# PS3.16

## DICOM PS3.16 2014c - Content Mapping Resource

### DICOM Standards Committee

Copyright © 2014 NEMA

List of Figures

A-3. Quantitative Ventricular Analysis Report SR IOD Template Structure
A-4. Quantitative Arterial Analysis Report SR IOD Template Structure
A-4b. Direction of Blood Flow
A-5. IVUS Report Template Hierarchy
A-6. Hemodynamic Report Template Hierarchy
A-7. Cardiac Catheterization Report Template Hierarchy
A-8. Mammography CAD SR IOD Template Structure
A-8a. Example of Breast Outline Including Pectoral Muscle Tissue
A-8b. Example of Pectoral Muscle Outline
A-9. Chest CAD SR IOD Template Structure
A-9b. Colon CAD SR IOD Template Structure
A-10. Breast Imaging Report Template Structure
A-11. Implantation Plan SR Document IOD Template Structure
A-12. Implant Assembly and Components Terminology
A-13. References to Registration Objects
A-14. X-Ray Radiation Dose SR IOD Template Structure
A-15. CT Radiation Dose SR IOD Template Structure
A-16. Spiral Acquisition Parameters
A-17. Radiopharmaceutical Radiation Dose SR IOD Template Structure

List of Tables

TID <#>. <SR Context Template Name>
TID <#>. <Acquisition Context Template Name>
TID <#>. <Protocol Context Template Name>
6.1.3-1. Syntax of Relationship Constraints
6.1.6-1. Permitted Values for VM
CID <#>. <Context Group Name>
7.2.1-1. Include Context Group Macro
8-1. Coding Schemes
TID 300. Parameters
TID 300. Measurement
TID 310. Parameters
TID 310. Measurement Properties
TID 311. Parameters
TID 311. Measurement Statistical Properties
TID 312. Parameters
TID 312. Normal Range Properties
TID 315. Parameters
TID 315. Equation or Table
TID 320. Parameters
TID 320. Image or Spatial Coordinates
TID 321. Parameters
TID 321. Waveform or Temporal Coordinates
TID 350. References to Supporting Evidence
TID 351. Previous Reports
TID 1000. Quotation
TID 1001. Observation Context
TID 1002. Observer Context
TID 1003. Person Observer Identifying Attributes
TID 1004. Device Observer Identifying Attributes
TID 1005. Procedure Context
TID 1006. Subject Context
TID 1007. Subject Context, Patient
TID 1008. Subject Context, Fetus
TID 1009. Subject Context, Specimen
TID 1010. Subject Context, Device
TID 1020. Parameters
TID 1020. Person Participant
TID 1021. Parameters
TID 1021. Device Participant
TID 1200. Language Designation
TID 1201. Language of Value
TID 1202. Language of Name and Value
TID 1204. Language of Content Item and Descendants
TID 1210. Equivalent Meaning(s) of Concept Name
TID 1211. Equivalent Meaning(s) of Value
TID 1350. Negation Modifier, Presence of Finding
TID 1400. Linear Measurement
TID 1401. Area Measurement
TID 1402. Volume Measurement
TID 1404. Numeric Measurement
TID 1406. Three Dimensional Linear Measurement
TID 1410. Parameters
TID 1410. Planar ROI Measurements
TID 1411. Parameters
TID 1411. Volumetric ROI Measurements
TID 1419. Parameters
TID 1419. ROI Measurements
TID 1420. Parameters
TID 1420. Measurements Derived From Multiple ROI Measurements
TID 1500. Measurement Report
TID 1501. Parameters
TID 1501. Measurement Group
TID 1502. Time Point Context
TID 1600. Image Library
TID 1601. Image Library Entry
TID 1602. Image Library Entry Descriptors
TID 1603. Image Library Entry Descriptors for Projection Radiography
TID 1604. Image Library Entry Descriptors for Cross-Sectional Modalities
TID 1605. Image Library Entry Descriptors for CT
TID 1606. Image Library Entry Descriptors for MR
TID 1607. Image Library Entry Descriptors for PET
TID 2000. Basic Diagnostic Imaging Report
TID 2001. Basic Diagnostic Imaging Report Observations
TID 2002. Report Narrative
TID 2005. Transcribed Diagnostic Imaging Report
TID 2006. Imaging Report With Conditional Radiation Exposure and Protection Information
TID 2007. Imaging Procedure Description
TID 2008. Radiation Exposure and Protection Information
TID 2010. Key Object Selection
TID 2020. Spectacle Prescription Report
TID 2021. Spectacle Prescription Details
TID 2100. Macular Grid Thickness and Volume Report
TID 2101. Parameters
TID 2101. Macular Grid Thickness and Volume Measurement
TID 2102. Quality Rating Identification
TID 3001. Procedure Log
TID 3010. Log Entry Qualifiers
TID 3100. Procedure Action
TID 3101. Image Acquisition
TID 3102. Waveform Acquisition
TID 3103. Referenced Object
TID 3104. Consumables
TID 3105. Lesion Identification and Properties
TID 3106. Drugs/Contrast Administered
TID 3107. Device Used
TID 3108. Intervention
TID 3109. Measurements
TID 3110. Impressions or Findings
TID 3111. Percutaneous Entry
TID 3112. Specimen Obtained
TID 3113. Patient Support
TID 3114. Patient Assessment
TID 3115. ECG ST Assessment
TID 3202. Ventricular Analysis
TID 3205. Parameters
TID 3205. Calibration
TID 3206. VA Main Results
TID 3207. AA Main Results
TID 3208. Frame-To-Frame Result
TID 3209. Centerline Wall Motion
TID 3210. Radial Based Wall Motion
TID 3211. Landmark Based Wall Motion
TID 3213. Quantitative Arterial Analysis
TID 3214. Analyzed Segment
TID 3215. Angiographic Lesion Analysis
TID 3216. Stenotic Flow Reserve
TID 3217. Sub-Segmental Data
TID 3218. Position in Arterial Segment
TID 3219. Segment Values
TID 3250. IVUS Report
TID 3251. IVUS Vessel
TID 3252. IVUS Lesion
TID 3253. IVUS Measurements
TID 3254. IVUS Qualitative Assessments
TID 3255. IVUS Volume Measurement
TID 3300. Stress Testing Report
TID 3301. Stress Test Procedure Description
TID 3303. Stress Test Phase Data
TID 3304. Stress Test Measurement Group
TID 3307. NM/PET Perfusion Measurement Group
TID 3309. Stress Echo Measurement Group
TID 3311. Stress Test Summary
TID 3312. Physiological Summary
TID 3313. Stress ECG Summary
TID 3317. Stress Imaging Summary
TID 3318. Comparison to Prior Stress Exam
TID 3320. Conclusions and Recommendations
TID 3500. Hemodynamics Report
TID 3501. Hemodynamics Measurement Group
TID 3504. Arterial Pressure Measurement
TID 3505. Atrial Pressure Measurement
TID 3506. Venous Pressure Measurement
TID 3507. Ventricular Pressure Measurement
TID 3508. Gradient Measurement
TID 3509. Blood Velocity Measurement
TID 3510. Vital Signs
TID 3515. Cardiac Output Measurement By Indicator Dilution
TID 3516. Blood Lab Measurements
TID 3520. Hemodynamic Clinical Context
TID 3521. Parameters
TID 3521. Relative Time
TID 3530. Parameters
TID 3530. Hemodynamic Acquisition Context
TID 3550. Pressure Waveform Measurements
TID 3560. Derived Hemodynamic Measurements
TID 3570. Summary, Hemodynamics
TID 3601. Procedure Context
TID 3602. Cardiovascular Patient Characteristics
TID 3603. Procedure Environmental Characteristics
TID 3700. ECG Report
TID 3702. Prior ECG Exam
TID 3704. Patient Characteristics for ECG
TID 3708. ECG Waveform Information
TID 3713. ECG Global Measurements
TID 3714. ECG Lead Measurements
TID 3715. ECG Measurement Source
TID 3717. Qualitative Analysis, ECG
TID 3719. Summary, ECG
TID 3800. Cardiac Catheterization Report Root
TID 3802. Cardiovascular Patient History
TID 3803. Patient Presentation, Cath
TID 3806. Cath Procedure
TID 3807. Percutaneous Coronary Intervention Procedure
TID 3808. Lesion Intervention Information
TID 3809. Other Interventional Procedures
TID 3810. Cardiac Catheterization Findings
TID 3812. Hemodynamic Findings
TID 3814. Left Ventriculography Findings
TID 3815. Right Ventriculography Findings
TID 3816. Ventricular Assessment
TID 3817. Coronary Arteriography Findings
TID 3818. Other Cardiographic Findings
TID 3819. Parameters
TID 3819. Common Findings
TID 3820. Adverse Outcomes, Cath
TID 3824. Summary, Cath
TID 3828. Discharge Summary, Cath
TID 3829. Parameters
TID 3829. Problem Properties
TID 3830. Parameters
TID 3830. Procedure Properties
TID 3831. Parameters
TID 3831. Medical Device Use
TID 3900. CT/MR Cardiovascular Analysis Report
TID 3901. Procedure Summary
TID 3902. Parameters
TID 3902. Vascular Analysis
TID 3905. Calcium Scoring Results
TID 3906. Parameters
TID 3906. Vascular Section Measurements
TID 3907. Vessel Measurements
TID 3908. Vascular Lesion
TID 3909. Best Illustration of Findings
TID 3910. Flow Quantification
TID 3911. Plaque Properties
TID 3912. Stenosis Properties
TID 3913. Aneurysm Properties
TID 3914. Arterial Dissection Properties
TID 3915. Vascular Occlusion Properties
TID 3916. Stent Properties
TID 3917. Aneurysm Measurements
TID 3920. Ventricular Analysis
TID 3921. Parameters
TID 3921. Ventricular Measurements
TID 3922. Absolute Values of Ventricular Measurements
TID 3923. BSA-Normalized Ventricular Measurements
TID 3924. Heart Rate-Normalized Ventricular Measurements
TID 3925. Thickening Analysis
TID 3926. Myocardial Perfusion Analysis
TID 3927. Report Summary
TID 3929. Cardiovascular Analysis Observation Context
TID 3990. Parameters
TID 3990. Two Dimensional Measurement Graph
TID 4000. Mammography CAD Document Root
TID 4001. Mammography CAD Overall Impression/Recommendation
TID 4002. Mammography CAD Impression/Recommendation Body
TID 4003. Mammography CAD Individual Impression/Recommendation
TID 4004. Mammography CAD Composite Feature
TID 4005. Mammography CAD Composite Feature Body
TID 4006. Mammography CAD Single Image Finding
TID 4007. Mammography CAD Breast Composition
TID 4008. Mammography CAD Breast Geometry
TID 4009. Mammography CAD Individual Calcification
TID 4010. Mammography CAD Calcification Cluster
TID 4011. Mammography CAD Density
TID 4012. Mammography CAD Non-Lesion
TID 4013. Mammography CAD Selected Region
TID 4014. Parameters
TID 4014. CAD Image Quality
TID 4015. Parameters
TID 4015. CAD Detections Performed
TID 4016. Parameters
TID 4016. CAD Analyses Performed
TID 4017. Parameters
TID 4017. CAD Detection Performed
TID 4018. Parameters
TID 4018. CAD Analysis Performed
TID 4019. CAD Algorithm Identification
TID 4020. Parameters
TID 4020. CAD Image Library Entry
TID 4021. Mammography CAD Geometry
TID 4022. CAD Observation Context
TID 4023. CAD Operating Points
TID 4100. Chest CAD Document Root
TID 4101. Chest CAD Findings Summary
TID 4102. Chest CAD Composite Feature
TID 4103. Chest CAD Composite Feature Body
TID 4104. Chest CAD Single Image Finding
TID 4105. Chest CAD Descriptors
TID 4106. Response Evaluation
