TID 3560 Derived Hemodynamic Measurements
Prev | Hemodynamics Report Templates | Next

### TID 3560 Derived Hemodynamic Measurements

The Derived Hemodynamic Measurements Template consists of a CONTAINER containing measurements derived from one or more other measurements. These measurements are associated with a particular procedure phase, but not necessarily with a particular anatomic location.

Type:

  Extensible

Order:

  Significant

Table TID 3560. Derived Hemodynamic Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122126, DCM, "Derived Hemodynamic Measurements") | 1 | U |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3614 “Valve Areas, Non-mitral” $Units = EV (cm2, UCUM, "cm2") $Equation = DT (122262, DCM, "Area = Flow / 44.5 * sqrt(Gradient[mmHg]) ")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-02320, SRT, "Mitral Valve Area") $Units = EV (cm2, UCUM, "cm2") $Equation = DT (122263, DCM, "MVA = Flow / 38.0 * sqrt(Gradient[mmHg]) ")
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3615 “Valve Areas” $ModType = EV (121425, DCM, "Index") $ModValue = EV (8277-6, LN, "Body Surface Area") $Units = DT (cm2/m2, UCUM, "cm2/m2")
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3616 “Hemodynamic Period Measurements” $Units = DT (s/min, UCUM, "s/min")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3617 “Valve Flows” $Units = DT (ml/s, UCUM, "ml/s")
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32110, SRT, "Cardiac Index") $Units = DT (l/min/m2, UCUM, "l/min/m2")
8 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3529 “Arterial / Venous Content” $Units = DT (ml/dl, UCUM, "ml/dl")
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3618 “Hemodynamic Flows” $Units = DT (l/min, UCUM, "l/min")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8736-1, LN, "FICK Cardiac Output") $Units = DT (l/min, UCUM, "l/min")
11 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8750-2, LN, "FICK Cardiac Index") $Units = DT (l/min/m2, UCUM, "l/min/m2")
12 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122229, DCM, "Arteriovenous difference") $Units = DT (ml/dl, UCUM, "ml/dl")
13 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = BCID 3620 “Hemodynamic Ratios” $Units = DT ({ratio}, UCUM, "ratio")
14 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122237, DCM, "Corrected Sinus Node Recovery Time") $Units = DT (ms, UCUM, "ms")
15 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8861-7, LN, "Left Ventricular Stroke Work") $Units = DT (gf.m, UCUM, "gf.m")
16 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8862-5, LN, "Right Ventricular Stroke Work") $Units = DT (gf.m, UCUM, "gf.m")
17 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8863-3, LN, "Left Ventricular Stroke Work Index") $Units = DT (gf.m/m2, UCUM, "gf.m/m2")
18 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8864-1, LN, "Right Ventricular Stroke Work Index") $Units = DT (gf.m/m2, UCUM, "gf.m/m2")
19 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122238, DCM, "Max volume normalized to 50mmHg pulse pressure") $Units = DT (ml, UCUM, "ml")
20 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122239, DCM, "Oxygen Consumption") $Units = DT (ml/min, UCUM, "ml/min") $Equation = BCID 3664 “Oxygen Consumption Equations and Tables”
21 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (19217-9, LN, "Oxygen partial pressure at 50% saturation (P50) ") $Units = DCID 3500 “Pressure Units” $Equation = BCID 3666 “P50 Equations”
22 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (11556-8, LN, "Blood Oxygen partial pressure") $Units = DCID 3500 “Pressure Units”
23 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3618 “Hemodynamic Flows” $ModType = EV (121425, DCM, "Index") $ModValue = EV (8277-6, LN, "Body Surface Area") $Units = DT (l/min/m2, UCUM, "l/min/m2")
24 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3619 “Hemodynamic Resistance Measurements” $Units = BCID 3502 “Hemodynamic Resistance Units”
25 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3619 “Hemodynamic Resistance Measurements” $ModType = EV (121425, DCM, "Index") $ModValue = EV (8277-6, LN, "Body Surface Area") $Units = BCID 3503 “Indexed Hemodynamic Resistance Units”
26 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122227, DCM, "Left to Right Flow") $Units = DT (l/min, UCUM, "l/min")
27 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122228, DCM, "Right to Left Flow") $Units = DT (l/min, UCUM, "l/min")
28 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $Units = DT (ml, UCUM, "ml")
29 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $ModType = EV (121425, DCM, "Index") $ModValue = EV (8277-6, LN, "Body Surface Area") $Units = DT (ml/m2, UCUM, "ml/m2")
30 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-042BA, SRT, "Total blood volume") $Units = DT (l, UCUM, "l")
31 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3667 “Framingham Scores” $Units = DT (%, UCUM, "%") $Equation = DCID 3668 “Framingham Tables”

Prev | Up | Next
TID 3550 Pressure Waveform Measurements | Home | TID 3570 Summary, Hemodynamics