TID 3107 Device Used
Prev | Procedure Log IOD Templates | Next

### TID 3107 Device Used

The Device Used Template allows recording of the use of interventional diagnostic and therapeutic devices.

The identification of one device used to deploy another device (e.g., a balloon catheter to deploy a stent) may be described with two entries, with one identified as a deployment device in the Concept Modifier of Row 6 of this template, and linked by the same Procedure Action ID in the Log Entry Qualifiers of the included TID 3010 “Log Entry Qualifiers”.

Type:

  Extensible

Order:

  Significant

Table TID 3107. Device Used

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DCID 3422 “Device Use Actions” | 1 | M |  | BCID 3429 “Catheterization Devices”
2 | > | HAS PROPERTIES | CODE | EV (121150, DCM, "Device Code") | 1-n | U |  | Vendor or local bar coded values
3 | > | HAS PROPERTIES | TEXT | EV (121145, DCM, "Description of Material") | 1 | U |  |
4 | > | HAS PROPERTIES | NUM | DCID 3423 “Numeric Device Characteristics” | 1-n | U |  |
5 | > | HAS PROPERTIES | CODE | EV (G-C0E9, SRT, "Procedure site") | 1 | U |  | BCID 3630 “Cardiovascular Anatomic Locations”
6 | > | HAS CONCEPT MOD | CODE | EV (G-C0E8, SRT, "Has Intent") | 1 | U |  | DT (121155, DCM, "Deployment")
7 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

Prev | Up | Next
TID 3106 Drugs/Contrast Administered | Home | TID 3108 Intervention