TID 4107. Chest CAD Geometry
TID 4108. Tracking Identifier
TID 4120. Colon CAD Document Root
TID 4121. Colon CAD Findings Summary
TID 4122. CAD Common Image Properties Entry
TID 4125. Colon CAD Composite Feature
TID 4126. Colon CAD Composite Feature Body
TID 4127. Colon CAD Single Image Finding
TID 4128. Colon CAD Descriptors
TID 4129. Colon CAD Geometry
TID 4200. Breast Imaging Report
TID 4201. Breast Imaging Procedure Reported
TID 4202. Breast Imaging Report Narrative
TID 4203. Breast Imaging Assessment
TID 4204. Breast Imaging Report Intervention Section
TID 4205. Breast Composition Section
TID 4206. Breast Imaging Report Finding Section
TID 4207. Breast Imaging Pathology Results
TID 4208. Breast Imaging Report Supplementary Data
TID 5000. OB-GYN Ultrasound Procedure Report
TID 5001. OB-GYN Patient Characteristics
TID 5002. OB-GYN Procedure Summary Section
TID 5003. OB-GYN Procedure Fetus Summary
TID 5004. Fetal Biometry Ratio Section
TID 5005. Fetal Biometry Section
TID 5006. Fetal Long Bones Section
TID 5007. Fetal Cranium Section
TID 5008. Parameters
TID 5008. Fetal Biometry Group
TID 5009. Fetal Biophysical Profile Section
TID 5010. Amniotic Sac Section
TID 5011. Early Gestation Section
TID 5012. Ovaries Section
TID 5013. Parameters
TID 5013. Follicles Section
TID 5014. Follicle Measurement Group
TID 5015. Pelvis and Uterus Section
TID 5016. Parameters
TID 5016. LWH Volume Group
TID 5025. Parameters
TID 5025. OB-GYN Fetal Vascular Ultrasound Measurement Group
TID 5026. Parameters
TID 5026. OB-GYN Pelvic Vascular Ultrasound Measurement Group
TID 5100. Vascular Ultrasound Report
TID 5101. Vascular Patient Characteristics
TID 5102. Vascular Procedure Summary Section
TID 5103. Parameters
TID 5103. Vascular Ultrasound Section
TID 5104. Parameters
TID 5104. Vascular Ultrasound Measurement Group
TID 5105. Ultrasound Graft Section
TID 5200. Echocardiography Procedure Report
TID 5201. Echocardiography Patient Characteristics
TID 5202. Parameters
TID 5202. Echo Section
TID 5203. Parameters
TID 5203. Echo Measurement
TID 5204. Parameters
TID 5204. Wall Motion Analysis
5204-1. Numeric Score Assignment for Segmental Findings
TID 5220. Pediatric, Fetal and Congenital Cardiac Ultrasound Reports
TID 5221. Cardiac Ultrasound Pediatric Echo Measurement Section
TID 5222. Parameters
TID 5222. Pediatric, Fetal and Congenital Cardiac Ultrasound Section
TID 5223. Parameters
TID 5223. Pediatric, Fetal and Congenital Cardiac Ultrasound Measurement
TID 5225. Cardiac Ultrasound Fetal Characteristics
TID 5226. Cardiac Ultrasound Summary Section
TID 5227. Cardiac Ultrasound Fetal Summary Section
TID 5228. Cardiac Ultrasound Fetal Measurement Section
TID 7000. Implantation Plan
TID 7001. Related Implantation Reports
TID 9000. Relevant Patient Information for Breast Imaging
TID 9001. Gynecological History
TID 9002. Parameters
TID 9002. Medication, Substance, Environmental Exposure
TID 9003. Parameters
TID 9003. Previous Procedure
TID 9004. Parameters
TID 9004. Indicated Problem
TID 9005. Parameters
TID 9005. Risk Factor
TID 9006. Obstetric History
TID 9007. General Relevant Patient Information
TID 10001. Projection X-Ray Radiation Dose
TID 10002. Parameters
TID 10002. Accumulated X-Ray Dose
TID 10003. Irradiation Event X-Ray Data
TID 10003A. Irradiation Event X-Ray Detector Data
TID 10003B. Irradiation Event X-Ray Source Data
TID 10003C. Irradiation Event X-Ray Mechanical Data
TID 10004. Accumulated Fluoroscopy and Acquisition Projection X-Ray Dose
TID 10005. Accumulated Mammography X-Ray Dose
TID 10006. Accumulated Cassette-Based Projection Radiography Dose
TID 10007. Accumulated Total Projection Radiography Dose
TID 10011. CT Radiation Dose
TID 10012. CT Accumulated Dose Data
TID 10013. CT Irradiation Event Data
TID 10014. Scanning Length
TID 10015. CT Dose Check Details
TID 10021. Radiopharmaceutical Radiation Dose
TID 10022. Radiopharmaceutical Administration Event Data
TID 10023. Organ Dose
TID 10024. Radiopharmaceutical Administration Patient Characteristics
CID 2. Anatomic Modifier
CID 4. Anatomic Region
CID 5. Transducer Approach
CID 6. Transducer Orientation
CID 7. Ultrasound Beam Path
CID 8. Angiographic Interventional Devices
CID 9. Image Guided Therapeutic Procedures
CID 10. Interventional Drug
CID 11. Route of Administration
CID 12. Radiographic Contrast Agent
CID 13. Radiographic Contrast Agent Ingredient
CID 18. Isotopes in Radiopharmaceuticals
CID 19. Patient Orientation
CID 20. Patient Orientation Modifier
CID 21. Patient Equipment Relationship
CID 23. Cranio-Caudad Angulation
CID 25. Radiopharmaceuticals
CID 26. Nuclear Medicine Projections
CID 27. Basic Cardiac Views
CID 29. Acquisition Modality
CID 30. DICOM Devices
CID 31. Abstract Priors
CID 42. Numeric Value Qualifier
CID 50. Instance Availability Status
CID 83. Units for Real World Value Mapping
CID 84. PET Units
CID 85. SUV Units
CID 91. Functional Condition Present During Acquisition
CID 92. Joint Position During Acquisition
CID 93. Joint Positioning Method
CID 94. Physical Force Applied During Acquisition
CID 100. Quantitative Diagnostic Imaging Procedures
CID 220. Level of Significance
CID 221. Measurement Range Concepts
CID 222. Normality Codes
CID 223. Normal Range Values
CID 224. Selection Method
CID 225. Measurement Uncertainty Concepts
CID 226. Population Statistical Descriptors
CID 227. Sample Statistical Descriptors
CID 228. Equation or Table
CID 230. Yes-No
CID 240. Present-Absent
CID 242. Normal-Abnormal
CID 244. Laterality
CID 250. Positive-Negative
CID 251. Severity of Complication
CID 252. S-M-L Size Descriptor
CID 270. Observer Type
CID 271. Observation Subject Class
CID 400. Audit Event Id
CID 401. Audit Event Type Code
CID 402. Audit Active Participant Role ID Code
CID 403. Security Alert Type Code
CID 404. Audit Participant Object ID Type Code
CID 405. Media Type Code
CID 3000. Audio Channel Source
CID 3001. ECG Leads
CID 3003. Hemodynamic Waveform Sources
CID 3004. Arterial Pulse Waveform
CID 3005. Respiration Waveform
CID 3010. Cardiovascular Anatomic Locations
CID 3011. Electrophysiology Anatomic Locations
CID 3014. Coronary Artery Segments
CID 3015. Coronary Arteries
CID 3016. Major Coronary Arteries
CID 3019. Cardiovascular Anatomic Location Modifiers
CID 3083. Units of Radioactivity
CID 3090. Time Synchronization Channel Types
CID 3101. Cardiac Procedural State Values
CID 3102. Rest-Stress
CID 3104. Cardiac Synchronization Technique
CID 3106. PET Cardiology Protocols
CID 3107. PET Cardiology Radiopharmaceuticals
CID 3108. NM/PET Procedures
CID 3110. Nuclear Cardiology Protocols
CID 3111. Nuclear Cardiology Radiopharmaceuticals
CID 3112. Attenuation Correction
CID 3113. Types of Perfusion Defects
CID 3114. Study Quality
CID 3115. Stress Imaging Quality Issues
CID 3116. NM Extracardiac Findings
CID 3117. Attenuation Correction Methods
CID 3118. Level of Risk
CID 3119. LV Function
CID 3120. Perfusion Findings
CID 3121. Perfusion Morphology
CID 3122. Ventricular Enlargement
CID 3200. Stress Test Procedure
CID 3201. Indications for Stress Test
CID 3202. Chest Pain
CID 3203. Exerciser Device
CID 3204. Stress Agents
CID 3205. Indications for Pharmacological Stress Test
CID 3206. Non-Invasive Cardiac Imaging Procedures
CID 3207. Stress Test Procedure Phases
CID 3208. Summary Codes Exercise ECG
CID 3209. Summary Codes Stress Imaging
CID 3210. Speed of Response
CID 3211. BP Response
CID 3212. Treadmill Speed
CID 3213. Stress Hemodynamic Findings
CID 3215. Perfusion Finding Method
CID 3217. Comparison Finding
CID 3220. Stress Symptoms
CID 3221. Stress Test Termination Reasons
CID 3227. QTc Measurements
CID 3228. ECG Timing Measurements
CID 3229. ECG Axis Measurements
CID 3230. ECG Findings
CID 3231. ST Segment Findings
CID 3232. ST Segment Location
CID 3233. ST Segment Morphology
CID 3234. Ectopic Beat Morphology
CID 3235. Perfusion Comparison Findings
CID 3236. Tolerance Comparison Findings
CID 3237. Wall Motion Comparison Findings
CID 3238. Stress Scoring Scales
CID 3239. Perceived Exertion Scales
CID 3240. Electrophysiology Measurement Functions and Techniques
CID 3241. Hemodynamic Measurement Techniques
CID 3250. Catheterization Procedure Phase
CID 3254. Electrophysiology Procedure Phase
CID 3261. Stress Protocols
CID 3262. ECG Patient State Values
CID 3263. Electrode Placement Values
CID 3271. Hemodynamic Physiological Challenges
CID 3335. ECG Annotations
CID 3337. Hemodynamic Annotations
CID 3339. Electrophysiology Annotations
CID 3400. Procedure Log Titles
CID 3401. Types of Log Notes
CID 3402. Patient Status and Events
CID 3403. Percutaneous Entry
CID 3404. Staff Actions
CID 3405. Procedure Action Values
CID 3406. Non-Coronary Transcatheter Interventions
CID 3407. Purpose of Reference to Object
CID 3408. Actions With Consumables
CID 3409. Administration of Drugs/Contrast
CID 3410. Numeric Parameters of Drugs/Contrast
CID 3411. Intracoronary Devices
CID 3412. Intervention Actions and Status
CID 3413. Adverse Outcomes
CID 3414. Procedure Urgency
CID 3415. Cardiac Rhythms
CID 3416. Respiration Rhythms
CID 3418. Lesion Risk
CID 3419. Findings Titles
CID 3421. Procedure Action
CID 3422. Device Use Actions
CID 3423. Numeric Device Characteristics
CID 3425. Intervention Parameters
CID 3426. Consumables Parameters
CID 3427. Equipment Events
CID 3428. Imaging Procedures
CID 3429. Catheterization Devices
CID 3430. DateTime Qualifiers
CID 3440. Peripheral Pulse Locations
CID 3441. Patient Assessments
CID 3442. Peripheral Pulse Methods
CID 3446. Skin Condition
CID 3448. Airway Assessment
CID 3451. Calibration Objects
CID 3452. Calibration Methods
CID 3453. Cardiac Volume Methods
CID 3455. Index Methods
CID 3456. Sub-Segment Methods
CID 3458. Contour Realignment
CID 3460. Circumferential Extent
CID 3461. Regional Extent
CID 3462. Chamber Identification
CID 3463. Ventricle Identification
CID 3465. QA Reference Methods
CID 3466. Plane Identification
CID 3467. Ejection Fraction
CID 3468. ED Volume
CID 3469. ES Volume
CID 3470. Vessel Lumen Cross-Sectional Area Calculation Methods
CID 3471. Estimated Volumes
