cattle breed | 131482006 | C1269182
SRT | L-80182 | Javanese Zebu cattle breed | 131483001 | C1269183
SRT | L-80183 | Jinnan cattle breed | 131484007 | C1295995
SRT | L-80184 | Kalmyk cattle breed | 131485008 | C1295996
SRT | L-80185 | Kaokoveld cattle breed | 131486009 | C1295997
SRT | L-80186 | Kazakh Whitehead cattle breed | 131487000 | C1295998
SRT | L-80187 | Kedah-Kelantan cattle breed | 131488005 | C1295999
SRT | L-80188 | Kigezi cattle breed | 131489002 | C1296000
SRT | L-80189 | Kisantu cattle breed | 131490006 | C1296001
SRT | L-8018A | Kolubara cattle breed | 131491005 | C1296002
SRT | L-8018B | Kurgan cattle breed | 131492003 | C1296003
SRT | L-8018C | Kyoga cattle breed | 131493008 | C1296004
SRT | L-8018D | Lucanian cattle breed | 131494002 | C1296005
SRT | L-8018E | Maremmana cattle breed | 131495001 | C1296006
SRT | L-8018F | Marianas cattle breed | 131496000 | C1296007
SRT | L-80190 | Maryuti cattle breed | 131497009 | C1296008
SRT | L-80191 | Mauritius Creole cattle breed | 131498004 | C1296009
SRT | L-80192 | Menufi cattle breed | 131499007 | C1296010
SRT | L-80193 | Mezzalina cattle breed | 131500003 | C1296011
SRT | L-80194 | Modicana cattle breed | 131501004 | C1296012
SRT | L-80195 | Moi cattle breed | 131502006 | C1296013
SRT | L-80196 | Nama cattle breed | 131503001 | C1296014
SRT | L-80197 | Nanyang cattle breed | 131504007 | C1296015
SRT | L-80198 | N'Dama Sanga cattle breed | 131505008 | C1296016
SRT | L-80199 | Nganda cattle breed | 131506009 | C1296017
SRT | L-8019A | Nilotic Sanga cattle breed | 131507000 | C1296018
SRT | L-8019B | Nkone cattle breed | 131508005 | C1296019
SRT | L-8019C | North Malawi Angoni cattle breed | 131509002 | C1269184
SRT | L-8019D | Nuer cattle breed | 131510007 | C1296020
SRT | L-8019E | Nuras cattle breed | 131511006 | C1296021
SRT | L-8019F | Nyoro cattle breed | 131512004 | C1296022
SRT | L-801A0 | Ovambo cattle breed | 131513009 | C1296023
SRT | L-801A1 | Pantelleria cattle breed | 131514003 | C1296024
SRT | L-801A2 | Pinzhou cattle breed | 131515002 | C1296025
SRT | L-801A3 | Porto Amboim cattle breed | 131516001 | C1296026
SRT | L-801A4 | Posavina cattle breed | 131517005 | C1296027
SRT | L-801A5 | Romanian Steppe cattle breed | 131518000 | C1269185
SRT | L-801A6 | Saidi cattle breed | 131519008 | C1296028
SRT | L-801A7 | Sardo-Modicana cattle breed | 131520002 | C1296029
SRT | L-801A8 | Sengologa cattle breed | 131521003 | C1296030
SRT | L-801A9 | Serere cattle breed | 131522005 | C1296031
SRT | L-801AA | Seshaga cattle breed | 131523000 | C1296032
SRT | L-801AB | Siberian Black Pied cattle breed | 131524006 | C1269186
SRT | L-801AC | Socotra cattle breed | 131525007 | C1296033
SRT | L-801AD | Southern Tswana cattle breed | 131526008 | C1269187
SRT | L-801AE | Spreca cattle breed | 131527004 | C1296034
SRT | L-801AF | Sunkuma cattle breed | 131528009 | C1296035
SRT | L-801B0 | Taiwan Zebu cattle breed | 131529001 | C1269188
SRT | L-801B1 | Thai cattle breed | 131530006 | C1296036
SRT | L-801B2 | Thailand Fighting Zebu cattle breed | 131531005 | C1269189
SRT | L-801B3 | Thanh-Hoa cattle breed | 131532003 | C1296037
SRT | L-801B4 | Tibetan cattle breed | 131533008 | C1296038
SRT | L-801B5 | Tonga cattle breed | 131534002 | C1296039
SRT | L-801B6 | Toro cattle breed | 131535001 | C1269190
SRT | L-801B7 | Tuni cattle breed | 131536000 | C1296040
SRT | L-801B8 | Turkish Gray Steppe cattle breed | 131537009 | C1269191
SRT | L-801B9 | Tuy-Hoa cattle breed | 131538004 | C1296041
SRT | L-801BA | Ujumqin cattle breed | 131539007 | C1296042
SRT | L-801BB | Abigar cattle breed | 131540009 | C1296043
SRT | L-801BC | Africangnus cattle breed | 131541008 | C1269101
SRT | L-801BD | Agerolese cattle breed | 131542001 | C1269102
SRT | L-801BE | Albese cattle breed | 131543006 | C1269103
SRT | L-801BF | Ukrainian Gray cattle breed | 131544000 | C1269104
SRT | L-801C0 | Vietnamese Yellow cattle breed | 131545004 | C1269105
SRT | L-801C1 | Watusi (USA) cattle breed | 131546003 | C1296044
SRT | L-801C2 | Wenshan cattle breed | 131547007 | C1296045
SRT | L-801C3 | Yakut cattle breed | 131548002 | C1296046
SRT | L-801C4 | Yunnan Zebu cattle breed | 131549005 | C1269106
SRT | L-801C5 | Zambia Angoni cattle breed | 131550005 | C1269107
SRT | L-801C6 | Drakensberger cattle breed | 131551009 | C1296047
SRT | L-801C7 | Modicana lowland cattle breed | 131552002 | C1269108
SRT | L-801C8 | Taiwan Yellow cattle breed | 131553007 | C1269109
SRT | L-801C9 | Menggu cattle breed | 131554001 | C1296048
SRT | L-801CA | Albères cattlebreed | 131555000 | C1321436
SRT | L-801CB | Alentejana cattlebreed | 131556004 | C1296049
SRT | L-801CC | American White Park cattle breed | 131557008 | C1269110
SRT | L-801CD | Amerifaxcattle breed | 131558003 | C1296050
SRT | L-801CE | Anatolian Black cattle breed | 131559006 | C1269111
SRT | L-801CF | Andalusian Black cattle breed | 131560001 | C1269112
SRT | L-801D0 | Andalusian Gray cattle breed | 131561002 | C1269113
SRT | L-801D1 | Angeln cattle breed | 131562009 | C1296051
SRT | L-801D2 | Asturian Mountain cattle breed | 131563004 | C1269114
SRT | L-801D3 | Asturian Valley cattle breed | 131564005 | C1269115
SRT | L-801D4 | Aubrac cattle breed | 131565006 | C1296052
SRT | L-801D5 | Aulie-Ata cattle breed | 131566007 | C1296053
SRT | L-801D6 | Australian Lowline cattle breed | 131567003 | C1269116
SRT | L-801D7 | Barzona cattle breed | 131568008 | C1296054
SRT | L-801D8 | Bazadais cattle breed | 131569000 | C1296055
SRT | L-801D9 | Beefmaker cattle breed | 131570004 | C1269117
SRT | L-801DA | Belarus Red cattle breed | 131571000 | C1269118
SRT | L-801DB | Belgian Blue cattle breed | 131572007 | C1269119
SRT | L-801DC | Belgian Red cattle breed | 131573002 | C1269120
SRT | L-801DD | Belmont Adaptaur cattle breed | 131574008 | C1269121
SRT | L-801DE | Berrendas cattle breed | 131575009 | C1269122
SRT | L-801DF | Blacksided Trondheim and Norland cattle breed | 131576005 | C1269123
SRT | L-801E0 | Blanco Orejinegro cattle breed | 131577001 | C1296056
SRT | L-801E1 | Braunvieh cattle breed | 131578006 | C1296057
SRT | L-801E2 | British White cattle breed | 131579003 | C1269124
SRT | L-801E3 | Cachena cattle breed | 131580000 | C1296058
SRT | L-801E4 | Canary Island cattle breed | 131581001 | C1269125
SRT | L-801E5 | Carinthian Blond cattle breed | 131582008 | C1269126
SRT | L-801E6 | Caucasian cattle breed | 131583003 | C1269127
SRT | L-801E7 | Charolais cattle breed | 131584009 | C1296059
SRT | L-801EA | Chinese Black-and-White cattle breed | 131585005 | C1269128
SRT | L-801EB | Corriente cattle breed | 131586006 | C1269129
SRT | L-801EC | Costeño con Cuernos cattle breed | 131587002 | C1321437
SRT | L-801ED | Damascus cattle breed | 131588007 | C1269130
SRT | L-801EE | Danish Red cattle breed | 131589004 | C1269131
SRT | L-801EF | Devon cattle breed | 131590008 | C0175926
SRT | L-801F0 | Dølafe cattle breed | 131591007 | C1321438
SRT | L-801F1 | Dutch Belted cattle breed | 131592000 | C1269132
SRT | L-801F2 | Dutch Friesian cattle breed | 131593005 | C1269133
SRT | L-801F3 | English Longhorn cattle breed | 131594004 | C1269134
SRT | L-801F4 | Estonian Red cattle breed | 131595003 | C1269135
SRT | L-801F5 | Evolène cattle breed | 131596002 | C1321439
SRT | L-801F6 | Fighting Bull cattle breed | 131597006 | C1269136
SRT | L-801F7 | Fjall cattle breed | 131598001 | C1296060
SRT | L-801F8 | Florida Cracker/Pineywoods cattle breed | 131599009 | C1269137
SRT | L-801F9 | Galician Blond cattle breed | 131600007 | C1269138
SRT | L-801FA | Gascon cattle breed | 131601006 | C1269139
SRT | L-801FB | German Red Pied cattle breed | 131602004 | C1269140
SRT | L-801FC | Glan cattle breed | 131603009 | C1296061
SRT | L-801FD | Gloucester cattle breed | 131604003 | C1296062
SRT | L-801FE | Groningen Whiteheaded cattle breed | 131605002 | C1296063
SRT | L-801FF | Hartón cattle breed | 131606001 | C1321440
SRT | L-8031A | Bündner Oberland sheep breed | 131699001 | C1321446
SRT | L-8031B | British Milk Sheep breed | 131700000 | C1296127
SRT | L-8031C | Brillenschaf sheep breed | 131701001 | C1296128
SRT | L-8031D | Brecknock Hill Cheviot sheep breed | 131702008 | C1296129
SRT | L-8031E | Cholistani sheep breed | 131703003 | C1296130
SRT | L-8031F | Bibrik sheep breed | 131704009 | C1296131
SRT | L-8032A | Columbia sheep breed | 131705005 | C1296132
SRT | L-8032B | Black Welsh Mountain Sheep breed | 131706006 | C1269165
SRT | L-8032C | Blackhead Persian sheep breed | 131707002 | C1269166
SRT | L-8032D | Bleu du Maine sheep breed | 131708007 | C1296133
SRT | L-8032E | Bluefaced Leicester sheep breed | 131709004 | C1269167
SRT | L-8032F | Bond sheep breed | 131710009 | C1296134
SRT | L-8033A | Border Leicester sheep breed | 131711008 | C1269168
SRT | L-8033B | Boreray sheep breed | 131712001 | C1296135
SRT | L-8033C | Bovska sheep breed | 131713006 | C1296136
SRT | L-8033D | Braunes Bergschaf sheep breed | 131714000 | C1296137
SRT | L-8033E | Brazilian Somali sheep breed | 131715004 | C1269169
SRT | L-8033F | Beulah Speckled-Face sheep breed | 131716003 | C1269170
SRT | L-8034A | Dartmoor sheep breed | 131717007 | C1296138
SRT | L-8034B | Fabrianese sheep breed | 131718002 | C1269171
SRT | L-8034C | Exmoor Horn sheep breed | 131719005 | C1296139
SRT | L-8034D | Elliottdale sheep breed | 131720004 | C1296140
SRT | L-8034E | Drysdale sheep breed | 131721000 | C1296141
SRT | L-8034F | Dorset Down sheep breed | 131722007 | C1296142
SRT | L-80351 | German Blackheaded Mutton sheep breed | 131723002 | C1296143
SRT | L-80352 | Kooka sheep breed | 131724008 | C1296144
SRT | L-80353 | Friesian Milk Sheep breed | 131725009 | C1296145
SRT | L-80354 | Gansu Alpine Fine-wool sheep breed | 131726005 | C1296146
SRT | L-80355 | German Whiteheaded Mutton sheep breed | 131727001 | C1296147
SRT | L-80356 | Graue Gehoernte Heidschnucke sheep breed | 131728006 | C1296148
SRT | L-80357 | Han sheep breed | 131729003 | C1296149
SRT | L-80358 | Gromark sheep breed | 131730008 | C1296150
SRT | L-80359 | Gulf Coast Native sheep breed | 131731007 | C1296151
SRT | L-8035A | Dorper sheep breed | 131732000 | C1296152
SRT | L-8035B | Devon Closewool sheep breed | 131733005 | C1296153
SRT | L-8035C | Deutsches Blaukoepfiges Fleischschaf sheep breed | 131734004 | C1296154
SRT | L-8035D | Derbyshire Gritstone sheep breed | 131735003 | C1296155
SRT | L-8035E | Coburger Fuchsschaf sheep breed | 131736002 | C1296156
SRT | L-8035F | Danish Landrace sheep breed | 131737006 | C1296157
SRT | L-80360 | Gute sheep breed | 131738001 | C1296158
SRT | L-80361 | Hampshire sheep breed | 131739009 | C1296159
SRT | L-80362 | Gentile di Puglia sheep breed | 131740006 | C1296160
SRT | L-80363 | German Mountain sheep breed | 131741005 | C1296161
SRT | L-80364 | Luzein sheep breed | 131742003 | C1296162
SRT | L-80365 | Katahdin sheep breed | 131743008 | C1296163
SRT | L-80366 | Leineschaf sheep breed | 131744002 | C1296164
SRT | L-80367 | Lincoln Longwool sheep breed | 131745001 | C1296165
SRT | L-80368 | Llanwenog sheep breed | 131746000 | C1296166
SRT | L-80369 | Lleyn sheep breed | 131747009 | C1296167
SRT | L-8036A | Damara sheep breed | 131748004 | C1296168
SRT | L-8036B | Damani sheep breed | 131749007 | C1296169
