DICOM PS3.17 2016e - Explanatory Information

AAA.3 Implantation Plan SR Document Total Hip Replacement Example
Prev | AAA Implantation Plan SR Document (Informative) | Next

## AAA.3 Implantation Plan SR Document Total Hip Replacement Example

The following example shows the planning result of a simple THR (Total Hip Replacement) without any registration information. One Patient Image was used and one visualization was produced. One Femoral Stem, one Femoral Head, one Acetabular Bearing Insert and one Acetabular Fixation Cup were selected to be implanted (see Figure AAA.3-1).

Figure AAA.3-1. Total Hip Replacement Components

Table AAA.3-1. Total Hip Replacement Example

Node | Code Meaning of Concept Name | Code Meaning or Example Value | TID
--- | --- | --- | ---
1 | Implantation Plan |  | TID 7000
1.1 | Language of Content Item and Descendants | English | TID 1204
1.2 | Observation Context |  | TID 1001
1.2.1 | Person Observer Name | Dr. Michael Mueller | TID 1003
1.2.2 | Subject Name | John Smith | TID 1007
1.2.3 | Subject ID | 1.2.3.4.5.6.7.8.9 | TID 1007
1.2.4 | Subject Species | L-85003, SRT, "homo sapiens" | TID 1007
1.3 | Implant Component List |  | TID 7000
1.3.1 | Implant Assembly Template | Reference to THR | TID 7000
1.3.2 | Selected Implant Component |  | TID 7000
1.3.2.1 | Component ID | 1 | TID 7000
1.3.2.2 | Component Type | 112310 | TID 7000
1.3.2.3 |  | Reference to Implant Template "FS1000" (derived) | TID 7000
1.3.2.4 | Frame Of Reference UID | 1.2.3.4.1 | TID 7000
1.3.2.5 | Manufacturer Implant Template | Reference to Implant Template "FS1000" (original) | TID 7000
1.3.3 | Selected Implant Component |  | TID 7000
1.3.3.1 | Component ID | 2 | TID 7000
1.3.3.2 | Component Type | A-04459 | TID 7000
1.3.3.3 |  | Reference to Implant Template "FH2000" (derived) | TID 7000
1.3.3.4 | Frame Of Reference UID | 1.2.3.4.2 | TID 7000
1.3.3.5 | Manufacturer Implant Template | Reference to Implant Template "FH2000" (original) | TID 7000
1.3.4 | Selected Implant Component |  | TID 7000
1.3.4.1 | Component ID | 3 | TID 7000
1.3.4.2 | Component Type | 112305 | TID 7000
1.3.4.3 |  | Reference to Implant Template "AFC3000" (derived) | TID 7000
1.3.4.4 | Frame Of Reference UID | 1.2.3.4.3 | TID 7000
1.3.4.5 | Manufacturer Implant Template | Reference to Implant Template "AFC3000" (original) | TID 7000
1.3.5 | Selected Implant Component |  | TID 7000
1.3.5.1 | Component ID | 4 | TID 7000
1.3.5.2 | Component Type | 112306 | TID 7000
1.3.5.3 |  | Reference to Implant Template "ABI4000" (derived) | TID 7000
1.3.5.4 | Frame Of Reference UID | 1.2.3.4.4 | TID 7000
1.3.5.5 | Manufacturer Implant Template | Reference to Implant Template "ABI4000" (original) | TID 7000
1.4. | Assembly |  | TID 7000
1.4.1 | Component Connection |  | TID 7000
1.4.1.1 | Connected Implantation Plan Component |  | TID 7000
1.4.1.1.1 | Component ID | 3 | TID 7000
1.4.1.1.2 | Mating Feature Set ID | 1 | TID 7000
1.4.1.1.3 | Mating Feature ID | 1 | TID 7000
1.4.1.2 | Connected Implantation Plan Component |  | TID 7000
1.4.1.2.1 | Component ID | 4 | TID 7000
1.4.1.2.2 | Mating Feature Set ID | 1 | TID 7000
1.4.1.2.3 | Mating Feature ID | 1 | TID 7000
1.4.2 | Component Connection |  | TID 7000
1.4.2.1 | Connected Implantation Plan Component |  | TID 7000
1.4.2.1.1 | Component ID | 2 | TID 7000
1.4.2.1.2 | Mating Feature Set ID | 1 | TID 7000
1.4.2.1.3 | Mating Feature ID | 1 | TID 7000
1.4.2.2 | Connected Implantation Plan Component |  | TID 7000
1.4.2.2.1 | Component ID | 1 | TID 7000
1.4.2.2.2 | Mating Feature Set ID | 1 | TID 7000
1.4.2.2.3 | Mating Feature ID | 2 | TID 7000
1.4.3 | Component Connection |  | TID 7000
1.4.3.1 | Connected Implantation Plan Component |  | TID 7000
1.4.3.1.1 | Component ID | 2 | TID 7000
1.4.3.1.2 | Mating Feature Set ID | 2 | TID 7000
1.4.3.1.3 | Mating Feature ID | 1 | TID 7000
1.4.3.2 | Connected Implantation Plan Component |  | TID 7000
1.4.3.2.1 | Component ID | 4 | TID 7000
1.4.3.2.2 | Mating Feature Set ID | 2 | TID 7000
1.4.3.2.3 | Mating Feature ID | 2 | TID 7000
1.5 | Information used for planning |  | TID 7000
1.5.1 | Patient Image | Reference to Image 01 | TID 7000
1.5.1.2 | Horizontal Pixel Spacing | 0.2 mm/pixel | TID 7000
1.5.1.3 | Vertical Pixel Spacing | 0.2 mm/pixel | TID 7000
1.6 | Planning Information for Intraoperative Usage |  | TID 7000
1.6.1 | Supporting Information | Reference to Encapsulated PDF-Document 01 | TID 7000
1.6.2 | Derived Images | Reference to Visualization 01 | TID 7000

Prev | Up | Next
AAA.2 Relationship Between Implant Template and Implantation Plan | Home | AAA.4 Implantation Plan SR Document Dental Drilling Template Example

DICOM PS3.17 2016e - Explanatory Information