CID 3472. Cardiac Contraction Phase
CID 3480. IVUS Procedure Phases
CID 3481. IVUS Distance Measurements
CID 3482. IVUS Area Measurements
CID 3483. IVUS Longitudinal Measurements
CID 3484. IVUS Indices and Ratios
CID 3485. IVUS Volume Measurements
CID 3486. Vascular Measurement Sites
CID 3487. Intravascular Volumetric Regions
CID 3488. Min/Max/Mean
CID 3489. Calcium Distribution
CID 3491. IVUS Lesion Morphologies
CID 3492. Vascular Dissection Classifications
CID 3493. IVUS Relative Stenosis Severities
CID 3494. IVUS Non Morphological Findings
CID 3495. IVUS Plaque Composition
CID 3496. IVUS Fiducial Points
CID 3497. IVUS Arterial Morphology
CID 3500. Pressure Units
CID 3502. Hemodynamic Resistance Units
CID 3503. Indexed Hemodynamic Resistance Units
CID 3510. Catheter Size Units
CID 3515. Specimen Collection
CID 3520. Blood Source Type
CID 3524. Blood Gas Pressures
CID 3525. Blood Gas Content
CID 3526. Blood Gas Saturation
CID 3527. Blood Base Excess
CID 3528. Blood pH
CID 3529. Arterial / Venous Content
CID 3530. Oxygen Administration Actions
CID 3531. Oxygen Administration
CID 3550. Circulatory Support Actions
CID 3551. Ventilation Actions
CID 3552. Pacing Actions
CID 3553. Circulatory Support
CID 3554. Ventilation
CID 3555. Pacing
CID 3560. Blood Pressure Methods
CID 3600. Relative Times
CID 3602. Hemodynamic Patient State
CID 3604. Arterial Lesion Locations
CID 3606. Arterial Source Locations
CID 3607. Venous Source Locations
CID 3608. Atrial Source Locations
CID 3609. Ventricular Source Locations
CID 3610. Gradient Source Locations
CID 3611. Pressure Measurements
CID 3612. Blood Velocity Measurements
CID 3613. Hemodynamic Time Measurements
CID 3614. Valve Areas, Non-Mitral
CID 3615. Valve Areas
CID 3616. Hemodynamic Period Measurements
CID 3617. Valve Flows
CID 3618. Hemodynamic Flows
CID 3619. Hemodynamic Resistance Measurements
CID 3620. Hemodynamic Ratios
CID 3621. Fractional Flow Reserve
CID 3627. Measurement Type
CID 3628. Cardiac Output Methods
CID 3629. Procedure Intent
CID 3630. Cardiovascular Anatomic Locations
CID 3640. Hypertension
CID 3641. Hemodynamic Assessments
CID 3642. Degree Findings
CID 3651. Hemodynamic Measurement Phase
CID 3663. Body Surface Area Equations
CID 3664. Oxygen Consumption Equations and Tables
CID 3666. P50 Equations
CID 3667. Framingham Scores
CID 3668. Framingham Tables
CID 3670. ECG Procedure Types
CID 3671. Reason for ECG Exam
CID 3676. Lead Measurement Technique
CID 3677. Summary Codes ECG
CID 3678. QT Correction Algorithms
CID 3680. ECG Lead Noise Descriptions
CID 3687. Electrophysiology Waveform Durations
CID 3688. Electrophysiology Waveform Voltages
CID 3689. ECG Global Waveform Durations
CID 3690. ECG Control Variables Numeric
CID 3691. ECG Control Variables Text
CID 3700. Cath Diagnosis
CID 3701. Cardiac Valves and Tracts
CID 3703. Wall Motion
CID 3704. Myocardium Wall Morphology Findings
CID 3705. Chamber Size
CID 3706. Overall Contractility
CID 3707. VSD Description
CID 3709. Aortic Root Description
CID 3710. Coronary Dominance
CID 3711. Valvular Abnormalities
CID 3712. Vessel Descriptors
CID 3713. TIMI Flow Characteristics
CID 3714. Thrombus
CID 3715. Lesion Margin
CID 3716. Severity
CID 3717. Myocardial Wall Segments
CID 3718. Myocardial Wall Segments in Projection
CID 3719. Canadian Clinical Classification
CID 3721. Cardiovascular Surgeries
CID 3722. Diabetic Therapy
CID 3723. MI Types
CID 3724. Smoking History
CID 3726. Indications for Coronary Intervention
CID 3727. Indications for Catheterization
CID 3728. Cath Findings
CID 3729. Admission Status
CID 3730. Insurance Payor
CID 3733. Primary Cause of Death
CID 3735. Acute Coronary Syndrome Time Period
CID 3736. NYHA Classification
CID 3737. Non-Invasive Test - Ischemia
CID 3738. Pre-Cath Angina Type
CID 3739. Cath Procedure Type
CID 3740. Thrombolytic Administration
CID 3741. Medication Administration, Lab Visit
CID 3742. Medication Administration, PCI
CID 3743. Clopidogrel/Ticlopidine Administration
CID 3744. EF Testing Method
CID 3745. Calculation Method
CID 3746. Percutaneous Entry Site
CID 3747. Percutaneous Closure
CID 3748. Angiographic EF Testing Method
CID 3749. PCI Procedure Result
CID 3750. Previously Dilated Lesion
CID 3752. Guidewire Crossing
CID 3754. Vascular Complications
CID 3755. Cath Complications
CID 3756. Cardiac Patient Risk Factors
CID 3757. Cardiac Diagnostic Procedures
CID 3758. Cardiovascular Family History
CID 3760. Hypertension Therapy
CID 3761. Antilipemic Agents
CID 3762. Antiarrhythmic Agents
CID 3764. Myocardial Infarction Therapies
CID 3769. Concern Types
CID 3770. Problem Status
CID 3772. Health Status
CID 3773. Use Status
CID 3774. Social History
CID 3777. Implanted Devices
CID 3778. Stages
CID 3802. Plaque Structures
CID 3804. Stenosis Measurement Methods
CID 3805. Stenosis Types
CID 3806. Stenosis Shape
CID 3807. Volume Measurement Methods
CID 3808. Aneurysm Types
CID 3809. Associated Conditions
CID 3810. Vascular Morphology
CID 3813. Stent Findings
CID 3814. Stent Composition
CID 3815. Source of Vascular Finding
CID 3817. Vascular Sclerosis Types
CID 3820. Non-Invasive Vascular Procedures
CID 3821. Papillary Muscle Included/Excluded
CID 3823. Respiratory Status
CID 3826. Heart Rhythm
CID 3827. Vessel Segments
CID 3829. Pulmonary Arteries
CID 3831. Stenosis Length
CID 3832. Stenosis Grade
CID 3833. Cardiac Ejection Fraction
CID 3835. Cardiac Volume Measurements
CID 3836. Time-Based Perfusion Measurements
CID 3837. Fiducial Feature
CID 3838. Diameter Derivation
CID 3839. Coronary Veins
CID 3840. Pulmonary Veins
CID 3843. Myocardial Subsegment
CID 3850. Intravascular OCT Flush Agent
CID 4005. Partial View Section for Mammography
CID 4009. DX Anatomy Imaged
CID 4010. DX View
CID 4011. DX View Modifier
CID 4012. Projection Eponymous Name
CID 4013. Anatomic Region for Mammography
CID 4014. View for Mammography
CID 4015. View Modifier for Mammography
CID 4016. Anatomic Region for Intra-Oral Radiography
CID 4017. Anatomic Region Modifier for Intra-Oral Radiography
CID 4018. Primary Anatomic Structure for Intra-Oral Radiography(Permanent Dentition - Designation of Teeth)
CID 4019. Primary Anatomic Structure for Intra-Oral Radiography (Deciduous Dentition - Designation of Teeth)
CID 4020. PET Radionuclide
CID 4021. PET Radiopharmaceutical
CID 4028. Craniofacial Anatomic Regions
CID 4030. CT, MR and PET Anatomy Imaged
CID 4031. Common Anatomic Regions
CID 4032. MR Spectroscopy Metabolites
CID 4033. MR Proton Spectroscopy Metabolites
CID 4040. Endoscopy Anatomic Regions
CID 4042. XA/XRF Anatomy Imaged
CID 4050. Drug or Contrast Agent Characteristics
CID 4051. General Devices
CID 4052. Phantom Devices
CID 4100. T1 Measurement Methods
CID 4101. Tracer Kinetic Models
CID 4102. Perfusion Measurement Methods
CID 4103. Arterial Input Function Measurement Methods
CID 4104. Bolus Arrival Time Derivation Methods
CID 4105. Perfusion Analysis Methods
CID 4106. Quantitative Methods used for Perfusion And Tracer Kinetic Models
CID 4107. Tracer Kinetic Model Parameters
CID 4108. Perfusion Model Parameters
CID 4109. Model-Independent Dynamic Contrast Analysis Parameters
CID 4110. Tracer Kinetic Modeling Covariates
CID 4111. Contrast Characteristics
CID 4200. Ophthalmic Imaging Agent
CID 4201. Patient Eye Movement Command
CID 4202. Ophthalmic Photography Acquisition Device
CID 4203. Ophthalmic Photography Illumination
CID 4204. Ophthalmic Filter
CID 4205. Ophthalmic Lens
CID 4206. Ophthalmic Channel Description
CID 4207. Ophthalmic Image Position
CID 4208. Mydriatic Agent
CID 4209. Ophthalmic Anatomic Structure Imaged
CID 4210. Ophthalmic Tomography Acquisition Device
CID 4211. Ophthalmic OCT Anatomic Structure Imaged
CID 4214. Ophthalmic Horizontal Directions
CID 4215. Ophthalmic Vertical Directions
CID 4216. Ophthalmic Visual Acuity Type
CID 4220. Visual Fixation Quality During Acquisition
CID 4221. Visual Fixation Quality Problem
CID 4222. Ophthalmic Macular Grid Problem
CID 4230. Ophthalmic Ultrasound Axial Measurements Type
CID 4231. Lens Status
CID 4232. Vitreous Status
CID 4233. Ophthalmic Axial Length Measurements Segment Names
CID 4234. Refractive Surgery Types
CID 4235. Keratometry Descriptors
CID 4236. IOL Calculation Formula
CID 4237. Lens Constant Type
CID 4238. Refractive Error Types
CID 4239. Anterior Chamber Depth Definition
CID 4240. Ophthalmic Measurement or Calculation Data Source
CID 4241. Ophthalmic Axial Length Selection Method
CID 4243. Ophthalmic Axial Length Quality Metric Type
CID 4244. Ophthalmic Agent Concentration Units
CID 4250. Visual Field Static Perimetry Test Patterns
CID 4251. Visual Field Static Perimetry Test Strategies
CID 4252. Visual Field Static Perimetry Screening Test Modes
CID 4253. Visual Field Static Perimetry Fixation Strategy
CID 4254. Visual Field Static Perimetry Test Analysis Results
CID 4255. Visual Field Illumination Color
CID 4256. Visual Field Procedure Modifier
CID 4257. Visual Field Global Index Name
CID 4260. Ophthalmic Mapping Units for Real World Value Mapping
CID 4261. Ophthalmic Mapping Acquisition Method
CID 4262. Retinal Thickness Definition
CID 4263. Ophthalmic Thickness Map Value Type
CID 4264. Ophthalmic Map Purposes of Reference
CID 4265. Ophthalmic Thickness Deviation Categories
CID 4266. Ophthalmic Anatomic Structure Reference Point
CID 4267. Corneal Topography Mapping Units for Real World Value Mapping
CID 4268. Corneal Topography Map Value Type
CID 6000. Overall Breast Composition
CID 6001. Overall Breast Composition From BI-RADS®
CID 6002. Change Since Last Mammogram or Prior Surgery
CID 6003. Change Since Last Mammogram or Prior Surgery From BI-RADS®
CID 6004. Mammography Characteristics of Shape
CID 6005. Characteristics of Shape From BI-RADS®
CID 6006. Mammography Characteristics of Margin
CID 6007. Characteristics of Margin From BI-RADS®
CID 6008. Density Modifier
CID 6009. Density Modifier From BI-RADS®
CID 6010. Mammography Calcification Types
CID 6011. Calcification Types From BI-RADS®
CID 6012. Calcification Distribution Modifier
CID 6013. Calcification Distribution Modifier From BI-RADS®