SRT | L-8036C | Dalesbred sheep breed | 131750007 | C1296170
SRT | L-8036D | Dala sheep breed | 131751006 | C1296171
SRT | L-8036E | Criollo sheep breed | 131752004 | C1296172
SRT | L-8036F | Cormo sheep breed | 131753009 | C1296173
SRT | L-80370 | Lati sheep breed | 131754003 | C1296174
SRT | L-80371 | Lonk sheep breed | 131755002 | C1296175
SRT | L-80372 | Langhe sheep breed | 131756001 | C1296176
SRT | L-80373 | Manx Loaghtan sheep breed | 131757005 | C1296177
SRT | L-80374 | Masai sheep breed | 131758000 | C1296178
SRT | L-80375 | Merinolandschaf sheep breed | 131759008 | C1296179
SRT | L-80376 | Lohi sheep breed | 131760003 | C1296180
SRT | L-80377 | Ile-de-France sheep breed | 131761004 | C1296181
SRT | L-80378 | Hasht Nagri sheep breed | 131762006 | C1296182
SRT | L-80379 | Hazaragie sheep breed | 131763001 | C1296183
SRT | L-8037A | Coopworth sheep breed | 131764007 | C1296184
SRT | L-8037B | Comisana sheep breed | 131765008 | C1296185
SRT | L-8037C | Comeback sheep breed | 131766009 | C1296186
SRT | L-8037D | Sicilian Barbary sheep breed | 131767000 | C1296187
SRT | L-8037E | Africana sheep breed | 131768005 | C1296188
SRT | L-8037F | Welsh Mountain Badger Faced sheep breed | 131769002 | C1296189
SRT | L-80380 | Hebridean sheep breed | 131770001 | C1296190
SRT | L-80381 | Heidschnucke sheep breed | 131771002 | C1296191
SRT | L-80382 | Herdwick sheep breed | 131772009 | C1296192
SRT | L-80383 | Hill Radnor sheep breed | 131773004 | C1296193
SRT | L-80384 | Icelandic sheep breed | 131774005 | C1296194
SRT | L-80385 | Harnai sheep breed | 131775006 | C1296195
SRT | L-80386 | Istrian Pramenka sheep breed | 131776007 | C1296196
SRT | L-80387 | Jacob sheep breed | 131777003 | C1296197
SRT | L-80388 | Jezerskosolcavska sheep breed | 131778008 | C1296198
SRT | L-80389 | Kachhi sheep breed | 131779000 | C1296199
SRT | L-8038A | Wensleydale sheep breed | 131780002 | C1296200
SRT | L-8038B | West African Dwarf sheep breed | 131781003 | C1296201
SRT | L-8038C | White Suffolk sheep breed | 131782005 | C1296202
SRT | L-8038D | Whiteface Dartmoor sheep breed | 131783000 | C1296203
SRT | L-8038E | Whiteface Woodland sheep breed | 131784006 | C1296204
SRT | L-8038F | Xinjiang Finewool sheep breed | 131785007 | C1296205
SRT | L-80390 | Kajli sheep breed | 131786008 | C1296206
SRT | L-80391 | Hog Island Sheep breed | 131787004 | C1296207
SRT | L-80392 | Biellese sheep breed | 131788009 | C1296208
SRT | L-80393 | Chios sheep breed | 131789001 | C1296209
SRT | L-80394 | Santa Cruz sheep breed | 131790005 | C1296210
SRT | L-80395 | Charollais sheep breed | 131791009 | C1296211
SRT | L-80396 | Castlemilk Moorit sheep breed | 131792002 | C1296212
SRT | L-80397 | Campanian Barbary sheep breed | 131793007 | C1296213
SRT | L-80398 | California Variegated Mutant sheep breed | 131794001 | C1296214
SRT | L-80399 | California Red sheep breed | 131795000 | C1296215
SRT | L-8039A | Sopravissana sheep breed | 131796004 | C1296216
SRT | L-8039B | Somali sheep breed | 131797008 | C1296217
SRT | L-8039C | Welsh Hill Speckled Face sheep breed | 131798003 | C1296218
SRT | L-8039D | Skudde sheep breed | 131799006 | C1296219
SRT | L-8039E | Waziri sheep breed | 131800005 | C1296220
SRT | L-8039F | Shetland sheep breed | 131801009 | C1296221
SRT | L-80403 | Cambridge sheep breed | 131802002 | C1296222
SRT | L-80404 | Solognote sheep breed | 131803007 | C1296223
SRT | L-8040A | Colombian Criollo horse breed | 131804001 | C1296224
SRT | L-8040B | Comtois horse breed | 131805000 | C1296225
SRT | L-8040C | Corsican horse breed | 131806004 | C1296226
SRT | L-8040D | Costa Rican Saddle Horse horse breed | 131807008 | C1296227
SRT | L-8040E | Costeno horse breed | 131808003 | C1296228
SRT | L-8040F | Cuban Paso horse breed | 131809006 | C1296229
SRT | L-80420 | Rough Fell sheep breed | 131816007 | C1296236
SRT | L-8042D | Danish Warmblood horse breed | 131819000 | C1296239
SRT | L-80432 | Swaledale sheep breed | 131822003 | C1296242
SRT | L-80434 | Polypay sheep breed | 131823008 | C1296243
SRT | L-80441 | Pagliarola sheep breed | 131830002 | C1296250
SRT | L-80442 | Pomeranian Coarsewool sheep breed | 131831003 | C1296251
SRT | L-80443 | Sheep, Breed Undetermined sheep breed | 131832005 | C1296252
SRT | L-80444 | Orkney sheep breed | 131833000 | C1296253
SRT | L-80445 | Old Norwegian sheep breed | 131834006 | C1296254
SRT | L-80446 | Old Format Sheep breed | 131835007 | C1296255
SRT | L-80447 | Norwegian Fur sheep breed | 131836008 | C1296256
SRT | L-80448 | Norfolk Horn sheep breed | 131837004 | C1296257
SRT | L-80449 | Navajo-Churro sheep breed | 131838009 | C1296258
SRT | L-80466 | Racka sheep breed | 131851004 | C1296270
SRT | L-80467 | Rasa Aragonesa sheep breed | 131852006 | C1296271
SRT | L-80468 | Red Engadine sheep breed | 131853001 | C1296272
SRT | L-80469 | Rhoenschaf sheep breed | 131854007 | C1296273
SRT | L-80470 | Hucul horse breed | 131861006 | C1296279
SRT | L-80471 | AraAppaloosa horse breed | 131862004 | C1296280
SRT | L-80472 | Argentine Criollo horse breed | 131863009 | C1296281
SRT | L-80473 | Argentine Polo Pony horse breed | 131864003 | C1296282
SRT | L-80474 | Australian Pony horse breed | 131865002 | C1296283
SRT | L-80475 | Auxois horse breed | 131866001 | C1296284
SRT | L-80476 | Avelignese horse breed | 131867005 | C1296285
SRT | L-80477 | Azerbaijan horse breed | 131868000 | C1296286
SRT | L-80478 | Azores horse breed | 131869008 | C1296287
SRT | L-80479 | Bali horse breed | 131870009 | C1296288
SRT | L-8047A | Balikun horse breed | 131871008 | C1296289
SRT | L-8047B | Waziri horse breed | 131872001 | C1296290
SRT | L-8047C | Banker Horse horse breed | 131873006 | C1296291
SRT | L-8047D | Bardigiano horse breed | 131874000 | C1296292
SRT | L-8047E | Batak horse breed | 131875004 | C1296293
SRT | L-8047F | Bavarian Warmblood horse breed | 131876003 | C1296294
SRT | L-80480 | Belgian Ardennais horse breed | 131877007 | C1296295
SRT | L-80481 | Belgian Halfblood horse breed | 131878002 | C1296296
SRT | L-80482 | Belgian Warmblood horse breed | 131879005 | C1296297
SRT | L-80483 | Bhutia horse breed | 131880008 | C1296298
SRT | L-80484 | Black Sea Horse horse breed | 131881007 | C1296299
SRT | L-80485 | Bosnian horse breed | 131882000 | C1296300
SRT | L-80486 | Boulonnais horse breed | 131883005 | C1296301
SRT | L-80487 | Brandenburg horse breed | 131884004 | C1296302
SRT | L-80488 | Brazilian Sport Horse horse breed | 131885003 | C1296303
SRT | L-80489 | British Appaloosa horse breed | 131886002 | C1296304
SRT | L-8048A | British Riding Pony horse breed | 131887006 | C1296305
SRT | L-8048B | British Spotted Pony horse breed | 131888001 | C1296306
SRT | L-8048C | Buohai horse breed | 131889009 | C1296307
SRT | L-8048D | Buryat horse breed | 131890000 | C1296308
SRT | L-8048E | Calabrian horse breed | 131891001 | C1296309
SRT | L-8048F | Camargue horse breed | 131892008 | C1320152
SRT | L-80490 | Canadian Cutting Horse horse breed | 131893003 | C1296310
SRT | L-80491 | Canadian Rustic Pony horse breed | 131894009 | C1296311
SRT | L-80492 | Canadian Sport Horse horse breed | 131895005 | C1296312
SRT | L-80493 | Canik horse breed | 131896006 | C1296313
SRT | L-80494 | Cape Horse horse breed | 131897002 | C1296314
SRT | L-80496 | Cerbat horse breed | 131898007 | C1296315
SRT | L-80497 | Chakouyi horse breed | 131899004 | C1296316
SRT | L-80498 | Chara Horse horse breed | 131900009 | C1296317
SRT | L-80499 | Chickasaw horse breed | 131901008 | C1296318
SRT | L-8049A | Chilote horse breed | 131902001 | C1296319
SRT | L-8049B | Chinese Kazakh horse breed | 131903006 | C1296320
SRT | L-8049C | Chinese Mongolian horse breed | 131904000 | C1296321
SRT | L-8049D | Chumbivilcas horse breed | 131905004 | C1296322
SRT | L-8049E | Chumysh horse breed | 131906003 | C1296323
SRT | L-8049F | Cirit horse breed | 131907007 | C1296324
SRT | L-804A1 | Irish Draft horse breed | 131908002 | C1296325
SRT | L-804A2 | Irish Hunter horse breed | 131909005 | C1296326
SRT | L-804A3 | Cuban Trotter horse breed | 131910000 | C1296327
SRT | L-804A4 | Italian Heavy Draft horse breed | 131911001 | C1296328
SRT | L-804A5 | Jabe horse breed | 131912008 | C1296329
SRT | L-804A6 | Java horse breed | 131913003 | C1296330
SRT | L-804A7 | Vendéen sheep breed | 131914009 | C1321447
SRT | L-804A8 | Czech Warmblood horse breed | 131915005 | C1296331
SRT | L-804A9 | Jinhong horse breed | 131916006 | C1296332
SRT | L-804AA | Jinzhou horse breed | 131917002 | C1296333
SRT | L-804AC | Danubian horse breed | 131919004 | C1296335
SRT | L-804AD | Karachai horse breed | 131920005 | C1296336
SRT | L-804AE | Karakacan horse breed | 131921009 | C1296337
SRT | L-804AF | Kathiawari horse breed | 131922002 | C1296338
SRT | L-804B1 | Ke-Er-Qin horse breed | 131923007 | C1296339
SRT | L-804B2 | Kirgiz horse breed | 131924001 | C1296340
SRT | L-804B3 | Kuznet horse breed | 131925000 | C1296341
SRT | L-804B4 | Landais horse breed | 131926004 | C1296342
SRT | L-804B5 | Lewitzer horse breed | 131927008 | C1296343
SRT | L-804B6 | Lichuan horse breed | 131928003 | C1296344
SRT | L-804B7 | Lijiang horse breed | 131929006 | C1296345
SRT | L-804B8 | Llanero horse breed | 131930001 | C1296346
SRT | L-804B9 | Lombok horse breed | 131931002 | C1296347
SRT | L-804BA | Lundy Pony horse breed | 131932009 | C1296348
SRT | L-804BB | Malakan horse breed | 131933004 | C1296349
SRT | L-804BC | Malopolski horse breed | 131934005 | C1296350
SRT | L-804BD | Datong horse breed | 131935006 | C1296351
SRT | L-804BE | Mangalarga Paulista horse breed | 131936007 | C1296352
SRT | L-804BF | Dulmen Pony horse breed | 131937003 | C1296353
SRT | L-804C1 | Maremmana horse breed | 131938008 | C1296354
SRT | L-804C2 | Marwari horse breed | 131939000 | C1296355
SRT | L-804C3 | Megezh horse breed | 131940003 | C1296356
SRT | L-804C4 | Megrel horse breed | 131941004 | C1296357
SRT | L-804C5 | Merens horse breed | 131942006 | C1296358
SRT | L-804C6 | Messara horse breed | 131943001 | C1296359
SRT | L-804C7 | Sumba horse breed | 131944007 | C1296360
SRT | L-804C8 | Sumbawa horse breed | 131945008 | C1296361
SRT | L-804C9 | Swedish Ardennes horse breed | 131946009 | C1296362
SRT | L-804CA | Dutch Tuigpaard horse breed | 131947000 | C1296363
SRT | L-804CB | East and Southeast Anadolu horse breed | 131948005 | C1296364
SRT | L-804CC | Thai Pony horse breed | 131949002 | C1296365
SRT | L-804CD | Thessalonian horse breed | 131950002 | C1296366
SRT | L-804CE | Tibetan horse breed | 131951003 | C1296367
SRT | L-804CF | Tieling horse breed | 131952005 | C1296368
SRT | L-804D1 | Timor horse breed | 131953000 | C1296369
SRT | L-804D2 | Trakya horse breed | 131954006 | C1296370
SRT | L-804D3 | Trote en Gallope horse breed | 131955007 | C1296371
SRT | L-804D4 | Turkoman horse breed | 131956008 | C1296372
SRT | L-804D5 | Tushin horse breed | 131957004 | C1296373
SRT | L-804D6 | Tuva horse breed | 131958009 | C1296374
SRT | L-804D7 | Uzunyayla horse breed | 131959001 | C1296375
SRT | L-804D9 | Voronezh Coach Horse horse breed | 131960006 | C1296376
SRT | L-804DA | Elegant Warmblood horse breed | 131961005 | C1296377
SRT | L-804DB | Welsh Cob horse breed | 131962003 | C1296378
SRT | L-804DC | Welsh Mountain Pony horse breed | 131963008 | C1296379
