V.4.3 Hanging Protocol Display Module
Prev | V.4 Neurosurgery Planning Hanging Protocol Example | Next

### V.4.3 Hanging Protocol Display Module

- Display Sets Sequence:

  - Figure V.4.3-1. Group #1 is CT only display (current CT)

  - Item 1:

    - Display Set Number: 1

    - Display Set Presentation Group: 1

    - Image Set Number: 2

    - Image Boxes Sequence:

      - Item 1: [lower left quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (0/3072, 512/2560), (512/3072,0/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Reformatting Operation Type: "MPR"

    - Reformatting Thickness: 5

    - Reformatting Interval: 5

    - Reformatting Operation Initial View Direction: "CORONAL"

    - Display Set Patient Orientation: "L\F"

    - VOI Type: BRAIN

    - Display Set Presentation Group Description: "Current CT only"

  - Item 2:

    - Display Set Number: 2

    - Display Set Presentation Group: 1

    - Image Set Number: 2

    - Image Boxes Sequence:

      - Item 1: [upper left quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (0/3072, 1024/2560), (512/3072, 512/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Reformatting Operation Type: "MPR"

    - Reformatting Thickness: 5

    - Reformatting Interval: 5

    - Reformatting Operation Initial View Direction: "SAGITTAL"

    - Display Set Patient Orientation: "P\F"

    - VOI Type: BRAIN

  - Item 3:

    - Display Set Number: 3

    - Display Set Presentation Group: 1

    - Image Set Number: 2

    - Image Boxes Sequence:

      - Item 1: [upper right quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (512/3072, 1024/2560), (1024/3072, 512/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Display Set Patient Orientation: "L\P"

    - VOI Type: BRAIN

    - Show Graphic Annotation Flag: "YES"

  - Item 4:

    - Display Set Number: 4

    - Display Set Presentation Group: 1

    - Image Set Number: 2

    - Image Boxes Sequence:

      - Item 1: [lower right quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (512/3072, 512/2560), (1024/3072, 0/2560)

        - Image Box Layout Type: "PROCESSED"

    - Filter Operations Sequence:

      - Item 1:

        - Selector Attribute: (0008,0008) [Image Type]

        - Selector Attribute VR: "CS"

        - Selector CS Value: "LOCALIZER "

        - Selector Value Number: 3

        - Filter-by Operator: "NOT_MEMBER_OF"

    - Sorting Operations Sequence: zero length

    - Reformatting Operation Type: "3D_RENDERING"

    - Reformatting Operation Initial View Direction: "CORONAL"

    - 3D Rendering Type: "VOLUME"

    - Display Set Patient Orientation: "X\F"

    - Show Graphic Annotation Flag: "NO"

  - Item 5:

    - Display Set Number: 5

    - Display Set Presentation Group: 1

    - Image Set Number: 2

    - Image Boxes Sequence:

      - Item 1: [entire 2048x2560 space]

        - Image Box Number: 1

        - Display Environment Spatial Position: (1024/3072, 2560/2560), (3072/3072, 0/2560)

        - Image Box Layout Type: "TILED"

        - Image Box Tile Horizontal Dimension: 3

        - Image Box Tile Vertical Dimension: 4

        - Image Box Scroll Direction: "VERTICAL"

        - Image Box Small Scroll Type: "ROW_COLUMN"

        - Image Box Small Scroll Amount: 1

        - Image Box Large Scroll Type: "PAGE"

        - Image Box Large Scroll Amount: 1

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Display Set Patient Orientation: "L\P"

    - VOI Type: BRAIN

    - Show Graphic Annotation Flag: "YES"

  - Figure V.4.3-2. Group #2 is MR only display

  - Item 6:

    - Display Set Number: 6

    - Display Set Presentation Group: 2

    - Image Set Number: 1

    - Image Boxes Sequence:

      - Item 1: [lower left quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (0/3072, 512/2560), (512/3072,0/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Reformatting Operation Type: "MPR"

    - Reformatting Thickness: 5

    - Reformatting Interval: 5

    - Reformatting Operation Initial View Direction: "CORONAL"

    - Display Set Patient Orientation: "P\F"

    - Display Set Presentation Group Description: "MR only"

  - Item 7:

    - Display Set Number: 7

    - Display Set Presentation Group: 2

    - Image Set Number: 1

    - Image Boxes Sequence:

      - Item 1: [upper left quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (0/3072, 1024/2560), (512/3072, 512/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Reformatting Operation Type: "MPR"