CID 6014. Mammography Single Image Finding
CID 6015. Single Image Finding From BI-RADS®
CID 6016. Mammography Composite Feature
CID 6017. Composite Feature From BI-RADS®
CID 6018. Clockface Location or Region
CID 6019. Clockface Location or Region From BI-RADS®
CID 6020. Quadrant Location
CID 6021. Quadrant Location From BI-RADS®
CID 6022. Side
CID 6023. Side From BI-RADS®
CID 6024. Depth
CID 6025. Depth From BI-RADS®
CID 6026. Mammography Assessment
CID 6027. Assessment From BI-RADS®
CID 6028. Mammography Recommended Follow-Up
CID 6029. Recommended Follow-Up From BI-RADS®
CID 6030. Mammography Pathology Codes
CID 6031. Benign Pathology Codes From BI-RADS®
CID 6032. High Risk Lesions Pathology Codes From BI-RADS®
CID 6033. Malignant Pathology Codes From BI-RADS®
CID 6034. Intended Use of CAD Output
CID 6035. Composite Feature Relations
CID 6036. Scope of Feature
CID 6037. Mammography Quantitative Temporal Difference Type
CID 6038. Mammography Qualitative Temporal Difference Type
CID 6039. Nipple Characteristic
CID 6040. Non-Lesion Object Type
CID 6041. Mammography Image Quality Finding
CID 6042. Status of Results
CID 6043. Types of Mammography CAD Analysis
CID 6044. Types of Image Quality Assessment
CID 6045. Mammography Types of Quality Control Standard
CID 6046. Units of Follow-Up Interval
CID 6047. CAD Processing and Findings Summary
CID 6048. CAD Operating Point Axis Label
CID 6050. Breast Procedure Reported
CID 6051. Breast Procedure Reason
CID 6052. Breast Imaging Report Section Title
CID 6053. Breast Imaging Report Elements
CID 6054. Breast Imaging Findings
CID 6055. Breast Clinical Finding or Indicated Problem
CID 6056. Associated Findings for Breast
CID 6057. Ductography Findings for Breast
CID 6058. Procedure Modifiers for Breast
CID 6059. Breast Implant Types
CID 6060. Breast Biopsy Techniques
CID 6061. Breast Imaging Procedure Modifiers
CID 6062. Interventional Procedure Complications
CID 6063. Interventional Procedure Results
CID 6064. Ultrasound Findings for Breast
CID 6065. Instrument Approach
CID 6066. Target Confirmation
CID 6067. Fluid Color
CID 6068. Tumor Stages From AJCC
CID 6069. Nottingham Combined Histologic Grade
CID 6070. Bloom-Richardson Histologic Grade
CID 6071. Histologic Grading Method
CID 6072. Breast Implant Findings
CID 6080. Gynecological Hormones
CID 6081. Breast Cancer Risk Factors
CID 6082. Gynecological Procedures
CID 6083. Procedures for Breast
CID 6084. Mammoplasty Procedures
CID 6085. Therapies for Breast
CID 6086. Menopausal Phase
CID 6087. General Risk Factors
CID 6088. OB-GYN Maternal Risk Factors
CID 6089. Substances
CID 6090. Relative Usage, Exposure Amount
CID 6091. Relative Frequency of Event Values
CID 6092. Quantitative Concepts for Usage, Exposure
CID 6093. Qualitative Concepts for Usage, Exposure Amount
CID 6094. Qualitative Concepts for Usage, Exposure Frequency
CID 6095. Numeric Properties of Procedures
CID 6096. Pregnancy Status
CID 6097. Side of Family
CID 6100. Chest Component Categories
CID 6101. Chest Finding or Feature
CID 6102. Chest Finding or Feature Modifier
CID 6103. Abnormal Lines Finding or Feature
CID 6104. Abnormal Opacity Finding or Feature
CID 6105. Abnormal Lucency Finding or Feature
CID 6106. Abnormal Texture Finding or Feature
CID 6107. Width Descriptor
CID 6108. Chest Anatomic Structure Abnormal Distribution
CID 6109. Radiographic Anatomy Finding or Feature
CID 6110. Lung Anatomy Finding or Feature
CID 6111. Bronchovascular Anatomy Finding or Feature
CID 6112. Pleura Anatomy Finding or Feature
CID 6113. Mediastinum Anatomy Finding or Feature
CID 6114. Osseous Anatomy Finding or Feature
CID 6115. Osseous Anatomy Modifiers
CID 6116. Muscular Anatomy
CID 6117. Vascular Anatomy
CID 6118. Size Descriptor
CID 6119. Chest Border Shape
CID 6120. Chest Border Definition
CID 6121. Chest Orientation Descriptor
CID 6122. Chest Content Descriptor
CID 6123. Chest Opacity Descriptor
CID 6124. Location in Chest
CID 6125. General Chest Location
CID 6126. Location in Lung
CID 6127. Segment Location in Lung
CID 6128. Chest Distribution Descriptor
CID 6129. Chest Site Involvement
CID 6130. Severity Descriptor
CID 6131. Chest Texture Descriptor
CID 6132. Chest Calcification Descriptor
CID 6133. Chest Quantitative Temporal Difference Type
CID 6134. Chest Qualitative Temporal Difference Type
CID 6135. Image Quality Finding
CID 6136. Chest Types of Quality Control Standard
CID 6137. Types of CAD Analysis
CID 6138. Chest Non-Lesion Object Type
CID 6139. Non-Lesion Modifiers
CID 6140. Calculation Methods
CID 6141. Attenuation Coefficient Measurements
CID 6142. Calculated Value
CID 6143. Lesion Response
CID 6144. RECIST Defined Lesion Response
CID 6145. Baseline Category
CID 6146. Time Point Types
CID 6147. Response Criteria
CID 6151. Background Echotexture
CID 6152. Orientation
CID 6153. Lesion Boundary
CID 6154. Echo Pattern
CID 6155. Posterior Acoustic Features
CID 6157. Vascularity
CID 6158. Correlation to Other Findings
CID 6159. Malignancy Type
CID 6160. Breast Primary Tumor Assessment From AJCC
CID 6161. Clinical Regional Lymph Node Assessment for Breast
CID 6162. Assessment of Metastasis for Breast
CID 6163. Menstrual Cycle Phase
CID 6164. Time Intervals
CID 6165. Breast Linear Measurements
CID 6166. CAD Geometry Secondary Graphical Representation
CID 6200. Colon Overall Assessment
CID 6201. Colon Finding or Feature
CID 6202. Colon Finding or Feature Modifier
CID 6203. Colon Non-Lesion Object Type
CID 6204. Anatomic Non-Colon Findings
CID 6205. Clockface Location for Colon
CID 6206. Recumbent Patient Orientation for Colon
CID 6207. Colon Quantitative Temporal Difference Type
CID 6208. Colon Types of Quality Control Standard
CID 6209. Colon Morphology Descriptor
CID 6210. Location in Intestinal Tract
CID 6211. Colon CAD Material Description
CID 6212. Calculated Value for Colon Findings
CID 7000. Diagnostic Imaging Report Document Titles
CID 7001. Diagnostic Imaging Report Headings
CID 7002. Diagnostic Imaging Report Elements
CID 7003. Diagnostic Imaging Report Purposes of Reference
CID 7004. Waveform Purposes of Reference
CID 7005. Contributing Equipment Purposes of Reference
CID 7006. SR Document Purposes of Reference
CID 7008. Media Import
CID 7009. Purpose of Reference to Predecessor Report
CID 7010. Key Object Selection Document Title
CID 7011. Rejected for Quality Reasons
CID 7012. Best in Set
CID 7021. Measurement Report Document Titles
CID 7030. Institutional Departments, Units and Services
CID 7040. Broselow-Luten Pediatric Size Categories
CID 7042. Calcium Scoring Patient Size Categories
CID 7050. De-Identification Method
CID 7100. RCS Registration Method Type
CID 7101. Brain Atlas Fiducials
CID 7140. Brain Structures for Volumetric Measurements
CID 7150. Segmentation Property Categories
CID 7151. Segmentation Property Types
CID 7152. Cardiac Structure Segmentation Types
CID 7153. CNS Tissue Segmentation Types
CID 7154. Abdominal Organ Segmentation Types
CID 7155. Thoracic Tissue Segmentation Types
CID 7156. Vascular Tissue Segmentation Types
CID 7157. Device Segmentation Types
CID 7158. Artifact Segmentation Types
CID 7159. Lesion Segmentation Types
CID 7160. Pelvic Organ Segmentation Types
CID 7161. Physiology Segmentation Types
CID 7162. Surface Processing Algorithm Families
CID 7165. Abstract Segmentation Types
CID 7166. Common Tissue Segmentation Types
CID 7167. Peripheral Nervous System Segmentation Types
CID 7180. Abstract Multi-Dimensional Image Model Component Semantics
CID 7181. Abstract Multi-Dimensional Image Model Component Units
CID 7182. Abstract Multi-Dimensional Image Model Dimension Semantics
CID 7183. Abstract Multi-Dimensional Image Model Dimension Units
CID 7184. Abstract Multi-Dimensional Image Model Axis Direction
CID 7185. Abstract Multi-Dimensional Image Model Axis Orientation
CID 7186. Abstract Multi-Dimensional Image Model Qualitative Dimension Sample Semantics
CID 7201. Referenced Image Purposes of Reference
CID 7202. Source Image Purposes of Reference
CID 7203. Image Derivation
CID 7205. Purpose of Reference to Alternate Representation
CID 7210. Related Series Purposes of Reference
CID 7215. Spectroscopy Purpose of Reference
CID 7220. RT Dose Derivation
CID 7221. RT Dose Purpose of Reference
CID 7222. Parametric Map Derivation Image Purpose of Reference
CID 7250. Multi-Frame Subset Type
CID 7300. Implant Materials
CID 7301. Intervention Types
CID 7302. Implant Templates View Orientations
CID 7303. Implant Templates Modified View Orientations
CID 7304. Implant Target Anatomy
CID 7305. Implant Planning Landmarks
CID 7306. Human Hip Implant Planning Landmarks
CID 7307. Implant Component Types
CID 7308. Human Hip Implant Component Types
CID 7309. Human Trauma Implant Component Types
CID 7310. Implant Fixation Method
CID 7320. Planning Methods
CID 7445. Device Participating Roles
CID 7450. Person Roles
CID 7451. Family Member
CID 7452. Organizational Roles
CID 7453. Performing Roles
CID 7454. Species
CID 7455. Sex
CID 7456. Units of Measure for Age
CID 7460. Units of Linear Measurement
CID 7461. Units of Area Measurement
CID 7462. Units of Volume Measurement
CID 7464. General Region of Interest Measurement Modifiers
CID 7465. Measurements Derived From Multiple ROI Measurements
CID 7466. PET Region of Interest Measurements
CID 7467. Grey Level Co-occurrence Matrix Measurements
CID 7468. Texture Measurements
CID 7469. Generic Intensity and Size Measurements
CID 7470. Linear Measurements
CID 7471. Area Measurements
CID 7472. Volume Measurements
CID 7473. General Area Calculation Methods
CID 7474. General Volume Methods
CID 7480. Breed
CID 7481. Breed Registry
CID 7482. DX Anatomy Imaged for Animals
CID 7483. Common Anatomic Regions for Animals
CID 7484. DX View for Animals
CID 7486. Mixed Breeds
CID 8101. Container Types
CID 8102. Container Component Types
CID 8103. Anatomic Pathology Specimen Types
CID 8104. Breast Tissue Specimen Types
CID 8109. Specimen Collection Procedure
CID 8110. Specimen Sampling Procedure
CID 8111. Specimen Preparation Procedure
CID 8112. Specimen Stains