SRT | L-804DE | English Hack horse breed | 131964002 | C1296380
SRT | L-804DF | Wurttemberg horse breed | 131965001 | C1296381
SRT | L-804E1 | Xilingol horse breed | 131966000 | C1296382
SRT | L-804E2 | Yanqi horse breed | 131967009 | C1296383
SRT | L-804E3 | Yemeni Horses horse breed | 131968004 | C1296384
SRT | L-804E4 | Yili horse breed | 131969007 | C1296385
SRT | L-804E5 | Yiwu horse breed | 131970008 | C1296386
SRT | L-804E6 | Yunnan horse breed | 131971007 | C1296387
SRT | L-804E7 | German Riding Pony horse breed | 131972000 | C1296388
SRT | L-804E8 | Guanzhong horse breed | 131973005 | C1296389
SRT | L-804E9 | Guizhou horse breed | 131974004 | C1296390
SRT | L-804EA | Guoxia horse breed | 131975003 | C1296391
SRT | L-804EB | Erlunchun horse breed | 131976002 | C1296392
SRT | L-804EC | Half Saddlebred horse breed | 131977006 | C1296393
SRT | L-804ED | Flores horse breed | 131978001 | C1296394
SRT | L-804EE | Freiberg horse breed | 131979009 | C1296395
SRT | L-804EF | Hessen horse breed | 131980007 | C1296396
SRT | L-804F1 | Hinis horse breed | 131981006 | C1296397
SRT | L-804F2 | Hirzai horse breed | 131982004 | C1296398
SRT | L-804F3 | Hungairan Coldblood horse breed | 131983009 | C1296399
SRT | L-804F4 | Hungarian Dun horse breed | 131984003 | C1296400
SRT | L-804F5 | Hungarian Sport Horse horse breed | 131985002 | C1296401
SRT | L-804F6 | International Striped Horse horse breed | 131986001 | C1296402
SRT | L-804F7 | Irish Cob horse breed | 131987005 | C1296403
SRT | L-804F8 | Mezen horse breed | 131988000 | C1296404
SRT | L-804F9 | Mezohegyes Sport Horse horse breed | 131989008 | C1296405
SRT | L-804FA | French Cob horse breed | 131990004 | C1296406
SRT | L-804FB | French Saddle pony horse breed | 131991000 | C1296407
SRT | L-804FC | Murakoz horse breed | 131992007 | C1296408
SRT | L-804FE | Finnhorse Draft horse breed | 131993002 | C1296409
SRT | L-804FF | Mecklenburg horse breed | 131994008 | C1296410
SRT | L-80504 | Catalana chicken breed | 131998006 | C1296414
SRT | L-80542 | Haiti Creole pig breed | 132009005 | C1296425
SRT | L-80543 | Manor Hybrid pig breed | 132010000 | C1296426
SRT | L-80544 | Hamline pig breed | 132011001 | C1296427
SRT | L-80545 | Manor Ranger pig breed | 132012008 | C1296428
SRT | L-80546 | Manor Meishan pig breed | 132013003 | C1296429
SRT | L-80547 | Cotswold Gold pig breed | 132014009 | C1296430
SRT | L-80548 | Cotswold Platinum pig breed | 132015005 | C1296431
SRT | L-80549 | Cotswold 16 pig breed | 132016006 | C1296432
SRT | L-8054A | Cotswold 29 pig breed | 132017002 | C1296433
SRT | L-8054B | Cotswold 90 pig breed | 132018007 | C1296434
SRT | L-8054C | Hampen pig breed | 132019004 | C1296435
SRT | L-8054D | SPM pig breed | 132020005 | C1296436
SRT | L-8054E | High Conformation White pig breed | 132021009 | C1296437
SRT | L-8054F | Line 32 pig breed | 132022002 | C1296438
SRT | L-80555 | Line 21 pig breed | 132023007 | C1296439
SRT | L-80556 | Meatline pig breed | 132024001 | C1296440
SRT | L-80557 | Hampline pig breed | 132025000 | C1296441
SRT | L-80558 | Euroline pig breed | 132026004 | C1296442
SRT | L-80559 | Norline pig breed | 132027008 | C1296443
SRT | L-8055A | Premier pig breed | 132028003 | C1296444
SRT | L-8055B | Tribred pig breed | 132029006 | C1296445
SRT | L-8055C | American Essex pig breed | 132030001 | C1296446
SRT | L-8055D | Sino-Gascony pig breed | 132031002 | C1296447
SRT | L-8055E | Guadeloupe Creole pig breed | 132032009 | C1296448
SRT | L-8055F | Managra pig breed | 132033004 | C1296449
SRT | L-8056A | Canadian Landrace pig breed | 132034005 | C1296450
SRT | L-8056B | Canadian Yorkshire pig breed | 132035006 | C1296451
SRT | L-8056C | Minnesota #4 pig breed | 132036007 | C0324271
SRT | L-8056D | Pineywoods pig breed | 132037003 | C1296453
SRT | L-8056E | Catalina Island pig breed | 132038008 | C1296454
SRT | L-8056F | Ras-n-Lansa pig breed | 132039000 | C1296455
SRT | L-8057B | Pitman-Moore Miniature pig breed | 132040003 | C1296456
SRT | L-8057C | Vita Vet Lab Minipig pig breed | 132041004 | C1296457
SRT | L-8057D | Hanford Miniature pig breed | 132042006 | C1296458
SRT | L-8057E | Black Hampshire pig breed | 132043001 | C1296459
SRT | L-8057F | Red Hamprace pig breed | 132044007 | C1269195
SRT | L-80583 | American Yorkshire pig breed | 132045008 | C1269196
SRT | L-80584 | American Berkshire pig breed | 132046009 | C1269197
SRT | L-80585 | Camborough Blue pig breed | 132047000 | C1269198
SRT | L-80586 | Camborough 12 pig breed | 132048005 | C1296460
SRT | L-80587 | Westrain pig breed | 132049002 | C1296461
SRT | L-80588 | Dalland 030 pig breed | 132050002 | C1296462
SRT | L-80589 | Razor-Back pig breed | 132051003 | C1296463
SRT | L-8058A | Macau pig breed | 132052005 | C1296464
SRT | L-8058B | Moura pig breed | 132053000 | C1296465
SRT | L-8058C | Canastra pig breed | 132054006 | C1296466
SRT | L-8058D | Pirapetinga pig breed | 132055007 | C1296467
SRT | L-8058E | Piau pig breed | 132056008 | C1296468
SRT | L-8058F | Nilo-Canastra pig breed | 132057004 | C1296469
SRT | L-80595 | Canastrão pig breed | 132058009 | C1321448
SRT | L-80596 | Canastrão, Junqueira pig breed | 132059001 | C1321449
SRT | L-80597 | Canastrão, Capitão Chico pig breed | 132060006 | C1321450
SRT | L-80598 | Canastrão, Zabumba pig breed | 132061005 | C1321451
SRT | L-80599 | Canastrão, Cabano pig breed | 132062003 | C1321452
SRT | L-8059A | Canastrão, Vermelho pig breed | 132063008 | C1321453
SRT | L-8059B | Piau, Caruncho Piau pig breed | 132064002 | C1296470
SRT | L-8059C | Canastrinho pig breed | 132065001 | C1296471
SRT | L-8059D | Honduras Switch-Tail pig breed | 132066000 | C1269199
SRT | L-8059E | Mastergilt pig breed | 132067009 | C1296472
SRT | L-8059F | Sovereign pig breed | 132068004 | C1269200
SRT | L-805A1 | Poltava pig breed | 132069007 | C1296473
SRT | L-805A2 | Lipetsk pig breed | 132070008 | C1296474
SRT | L-805A3 | Soviet Meat pig breed | 132071007 | C1269201
SRT | L-805A4 | Central Russian pig breed | 132072000 | C1269202
SRT | L-805A5 | Steppe Meat pig breed | 132073005 | C1269203
SRT | L-805A6 | Kharkov pig breed | 132074004 | C1296475
SRT | L-805A7 | Dnepropetrovsk pig breed | 132075003 | C1296476
SRT | L-805A8 | Russian Large White pig breed | 132076002 | C1269204
SRT | L-805A9 | Forest Mountain pig breed | 132077006 | C1269205
SRT | L-805AA | Dnieper pig breed | 132078001 | C1296477
SRT | L-805AB | Iberian pig breed | 132079009 | C1296478
SRT | L-805AC | Iberian, Extremadura Red pig breed | 132080007 | C1269206
SRT | L-805AD | Iberian, Jabugo Spotted pig breed | 132081006 | C1269207
SRT | L-805AE | Iberian, Black Iberian pig breed | 132082004 | C1269208
SRT | L-805AF | Philippine Native, Ilocos pig breed | 132083009 | C1269209
SRT | L-805B1 | Philippine Native, Jalajala pig breed | 132084003 | C1269210
SRT | L-805B2 | Mangalista pig breed | 132085002 | C1269211
SRT | L-805B3 | Alentejana pig breed | 132086001 | C1269212
SRT | L-805B4 | Belgian Landrace, BN pig breed | 132087005 | C1269213
SRT | L-805B5 | French Large White pig breed | 132088000 | C1269214
SRT | L-805B6 | Hyper Large White pig breed | 132089008 | C1269215
SRT | L-805B7 | Tia Meslan pig breed | 132090004 | C1269216
SRT | L-805B8 | Pen ar Lan 77 pig breed | 132091000 | C1296479
SRT | L-805B9 | Penshire pig breed | 132092007 | C1296480
SRT | L-805BA | Laconie pig breed | 132093002 | C1269217
SRT | L-805BB | Murcian pig breed | 132094008 | C1269218
SRT | L-805BC | Cavallino pig breed | 132095009 | C1269219
SRT | L-805BD | Calabrian pig breed | 132096005 | C1296481
SRT | L-805BE | Apulian pig breed | 132097001 | C1269220
SRT | L-805BF | Siena Belted pig breed | 132098006 | C1269221
SRT | L-805C1 | Calascibetta pig breed | 132099003 | C1269222
SRT | L-805C2 | Güssing Forest Pig pig breed | 132100006 | C1321454
SRT | L-805C3 | Swiss Edelschwein pig breed | 132101005 | C1269223
SRT | L-805C4 | North Caucasus pig breed | 132102003 | C1296482
SRT | L-805C5 | Don pig breed | 132103008 | C1269224
SRT | L-805C6 | Rostov pig breed | 132104002 | C1296483
SRT | L-805C7 | Russian Long-Eared White pig breed | 132105001 | C1269225
SRT | L-805C8 | Russian Short-Eared White pig breed | 132106000 | C1269226
SRT | L-805C9 | Prisheksninsk pig breed | 132107009 | C1296484
SRT | L-805CA | Breitov pig breed | 132108004 | C1296485
SRT | L-805CB | Livny pig breed | 132109007 | C1296486
SRT | L-805CC | Tsivilsk pig breed | 132110002 | C1296487
SRT | L-805CD | Urzhum pig breed | 132111003 | C1296488
SRT | L-805CE | Minisib pig breed | 132112005 | C1296489
SRT | L-805CF | Sakhalin White pig breed | 132113000 | C1269227
SRT | L-805D0 | North Siberian pig breed | 132114006 | C1296490
SRT | L-805D1 | Siberian Black Pied pig breed | 132115007 | C1269228
SRT | L-805D2 | Kemerovo pig breed | 132116008 | C1296491
SRT | L-805D3 | KM-1 pig breed | 132117004 | C1296492
SRT | L-805D4 | Aksaî Black Pied pig breed | 132118009 | C1321455
SRT | L-805D5 | Semirechensk pig breed | 132119001 | C1296493
SRT | L-805D6 | Min pig breed | 132120007 | C1296494
SRT | L-805D7 | Sanjiang White pig breed | 132121006 | C1269229
SRT | L-805D8 | Basque Black Pied pig breed | 132122004 | C1269230
SRT | L-805D9 | Corsican pig breed | 132123009 | C1296495
SRT | L-805DA | Créole pig breed | 132124003 | C1321456
SRT | L-805DB | Gascony pig breed | 132125002 | C1296496
SRT | L-805DC | Limousin pig breed | 132126001 | C1296497
SRT | L-805DD | Harbin White pig breed | 132127005 | C1269231
SRT | L-805DE | Heilongjiang Spotted pig breed | 132128000 | C1269232
SRT | L-805DF | Liaoning Black pig breed | 132129008 | C1269233
SRT | L-805E1 | Huang-Huai-Hai Black, Shenxian pig breed | 132130003 | C1269234
SRT | L-805E2 | Huang-Huai-Hai Black pig breed | 132131004 | C1269235
SRT | L-805E3 | Bamei pig breed | 132132006 | C1296498
SRT | L-805E4 | Hanjiang Black pig breed | 132133001 | C1269236
SRT | L-805E5 | Ding pig breed | 132134007 | C1296499
SRT | L-805E6 | Huai pig breed | 132135008 | C1296500
SRT | L-805E7 | New Huai pig breed | 132136009 | C1296501
SRT | L-805E8 | Mashen pig breed | 132137000 | C1296502
SRT | L-805E9 | Yimeng Black pig breed | 132138005 | C1269237
SRT | L-805EB | Hetao Lop-Ear pig breed | 132139002 | C1269238
SRT | L-805EC | Korean Native pig breed | 132140000 | C1269239
SRT | L-805ED | Korean Improved pig breed | 132141001 | C1269240
SRT | L-805EE | Penbuk pig breed | 132142008 | C1296503
SRT | L-805EF | Beijing Black pig breed | 132143003 | C1269241
SRT | L-805F1 | Chenghua pig breed | 132144009 | C1296504
SRT | L-805F2 | Taoyuan pig breed | 132145005 | C1296505
SRT | L-805F3 | Taiwan Small Black pig breed | 132146006 | C1269242
SRT | L-805F4 | Taiwan Small Red pig breed | 132147002 | C1269243
SRT | L-805F5 | Guanling pig breed | 132148007 | C1296506
SRT | L-805F6 | Huchuan Mountain pig breed | 132149004 | C1269244
SRT | L-805F7 | Rongchang pig breed | 132150004 | C1296507
SRT | L-805F8 | Wujin pig breed | 132151000 | C1296508
SRT | L-805F9 | Dahe pig breed | 132152007 | C1296509
SRT | L-805FA | Yanan pig breed | 132153002 | C1296510
SRT | L-805FB | South Yunnan Short-Eared pig breed | 132154008 | C1269245
SRT | L-805FC | Hainan, Lingao pig breed | 132155009 | C1269246
SRT | L-805FD | Hainan, Tunchang pig breed | 132156005 | C1269247