    - Reformatting Thickness: 5

    - Reformatting Interval: 5

    - Reformatting Operation Initial View Direction: "SAGITTAL"

    - Display Set Patient Orientation: "P\F"

  - Item 8:

    - Display Set Number: 8

    - Display Set Presentation Group: 2

    - Image Set Number: 1

    - Image Boxes Sequence:

      - Item 1: [upper right quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (512/3072, 1024/2560), (1024/3072, 512/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Display Set Patient Orientation: "L\P"

  - Item 9:

    - Display Set Number: 9

    - Display Set Presentation Group: 2

    - Image Set Number: 1

    - Image Boxes Sequence:

      - Item 1: [lower right quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (512/3072, 512/2560), (1024/3072, 0/2560)

        - Image Box Layout Type: "PROCESSED"

    - Filter Operations Sequence: zero length

    - Sorting Operations Sequence: zero length

    - Reformatting Operation Type: "3D_RENDERING"

    - Reformatting Operation Initial View Direction: "CORONAL"

    - 3D Rendering Type: "VOLUME"

    - Display Set Patient Orientation: "X\F"

  - Item 10:

    - Display Set Number: 10

    - Display Set Presentation Group: 2

    - Image Set Number: 1

    - Image Boxes Sequence:

      - Item 1: [entire 2048x2560 space]

        - Image Box Number: 1

        - Display Environment Spatial Position: (1024/3072, 2560/2560), (3072/3072, 0/2560)

        - Image Box Layout Type: "TILED"

        - Image Box Tile Horizontal Dimension: 3

        - Image Box Tile Vertical Dimension: 4

        - Image Box Scroll Direction: "VERTICAL"

        - Image Box Small Scroll Type: "ROW_COLUMN"

        - Image Box Small Scroll Amount: 1

        - Image Box Large Scroll Type: "PAGE"

        - Image Box Large Scroll Amount: 1

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Display Set Patient Orientation: "L\P"

  - Figure V.4.3-3. Group #3 is combined MR & CT

  - Item 11: [MR coronal]

    - Display Set Number: 11

    - Display Set Presentation Group: 3

    - Image Set Number: 1

    - Image Boxes Sequence:

      - Item 1: [lower left quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (0/3072, 512/2560), (512/3072,0/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Reformatting Operation Type: "MPR"

    - Reformatting Thickness: 5

    - Reformatting Interval: 5

    - Reformatting Operation Initial View Direction: "CORONAL"

    - Display Set Patient Orientation: "L\F"

    - Show Graphic Annotation Flag: "NO"

    - Display Set Presentation Group Description: "MR & CT combined"

  - Item 12: [CT coronal]

    - Display Set Number: 12

    - Display Set Presentation Group: 3

    - Image Set Number: 2

    - Image Boxes Sequence:

      - Item 1: [upper left quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (0/3072, 1024/2560), (512/3072, 512/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Reformatting Operation Type: "MPR"

    - Reformatting Thickness: 5

    - Reformatting Interval: 5

    - Reformatting Operation Initial View Direction: "CORONAL"

    - Display Set Patient Orientation: "L\F"

    - VOI Type: BRAIN

    - Show Graphic Annotation Flag: "NO"

  - Item 13: [CT transverse]

    - Display Set Number: 13

    - Display Set Presentation Group: 3

    - Image Set Number: 2

    - Image Boxes Sequence:

      - Item 1: [upper right quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (512/3072, 1024/2560), (1024/3072, 512/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Display Set Patient Orientation: "L\P"

    - VOI Type: BRAIN

    - Show Graphic Annotation Flag: "YES"

  - Item 14: [MR transverse]

    - Display Set Number: 14

    - Display Set Presentation Group: 3

    - Image Set Number: 1

    - Image Boxes Sequence:

      - Item 1: [lower right quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (512/3072, 512/2560), (1024/3072, 0/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Display Set Patient Orientation: "L\P"

    - Show Graphic Annotation Flag: "NO"

  - Item 15: [CT two part scrolled, rows 1 & 3]

    - Display Set Number: 15

    - Display Set Presentation Group: 3

    - Image Set Number: 2

    - Image Boxes Sequence:

      - Item 1: [row 1 (top row) of 2048x2560 space]

        - Image Box Number: 1

        - Display Environment Spatial Position: (1024/3072, 2048/2560), (3072/3072, 1536/2560)

        - Image Box Layout Type: "TILED"

        - Image Box Tile Horizontal Dimension: 3

        - Image Box Tile Vertical Dimension: 1

        - Image Box Scroll Direction: "HORIZONTAL"

