DICOM PS3.3 2016e - Information Object Definitions

C.34.5 Patient Specification Module
Prev | C.34 Procedure Protocol Modules | Next

### C.34.5 Patient Specification Module

Table C.34.5-1 contains patient characteristics for which a Protocol may be considered applicable.

Table C.34.5-1. Patient Specification Module Attributes

Attribute Name | Tag | Type | Attribute Description
--- | --- | --- | ---
Patient Specification Sequence | (0018,9911) | 1 | Constraints on the characteristics of patients to which the protocol is applicable. One or more Items shall be included in this Sequence.
>Include Table 10.25-1 “Attribute Value Constraint Macro Attributes” | Only Attributes defined in the Patient Module (See Table C.7-1) or the Patient Study Module (See Table C.7-4a) may be specified as Selector Attributes. See Section C.34.9.3. The same Attribute shall not be constrained in more than one item in this sequence. ### Note Since these constraints are being placed on the attributes that will appear in the Patient Module or Patient Study Module of the resulting instances, the Patient Specification Sequence will not appear in the Selector Sequence Pointer (0072,0052).

Attributes that might commonly be specified in this sequence include:

- Patient's Age (0010,1010), to indicate Protocols for age ranges like adults, pediatrics, infants

- Patient's Sex (0010,0040), to indicate Protocols for males, females, both

- Patient's Weight (0010,1030), to indicate Protocols for different weight ranges

Prev | Up | Next
C.34.4 Clinical Trial Context Module | Home | C.34.6 Equipment Specification Module

DICOM PS3.3 2016e - Information Object Definitions