SRT | L-805FE | Hainan, Wenchang pig breed | 132157001 | C1269248
SRT | L-805FF | Liang Guang Small Spotted pig breed | 132158006 | C1269249
SRT | L-8060A | German Pasture pig breed | 132159003 | C1296511
SRT | L-8060B | Piau, Sorocaba pig breed | 132160008 | C1269250
SRT | L-8060C | Nilo pig breed | 132161007 | C1296512
SRT | L-8060D | Bahia pig breed | 132162000 | C1296513
SRT | L-8060E | Perna-Curta pig breed | 132163005 | C1296514
SRT | L-8060F | Carunchinho pig breed | 132164004 | C1296515
SRT | L-80613 | Mandi pig breed | 132165003 | C1296516
SRT | L-80614 | Orehla de Colher pig breed | 132166002 | C1296517
SRT | L-80615 | Venezuelan Black pig breed | 132167006 | C1296518
SRT | L-80616 | Bolivian pig breed | 132168001 | C1296519
SRT | L-80617 | Pelón pig breed | 132169009 | C1321457
SRT | L-80618 | Mexican Wattled pig breed | 132170005 | C1269251
SRT | L-80619 | Dalland 080 pig breed | 132171009 | C1296520
SRT | L-8061B | Monarch pig breed | 132173007 | C1269252
SRT | L-8061C | Bisaro pig breed | 132174001 | C1296521
SRT | L-8061D | Black Hairless pig breed | 132175000 | C1269253
SRT | L-8061E | Black Mangalitsa pig breed | 132176004 | C1269254
SRT | L-80623 | Borghigiana pig breed | 132178003 | C1296523
SRT | L-80624 | Chianina pig breed | 132179006 | C1296524
SRT | L-80625 | Cosentina pig breed | 132180009 | C1296525
SRT | L-80626 | Cuino pig breed | 132181008 | C1296526
SRT | L-80627 | Friuli Black pig breed | 132182001 | C1269255
SRT | L-80628 | Fumati pig breed | 132183006 | C1296527
SRT | L-80629 | Galician pig breed | 132184000 | C1296528
SRT | L-8062A | German Berkshire pig breed | 132185004 | C1296529
SRT | L-8062B | Ghori pig breed | 132186003 | C1296530
SRT | L-8062C | Jianli pig breed | 132187007 | C1296531
SRT | L-8062D | Lucanian pig breed | 132188002 | C1269256
SRT | L-8062E | Maremmana pig breed | 132189005 | C1296532
SRT | L-8062F | Miami pig breed | 132190001 | C1296533
SRT | L-80634 | Montmorillon pig breed | 132191002 | C1296534
SRT | L-80635 | Old Swedish Spotted pig breed | 132192009 | C1269257
SRT | L-80636 | Oliventina pig breed | 132193004 | C1296535
SRT | L-80637 | Parmense pig breed | 132194005 | C1296536
SRT | L-80638 | Romagnola pig breed | 132195006 | C1296537
SRT | L-80639 | Siberian pig breed | 132196007 | C1296538
SRT | L-8063A | Small White pig breed | 132197003 | C1269258
SRT | L-8063B | Baltaret pig breed | 132198008 | C1296539
SRT | L-8063C | Tungchang pig breed | 132199000 | C1296540
SRT | L-8063D | Sterling pig breed | 132200002 | C1296541
SRT | L-8063E | Vich pig breed | 132201003 | C1296542
SRT | L-8063F | Vietnamese pig breed | 132202005 | C1296543
SRT | L-80645 | Vitoria pig breed | 132203000 | C1296544
SRT | L-80646 | Wai Chow pig breed | 132204006 | C1296545
SRT | L-80647 | Yorkshire Blue and White pig breed | 132205007 | C1269259
SRT | L-80648 | Dalland 020 pig breed | 132206008 | C1296546
SRT | L-80649 | Wiltshire pig breed | 132207004 | C1296547
SRT | L-8064A | Hamroc pig breed | 132208009 | C1296548
SRT | L-8064B | DRU™ Terminals pig breed | 132209001 | C1269260
SRT | L-8064C | Camborough 22 pig breed | 132210006 | C1296549
SRT | L-8064D | Camborough 15 pig breed | 132211005 | C1296550
SRT | L-8064E | PR 1050 pig breed | 132212003 | C1296551
SRT | L-8064F | PR 1075 pig breed | 132213008 | C1296552
SRT | L-8065A | Chryak PIC pig breed | 132214002 | C1296553
SRT | L-8065B | Canadian Royal Blue pig breed | 132215001 | C1269261
SRT | L-8065C | Line 500 Duroc pig breed | 132216000 | C1269262
SRT | L-8065D | Bodmin 950 pig breed | 132217009 | C1296554
SRT | L-8065E | Canadian Duroc pig breed | 132218004 | C1296555
SRT | L-8065F | Canadian Hampshire pig breed | 132219007 | C1296556
SRT | L-80664 | Ba Xuyen pig breed | 132220001 | C1296557
SRT | L-80665 | Arapawa Island pig breed | 132221002 | C1296558
SRT | L-80666 | Wuzhishan pig breed | 132222009 | C1296559
SRT | L-80667 | Philippine Native pig breed | 132223004 | C1269263
SRT | L-80668 | Sinclair Miniature pig breed | 132224005 | C1269264
SRT | L-80669 | Saddleback pig breed | 132225006 | C1296560
SRT | L-8066A | Yucatan Minature pig breed | 132226007 | C1269265
SRT | L-8066B | Bantu pig breed | 132227003 | C1296561
SRT | L-8066C | Tibetan pig breed | 132228008 | C1296562
SRT | L-8066D | Turopolje pig breed | 132229000 | C1296563
SRT | L-8066E | Vietnamese Pot-Bellied Pig pig breed | 132230005 | C1296564
SRT | L-8066F | American Landrace pig breed | 132231009 | C1269266
SRT | L-80670 | Swallow Belied Mangalitza pig breed | 132232002 | C1269267
SRT | L-80671 | Fengjing pig breed | 132233007 | C1296565
SRT | L-80672 | Finnish Landrace pig breed | 132234001 | C1269268
SRT | L-80673 | Guinea Hog pig breed | 132235000 | C1296566
SRT | L-80674 | Hezuo pig breed | 132236004 | C1296567
SRT | L-80675 | Ossabaw Island pig breed | 132237008 | C1296568
SRT | L-80676 | Kele pig breed | 132238003 | C1296569
SRT | L-80677 | Krskopolje pig breed | 132239006 | C1296570
SRT | L-80678 | Kunekune pig breed | 132240008 | C1296571
SRT | L-80679 | Large Black-White pig breed | 132241007 | C1269269
SRT | L-8067A | Lithuanian Native pig breed | 132242000 | C1269270
SRT | L-8067B | Meishan pig breed | 132243005 | C1296572
SRT | L-8067C | Jinhua pig breed | 132244004 | C1296573
SRT | L-8067D | Ningxiang pig breed | 132245003 | C1296574
SRT | L-8067E | Mora Romagnola pig breed | 132246002 | C1296575
SRT | L-8067F | Mukota pig breed | 132247006 | C1296576
SRT | L-80680 | Minzhu pig breed | 132248001 | C1296577
SRT | L-80681 | Neijiang pig breed | 132249009 | C1296578
SRT | L-80682 | Mulefoot pig breed | 132250009 | C1269271
SRT | L-80683 | Normand pig breed | 132251008 | C1296579
SRT | L-80684 | Angeln Saddleback pig breed | 132252001 | C1269192
SRT | L-80685 | Greek Local pig breed | 132253006 | C1269193
SRT | L-80686 | Icelandic pig breed | 132254000 | C1296580
SRT | L-80687 | Casertana pig breed | 132255004 | C1296581
SRT | L-80688 | Madonie-Sicilian pig breed | 132256003 | C1269194
SRT | L-80689 | Sardinian pig breed | 132257007 | C1296582
SRT | L-8068A | Sicilian pig breed | 132258002 | C1296583
SRT | L-8068B | Zlotniki Spotted pig breed | 132259005 | C1269272
SRT | L-8068C | Zlotniki White pig breed | 132260000 | C1269273
SRT | L-8068D | Siska pig breed | 132261001 | C1296584
SRT | L-8068E | Sumadija pig breed | 132262008 | C1296585
SRT | L-8068F | Froxfield Pygmy pig breed | 132263003 | C1269274
SRT | L-80690 | Danish Large White pig breed | 132264009 | C1269275
SRT | L-80691 | Danish Duroc pig breed | 132265005 | C1296586
SRT | L-80692 | Danish Hampshire pig breed | 132266006 | C1296587
SRT | L-80693 | Piggham pig breed | 132267002 | C1296588
SRT | L-80694 | New York Red pig breed | 132268007 | C1269276
SRT | L-80695 | Finnish Yorkshire pig breed | 132269004 | C1296589
SRT | L-80696 | Dutch Yorkshire pig breed | 132270003 | C1296590
SRT | L-80697 | Pulawy pig breed | 132271004 | C1296591
SRT | L-80698 | Pomeranian pig breed | 132272006 | C1296592
SRT | L-80699 | Polish Landrace pig breed | 132273001 | C1269277
SRT | L-8069A | Estonian Bacon pig breed | 132274007 | C1269278
SRT | L-8069B | Latvian White pig breed | 132275008 | C1269279
SRT | L-8069C | Lithuanian White pig breed | 132276009 | C1269280
SRT | L-8069D | BKB-1 pig breed | 132277000 | C1296593
SRT | L-8069E | Belorus Black Pied pig breed | 132278005 | C1269281
SRT | L-8069F | Mirgorod pig breed | 132279002 | C1296594
SRT | L-806A1 | Liang Guang Small Spotted, Luchuan pig breed | 132280004 | C1269282
SRT | L-806A2 | Fujian Small pig breed | 132281000 | C1269283
SRT | L-806A3 | North Fujian Black-and-White pig breed | 132282007 | C1269284
SRT | L-806A4 | Fuan Spotted pig breed | 132283002 | C1269285
SRT | L-806A5 | Putian pig breed | 132284008 | C1269286
SRT | L-806A6 | Fuzhou Black pig breed | 132285009 | C1269287
SRT | L-806A7 | Minbei Spotted pig breed | 132286005 | C1269288
SRT | L-806A8 | Lantang pig breed | 132287001 | C1296595
SRT | L-806A9 | Liang Guang Small Spotted, Guangdong Small Ear pig breed | 132288006 | C1269289
SRT | L-806AA | Longlin pig breed | 132289003 | C1296596
SRT | L-806AB | Yuedong Black pig breed | 132290007 | C1269290
SRT | L-806AC | Xiang pig breed | 132291006 | C1296597
SRT | L-806AD | Cantonese pig breed | 132292004 | C1296598
SRT | L-806AE | Jinhua, Dongyang pig breed | 132293009 | C1269291
SRT | L-806AF | Jinhua, Yongkang pig breed | 132294003 | C1269292
SRT | L-806B1 | Daweizi pig breed | 132295002 | C1296599
SRT | L-806B2 | Huazhong Two-End Black pig breed | 132296001 | C1269293
SRT | L-806B3 | Huazhong Two-End Black, Jianli pig breed | 132297005 | C1269294
SRT | L-806B4 | Huazhong Two-End Black, Tongcheng pig breed | 132298000 | C1269295
SRT | L-806B5 | Huazhong Two-End Black, Satzeling pig breed | 132299008 | C1269296
SRT | L-806B6 | Ganzhongnan Spotted pig breed | 132300000 | C1269297
SRT | L-806B7 | Hang pig breed | 132301001 | C1296600
SRT | L-806B8 | Leping pig breed | 132302008 | C1296601
SRT | L-806B9 | Longyou Black pig breed | 132303003 | C1269298
SRT | L-806BA | Wuyi Black pig breed | 132304009 | C1269299
SRT | L-806BB | Lee-Sung pig breed | 132305005 | C1296602
SRT | L-806BC | Lan-Yu pig breed | 132306006 | C1296603
SRT | L-806BD | Vietnamese Yorkshire pig breed | 132307002 | C1296604
SRT | L-806BE | Yujiang pig breed | 132308007 | C1296605
SRT | L-806BF | Wanzhe Spotted pig breed | 132309004 | C1269300
SRT | L-806C1 | Wanzhe Spotted, Chunan Spotted pig breed | 132310009 | C1269301
SRT | L-806C2 | Wanzhe Spotted, Wannan Spotted pig breed | 132311008 | C1296606
SRT | L-806C3 | Shengxian Spotted pig breed | 132312001 | C1296607
SRT | L-806C4 | Qingping pig breed | 132313006 | C1296608
SRT | L-806C5 | Xiangxi Black pig breed | 132314000 | C1296609
SRT | L-806C6 | Bamaxiang pig breed | 132315004 | C1296610
SRT | L-806C7 | Taihu pig breed | 132316003 | C1296611
SRT | L-806C8 | Erhulian pig breed | 132317007 | C1296612
SRT | L-806C9 | Jiaxing Black pig breed | 132318002 | C1296613
SRT | L-806CA | Mi pig breed | 132319005 | C1296614
SRT | L-806CB | Shahutou pig breed | 132320004 | C1296615
SRT | L-806CC | Jiaoxi pig breed | 132321000 | C1296616
SRT | L-806CD | Shanghai White pig breed | 132322007 | C1296617
SRT | L-806CE | Hubei White pig breed | 132323002 | C1296618
SRT | L-806CF | Xinjin pig breed | 132324008 | C1296619
SRT | L-806D1 | Xinjin, Jilin Black pig breed | 132325009 | C1296620
SRT | L-806D2 | Xinjin, Ning-an pig breed | 132326005 | C1296621
SRT | L-806D3 | Í pig breed | 132327001 | C1321458
SRT | L-806D4 | DBI pig breed | 132328006 | C1296622
SRT | L-806D5 | Xinjin, Xinjin pig breed | 132329003 | C1296623
SRT | L-806D6 | Meixin pig breed | 132330008 | C1296624
SRT | L-806D7 | North East China Spotted pig breed | 132331007 | C1296625
SRT | L-806D8 | Fannong Spotted pig breed | 132332000 | C1296626
SRT | L-806D9 | Laoshan pig breed | 132333005 | C1296627
SRT | L-806DA | Nanjing Black pig breed | 132334004 | C1296628
SRT | L-806DB | Shanxi Black pig breed | 132335003 | C1296629
SRT | L-806DC | Ganzhou White pig breed | 132336002 | C1296630
SRT | L-806DD | Guangxi White pig breed | 132337006 | C1296631
SRT | L-806DE | Hanzhong White pig breed | 132338001 | C1296632
SRT | L-806DF | Lutai White pig breed | 132339009 | C1296633
SRT | L-806E1 | Yili White pig breed | 132340006 | C1296634