CID 8113. Specimen Preparation Steps
CID 8114. Specimen Fixatives
CID 8115. Specimen Embedding Media
CID 8120. WSI Referenced Image Purposes of Reference
CID 8121. Microscopy Lens Type
CID 8122. Microscopy Illuminator and Sensor Color
CID 8123. Microscopy Illumination Method
CID 8124. Microscopy Filter
CID 8125. Microscopy Illuminator Type
CID 8130. Staining Protocols
CID 8131. Pathology Imaging Protocols
CID 8132. Magnification Selection
CID 8133. Tissue Selection
CID 8201. Surface Scan Acquisition Types
CID 8202. Surface Scan Mode Types
CID 8203. Surface Scan Registration Method Types
CID 8300. Visual Evaluation Methods
CID 8301. Test Pattern Codes
CID 8302. Measurement Pattern Codes
CID 8303. Display Device Type
CID 9000. Physical Unit Descriptors
CID 9231. Workitem Definition
CID 9241. Radiotherapy General Workitem Definition
CID 9242. Radiotherapy Acquisition Workitem Definition
CID 9243. Radiotherapy Registration Workitem Definition
CID 9250. Scheduled Processing Parameter Concept Codes for RT Treatment
CID 9300. Procedure Discontinuation Reasons
CID 9301. Modality PPS Discontinuation Reasons
CID 9302. Media Import PPS Discontinuation Reasons
CID 10000. Scope of Accumulation
CID 10001. UID Types
CID 10002. Irradiation Event Types
CID 10003. Equipment Plane Identification
CID 10004. Fluoro Modes
CID 10006. X-Ray Filter Materials
CID 10007. X-Ray Filter Types
CID 10008. Dose Related Distance Measurements
CID 10009. Measured/Calculated
CID 10010. Dose Measurement Devices
CID 10011. Effective Dose Evaluation Method
CID 10013. CT Acquisition Type
CID 10014. Contrast Imaging Technique
CID 10015. CT Dose Reference Authorities
CID 10016. Anode Target Material
CID 10017. X-Ray Grid
CID 10020. Source of Projection X-Ray Dose Information
CID 10021. Source of CT Dose Information
CID 10022. Label Types
CID 10023. Size Specific Dose Estimation Method for CT
CID 10025. Radiation Dose Reference Points
CID 10030. Detector Types
CID 10031. CR/DR Mechanical Configuration
CID 10032. Projection X-Ray Acquisition Device Types
CID 10033. CT Reconstruction Algorithm
CID 10040. Radiopharmaceutical Organ Dose Reference Authority
CID 10041. Source of Radioisotope Activity Information
CID 10043. Intravenous Extravasation Symptoms
CID 10044. Radiosensitive Organs
CID 10045. Radiopharmaceutical Patient State
CID 10046. GFR Measurements
CID 10047. GFR Measurement Methods
CID 12001. Ultrasound Protocol Types
CID 12002. Ultrasound Protocol Stage Types
CID 12003. OB-GYN Dates
CID 12004. Fetal Biometry Ratios
CID 12005. Fetal Biometry Measurements
CID 12006. Fetal Long Bones Measurements
CID 12007. Fetal Cranium
CID 12008. OB-GYN Amniotic Sac
CID 12009. Early Gestation Biometry Measurements
CID 12011. Ultrasound Pelvis and Uterus
CID 12012. OB Equations and Tables
CID 12013. Gestational Age Equations and Tables
CID 12014. OB Fetal Body Weight Equations and Tables
CID 12015. Fetal Growth Equations and Tables
CID 12016. Estimated Fetal Weight Percentile Equations and Tables
CID 12017. Growth Distribution Rank
CID 12018. OB-GYN Summary
CID 12019. OB-GYN Fetus Summary
CID 12020. Fetal Biometry Anatomic Sites
CID 12021. Fetal Long Bone Anatomic Sites
CID 12022. Fetal Cranium Anatomic Sites
CID 12023. Pelvis and Uterus Anatomic Sites
CID 12030. Ultrasound Contrast/Bolus Agents
CID 12031. Protocol Interval Events
CID 12032. Transducer Scan Pattern
CID 12033. Ultrasound Transducer Geometry
CID 12034. Ultrasound Transducer Beam Steering
CID 12035. Ultrasound Transducer Application
CID 12101. Vascular Summary
CID 12102. Temporal Periods Relating to Procedure or Therapy
CID 12103. Vascular Ultrasound Anatomic Location
CID 12104. Extracranial Arteries
CID 12105. Intracranial Cerebral Vessels
CID 12106. Intracranial Cerebral Vessels (Unilateral)
CID 12107. Upper Extremity Arteries
CID 12108. Upper Extremity Veins
CID 12109. Lower Extremity Arteries
CID 12110. Lower Extremity Veins
CID 12111. Abdominal Arteries (Lateral)
CID 12112. Abdominal Arteries (Unilateral)
CID 12113. Abdominal Veins (Lateral)
CID 12114. Abdominal Veins (Unilateral)
CID 12115. Renal Vessels
CID 12116. Vessel Segment Modifiers
CID 12117. Vessel Branch Modifiers
CID 12118. Measurement Orientation
CID 12119. Vascular Ultrasound Property
CID 12120. Blood Velocity Measurements by Ultrasound
CID 12121. Vascular Indices and Ratios
CID 12122. Other Vascular Properties
CID 12123. Carotid Ratios
CID 12124. Renal Ratios
CID 12140. Pelvic Vasculature Anatomical Location
CID 12141. Fetal Vasculature Anatomical Location
CID 12200. Echocardiography Left Ventricle
CID 12201. Left Ventricle Linear
CID 12202. Left Ventricle Volume
CID 12203. Left Ventricle Other
CID 12204. Echocardiography Right Ventricle
CID 12205. Echocardiography Left Atrium
CID 12206. Echocardiography Right Atrium
CID 12207. Echocardiography Mitral Valve
CID 12208. Echocardiography Tricuspid Valve
CID 12209. Echocardiography Pulmonic Valve
CID 12210. Echocardiography Pulmonary Artery
CID 12211. Echocardiography Aortic Valve
CID 12212. Echocardiography Aorta
CID 12214. Echocardiography Pulmonary Veins
CID 12215. Echocardiography Vena Cavae
CID 12216. Echocardiography Hepatic Veins
CID 12217. Echocardiography Cardiac Shunt
CID 12218. Echocardiography Congenital
CID 12219. Pulmonary Vein Modifiers
CID 12220. Echocardiography Common Measurements
CID 12221. Flow Direction
CID 12222. Orifice Flow Properties
CID 12223. Echocardiography Stroke Volume Origin
CID 12224. Ultrasound Image Modes
CID 12226. Echocardiography Image View
CID 12227. Echocardiography Measurement Method
CID 12228. Echocardiography Volume Methods
CID 12229. Echocardiography Area Methods
CID 12230. Gradient Methods
CID 12231. Volume Flow Methods
CID 12232. Myocardium Mass Methods
CID 12233. Cardiac Phase
CID 12234. Respiration State
CID 12235. Mitral Valve Anatomic Sites
CID 12236. Echo Anatomic Sites
CID 12237. Echocardiography Anatomic Site Modifiers
CID 12238. Wall Motion Scoring Schemes
CID 12239. Cardiac Output Properties
CID 12240. Left Ventricle Area
CID 12241. Tricuspid Valve Finding Sites
CID 12242. Aortic Valve Finding Sites
CID 12243. Left Ventricle Finding Sites
CID 12244. Congenital Finding Sites
CID 12245. Cardiac Ultrasound Report Titles
CID 12246. Cardiac Ultrasound Indication for Study
CID 12247. Pediatric, Fetal and Congenital Cardiac Surgical Interventions
CID 12248. Cardiac Ultrasound Summary Codes
CID 12249. Cardiac Ultrasound Fetal Summary Codes
CID 12250. Cardiac Ultrasound Common Linear Measurements
CID 12251. Cardiac Ultrasound Linear Valve Measurements
CID 12252. Cardiac Ultrasound Cardiac Function
CID 12253. Cardiac Ultrasound Area Measurements
CID 12254. Cardiac Ultrasound Hemodynamic Measurements
CID 12255. Cardiac Ultrasound Myocardium Measurements
CID 12257. Cardiac Ultrasound Left Ventricle
CID 12258. Cardiac Ultrasound Right Ventricle
CID 12259. Cardiac Ultrasound Ventricles Measurements
CID 12260. Cardiac Ultrasound Pulmonary Artery
CID 12261. Cardiac Ultrasound Pulmonary Vein
CID 12262. Cardiac Ultrasound Pulmonary Valve
CID 12263. Cardiac Ultrasound Venous Return Pulmonary Measurements
CID 12264. Cardiac Ultrasound Venous Return Systemic Measurements
CID 12265. Cardiac Ultrasound Atria and Atrial Septum Measurements
CID 12266. Cardiac Ultrasound Mitral Valve
CID 12267. Cardiac Ultrasound Tricuspid Valve
CID 12268. Cardiac Ultrasound Atrioventricular Valves Measurements
CID 12269. Cardiac Ultrasound Interventricular Septum Measurements
CID 12270. Cardiac Ultrasound Aortic Valve
CID 12271. Cardiac Ultrasound Outflow Tracts Measurements
CID 12272. Cardiac Ultrasound Semilunar Valves, Annulate and Sinuses Measurements
CID 12273. Cardiac Ultrasound Aortic Sinotubular Junction
CID 12274. Cardiac Ultrasound Aorta Measurements
CID 12275. Cardiac Ultrasound Coronary Arteries Measurements
CID 12276. Cardiac Ultrasound Aorto Pulmonary Connections Measurements
CID 12277. Cardiac Ultrasound Pericardium and Pleura Measurements
CID 12279. Cardiac Ultrasound Fetal General Measurements
CID 12280. Cardiac Ultrasound Target Sites
CID 12281. Cardiac Ultrasound Target Site Modifiers
CID 12282. Cardiac Ultrasound Venous Return Systemic Finding Sites
CID 12283. Cardiac Ultrasound Venous Return Pulmonary Finding Sites
CID 12284. Cardiac Ultrasound Atria and Atrial Septum Finding Sites
CID 12285. Cardiac Ultrasound Atrioventricular Valves Finding Sites
CID 12286. Cardiac Ultrasound Interventricular Septum Finding Sites
CID 12287. Cardiac Ultrasound Ventricles Finding Sites
CID 12288. Cardiac Ultrasound Outflow Tracts Finding Sites
CID 12289. Cardiac Ultrasound Semilunar Valves, Annulus and Sinuses Finding Sites
CID 12290. Cardiac Ultrasound Pulmonary Arteries Finding Sites
CID 12291. Cardiac Ultrasound Aorta and Coronary Arteries Finding Sites
CID 12292. Cardiac Ultrasound Aorta and Coronary Arteries Finding Sites
CID 12293. Cardiac Ultrasound Aorto Pulmonary Connections Finding Sites
CID 12294. Cardiac Ultrasound Pericardium and Pleura Finding Sites
TID 3401. ECG Acquisition Context
TID 3403. Catheterization Acquisition Context
TID 3450. Cardiac Electrophysiology Acquisition Context
TID 3460. Projection Radiography Acquisition Context
TID 3470. NM/PET Acquisition Context
TID 3471. PET Covariates Acquisition Context
TID 8001. Specimen Preparation
TID 8002. Specimen Sampling
TID 8003. Specimen Staining
TID 8004. Specimen Localization
TID 8010. Slide Imaging Parameters
TID 15100. Contrast Agent/Pre-Medication Protocol Context
TID 15101. NM/PET Protocol Context
TID 15200. JJ1017 Protocol Context
D-1. DICOM Controlled Terminology Definitions
E-1. French Language Meanings of Selected Codes
E-2. Mapping of Pathology Codes used in DICOM to ADICAP
F-1. Japanese Language Meanings of Selected Codes
G-1. English Code Meanings of Selected Codes
H-1. Code Meanings of LOINC Codes
I-1. Examples of the Common Nomenclature for the Type of Endoscopy Performed
J-1. SNOMED DICOM Microglossary Retired Codes
L-1. Corresponding SNOMED Terms for Human Use
L-2. Corresponding SNOMED Terms for Animals
M-1. German Language Meanings of Selected Codes