        - Image Box Small Scroll Type: "IMAGE"

        - Image Box Small Scroll Amount: 1

        - Image Box Large Scroll Type: "ROW_COLUMN"

        - Image Box Large Scroll Amount: 1

      - Item 2: [row 3 of 2048x2560 space]

        - Image Box Number: 2

        - Display Environment Spatial Position: (1024/3072, 1024/2560), (3072/3072, 512/2560)

        - Image Box Layout Type: "TILED"

        - Image Box Tile Horizontal Dimension: 3

        - Image Box Tile Vertical Dimension: 1

        - Image Box Scroll Direction: "HORIZONTAL"

        - Image Box Small Scroll Type: "IMAGE"

        - Image Box Small Scroll Amount: 1

        - Image Box Large Scroll Type: "ROW_COLUMN"

        - Image Box Large Scroll Amount: 1

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Display Set Patient Orientation: "L\P"

    - VOI Type: BRAIN

    - Show Graphic Annotation Flag: "YES"

  - Item 16: [MR two part scrolled, rows 2 & 4]

    - Display Set Number: 16

    - Display Set Presentation Group: 3

    - Image Set Number: 1

    - Image Boxes Sequence:

      - Item 1: [row 2 of 2048x2560 space]

        - Image Box Number: 1

        - Display Environment Spatial Position: (1024/3072, 1536/2560), (3072/3072, 1024/2560)

        - Image Box Layout Type: "TILED"

        - Image Box Tile Horizontal Dimension: 3

        - Image Box Tile Vertical Dimension: 1

        - Image Box Scroll Direction: "HORIZONTAL"

        - Image Box Small Scroll Type: "IMAGE"

        - Image Box Small Scroll Amount: 1

        - Image Box Large Scroll Type: "ROW_COLUMN"

        - Image Box Large Scroll Amount: 1

      - Item 2: [row 4 (bottom row) of 2048x2560 space]

        - Image Box Number: 2

        - Display Environment Spatial Position: (1024/3072, 512/2560), (3072/3072, 0/2560)

        - Image Box Layout Type: "TILED"

        - Image Box Tile Horizontal Dimension: 3

        - Image Box Tile Vertical Dimension: 1

        - Image Box Scroll Direction: "HORIZONTAL"

        - Image Box Small Scroll Type: "IMAGE"

        - Image Box Small Scroll Amount: 1

        - Image Box Large Scroll Type: "ROW_COLUMN"

        - Image Box Large Scroll Amount: 1

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Display Set Patient Orientation: "L\P"

    - Show Graphic Annotation Flag: "NO"

  - Figure V.4.3-4. Group #4 is combined CT new & CT old

  - Item 17: [CT old coronal]

    - Display Set Number: 17

    - Display Set Presentation Group: 4

    - Image Set Number: 3

    - Image Boxes Sequence:

      - Item 1: [lower left quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (0/3072, 512/2560), (512/3072,0/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Reformatting Operation Type: "MPR"

    - Reformatting Thickness: 5

    - Reformatting Interval: 5

    - Reformatting Operation Initial View Direction: "CORONAL"

    - Display Set Patient Orientation: "L\F"

    - VOI Type: BRAIN

    - Display Set Presentation Group Description: "CT old & CT new combined"

  - Item 18: [CT new coronal]

    - Display Set Number: 18

    - Display Set Presentation Group: 4

    - Image Set Number: 2

    - Image Boxes Sequence:

      - Item 1: [upper left quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (0/3072, 1024/2560), (512/3072, 512/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Reformatting Operation Type: "MPR"

    - Reformatting Thickness: 5

    - Reformatting Interval: 5

    - Reformatting Operation Initial View Direction: "CORONAL"

    - Display Set Patient Orientation: "L\F"

    - VOI Type: BRAIN

  - Item 19: [CT new transverse]

    - Display Set Number: 19

    - Display Set Presentation Group: 4

    - Image Set Number: 2

    - Image Boxes Sequence:

      - Item 1: [upper right quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (512/3072, 1024/2560), (1024/3072, 512/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Display Set Patient Orientation: "L\P"

    - VOI Type: BRAIN

    - Show Graphic Annotation Flag: "YES"

  - Item 20: [CT old transverse]

    - Display Set Number: 20

    - Display Set Presentation Group: 4

    - Image Set Number: 3

    - Image Boxes Sequence:

      - Item 1: [lower right quadrant of 1024x1024]