SRT | L-806E2 | Xinjiang White pig breed | 132341005 | C1296635
SRT | L-806E3 | BSI pig breed | 132342003 | C1296636
SRT | L-806E4 | Mong Cai pig breed | 132343008 | C1296637
SRT | L-806E5 | Lang Hong pig breed | 132344002 | C1296638
SRT | L-806E6 | Muong Khuong pig breed | 132345001 | C1296639
SRT | L-806E7 | Meo pig breed | 132346000 | C1296640
SRT | L-806E8 | Tong Con pig breed | 132347009 | C1296641
SRT | L-806E9 | Ha Bac pig breed | 132348004 | C1296642
SRT | L-806EA | Thai Binh pig breed | 132349007 | C1296643
SRT | L-806EB | Co pig breed | 132350007 | C1296644
SRT | L-806EC | Swiss Improved Landrace pig breed | 132351006 | C1296645
SRT | L-806ED | German Landrace B pig breed | 132352004 | C1296646
SRT | L-806EE | Edelschwein pig breed | 132353009 | C1296647
SRT | L-806EF | Swabian-Hall pig breed | 132354003 | C1296648
SRT | L-806F1 | Bentheim Black Pied pig breed | 132355002 | C1296649
SRT | L-806F2 | Baldinger Spotted pig breed | 132356001 | C1296650
SRT | L-806F3 | German Red Pied pig breed | 132357005 | C1296651
SRT | L-806F4 | German Cornwall pig breed | 132358000 | C1296652
SRT | L-806F5 | Göttingen Miniature pig breed | 132359008 | C1321459
SRT | L-806F6 | Munich Miniature pig breed | 132360003 | C1296653
SRT | L-806F8 | Leicoma pig breed | 132361004 | C1296654
SRT | L-806F9 | Schwerfurt Meat pig breed | 132362006 | C1296655
SRT | L-806FA | Hungarian White pig breed | 132363001 | C1296656
SRT | L-806FB | Hungahyb pig breed | 132364007 | C1296657
SRT | L-806FC | Bulgarian Native pig breed | 132365008 | C1296658
SRT | L-806FD | East Balkan pig breed | 132366009 | C1296659
SRT | L-806FE | Kula pig breed | 132367000 | C1296660
SRT | L-806FF | Nghia Binh pig breed | 132368005 | C1296661
SRT | L-807E2 | Bichon Teneriffe dog breed | 132371002 | C1296664
SRT | L-807E3 | Bizanian Hound dog breed | 132372009 | C1296663
SRT | L-807E4 | Bloodhound, St. Hubert dog breed | 132373004 | C1296665
SRT | L-807E5 | Bloodhound, Southern Hound dog breed | 132374005 | C1296666
SRT | L-808A3 | Bordeaux Dog breed | 132389001 | C1296679
SRT | L-807E7 | Brandlbracke dog breed | 132376007 | C1296668
SRT | L-807E8 | Braque d'Ariége dog breed | 132377003 | C1321460
SRT | L-807E9 | Portuguese Guard Dog breed | 132378008 | C1296669
SRT | L-807EA | Great Münsterländer dog breed | 132379000 | C1321461
SRT | L-807EB | Beagle, Smooth dog breed | 132380002 | C1296670
SRT | L-807EC | Beagle, Rough dog breed | 132381003 | C1296671
SRT | L-807ED | Belgian Griffon, Rough dog breed | 132382005 | C1296672
SRT | L-807EE | Belgian Griffon, Smooth dog breed | 132383000 | C1296673
SRT | L-807EF | Braque Belge dog breed | 132384006 | C1296674
SRT | L-807F1 | Belgian Street Dog breed | 132385007 | C1296675
SRT | L-807F2 | Bernese Hound dog breed | 132386008 | C1296676
SRT | L-808A1 | Eurasier dog breed | 132387004 | C1296677
SRT | L-808A2 | English Bulldog breed | 132388009 | C1296678
SRT | L-808A3 | Dogue de Bourdeaux dog breed | 132389001 | C1296679
SRT | L-808A4 | Kai Ken dog breed | 132390005 | C1296680
SRT | L-808A5 | Kui Mlk dog breed | 132391009 | C1296681
SRT | L-808A6 | Argentine Dogo dog breed | 132392002 | C1296682
SRT | L-808A7 | Alentejo herder dog breed | 132393007 | C1296683
SRT | L-808A8 | Saint Bernard, Long-haired dog breed | 132394001 | C1296684
SRT | L-808A9 | Saint Bernard, Short-haired dog breed | 132395000 | C1296685
SRT | L-808AA | West Siberian Laika dog breed | 132396004 | C1296686
SRT | L-808AB | Basset Fauve de Bretagne dog breed | 132397008 | C1296687
SRT | L-808AC | Japanese Retriever dog breed | 132398003 | C1296688
SRT | L-808AD | Kai Dog breed | 132399006 | C1296689
SRT | L-808AE | American Blue Gascon Hound dog breed | 132400004 | C1296690
SRT | L-808AF | Beagle Harrier dog breed | 132401000 | C1296691
SRT | L-808B1 | Kangal Dog breed | 132402007 | C1296692
SRT | L-808B2 | Leopard Cur dog breed | 132403002 | C1296693
SRT | L-808B3 | Patterdale Terrier dog breed | 132404008 | C1296694
SRT | L-808B4 | Petit Brabaçon dog breed | 132405009 | C1296695
SRT | L-808B5 | Aidi dog breed | 132406005 | C1296696
SRT | L-808B6 | American Indian Dog breed | 132407001 | C1296697
SRT | L-808B7 | Austrian Pinscher dog breed | 132408006 | C1296698
SRT | L-808B8 | American Eskimo, standard dog breed | 132409003 | C1296699
SRT | L-808B9 | American Eskimo, Miniature dog breed | 132410008 | C1296700
SRT | L-808BA | American Eskimo, Toy dog breed | 132411007 | C1296701
SRT | L-808BB | Basset Griffon Vendéen dog breed | 132412000 | C1296702
SRT | L-808BC | Batard dog breed | 132413005 | C1296703
SRT | L-808BD | Basset Bleu de Gascogne dog breed | 132414004 | C1296704
SRT | L-808BE | Braque Dupuy dog breed | 132415003 | C1296705
SRT | L-808BF | Bruno de Jura dog breed | 132416002 | C1296706
SRT | L-808C1 | Cão da Serra de Aires dog breed | 132417006 | C1296707
SRT | L-808C2 | Cão de Castro Laboreiro dog breed | 132418001 | C1296708
SRT | L-808C3 | Cão de Fila Miguel dog breed | 132419009 | C1296709
SRT | L-808C4 | Catalan Sheepdog breed | 132420003 | C1296710
SRT | L-808C5 | Caucasian Shepherd Dog breed | 132421004 | C1296711
SRT | L-808C6 | Cirneco dell'Etna dog breed | 132422006 | C1296712
SRT | L-808C7 | English Toy Terrier dog breed | 132423001 | C1296713
SRT | L-808C8 | German Spitz dog breed | 132424007 | C1296714
SRT | L-DA692 | Dingo dog breed | 709853007 | C1296715
SRT | L-808CA | Fauve de Bretagne dog breed | 132426009 | C1296716
SRT | L-808CB | Hellenic Hound dog breed | 132427000 | C1296717
SRT | L-808CC | Holland Shepherd dog breed | 132428005 | C1296718
SRT | L-808CD | Japanese Spitz dog breed | 132429002 | C1296719
SRT | L-808CE | Jämthund dog breed | 132430007 | C1296720
SRT | L-808CF | Jindo dog breed | 132431006 | C1296721
SRT | L-808D1 | Karelo-Finnish Laika dog breed | 132432004 | C1296722
SRT | L-808D2 | King Shepherd dog breed | 132433009 | C1296723
SRT | L-808D3 | Kishu dog breed | 132434003 | C1296724
SRT | L-808D4 | Kirhiz dog breed | 132435002 | C1296725
SRT | L-808D5 | Magyar Agår dog breed | 132436001 | C1296726
SRT | L-808D6 | Middle Asian Ovtcharka dog breed | 132437005 | C1296727
SRT | L-808D7 | Mi-Ki dog breed | 132438000 | C1296728
SRT | L-808D8 | Miniature Australian Shepherd dog breed | 132439008 | C1296729
SRT | L-808D9 | Min-pei dog breed | 132440005 | C1296730
SRT | L-808DA | Mountain Cur dog breed | 132441009 | C1296731
SRT | L-808DB | Moscow Longhaired Toy Terrier dog breed | 132442002 | C1296732
SRT | L-808DC | Perdigueiro Portuguese dog breed | 132443007 | C1296733
SRT | L-808DD | Podengo Canario dog breed | 132444001 | C1296734
SRT | L-808DE | Podengo Pequeno dog breed | 132445000 | C1296735
SRT | L-808DF | Pressa Mallorquin dog breed | 132446004 | C1296736
SRT | L-808E1 | Pyrenean Mastiff dog breed | 132447008 | C1296737
SRT | L-808E2 | Rastreador Brasileiro dog breed | 132448003 | C1296738
SRT | L-808E3 | Sabuesos Españoles dog breed | 132449006 | C1296739
SRT | L-808E4 | Schiller Hound dog breed | 132450006 | C1296740
SRT | L-808E5 | South Russian Steppe Hound dog breed | 132451005 | C1296741
SRT | L-808E6 | Styrian Mountain dog breed | 132452003 | C1296742
SRT | L-808E7 | Berger du Languedoc dog breed | 132453008 | C1296743
SRT | L-808E8 | Teddy Roosevelt Terrier dog breed | 132454002 | C1296744
SRT | L-808E9 | Transylvanian Hound dog breed | 132455001 | C1296745
SRT | L-808EA | Trigg Hound dog breed | 132456000 | C1296746
SRT | L-808EB | Tyrolean Hound dog breed | 132457009 | C1296747
SRT | L-808EC | White Shepherd dog breed | 132458004 | C1296748
SRT | L-808ED | Wirehair Styrian mountain dog breed | 132459007 | C1296749
SRT | L-808EE | Yugoslavian Hound dog breed | 132460002 | C1296750
SRT | L-808EF | Old Farm Collie dog breed | 132461003 | C1296751
SRT | L-808F1 | Old German Shepherd dog breed | 132462005 | C1296752
SRT | L-808F2 | New Zealand Heading Dog breed | 132463000 | C1296753
SRT | L-808F3 | German Koolie dog breed | 132464006 | C1296754
SRT | L-808F4 | Smithfield dog breed | 132465007 | C1296755
SRT | L-808F5 | Spanish Greyhound dog breed | 132466008 | C1296756
SRT | L-808F6 | Armant dog breed | 132467004 | C1296757
SRT | L-808F8 | Australian Greyhound dog breed | 132468009 | C1296758
SRT | L-808F9 | Australian Terrier, rough-coated dog breed | 132469001 | C1296759
SRT | L-808FA | Australian Terrier, silky dog breed | 132470000 | C1296760
SRT | L-808FB | Austrian Hound dog breed | 132471001 | C1296761
SRT | L-808FC | Austrian Smooth-Haired Bracke dog breed | 132472008 | C1296762
SRT | L-808FD | Balkan Hound dog breed | 132473003 | C1296763
SRT | L-808FE | Banjara greyhound dog breed | 132474009 | C1296764
SRT | L-808FF | Beagle, Standard dog breed | 132475005 | C1296765
SRT | L-80916 | Estrela Mountain Dog breed | 132476006 | C1296766
SRT | L-80917 | Epagneul Picard dog breed | 132477002 | C1296767
SRT | L-80918 | Epagneul Bleu de Picardie dog breed | 132478007 | C1296768
SRT | L-80919 | Estonian Hound dog breed | 132479004 | C1296769
SRT | L-80920 | Epagneul Pont-Audemer dog breed | 132480001 | C1296770
SRT | L-80921 | Eurasian dog breed | 132481002 | C1296771
SRT | L-80922 | Fell Terrier dog breed | 132482009 | C1296772
SRT | L-80923 | Fila Brasileiro dog breed | 132483004 | C1296773
SRT | L-80924 | Finnish Hound dog breed | 132484005 | C1296774
SRT | L-80925 | Finnish Lapphund dog breed | 132485006 | C1296775
SRT | L-80926 | Entlebucher dog breed | 132486007 | C1296776
SRT | L-80927 | French Guard Dog breed | 132487003 | C1296777
SRT | L-80928 | French Spaniel dog breed | 132488008 | C1296778
SRT | L-80929 | Coton de Tuléar dog breed | 132489000 | C1296779
SRT | L-80930 | Hamiltonstövare dog breed | 132490009 | C1296780
SRT | L-80931 | Danish Broholmer dog breed | 132491008 | C1296781
SRT | L-80932 | English Shepherd dog breed | 132492001 | C1296782
SRT | L-80933 | Drentse Patrijshond dog breed | 132493006 | C1296783
SRT | L-80934 | Dunker dog breed | 132494000 | C1296784
SRT | L-80935 | Dutch Kooiker Dog breed | 132495004 | C1296785
SRT | L-80936 | Dutch Shepherd dog breed | 132496003 | C1296786
SRT | L-80937 | East Siberian Laika dog breed | 132497007 | C1296787
SRT | L-80938 | Deutsche bracke dog breed | 132498002 | C1296788
SRT | L-80939 | Hanoverian Hound dog breed | 132499005 | C1296789
SRT | L-80940 | Hovawart dog breed | 132500001 | C1296790
SRT | L-80941 | Icelandic Sheepdog breed | 132501002 | C1296791
SRT | L-80942 | Inca Hairless Dog breed | 132502009 | C1296792
SRT | L-80943 | Irish Red and White Setter dog breed | 132503004 | C1296793
SRT | L-80944 | Jagdterrier dog breed | 132504005 | C1296794
SRT | L-80945 | German Spaniel dog breed | 132505006 | C1296795
SRT | L-80946 | Grand Anglo-Français dog breed | 132506007 | C1296796
SRT | L-80947 | Grand Bassett Griffon Vendeen dog breed | 132507003 | C1296797
SRT | L-80948 | Grand Bleu de Gascogne dog breed | 132508008 | C1296798
SRT | L-80949 | Grand Gascon-Saintongeois dog breed | 132509000 | C1296799
SRT | L-80950 | German Pinscher dog breed | 132510005 | C1296800
SRT | L-80951 | Greater Swiss Mountain Dog breed | 132511009 | C1296801
SRT | L-80952 | Greenland Dog breed | 132512002 | C1296802
SRT | L-80953 | Griffon Fauve de Bretegne dog breed | 132513007 | C1296803
SRT | L-80954 | Griffon Nivernais dog breed | 132514001 | C1296804