# Notice and Disclaimer

The information in this publication was considered technically sound by the consensus of persons engaged in the development and approval of the document at the time it was developed. Consensus does not necessarily mean that there is unanimous agreement among every person participating in the development of this document.

NEMA standards and guideline publications, of which the document contained herein is one, are developed through a voluntary consensus standards development process. This process brings together volunteers and/or seeks out the views of persons who have an interest in the topic covered by this publication. While NEMA administers the process and establishes rules to promote fairness in the development of consensus, it does not write the document and it does not independently test, evaluate, or verify the accuracy or completeness of any information or the soundness of any judgments contained in its standards and guideline publications.

NEMA disclaims liability for any personal injury, property, or other damages of any nature whatsoever, whether special, indirect, consequential, or compensatory, directly or indirectly resulting from the publication, use of, application, or reliance on this document. NEMA disclaims and makes no guaranty or warranty, expressed or implied, as to the accuracy or completeness of any information published herein, and disclaims and makes no warranty that the information in this document will fulfill any of your particular purposes or needs. NEMA does not undertake to guarantee the performance of any individual manufacturer or seller's products or services by virtue of this standard or guide.

In publishing and making this document available, NEMA is not undertaking to render professional or other services for or on behalf of any person or entity, nor is NEMA undertaking to perform any duty owed by any person or entity to someone else. Anyone using this document should rely on his or her own independent judgment or, as appropriate, seek the advice of a competent professional in determining the exercise of reasonable care in any given circumstances. Information and other standards on the topic covered by this publication may be available from other sources, which the user may wish to consult for additional views or information not covered by this publication.

NEMA has no power, nor does it undertake to police or enforce compliance with the contents of this document. NEMA does not certify, test, or inspect products, designs, or installations for safety or health purposes. Any certification or other statement of compliance with any health or safety-related information in this document shall not be attributable to NEMA and is solely the responsibility of the certifier or maker of the statement.

# Foreword

This DICOM Standard was developed according to the procedures of the DICOM Standards Committee.

