DICOM PS3.17 2016d - Explanatory Information

E.3 Mammography CAD SR Examples
Prev | E Mammography CAD (Informative) | Next

## E.3 Mammography CAD SR Examples

The following is a simple and non-comprehensive illustration of an encoding of the Mammography CAD SR IOD for Mammography computer aided detection results. For brevity, some Mandatory content items are not included, such as several acquisition context content items for the images in the Image Library.

### E.3.1 Example 1: Calcification and Mass Detection With No Findings

A mammography CAD device processes a typical screening mammography case, i.e., there are four films and no cancer. Mammography CAD runs both density and calcification detection successfully and finds nothing. The mammograms resemble:

Figure E.3-1. Mammograms as Described in Example 1

The content tree structure would resemble:

Node | Code Meaning of Concept Name | Code Meaning or Example Value | TID
--- | --- | --- | ---
1 | Mammography CAD Report |  | TID 4000
1.1 | Image Library |  | TID 4000
1.1.1 |  | IMAGE 1 | TID 4020
1.1.1.1 | Image Laterality | Right | TID 4020
1.1.1.2 | Image View | Cranio-caudal | TID 4020
1.1.1.3 | Study Date | 19980101 | TID 4020
1.1.2 |  | IMAGE 2 | TID 4020
1.1.2.1 | Image Laterality | Left | TID 4020
1.1.2.2 | Image View | Cranio-caudal | TID 4020
1.1.2.3 | Study Date | 19980101 | TID 4020
1.1.3 |  | IMAGE 3 | TID 4020
1.1.3.1 | Image Laterality | Right | TID 4020
1.1.3.2 | Image View | Medio-lateral oblique | TID 4020
1.1.3.3 | Study Date | 19980101 | TID 4020
1.1.4 |  | IMAGE 4 | TID 4020
1.1.4.1 | Image Laterality | Left | TID 4020
1.1.4.2 | Image View | Medio-lateral oblique | TID 4020
1.1.4.3 | Study Date | 19980101 | TID 4020
1.2 | CAD Processing and Findings Summary | All algorithms succeeded; without findings | TID 4001
1.3 | Summary of Detections | Succeeded | TID 4000
1.3.1 | Successful Detections |  | TID 4015
1.3.1.1 | Detection Performed | Mammography breast density | TID 4017
1.3.1.1.1 | Algorithm Name | "Density Detector" | TID 4019
1.3.1.1.2 | Algorithm Version | "V3.7" | TID 4019
1.3.1.1.3 |  | Reference to node 1.1.1 | TID 4017
1.3.1.1.4 |  | Reference to node 1.1.2 | TID 4017
1.3.1.1.5 |  | Reference to node 1.1.3 | TID 4017
1.3.1.1.6 |  | Reference to node 1.1.4 | TID 4017
1.3.1.2 | Detection Performed | Individual Calcification | TID 4017
1.3.1.2.1 | Algorithm Name | "Calc Detector" | TID 4019
1.3.1.2.2 | Algorithm Version | "V2.4" | TID 4019
1.3.1.2.3 |  | Reference to node 1.1.1 | TID 4017
1.3.1.2.4 |  | Reference to node 1.1.2 | TID 4017
1.3.1.2.5 |  | Reference to node 1.1.3 | TID 4017
1.3.1.2.6 |  | Reference to node 1.1.4 | TID 4017
1.4 | Summary of Analyses | Not Attempted | TID 4000

Prev | Up | Next
E.2 Mammography CAD SR Observation Context Encoding | Home | E.3.2 Example 2: Calcification and Mass Detection With Findings

DICOM PS3.17 2016d - Explanatory Information