SRT | L-80955 | Grand Griffon Vendeen dog breed | 132515000 | C1296805
SRT | L-80956 | Ainu dog breed | 132516004 | C1296806
SRT | L-80957 | Basset Artésian Normand dog breed | 132517008 | C1296807
SRT | L-80958 | Bavarian Mountain Hound dog breed | 132518003 | C1296808
SRT | L-80959 | Beauceron dog breed | 132519006 | C1296809
SRT | L-80960 | Azawakh dog breed | 132520000 | C1296810
SRT | L-80961 | Australian Shepherd dog breed | 132521001 | C1296811
SRT | L-80962 | Belgian Wolfhound dog breed | 132522008 | C1296812
SRT | L-80963 | Bergamasco dog breed | 132523003 | C1296813
SRT | L-80964 | Berger de Picard dog breed | 132524009 | C1296814
SRT | L-80965 | Berger de Pyrenees dog breed | 132525005 | C1296815
SRT | L-80966 | Billy dog breed | 132526006 | C1296816
SRT | L-80967 | Belgian Griffon dog breed | 132527002 | C0324378
SRT | L-80968 | American Hairless Terrier dog breed | 132528007 | C1296817
SRT | L-80969 | Beagle, Elizabethan dog breed | 132529004 | C1296818
SRT | L-80970 | Japanese Pointer dog breed | 132530009 | C1296819
SRT | L-80971 | Akbash dog breed | 132531008 | C1296820
SRT | L-80972 | Alapaha blueblood bullDog breed | 132532001 | C1296821
SRT | L-80973 | Barbet dog breed | 132533006 | C1296822
SRT | L-80974 | American Bulldog breed | 132534000 | C1296823
SRT | L-80975 | Black Russian Terrier dog breed | 132535004 | C1296824
SRT | L-80976 | Anglo-Francais de moyen venerie dog breed | 132536003 | C1296825
SRT | L-80977 | Anglo-Francais de petit venerie dog breed | 132537007 | C1296826
SRT | L-80978 | Appenzeller dog breed | 132538002 | C1296827
SRT | L-80979 | Ariégeois dog breed | 132539005 | C1321491
SRT | L-80980 | Alano Español dog breed | 132540007 | C1321462
SRT | L-80981 | Australian Kelpie dog breed | 132541006 | C1296828
SRT | L-80982 | Alpine dachsbracke dog breed | 132542004 | C1296829
SRT | L-80983 | Chien Français Blanc et Noir dog breed | 132543009 | C1321463
SRT | L-80984 | Carolina Dog breed | 132544003 | C1296830
SRT | L-80985 | Catahoula Leopard dog breed | 132545002 | C1296831
SRT | L-80986 | Caucasian Mountain Dog breed | 132546001 | C1296832
SRT | L-80987 | Cesky Fousek dog breed | 132547005 | C1296833
SRT | L-80988 | Cesky Terrier dog breed | 132548000 | C1296834
SRT | L-80989 | Chart Polski dog breed | 132549008 | C1296835
SRT | L-80990 | Black Forest Hound dog breed | 132550008 | C1296836
SRT | L-80991 | Chien d'Artois dog breed | 132551007 | C1296837
SRT | L-80992 | Canaan dog breed | 132552000 | C1296838
SRT | L-80993 | Chien Français Tricolore dog breed | 132553005 | C1321464
SRT | L-80994 | Chinese Crested dog breed | 132554004 | C1296839
SRT | L-80995 | Chinese Foo Dog breed | 132555003 | C1296840
SRT | L-80996 | Chinese Imperial ch'in dog breed | 132556002 | C1296841
SRT | L-80997 | Chinook dog breed | 132557006 | C1296842
SRT | L-80998 | Chien Français Blanc et Orange dog breed | 132558001 | C1321465
SRT | L-80999 | Braque Français de Grand Taille dog breed | 132559009 | C1296843
SRT | L-809A1 | Bolognese dog breed | 132560004 | C1296844
SRT | L-809A2 | Border Collie dog breed | 132561000 | C1296845
SRT | L-809A3 | Bracco Italiano dog breed | 132562007 | C1296846
SRT | L-809A4 | Cane Corso dog breed | 132563002 | C1296847
SRT | L-809A5 | Braque du Bourbonnais dog breed | 132564008 | C1296848
SRT | L-809A6 | Braque Francais de Petite Taille dog breed | 132565009 | C1296849
SRT | L-809A7 | Braque Saint-Germain dog breed | 132566005 | C1296850
SRT | L-809A8 | Briquet Basset Griffon Vendeen dog breed | 132567001 | C1296851
SRT | L-809A9 | Black Mouth Cur dog breed | 132568006 | C1296852
SRT | L-809AA | Braque d'Auvergne dog breed | 132569003 | C1296853
SRT | L-809AB | Schapendoes dog breed | 132570002 | C1296854
SRT | L-809AC | Sarplaninac dog breed | 132571003 | C1296855
SRT | L-809AD | Russo-Laika dog breed | 132572005 | C1296856
SRT | L-809AE | Bosnian Hound dog breed | 132573000 | C1296857
SRT | L-809AF | Rat Terrier dog breed | 132574006 | C1296858
SRT | L-809B1 | Pumi dog breed | 132575007 | C1296859
SRT | L-809B2 | Presa Canario dog breed | 132576008 | C1296860
SRT | L-809B3 | Portuguese Pointer dog breed | 132577004 | C1296861
SRT | L-809B4 | Porcelaine dog breed | 132578009 | C1296862
SRT | L-809B5 | Shropshire Terrier dog breed | 132579001 | C1296863
SRT | L-809B6 | Boykin Spaniel dog breed | 132580003 | C1296864
SRT | L-809B7 | Southern Blackmouth Cur dog breed | 132581004 | C1296865
SRT | L-809B8 | South Russian Ovcharka dog breed | 132582006 | C1296866
SRT | L-809B9 | Small Spanish Hound dog breed | 132583001 | C1296867
SRT | L-809BA | Small Münsterländer dog breed | 132584007 | C1321466
SRT | L-809BB | Slovak Cuvak dog breed | 132585008 | C1296868
SRT | L-809BC | Shiloh Shepherd dog breed | 132586009 | C1296869
SRT | L-809BD | Shiba Inu dog breed | 132587000 | C1296870
SRT | L-809BE | Welsh Sheepdog breed | 132588005 | C1296871
SRT | L-809BF | Shar-pei dog breed | 132589002 | C1296872
SRT | L-809C1 | Sloughi dog breed | 132590006 | C1296873
SRT | L-809C2 | Owczarek Podhalandski dog breed | 132591005 | C1296874
SRT | L-809C3 | Norbottenspets dog breed | 132592003 | C1296875
SRT | L-809C4 | Norwegian Dunkerhound dog breed | 132593008 | C1296876
SRT | L-809C5 | Old Danish Bird Dog breed | 132594002 | C1269305
SRT | L-809C6 | Old Format Dachsund dog breed | 132595001 | C1269306
SRT | L-809C7 | Old Format Manchester Terrier dog breed | 132596000 | C1269307
SRT | L-809C8 | Old Format Min/Toy Poodle dog breed | 132597009 | C1269308
SRT | L-809C9 | Old Format Welsh Corgi dog breed | 132598004 | C1269309
SRT | L-809CA | Neopolitan Mastiff dog breed | 132599007 | C1269310
SRT | L-809CB | Perdiguero de Burgos dog breed | 132600005 | C1296877
SRT | L-809CC | Perdiguero Navarro dog breed | 132601009 | C1296878
SRT | L-809CD | Peruvian Inca Orchid dog breed | 132602002 | C1269311
SRT | L-809CE | Petit Bleu de Gascogne dog breed | 132603007 | C1296879
SRT | L-809CF | Petit Gascon-Saintongeois dog breed | 132604001 | C1296880
SRT | L-809D1 | Petit Griffon Bleu de Gascogne dog breed | 132605000 | C1296881
SRT | L-809D2 | Olde English Bulldogge dog breed | 132606004 | C1296882
SRT | L-809D3 | Löwchen dog breed | 132607008 | C1321467
SRT | L-809D4 | Polski Owczarek Nizinny dog breed | 132608003 | C1296883
SRT | L-809D5 | Polish Hound dog breed | 132609006 | C1296884
SRT | L-809D6 | Poitevin dog breed | 132610001 | C1296885
SRT | L-809D7 | Spanish Pointer dog breed | 132611002 | C1296886
SRT | L-809D8 | Kyi-Leo dog breed | 132612009 | C1296887
SRT | L-809D9 | Large Spanish Hound dog breed | 132613004 | C1269312
SRT | L-809DA | Lundehund dog breed | 132614005 | C1296888
SRT | L-809DB | Lurcher Hound dog breed | 132615006 | C1269313
SRT | L-809DC | Maremma Sheepdogs dog breed | 132616007 | C1269314
SRT | L-809DD | McNab dog breed | 132617003 | C1296889
SRT | L-809DE | Miniature Bull Terrier dog breed | 132618008 | C1269315
SRT | L-809E1 | Mudi dog breed | 132620006 | C1296890
SRT | L-809E2 | Munster Lander Pointer dog breed | 132621005 | C1269317
SRT | L-809E3 | Loenberger dog breed | 132622003 | C1562740
SRT | L-809E4 | Chi Terrier dog breed | 132623008 | C1296892
SRT | L-809E5 | Krasky Ovcar dog breed | 132624002 | C1296893
SRT | L-809E6 | Kromfohrländer dog breed | 132625001 | C1321468
SRT | L-809E7 | Havanese dog breed | 132626000 | C1296894
SRT | L-809E8 | American lamalese dog breed | 132627009 | C1269318
SRT | L-809EA | Norwegian Lundehund dog breed | 132629007 | C1269320
SRT | L-809EB | North American Shepherd dog breed | 132630002 | C1296895
SRT | L-809EC | Kyi Apso dog breed | 132631003 | C1296896
SRT | L-809ED | Swedish Lapphund dog breed | 132632005 | C1269321
SRT | L-809EE | Treeing Tennessee Brindle dog breed | 132633000 | C1296897
SRT | L-809EF | Telomain dog breed | 132634006 | C1296898
SRT | L-809F1 | Swedish Vallhund dog breed | 132635007 | C1269322
SRT | L-809F2 | Stumpy Tail Cattle Dog breed | 132636008 | C1269323
SRT | L-809F3 | Stabyhoun dog breed | 132637004 | C1296899
SRT | L-809F4 | Spinone Italiano dog breed | 132638009 | C1296900
SRT | L-809F5 | Spanish Mastiff dog breed | 132639001 | C1296901
SRT | L-809F6 | Berger Shetland dog breed | 132640004 | C1296902
SRT | L-809F7 | Thai Ridgeback dog breed | 132641000 | C1296903
SRT | L-809F8 | Swiss Mountain Dog breed | 132642007 | C1269324
SRT | L-809F9 | Tibetan Mastiff dog breed | 132643002 | C1296904
SRT | L-809FA | Glen of Imaal Terrier dog breed | 132644008 | C1296905
SRT | L-809FB | Tosa Inu dog breed | 132645009 | C1296906
SRT | L-809FC | Toy Havanese Terrier dog breed | 132646005 | C1296907
SRT | L-809FD | Treeing Cur dog breed | 132647001 | C1296908
SRT | L-809FE | Treeing Feist dog breed | 132648006 | C1296909
SRT | L-809FF | Greater Swiss Mountain Hound dog breed | 132649003 | C1269325
SRT | L-80A70 | Harlequin cat breed | 132650003 | C1269326
SRT | L-80A71 | Manxamese cat breed | 132651004 | C1296910
SRT | L-80A73 | Maltese cat breed | 132652006 | C1296911
SRT | L-80A75 | Ragdoll cat breed | 132654007 | C1296912
SRT | L-80A76 | Turkish van cat breed | 132655008 | C1269328
SRT | L-80A77 | British Blue cat breed | 132656009 | C1269329
SRT | L-80A78 | American Bobtail Shorthair cat breed | 132657000 | C1296913
SRT | L-80A79 | American Bobtail Longhair cat breed | 132658005 | C1296914
SRT | L-80A80 | American Curl cat breed | 132659002 | C1269330
SRT | L-80A81 | Australian Mist cat breed | 132660007 | C1269331
SRT | L-80A83 | Bengal cat breed | 132661006 | C1296915
SRT | L-80A84 | Brazilian Shorthair cat breed | 132662004 | C1296916
SRT | L-80A85 | California Spangled cat breed | 132663009 | C1269332
SRT | L-80A86 | Chantilly/Tiffany cat breed | 132664003 | C1296917
SRT | L-80A87 | Shorthair cat breed | 132665002 | C1296918
SRT | L-80A88 | German Rex cat breed | 132666001 | C1269333
SRT | L-80A89 | LaPerm Shorthair cat breed | 132667005 | C1296919
SRT | L-80A90 | LaPerm Longhair cat breed | 132668000 | C1296920
SRT | L-80A91 | Munchkin Shorthair cat breed | 132669008 | C1296921
SRT | L-80A92 | Munchkin Longhair cat breed | 132670009 | C1296922
SRT | L-80A93 | Nebelung cat breed | 132671008 | C1296923
SRT | L-80A94 | Norwegian Forest cat breed | 132672001 | C1269334
SRT | L-80A95 | Oriental Longhair cat breed | 132673006 | C1296924
SRT | L-80A97 | Ragamuffin cat breed | 132675004 | C1296926
SRT | L-80A99 | Selkirk Rex cat breed | 132676003 | C1296927
SRT | L-80AA1 | Siberian cat breed | 132677007 | C1296928
SRT | L-80AA2 | Snowshoe cat breed | 132678002 | C1269335
SRT | L-80AA3 | Sokoke cat breed | 132679005 | C1296929
SRT | L-80AA4 | Sphynx cat breed | 132680008 | C1269336
SRT | L-80B01 | Bergamasca sheep breed | 132681007 | C1296930
SRT | L-80B02 | Portland sheep breed | 132682000 | C1296931
SRT | L-80B04 | Weisse Hornlose Heidschnucke sheep breed | 132684004 | C1296932
SRT | L-80B05 | Drents Heideschaap sheep breed | 132685003 | C1296933
SRT | L-80B06 | Kameroen sheep breed | 132686002 | C1296934
SRT | L-80B07 | Mergelland sheep breed | 132687006 | C1296935
SRT | L-80B08 | Ouessant sheep breed | 132688001 | C1296936
SRT | L-80B09 | Canadian Arcott sheep breed | 132689009 | C1296937
SRT | L-80B10 | Noordhollander sheep breed | 132690000 | C1296938