        - Image Box Number: 1

        - Display Environment Spatial Position: (512/3072, 512/2560), (1024/3072, 0/2560)

        - Image Box Layout Type: "STACK"

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Display Set Patient Orientation: "L\P"

    - VOI Type: BRAIN

    - Show Graphic Annotation Flag: "YES"

  - Item 21: [CT new two part scrolled, rows 1 & 3]

    - Display Set Number: 21

    - Display Set Presentation Group: 4

    - Image Set Number: 2

    - Image Boxes Sequence:

      - Item 1: [row 1 (top row) of 2048x2560 space]

        - Image Box Number: 1

        - Display Environment Spatial Position: (1024/3072, 2048/2560), (3072/3072, 1536/2560)

        - Image Box Layout Type: "TILED"

        - Image Box Tile Horizontal Dimension: 3

        - Image Box Tile Vertical Dimension: 1

        - Image Box Scroll Direction: "HORIZONTAL"

        - Image Box Small Scroll Type: "IMAGE"

        - Image Box Small Scroll Amount: 1

        - Image Box Large Scroll Type: "ROW_COLUMN"

        - Image Box Large Scroll Amount: 1

      - Item 2: [row 3 of 2048x2560 space]

        - Image Box Number: 2

        - Display Environment Spatial Position: (1024/3072, 1024/2560), (3072/3072, 512/2560)

        - Image Box Layout Type: "TILED"

        - Image Box Tile Horizontal Dimension: 3

        - Image Box Tile Vertical Dimension: 1

        - Image Box Scroll Direction: "HORIZONTAL"

        - Image Box Small Scroll Type: "IMAGE"

        - Image Box Small Scroll Amount: 1

        - Image Box Large Scroll Type: "ROW_COLUMN"

        - Image Box Large Scroll Amount: 1

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Display Set Patient Orientation: "L\P"

    - VOI Type: BRAIN

    - Show Graphic Annotation Flag: "YES"

  - Item 22: [CT old two part scrolled, rows 2 & 4]

    - Display Set Number: 22

    - Display Set Presentation Group: 4

    - Image Set Number: 3

    - Image Boxes Sequence:

      - Item 1: [row 2 of 2048x2560 space]

        - Image Box Number: 1

        - Display Environment Spatial Position: (1024/3072, 1536/2560), (3072/3072, 1024/2560)

        - Image Box Layout Type: "TILED"

        - Image Box Tile Horizontal Dimension: 3

        - Image Box Tile Vertical Dimension: 1

        - Image Box Scroll Direction: "HORIZONTAL"

        - Image Box Small Scroll Type: "IMAGE"

        - Image Box Small Scroll Amount: 1

        - Image Box Large Scroll Type: "ROW_COLUMN"

        - Image Box Large Scroll Amount: 1

      - Item 2: [row 4 (bottom row) of 2048x2560 space]

        - Image Box Number: 2

        - Display Environment Spatial Position: (1024/3072, 512/2560), (3072/3072, 0/2560)

        - Image Box Layout Type: "TILED"

        - Image Box Tile Horizontal Dimension: 3

        - Image Box Tile Vertical Dimension: 1

        - Image Box Scroll Direction: "HORIZONTAL"

        - Image Box Small Scroll Type: "IMAGE"

        - Image Box Small Scroll Amount: 1

        - Image Box Large Scroll Type: "ROW_COLUMN"

        - Image Box Large Scroll Amount: 1

    - Filter Operations Sequence:

      - Item 1:

        - Filter-by Category: "IMAGE_PLANE"

        - Selector Attribute VR: "CS"

        - Selector CS Value: "TRANSVERSE"

        - Filter-by Operator: "MEMBER_OF"

    - Sorting Operations Sequence:

      - Item 1:

        - Sort-by Category: "ALONG_AXIS"

        - Sorting Direction: "INCREASING"

    - Display Set Patient Orientation: "L\P"

    - VOI Type: BRAIN

    - Show Graphic Annotation Flag: "YES"

- Partial Data Display Handling: "MAINTAIN_LAYOUT"

- Synchronized Scrolling Sequence: [Link up (synchronize) the MR and CT tiled scroll panes in Display Sets 15 and 16, and the CT new and CT old tiled scroll panes in Display Sets 21 and 22]

  - Item 1:

    - Display Set Scrolling Group: 15\16

  - Item 2:

    - Display Set Scrolling Group: 21\22

Prev | Up | Next
V.4.2 Hanging Protocol Environment Module | Home | V.5 Hanging Protocol Query Example