DICOM PS3.3 2016d - Information Object Definitions

A.2.3 CR Image IOD Module Table
Prev | A.2 Computed Radiography Image IOD | Next

### A.2.3 CR Image IOD Module Table

Table A.2-1 specifies the Modules of the CR Image IOD.

Table A.2-1. CR Image IOD Modules

IE | Module | Reference | Usage
--- | --- | --- | ---
Patient | Patient | C.7.1.1 | M
Clinical Trial Subject | C.7.1.3 | U
Study | General Study | C.7.2.1 | M
Patient Study | C.7.2.2 | U
Clinical Trial Study | C.7.2.3 | U
Series | General Series | C.7.3.1 | M
CR Series | C.8.1.1 | M
Clinical Trial Series | C.7.3.2 | U
Equipment | General Equipment | C.7.5.1 | M
Image | General Image | C.7.6.1 | M
General Reference | C.12.4 | U
Image Pixel | C.7.6.3 | M
Contrast/Bolus | C.7.6.4 | C - Required if contrast media was used in this image
Display Shutter | C.7.6.11 | U
Device | C.7.6.12 | U
Specimen | C.7.6.22 | U
CR Image | C.8.1.2 | M
Overlay Plane | C.9.2 | U
Modality LUT | C.11.1 | U
VOI LUT | C.11.2 | U
SOP Common | C.12.1 | M
Common Instance Reference | C.12.2 | U

### Note

The Curve Module (Retired) was previously included in the Image IE for this IOD but has been retired. See PS3.3-2004.

Prev | Up | Next
A.2.2 CR Image IOD Entity-Relationship Model | Home | A.3 Computed Tomography Image IOD

DICOM PS3.3 2016d - Information Object Definitions