SRT | L-80B17 | Rijnlam-A sheep breed | 132697002 | C1296940
SRT | L-80B18 | Schoonebeker sheep breed | 132698007 | C1296941
SRT | L-80B19 | Wallis Blacknosed Sheep breed | 132699004 | C1269341
SRT | L-80B22 | Newfoundland sheep breed | 132701004 | C1296943
SRT | L-80B23 | Wallis Country Sheep breed | 132702006 | C1269342
SRT | L-80B24 | Rideau Arcott sheep breed | 132703001 | C1296944
SRT | L-80B25 | Tukidale sheep breed | 132704007 | C1296945
SRT | L-80B26 | Polwarth sheep breed | 132705008 | C1296946
SRT | L-80B27 | Ryeland sheep breed | 132706009 | C1296947
SRT | L-80B2A | Thalli sheep breed | 132707000 | C1296948
SRT | L-80B2B | Tong sheep breed | 132708005 | C1296949
SRT | L-80B2C | Touabire sheep breed | 132709002 | C1296950
SRT | L-80B2D | Tunis sheep breed | 132710007 | C1296951
SRT | L-80B2E | Tyrol Mountain sheep breed | 132711006 | C1269343
SRT | L-80B2F | Uda sheep breed | 132712004 | C1296952
SRT | L-80B33 | German Mutton Merino sheep breed | 132716001 | C1296955
SRT | L-80B34 | Medium-Wool Merino sheep breed | 132717005 | C1269345
SRT | L-80B35 | Fonthill Merino sheep breed | 132718000 | C1296956
SRT | L-80B36 | South African Mutton Merino sheep breed | 132719008 | C1269346
SRT | L-80B37 | Strong Wool Merino sheep breed | 132720002 | C1269347
SRT | L-80B38 | Poll Merino sheep breed | 132721003 | C1296957
SRT | L-80B39 | Fine Merino sheep breed | 132722005 | C1296958
SRT | L-80B3A | South African Merino sheep breed | 132723000 | C1296959
SRT | L-80B40 | Superfine Merino sheep breed | 132724006 | C1296960
SRT | L-80B47 | Baden Wurttemburg horse breed | 132731005 | C1296962
SRT | L-80B48 | British Warmblood horse breed | 132732003 | C1296963
SRT | L-80B49 | Israeli horse breed | 132733008 | C1296964
SRT | L-80B4A | French Ardennais horse breed | 132734002 | C1296965
SRT | L-80B4B | Booroola Merino sheep breed | 132735001 | C1296966
SRT | L-80B50 | Cukurova horse breed | 132736000 | C1296967
SRT | L-80B51 | Czech Coldblood horse breed | 132737009 | C1296968
SRT | L-80B52 | Czechoslovakian Small Riding Horse horse breed | 132738004 | C1269353
SRT | L-80B53 | Jianchang horse breed | 132739007 | C1296969
SRT | L-80B54 | Jielin horse breed | 132740009 | C1296970
SRT | L-80B55 | Wielkopolski horse breed | 132741008 | C1296971
SRT | L-80B56 | Eleia horse breed | 132742001 | C1296972
SRT | L-80B57 | English Cob horse breed | 132743006 | C1269354
SRT | L-80B58 | Welsh Pony horse breed | 132744000 | C1296973
SRT | L-80B59 | Welsh Pony of Cob Type horse breed | 132745004 | C1269355
SRT | L-80B5A | English Hunter horse breed | 132746003 | C1269356
SRT | L-80B5B | Eriskay Pony horse breed | 132747007 | C1296974
SRT | L-80B5C | Hackney Pony horse breed | 132748002 | C1296975
SRT | L-80B5D | Estonian Draft horse breed | 132749005 | C1296976
SRT | L-80B5E | Heihe horse breed | 132750005 | C1296977
SRT | L-80B5F | Heilongkaing horse breed | 132751009 | C1296978
SRT | L-80B65 | Danish Sport Pony horse breed | 132757008 | C1269357
SRT | L-80B66 | Kabarda horse breed | 132758003 | C1296983
SRT | L-80B67 | Kalmyk horse breed | 132759006 | C1296984
SRT | L-80B68 | Mangalarga Marchador horse breed | 132760001 | C1296985
SRT | L-80B69 | Don horse breed | 132761002 | C1296986
SRT | L-80B6A | Manipuri horse breed | 132762009 | C1296987
SRT | L-80B6B | Swiss Warmblood horse breed | 132763004 | C1296988
SRT | L-80B6C | Tavda horse breed | 132764005 | C1296989
SRT | L-80B6D | East Bulgarian horse breed | 132765006 | C1269358
SRT | L-80B6E | East Friesian (Old Type) horse breed | 132766007 | C1269359
SRT | L-80B6F | East Friesian Warmblood (Modern Type) horse breed | 132767003 | C1269360
SRT | L-80B70 | Kakhetian pig breed | 132768008 | C1296990
SRT | L-80B71 | West French White pig breed | 132769000 | C1269361
SRT | L-80B80 | Miniature Hereford cattle breed | 132770004 | C1269362
SRT | L-80B81 | Jem-Jem Zebu cattle breed | 132771000 | C1296991
SRT | L-80B82 | Minusin horse breed | 132772007 | C1296992
SRT | L-80B83 | Morochuco horse breed | 132773002 | C1296993
SRT | L-80B84 | French Trotter horse breed | 132774008 | C1296994
SRT | L-80B85 | Furioso horse breed | 132775009 | C1296995
SRT | L-80B86 | Murghese horse breed | 132776005 | C1269363
SRT | L-80B87 | Mytilene horse breed | 132777001 | C1269364
SRT | L-80B88 | Namib Desert Horse horse breed | 132778006 | C1296996
SRT | L-80B89 | Danish Oldenborg horse breed | 132779003 | C1296997
SRT | L-80B8A | Volynsk cattle breed | 132780000 | C1296998
SRT | L-80B8B | Senepol cattle breed | 132781001 | C1296999
SRT | L-80B8C | Shilluk cattle breed | 132782008 | C1297000
SRT | L-80B8D | Sar Planina sheep breed | 132783003 | C1297001
SRT | L-80B8E | Santa Inês sheep breed | 132784009 | C1321470
SRT | L-80B8F | Sahel-type sheep breed | 132785005 | C1297002
SRT | L-80B90 | Rygja sheep breed | 132786006 | C1297003
SRT | L-80B91 | Rya sheep breed | 132787002 | C1297004
SRT | L-80B92 | Moghani sheep breed | 132788007 | C1297005
SRT | L-80B93 | Rouge de l'Quest sheep breed | 132789004 | C1297006
SRT | L-80B94 | Soay sheep breed | 132790008 | C1297007
SRT | L-80B95 | South Suffolk sheep breed | 132791007 | C1269365
SRT | L-80B96 | South Wales Mountain sheep breed | 132792000 | C1269366
SRT | L-80B97 | Spælsau sheep breed | 132793005 | C1321471
SRT | L-80B98 | Spiegel sheep breed | 132794004 | C1297008
SRT | L-80B99 | St. Croix sheep breed | 132795003 | C1297009
SRT | L-80B9A | Steigar sheep breed | 132796002 | C1297010
SRT | L-80B9B | Steinschaf sheep breed | 132797006 | C1297011
SRT | L-80B9C | Welsh Mountain sheep breed | 132798001 | C1269367
SRT | L-80B9D | Swedish Fur Sheep breed | 132799009 | C1269368
SRT | L-80B9E | Teeswater sheep breed | 132800008 | C1297012
SRT | L-80B9F | Texel sheep breed | 132801007 | C1297013
SRT | L-80BA1 | Pelibüey sheep breed | 132802000 | C1321472
SRT | L-80BA2 | Morada Nova sheep breed | 132803005 | C1297014
SRT | L-80BA3 | Balkhi sheep breed | 132804004 | C1297015
SRT | L-80BA4 | Bavarian Forest sheep breed | 132805003 | C1269369
SRT | L-80BA5 | Barbados Blackbelly sheep breed | 132806002 | C1269370
SRT | L-80BA6 | Romney sheep breed | 132807006 | C1297016
SRT | L-80BA7 | Awassi sheep breed | 132808001 | C1297017
SRT | L-80BA8 | Arapawa Island sheep breed | 132809009 | C1297018
SRT | L-80BA9 | Arabi sheep breed | 132810004 | C1297019
SRT | L-80BB1 | Apennine sheep breed | 132811000 | C1269371
SRT | L-80BB2 | American Tunis sheep breed | 132812007 | C1269372
SRT | L-80BB3 | Balwen Welsh Mountain sheep breed | 132813002 | C1269373
SRT | L-80BB4 | Priangan sheep breed | 132814008 | C1269374
SRT | L-80BB5 | Rabo Largo sheep breed | 132815009 | C1297020
SRT | L-80BE6 | Muban pig breed | 132843000 | C1297039
SRT | L-80BE7 | Iban pig breed | 132844006 | C1297040
SRT | L-80BE8 | Altay sheep breed | 132845007 | C1297041
SRT | L-80BE9 | Faeroes sheep breed | 132846008 | C1297042
SRT | L-80BF6 | Pitt Island sheep breed | 132849001 | C1269382
SRT | L-80BF8 | Pinzirita sheep breed | 132851002 | C1297044
SRT | L-80BF9 | Sardinian sheep breed | 132852009 | C1297045
SRT | L-80C01 | East Friesian sheep breed | 132853004 | C1269384
SRT | L-80C02 | Ujumqin sheep breed | 132854005 | C1297046
SRT | L-80C22 | DLS sheep breed | 132855006 | C1297047
SRT | L-80C23 | Walachenschaf sheep breed | 132856007 | C1297048
SRT | L-80C24 | Outaouais Arcott sheep breed | 132857003 | C1297049
SRT | L-80C25 | Ossimi sheep breed | 132858008 | C1297050
SRT | L-80C29 | Bentheimer Landschaf sheep breed | 132859000 | C1297051
SRT | L-80C30 | Barbado sheep breed | 132860005 | C1297052
SRT | L-80C31 | Baluchi sheep breed | 132861009 | C1297053
SRT | L-86B36 | Blanc de Bouscat rabbit breed | 132888004 | C1297065
SRT | L-8A111 | American Indian Horse horse breed | 132951001 | C1297111
SRT | L-8A112 | American Mustang horse breed | 132952008 | C1297112
SRT | L-8A113 | American Quarter Horse horse breed | 132953003 | C1297113
SRT | L-8A115 | American Shetland pony horse breed | 132954009 | C1297114
SRT | L-8A116 | Anadolu horse breed | 132955005 | C1297115
SRT | L-8A117 | Andean horse breed | 132956006 | C1297116
SRT | L-8A118 | Anglo-Kabarda horse breed | 132957002 | C1297117
SRT | L-8A125 | Narym horse breed | 132960009 | C1297120
SRT | L-8A126 | National Spotted Saddle Horse horse breed | 132961008 | C1297121
SRT | L-8A127 | Nigerian horse breed | 132962001 | C1297122
SRT | L-8A128 | North Swedish Trotter horse breed | 132963006 | C1297123
SRT | L-8A129 | Oriental Horse horse breed | 132964000 | C1297124
SRT | L-8A12A | Rhineland Heavy Draft horse breed | 132965004 | C1297125
SRT | L-8A12B | Romanian Saddle Horse horse breed | 132966003 | C1297126
SRT | L-8A12C | Rottal horse breed | 132967007 | C1297127
SRT | L-8A12D | Royal Canadian Mounted Police Horse horse breed | 132968002 | C1297128
SRT | L-8A12E | Russian Saddle Horse horse breed | 132969005 | C1297129
SRT | L-8A12F | Sable Island Horse horse breed | 132970006 | C1297130
SRT | L-8A130 | Panje horse breed | 132971005 | C1297131
SRT | L-8A131 | Patibarcina horse breed | 132972003 | C1297132
SRT | L-8A132 | Pechora horse breed | 132973008 | C1297133
SRT | L-8A133 | Peneia horse breed | 132974002 | C1297134
SRT | L-8A134 | Periangan horse breed | 132975001 | C1297135
SRT | L-8A135 | Persian Arab horse breed | 132976000 | C1297136
SRT | L-8A136 | Petiso Argentino horse breed | 132977009 | C1297137
SRT | L-8A137 | Polish Draft horse breed | 132978004 | C1297138
SRT | L-8A138 | Priob horse breed | 132979007 | C1297139
SRT | L-8A139 | Rahvan horse breed | 132980005 | C1297140
SRT | L-8A13A | Salerno horse breed | 132981009 | C1297141
SRT | L-8A13B | Sandalwood horse breed | 132982002 | C1297142
SRT | L-8A13C | Sandan horse breed | 132983007 | C1297143
SRT | L-8A13D | Pindos horse breed | 132984001 | C1297144
SRT | L-8A13E | Piquira Pony horse breed | 132985000 | C1297145
SRT | L-8A13F | Pleven horse breed | 132986004 | C1297146
SRT | L-8A14A | Garrano tarpan horse X domestic horse breed | 132990002 | C1297150
SRT | L-8A14B | Konink tarpan horse X domestic horse breed | 132991003 | C1297151
SRT | L-8A14C | Asturian tarpan horse X domestic horse breed | 132992005 | C1297152
SRT | L-8A14D | Pottok tarpan horse X domestic horse breed | 132993000 | C1297153
SRT | L-8A150 | Russian Trotter horse breed | 132994006 | C1297154
SRT | L-8A151 | West African Barb horse breed | 132995007 | C1297155
SRT | L-8A152 | Fell Pony horse breed | 132996008 | C1297156
SRT | L-8A153 | National Show Horse horse breed | 132997004 | C1297157
SRT | L-8A154 | Zhemaichu horse breed | 132998009 | C1297158
SRT | L-8A155 | Yonaguni horse breed | 132999001 | C1297159
SRT | L-8A156 | Yakut horse breed | 133000000 | C1297160
SRT | L-8A157 | Tawleed horse breed | 133001001 | C1297161
SRT | L-8A158 | Western Sudan Pony horse breed | 133002008 | C1297162
SRT | L-8A159 | Welera Pony horse breed | 133003003 | C1297163
SRT | L-8A15A | Vyatka horse breed | 133004009 | C1297164
SRT | L-8A15B | Vladimir Heavy Draft horse breed | 133005005 | C1297165
SRT | L-8A15C | Vlaamperd horse breed | 133006006 | C1297166
SRT | L-8A15D | Ukrainian Saddle Horse horse breed | 133007002 | C1297167
SRT | L-8A15E | Tori horse breed | 133008007 | C1297168
SRT | L-8A15F | Tokara horse breed | 133009004 | C1297169
SRT | L-8A160 | New Kirgiz horse breed | 133010009 | C1297170