The DICOM Standard is structured as a multi-part document using the guidelines established in [ISO/IEC Directives, Part 3].

PS3.1 should be used as the base reference for the current parts of this standard.

# 1 Scope and Field of Application

This part of the DICOM Standard specifies the DICOM Content Mapping Resource (DCMR), which defines the templates and context groups used elsewhere in the standard.

# 2 Normative References

The following standards contain provisions that, through reference in this text, constitute provisions of this Standard. At the time of publication, the editions indicated were valid. All standards are subject to revision, and parties to agreements based on this Standard are encouraged to investigate the possibilities of applying the most recent editions of the standards indicated below.

Extracts of ISO/IEEE 11073 reprinted by permission of IEEE, 3 Park Avenue, New York, NY 10016-5997 USA. Copyright 2004 by IEEE. http://standards.ieee.org/.

Under license from IEEE, the term codes and descriptions of the ISO/IEEE 11073 Nomenclature are available at no cost through the Rosetta Terminology Mapping Management System of the U.S. National Institute of Standards and Technology. http://hit-testing.nist.gov:13110/rtmms/index.htm

This material includes an extract of SNOMED Clinical Terms® (SNOMED CT®), which is used by permission of the International Health Terminology Standards Development Organisation (IHTSDO). All rights reserved. SNOMED CT®, was originally created by The College of American Pathologists. "SNOMED" and "SNOMED CT" are registered trademarks of the IHTSDO.

All the SNOMED terms used in this Standard are the subject of a royalty pre-paid licensing agreement between NEMA and CAP that allows their use in DICOM applications without further license or payment of fee

## 2.1 General

[ISO/IEC Directives, Part 3] ISO/IEC. 1989. Drafting and presentation of International Standards.

[UCUM] Regenstrief Institute for Health Care, Indianapolis. 2000. UCUM Unified Code for Units of Measure.

[LOINC] Regenstrief Institute for Health Care, Indianapolis. 2000. LOINC® Logical Observation Identifier Names and Codes.

[SNOMED] International Health Terminology Standards Development Organisation (IHTSDO). . SNOMED CT Systematized Nomenclature of Medicine - Clinical Terms.

[JJ1017] Japan Medical Imaging and Radiological Systems Industries Association (JIRA) and Japanese Association of Healthcare Information Systems Industry (JAHIS). Guidelines for HIS, RIS, PACS - Modality Data Communication on Scheduling, Billing, and Examination Records. http://www.jira-net.or.jp/commission/system/04_information/files/JJ1017VER3_20051005.doc .

[RFC 3066] IETF. . Tags for the Identification of Languages.

[IEC 60601-2-44] IEC. . Medical Electrical Equipment - Part 2-44: Particular Requirements for the Safety of X-Ray Equipment for Computed Tomography.

[ISO 639] ISO. . Codes for the representation of names of languages, International Organization for Standardization.

[ISO 3166] ISO. . Codes for the representation of names of countries, International Organization for Standardization.

[ASTM E 1762-04] ASTM. . Standard Guide for Electronic Authentication of Health Care Information, ASTM International.

[ASTM E 2084-00] ASTM. . Standard Specification for Authentication of Healthcare Information Using Digital Signatures, ASTM International.

[HL7 v3 CMET] HL7. . Version 3 Standard: Common Message Element Types.

[RadLex] RSNA, Chicago. 2006. A Lexicon for Uniform Indexing and Retrieval of Radiology Information Resources. http://www.radlex.org/ .

[ETDRS Report Number 10] ETDRS. . Report Number 10, Grading Diabetic Retinopathy from Stereoscopic Color Fundus Photographs- An Extension of the Modified Airlie House Classification. Ophthalmology, May 1991, vol98 (p786-805), Supplement.

[NEMA XR 25-2010] National Electrical Manufacturers Association, Rosslyn, Virginia. 2010. Computed Tomography Dose Check Standard. http://www.nema.org/stds/xr25.cfm .

[ISO/IEEE 11073-10101] ISO/IEEE. . Health informatics - Point-of-care medical device communication - Nomenclature.

[ISO/IEEE 11073-10102] ISO/IEEE. . Health informatics - Point-of-care medical device communication - Annotated ECG - Nomenclature.

[AAPM Report 204] American Association of Physicists in Medicine, College Park, Maryland. 2011. Size-Specific Dose Estimates (SSDE) in Pediatric and Adult Body CT Examinations. http://www.aapm.org/pubs/reports/RPT_204.pdf .

[AAPM OR 03] American Association of Physicists in Medicine. 2005. Assessment of Display Performance for Medical Imaging Systems. http://www.aapm.org/pubs/reports/OR_03.pdf .

[DIN 6868-57] Deutsches Institut für Normung. 2001. Image quality assurance in diagnostic X-ray departments - Acceptance testing for image display devices.

[IEC 62563-1] IEC. 2009. Ed 1.0. Medical Electrical Equipment - Medical image display systems - Part 1: Evaluation methods.

[SMPTE RP133] Society of Motion Picture and Television Engineers (SMPTE). 1991. Specifications for Medical Diagnostic Imaging Test Pattern for Television Monitors and Hard-copy Recording Cameras.

## 2.2 BI-RADS® Terminology and Nomenclature

A portion of the terminology used within the Mammography CAD SR SOP Class and the Breast Imaging Report and Relevant Patient Information for Breast Imaging templates is derived from BI-RADS®, a copyrighted lexicon of breast imaging terminology and nomenclature licensed by DICOM from the American College of Radiology.

[BI-RADS®] American College of Radiology, Reston, Virginia. 1998. 3.0. Breast Imaging Reporting and Data System Atlas. http://www.acr.org/Quality-Safety/Resources/BIRADS .

## 2.3 MQCM 1999 Terminology and Nomenclature

References to MQCM 1999 are made in the description of the Mammography CAD SR SOP Class. In this MQCM 1999 refers to the Mammography Quality Control Manual 1999, available from the American College of Radiology. This document describes a standardized approach to mammographic acquisition standards, patient positioning, and so on. The DICOM standard does not require Mammography CAD SR SOP Class implementations to adhere to MQCM 1999.

[MQCM] American College of Radiology, Reston, Virginia. 1999. Mammography Quality Control Manual. http://www.acr.org/Education/Education-Catalog/Products/639 .

## 2.4 MQSA Terminology and Nomenclature

References to MQSA are made in the description of the Mammography CAD SR SOP Class. In this MQSA refers to the Mammography Quality Standards Act final rules. While MQSA is a federal regulation of the United States government, it provides the only widely published standards for mammographic quality and is incorporated in this document for that reason. The DICOM standard does not require Mammography CAD SR SOP Class implementations to adhere to MQSA.

[MQSA] U.S. Food and Drug Administration, Silver Spring, Maryland. 2002. Mammography Quality Standards Act Regulations. http://www.fda.gov/Radiation-EmittingProducts/MammographyQualityStandardsActandProgram/Regulations/ucm110906.htm .

## 2.5 ACR Position Statement

[ACR Position Statement] American College of Radiology, Reston, Virginia. 2001. Quality Control and Improvement, Safety, Infection Control, and Patient Education. http://www.acr.org/Quality-Safety/Radiology-Safety .

## 2.6 Chest Radiology and CT

References are made in the description of the Chest CAD SR templates and context groups.

[Fraser and Pare] Fraser, Muller, Colman, and Pare. 1999. 4th. I. xvii-xxxi. Diagnosis of Diseases of the Chest. Terms Used in Chest Radiology.

[Fraser and Pare] Fraser, Muller, Colman, and Pare. 1999. 4th. I. xxxiii-xxxvi. Diagnosis of Diseases of the Chest. Terms for CT of the Lungs.

[ACR CT PE] American College of Radiology, Reston, Virginia. 2001. 109-113. ACR Standards. ACR Standard for the Performance of Computed Tomography for the Detection of Pulmonary Embolism in Adults.

[ACR HR CT] American College of Radiology, Reston, Virginia. 2001. 115-118. ACR Standards. ACR Standard for the Performance of High-Resolution Computed Tomography (HRCT) of the Lungs in Adults.

[ACR Radiography] American College of Radiology, Reston, Virginia. 2001. 95-98. ACR Standards. ACR Standard for the Performance of Pediatric and Adult Chest Radiography.

[ACR Thoracic CT] American College of Radiology, Reston, Virginia. 2001. 103-107. ACR Standards. ACR Standard for the Performance of Pediatric and Adult Thoracic Computed Tomography (CT).

## 2.7 Response Evaluation Criteria

References to Response Evaluation Criteria are made from the Chest CAD SR templates and context groups

[RECIST] Journal of the National Cancer Institute. Therasse. February 2, 2000. 92. 3. 205-216. “New Guidelines to Evaluate the Response to Treatment in Solid Tumors”. http://www.eortc.be/recist/ .

[WHO] World Health Organization, Geneva. 1979. WHO Handbook for Reporting Results for Cancer Treatment. WHO Offset Publication No. 48. http://whqlibdoc.who.int/publications/9241700483.pdf .

## 2.8 Myocardial Segmentation

[Cerqueira 2002] Circulation. Cerqueira MD. 2002. 105. 4. 539. “AHA Scientific Statement: Standardized Myocardial Segmentation and Nomenclature for Tomographic Imaging of the Heart”. doi:10.1161/hc0402.102975

## 2.9 Quantitation of the Left Ventricle

[Schiller 1989] Journal of the American Society of Echocardiography. Schiller. Oct 1989. 2. 5. 358-367. “Recommendations for Quantitation of the Left Ventricle by Two-Dimensional Echocardiography”.

## 2.10 Cancer Staging

[AJCC] American Joint Committee on Cancer. 2002. Sixth. AJCC Cancer Staging Handbook: From the AJCC Cancer Staging Manual. Springer-Verlag. 0-387-95270-5.

## 2.11 Quantitative Arteriography and Ventriculography

[Sheehan, 1986] Circulation. Sheehan FH, Bolson EL, Dodge HT, Mathey DG, Schofer J, and Woo HW. 1986. 74. 2. 293-305. “Advantages and applications of the centerline method for characterizing regional ventricular function”. doi:10.1161/01.CIR.74.2.293

[Slager, 1986] J Am Coll Cardiol.. Slager CJ, Hooghoudt TE, Serruys PW, Schuurbiers JC, Reiber JH, Meester GT, Verdouw PD, and Hugenholtz PG. 1986. 6. 2. 317-26. “Quantitative assessment of regional left ventricular motion using endocardial landmarks”. doi:10.1016/S0735-1097(86)80498-3

[Kennedy, 1970] Am Heart J. Kennedy JW, Trenholme SE, and Kasser IS. 1970. 80. 3. 343. “Left ventricular volume and mass from single-plane cineangiocardiogram. A comparison of anteroposterior and right anterior oblique methods”.