SRT | L-8A161 | Oldenburg horse breed | 133011008 | C1297171
SRT | L-8A162 | Misaki horse breed | 133012001 | C1297172
SRT | L-8A163 | Miyako horse breed | 133013006 | C1297173
SRT | L-8A164 | Mongolian horse breed | 133014000 | C1321685
SRT | L-8A165 | Waler horse breed | 133015004 | C1297174
SRT | L-8A166 | Dutch Draft horse breed | 133016003 | C1297175
SRT | L-8A167 | Egyptian horse breed | 133017007 | C1297176
SRT | L-8A168 | Estonian Native horse breed | 133018002 | C1297177
SRT | L-8A169 | Exmoor Pony horse breed | 133019005 | C1297178
SRT | L-8A16A | Faeroes Island Horse horse breed | 133020004 | C1297179
SRT | L-8A16B | Falabella horse breed | 133021000 | C1297180
SRT | L-8A16C | Dutch Warmblood horse breed | 133022007 | C1297181
SRT | L-8A16D | Dongola horse breed | 133023002 | C1297182
SRT | L-8A16E | Døle horse breed | 133024008 | C1321476
SRT | L-8A16F | Djerma horse breed | 133025009 | C1297183
SRT | L-8A170 | Deliboz horse breed | 133026005 | C1297184
SRT | L-8A171 | Dartmoor Pony horse breed | 133027001 | C1297185
SRT | L-8A172 | Crioulo horse breed | 133028006 | C1297186
SRT | L-8A173 | Finnhorse horse breed | 133029003 | C1297187
SRT | L-8A174 | Sanfratello horse breed | 133030008 | C1297188
SRT | L-8A175 | Morab horse breed | 133031007 | C1297189
SRT | L-8A176 | Moyle horse breed | 133032000 | C1297190
SRT | L-8A177 | Mustang horse breed | 133033005 | C1297191
SRT | L-8A178 | M'Bayar horse breed | 133034004 | C1297192
SRT | L-8A179 | Lusitano horse breed | 133035003 | C1297193
SRT | L-8A17A | Newfoundland Pony horse breed | 133036002 | C1297194
SRT | L-8A17B | Noma horse breed | 133037006 | C1297195
SRT | L-8A17C | Nooitgedacht Pony horse breed | 133038001 | C1297196
SRT | L-8A17D | Nordland horse breed | 133039009 | C1297197
SRT | L-8A17E | Noric horse breed | 133040006 | C1297198
SRT | L-8A17F | North Swedish Horse horse breed | 133041005 | C1297199
SRT | L-8A180 | Northeastern horse breed | 133042003 | C1297200
SRT | L-8A181 | Kisber Felver horse breed | 133043008 | C1297201
SRT | L-8A182 | Anglo-Arab horse breed | 133044002 | C1297202
SRT | L-8A183 | Nonius horse breed | 133045001 | C1297203
SRT | L-8A184 | Nooitgedacht horse breed | 133046000 | C1297204
SRT | L-8A185 | Iomud horse breed | 133047009 | C1297205
SRT | L-8A186 | Jutland horse breed | 133048004 | C1297206
SRT | L-8A187 | Karabair horse breed | 133049007 | C1297207
SRT | L-8A188 | Karabakh horse breed | 133050007 | C1297208
SRT | L-8A189 | Kazakh horse breed | 133051006 | C1297209
SRT | L-8A18A | Mangalarga horse breed | 133052004 | C1297210
SRT | L-8A18B | Kirdi Pony horse breed | 133053009 | C1297211
SRT | L-8A18C | Kiso horse breed | 133054003 | C1297212
SRT | L-8A18D | Kladruby horse breed | 133055002 | C1297213
SRT | L-8A18E | Knabstrup horse breed | 133056001 | C1297214
SRT | L-8A18F | Kushum horse breed | 133057005 | C1297215
SRT | L-8A190 | Kustanai horse breed | 133058000 | C1297216
SRT | L-8A191 | Latvian horse breed | 133059008 | C1297217
SRT | L-8A192 | Lithuanian Heavy Draft horse breed | 133060003 | C1297218
SRT | L-8A193 | Lokai horse breed | 133061004 | C1297219
SRT | L-8A194 | Kiger Mustang horse breed | 133062006 | C1297220
SRT | L-8A195 | Pony of the Americas horse breed | 133063001 | C1297221
SRT | L-8A196 | Pintabian horse breed | 133064007 | C1297222
SRT | L-8A197 | Pantaneiro horse breed | 133065008 | C1297223
SRT | L-8A198 | Orlov Trotter horse breed | 133066009 | C1297224
SRT | L-8A199 | Northern Ardennais horse breed | 133067000 | C1297225
SRT | L-8A19A | Abtenauer horse breed | 133068005 | C1297226
SRT | L-8A19B | Adaev horse breed | 133069002 | C1297227
SRT | L-8A19C | Albanian horse breed | 133070001 | C1297228
SRT | L-8A19E | Alter Real horse breed | 133071002 | C1297229
SRT | L-8A19F | American Bashkir Curly horse breed | 133072009 | C1297230
SRT | L-8A1A1 | Poitou Mule Producer horse breed | 133073004 | C1297231
SRT | L-8A1A2 | Polesian horse breed | 133074005 | C1297232
SRT | L-8A1A3 | Sardinian Anglo-Arab horse breed | 133075006 | C1297233
SRT | L-8A1A4 | Sardinian Pony horse breed | 133076007 | C1297234
SRT | L-8A1A5 | Sarvar horse breed | 133077003 | C1297235
SRT | L-8A1A6 | Schleswig horse breed | 133078008 | C1297236
SRT | L-8A1A7 | Schwarzwalder Fuchse horse breed | 133079000 | C1297237
SRT | L-8A1A8 | Senne horse breed | 133080002 | C1297238
SRT | L-8A1A9 | Shan horse breed | 133081003 | C1297239
SRT | L-8A1AA | Silesian horse breed | 133082005 | C1297240
SRT | L-8A1AB | Sini horse breed | 133083000 | C1297241
SRT | L-8A1AC | Skyros horse breed | 133084006 | C1297242
SRT | L-8A1AD | Slovak Warmblood horse breed | 133085007 | C1297243
SRT | L-8A1AE | Sokolka horse breed | 133086008 | C1297244
SRT | L-8A1AF | South African Miniature horse breed | 133087004 | C1297245
SRT | L-8A1B1 | South German Coldblood horse breed | 133088009 | C1297246
SRT | L-8A1B2 | Southwest Spanish Mustang horse breed | 133089001 | C1297247
SRT | L-8A1B4 | Spanish-American Horse horse breed | 133090005 | C1297248
SRT | L-8A1B5 | Spanish Anglo-Arab horse breed | 133091009 | C1297249
SRT | L-8A1B6 | Spanish Colonial Horse horse breed | 133092002 | C1297250
SRT | L-8A1B7 | Spiti horse breed | 133093007 | C1297251
SRT | L-8A1B8 | Sulawesi horse breed | 133094001 | C1297252
SRT | L-8A1B9 | Criollo horse breed | 133095000 | C1297253
SRT | L-8A1BA | Hequ horse breed | 133096004 | C1297254
SRT | L-8A1BB | Connemara Pony horse breed | 133097008 | C1297255
SRT | L-8A1BC | Colorado Ranger horse breed | 133098003 | C1297256
SRT | L-8A1BD | Dales Pony horse breed | 133099006 | C1297257
SRT | L-8A1BE | Gotland horse breed | 133100003 | C1297258
SRT | L-8A1BF | Chincoteague Pony horse breed | 133101004 | C1297259
SRT | L-8A1C1 | Hokkaido horse breed | 133102006 | C1297260
SRT | L-8A1C2 | Highland Pony horse breed | 133103001 | C1297261
SRT | L-8A1C3 | Groningen horse breed | 133104007 | C1297262
SRT | L-8A1C4 | Cuban Pinto horse breed | 133105008 | C1297263
SRT | L-8A1C5 | Fleuve horse breed | 133106009 | C1297264
SRT | L-8A1C6 | Golden American Saddlebred horse breed | 133107000 | C1297265
SRT | L-8A1C7 | Gidran horse breed | 133108005 | C1297266
SRT | L-8A1C8 | Gelderland horse breed | 133109002 | C1320153
SRT | L-8A1C9 | Galician Pony horse breed | 133110007 | C1297267
SRT | L-8A1CA | Friesian horse breed | 133111006 | C1297268
SRT | L-8A1CB | Frederiksborg horse breed | 133112004 | C1297269
SRT | L-8A1CC | Fouta horse breed | 133113009 | C1297270
SRT | L-8A1CD | Florida Cracker horse breed | 133114003 | C1297271
SRT | L-8A1CE | Guangxi horse breed | 133115002 | C1297272
SRT | L-8A1CF | Ardennes horse breed | 133116001 | C1297273
SRT | L-8A1D1 | American Walking Pony horse breed | 133117005 | C1297274
SRT | L-8A1D2 | Azteca horse breed | 133118000 | C1297275
SRT | L-8A1D3 | American Cream Draft horse breed | 133119008 | C1297276
SRT | L-8A1D4 | Altai horse breed | 133120002 | C1297277
SRT | L-8A1D5 | Akhal-Teke horse breed | 133121003 | C1297278
SRT | L-8A1D6 | Abyssinian horse breed | 133122005 | C1297279
SRT | L-8A1D7 | Bhirum Pony horse breed | 133123000 | C1297280
SRT | L-8A1D8 | Cheju horse breed | 133124006 | C1297281
SRT | L-8A1D9 | Cayuse horse breed | 133125007 | C1297282
SRT | L-8A1DA | Caspian horse breed | 133126008 | C1297283
SRT | L-8A1DB | Carthusian horse breed | 133127004 | C1297284
SRT | L-8A1DC | Campolina horse breed | 133128009 | C1297285
SRT | L-8A1DD | Byelorussian Harness horse breed | 133129001 | C1297286
SRT | L-8A1DE | Budyonny horse breed | 133130006 | C1297287
SRT | L-8A1DF | Australian Brumby horse breed | 133131005 | C1297288
SRT | L-8A1E1 | Australian Stock Horse horse breed | 133132003 | C1297289
SRT | L-8A1E2 | Basuto Pony horse breed | 133133008 | C1297290
SRT | L-8A1E3 | Bashkir Curly horse breed | 133134002 | C1297291
SRT | L-8A1E4 | Bashkir horse breed | 133135001 | C1297292
SRT | L-8A1E5 | Barb horse breed | 133136000 | C1297293
SRT | L-8A1E6 | Ban-ei horse breed | 133137009 | C1297294
SRT | L-8A1E7 | Carpathian Pony horse breed | 133138004 | C1297295
SRT | L-8A1E8 | Baluchi horse breed | 133139007 | C1297296
SRT | L-8A1E9 | Balearic horse breed | 133140009 | C1297297
SRT | L-8A1EA | Chilean Corralero horse breed | 133141008 | C1297298
SRT | L-8A1EB | Breton horse breed | 133142001 | C1297299
SRT | L-8A1EC | Taishuh horse breed | 133143006 | C1297300
SRT | L-8A1ED | Swedish Warmblood horse breed | 133144000 | C1297301
SRT | L-8A1EE | Sudan Country-Bred horse breed | 133145004 | C1297302
SRT | L-8A1EF | Spanish-Norman horse breed | 133146003 | C1297303
SRT | L-8A1F1 | Spanish Barb horse breed | 133147007 | C1297304
SRT | L-8A1F2 | Soviet Heavy Draft horse breed | 133148002 | C1297305
SRT | L-8A1F3 | Sorraia horse breed | 133149005 | C1297306
SRT | L-8A1F4 | Somali Pony horse breed | 133150005 | C1297307
SRT | L-8A1F5 | Tersk horse breed | 133151009 | C1297308
SRT | L-8A1F6 | Shagya horse breed | 133152002 | C1297309
SRT | L-8A1F7 | Selle Francais horse breed | 133153007 | C1297310
SRT | L-8A1F8 | Sanhe horse breed | 133154001 | C1297311
SRT | L-8A1FA | Russian Heavy Draft horse breed | 133155000 | C1297312
SRT | L-8A1FB | Rocky Mountain Horse horse breed | 133156004 | C1297313
SRT | L-8A1FC | Racking Horse horse breed | 133157008 | C1297314
SRT | L-8A1FD | Quarter Pony horse breed | 133158003 | C1297315
SRT | L-8A1FE | Quarab horse breed | 133159006 | C1297316
SRT | L-8A1FF | Single-Footing Horse horse breed | 133160001 | C1297317
SRT | L-8B105 | Tuy Hoa Hairless pig breed | 133161002 | C1297318
SRT | L-8B106 | Hainan pig breed | 133162009 | C1297319
SRT | L-8B107 | Sino-Vietnamese pig breed | 133163004 | C1297320
SRT | L-8B108 | Bo Xu pig breed | 133164005 | C1297321
SRT | L-8B109 | Thuoc Nhieu pig breed | 133165006 | C1297322
SRT | L-8B111 | Burmese pig breed | 133166007 | C1297323
SRT | L-8B112 | Chin pig breed | 133167003 | C1297324
SRT | L-8B113 | Siamese pig breed | 133168008 | C1297325
SRT | L-8B114 | Hailum pig breed | 133169000 | C1297326
SRT | L-8B115 | Kwai pig breed | 133170004 | C1297327
SRT | L-8B116 | Raad pig breed | 133171000 | C1297328
SRT | L-8B117 | Akha pig breed | 133172007 | C1297329
SRT | L-8B118 | South China pig breed | 133173002 | C1297330
SRT | L-8B119 | South China Black pig breed | 133174008 | C1297331
SRT | L-8B121 | Balinese pig breed | 133175009 | C1297332
SRT | L-8B122 | Diani pig breed | 133176005 | C1297333
SRT | L-8B123 | Kaman pig breed | 133177001 | C1297334
SRT | L-8B124 | Ashanti Dwarf pig breed | 133178006 | C1297335
SRT | L-8B125 | Koronadal pig breed | 133179003 | C1297336
SRT | L-8B126 | Ohmini pig breed | 133180000 | C1297337
SRT | L-8B127 | Clawn pig breed | 133181001 | C1297338
SRT | L-8B128 | Inobuta (inter-species hybrid) pig breed | 133182008 | C1297339
SRT | L-8B129 | Kangaroo Island pig breed | 133183003 | C1297340
SRT | L-8B130 | Captain Cooker pig breed | 133184009 | C1297341
SRT | L-8B131 | West African pig breed | 133185005 | C1297342
SRT | L-8B132 | Nigerian pig breed | 133186006 | C1297343
SRT | L-8B133 | Bakosi pig breed | 133187002 | C1297344
SRT | L-8B134 | Windsnyer pig breed | 133188007 | C1297345
SRT | L-8B135 | Kolbroek pig breed | 133189004 | C1297346
SRT | L-8B136 | South African Landrace pig breed | 133190008