[Dodge, 1960] Am Heart J. Dodge HT, Sandler H, Ballew DW, and Lord JD. 1960. 60. 5. 762. “The use of biplane angiocardiography for the measurement of left ventricular volume in man”. http://www.sciencedirect.com/science/article/pii/0002870360903598 .

[Wynne, 1978] Am J Cardiol. Wynne J, Green LH, Mann T, Levin D, and Grossman W. 1978. 41. 4. 726. “Estimation of left ventricular volumes in man from biplane cineangiograms filmed in oblique projections”.

[Boak, 1977] Cathet Cardiovasc Diagn. Boak, JG, Bove AA, Kreulen T, and Spann JF. 1977. 3. 3. 217-30. “A geometric basis for calculation of right ventricular volume in man”. doi:10.1002/ccd.1810030305

[Ferlinz, 1977] Am Heart J. Ferlinz J. 1977. 94. 1. 87-90. “Measurements of right ventricular volumes in man from single plane cineangiograms. A comparison to the biplane approach”. http://www.sciencedirect.com/science/article/pii/S0002870377803487 .

[Graham, 1973] Circulation. Graham TP, Jarmakani JM, Atwood GF, and Canent RV. 1973. 47. 1. 144-53. “Right ventricular volume determinations in children. Normal values and observations with volume or pressure overload”. doi:10.1161/01.CIR.47.1.144

[Arcilla, 1971] Chest. Arcilla RA, Tsai P, Thilenius O, and Ranniger K. 1971. 60. 5. 446. “Angiographic method for volume estimation of right and left ventricles”. doi:10.1378/chest.60.5.446

## 2.12 IVUS

[Mintz, 2001] Journal of the American College of Cardiology. Mintz GS. 2001. 37. 5. 1478-1492. “American College of Cardiology Clinical Expert Consensus Document on Standards for Acquisition, Measurement and Reporting of Intravascular Ultrasound Studies (IVUS)”. doi:10.1016/S0735-1097(01)01175-5

[Di Mario, 1998] European Heart Journal. Di Mario C. 1998. 19. 2. 207-229. “Clinical Application and Image Interpretation in Intravascular Ultrasound”. doi:10.1053/euhj.1996.0433

[Tobis and Yock] Tobis JM and Yock PC. 1992. Intravascular Ultrasound Imaging. 0443088098.

## 2.13 C-RADS CT Colonography Reporting and Data System

[Zalis, 2005] Radiology. Zalis ME. 2005. 236. 1. 3-9. “CT Colonography Reporting and Data System: A Consensus Proposal”. doi:10.1148/radiol.2361041926

## 2.14 Implants

[Eggli, 1998] J Bone Joint Surg Br. Eggli S, Pisan M, and Müller ME. 1998. 80-B. 3. 382-390. “The value of preoperative planning for total hip arthroplasty”. http://www.bjj.boneandjoint.org.uk/content/80-B/3/382 .

# 3 Definitions

For the purposes of this Standard the following definitions apply.

## 3.1 Codes and Controlled Terminology Definitions:

The following definitions are commonly used in this Part of the DICOM Standard:

Baseline Context Group Identifier (BCID)

  Identifier that specifies the suggested Context Group for a Code Sequence Attribute.

Defined Context Group Identifier (DCID)

  Identifier that specifies the Context Group for a Code Sequence Attribute that shall be used.

Extensible Context Group

  Context Group that may be extended by a particular application by inclusion of additional concepts.

Non-Extensible Context Group

  Context Group whose defined set of concepts shall not be extended by an application.

Context Group

  A set of coded concepts defined by a Mapping Resource forming a set appropriate to use in a particular context.

Context Group Version

  Version of a Context Group.

Context ID (CID)

  Identifier of a Context Group.

Mapping Resource

  A resource that defines context-dependent usage constraints (i.e., Value Set or Relationship Type restrictions) for Attributes. A resource that specifies the mapping of the content of an external controlled terminology to the components of a message standard.

Relationship Type

  The association between two Concepts. Examples: "HAS PROPERTIES", "CONTAINS", "INFERRED FROM".

DICOM Content Mapping Resource (DCMR)

  A Mapping Resource that defines Templates and Context Groups for use in DICOM IODs.

Template

  A pattern that describes the Content Items, Value Types, Relationship Types and Value Sets that may be used in part of a Structured Report content tree, or in other Content Item constructs, such as Acquisition Context or Protocol Context. Analogous to a Module of an Information Object Definition.

Template ID (TID)

  Identifier of a Template.

Value Set

  The allowed values of a Code Sequence Attribute in a given context. Specified either as one or more individual values or by reference to a Context Group.

Baseline Template Identifier (BTID)

  Identifier that specifies a template suggested to be used in the creation of a set of Content Items.

Defined Template Identifier (DTID)

  Identifier that specifies a template that shall be used in the creation of a set of Content Items.

Extensible Template

  A template that may be extended by a particular application by inclusion of additional Content Items beyond those specified in the template.

Non-Extensible Template

  A template that specifies the exact set of Content Items and corresponding Value Sets that shall be used and that shall not be extended by an application.

Coding schemes

  Dictionaries (lexicons) of concepts (terms) with assigned codes and well defined meanings.

  ### Note

  Examples of coding schemes include SNOMED and LOINC.

## 3.2 Information Object Definitions:

This Part of the Standard makes use of the following terms defined in PS3.3:

Code Sequence Attribute

  See PS3.3.

# 4 Symbols and Abbreviations

The following symbols and abbreviations are used in this Part of the Standard.

Mammography CAD

  Computer-Aided Detection and/or Computer-Aided Diagnosis for Mammography

Chest CAD

  Computer-Aided Detection and/or Computer-Aided Diagnosis for chest radiography

Colon CAD

  Computer-Aided Detection and/or Computer-Aided Diagnosis for colon radiography

ACR

  American College of Radiology

ASE

  American Society of Echocardiography

CAP

  College of American Pathologists

DCMR

  DICOM Content Mapping Resource

NEMA

  National Electrical Manufacturers Association

RECIST

  Response Evaluation Criteria In Solid Tumors

SNOMED

  Systematized Nomenclature of Medicine

UCUM

  Unified Code for Units of Measure

WHO

  World Health Organization

EV

  Enumerated Value

DT

  Defined Term

CNAME

  Context Group Name

TNAME

  Template Name

BCID

  Baseline Context Group ID

DCID

  Defined Context Group ID

ECID

  Enumerated Context Group ID

BTID

  Baseline Template ID

DTID

  Defined Template ID

ETID

  Enumerated Template ID

The following upper-case abbreviations represent specific Attributes:

CV

  Code Value (0008,0100)

CSD

  Coding Scheme Designator (0008,0102)

CM

  Code Meaning (0008,0104)

CSV

  Coding Scheme Version (0008,0103)

# 5 Conventions

Terms listed in Section 3 are capitalized throughout the document.

# 6 Form of Template Specifications

Templates are patterns that specify the Concept Names, Requirements, Conditions, Value Types, Value Multiplicity, Value Set restrictions, Relationship Types and other attributes of Content Items for a particular application.

An IOD may specify that particular Standard Templates shall be used or may be used to define or constrain the content of a Content Item construct. A Content Item construct includes a coded concept name and one of several types of coded values. Content Item constructs are used in:

- the main Data Set and recursively nested Content Sequences (0040,A730) of the SR Document Content Module

- the Acquisition Context Sequence(0040,0555) of the Acquisition Context Module,

- the Protocol Context Sequence (0040,0440) and Content Item Modifier Sequence (0040,0441) of the Scheduled Procedure Step Module, Image Acquisition Results Module, and others.

- the Specimen Preparation Step Content Item Sequence (0040,0612) of the Specimen Module.

Annexes A and C of this Part define Standard Templates.

### Note

Standard Extended and Private Templates may be defined by implementers of the Standard. The rules for definition of Standard Extended and Private SR Templates are similar to the rules for definition of Standard Extended and Private SOP Classes. One row of a Template definition table corresponds to one row of a Module table.

Each Standard Template is specified by a Template table in this Part. Each Template table specifies exactly one Template, corresponding to a pattern of content within a Content Item construct.

Each Template table identifies whether the order of Content Items is significant or not significant. SOP Instances whose content is based on a Template where the order is significant shall encode the top level Content Items in the order they are specified in the Template, and the subsidiary Content Items under each parent item in the order they are specified, and so on for each Nesting Level. The significance of the order applies only to the Template itself; subsidiary included Templates may have a different order significance.

### Note

Even if a template specifies that the order is not significant, there may be significance to the order in which Content Items are encoded in a SOP Instance. For example, CONTAINER Content Items with attribute Continuity of Content (0040,A050) value CONTINUOUS encode Content Items in narrative sequence, and procedure logs encode Content Items in time order.

The Content Items from subsidiary templates may be intermingled if and only if the parent and subsidiary all specify that the order is not significant. This permits later refactoring into reusable templates.

The range of concepts and the options that are permitted in a family of SR Documents vary inversely with the level of constraint that is applied by the corresponding SR Template. The more narrow the range of concepts and the more restricted the options permitted by a Template, the more predictable the content of the SR Documents will be.

### Note

1. A very specific Template defines a family of SR Documents that are very similar to each other. They have a narrow range of content options (e.g., high level of constraint of Content Item values; use of CODE or NUM with Enumerated Context Groups) and their content is therefore highly predictable. A very general (e.g., permissive or broad) Template defines a family of SR Documents that may differ considerably from one another. They have a broader range of content options (e.g., low level of constraint of Content Item values; use of TEXT and relatively little restriction of Content Item values) and their content is less predictable.

2. The degree of interoperability that may be achieved with a family of SR Documents generated from a Template may be determined intentionally and precisely at a desired level by appropriate Template design to achieve the necessary degree of predictability of SR Document contents.

## 6.1 Template Table Field Definition

SR Templates are described using tables of the following form:

Type:

  (Non-) Extensible

Order:

  (Non-) Significant

Table TID <#>. <SR Context Template Name>

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  |  |  |  |  |  |
2 |  |  |  |  |  |  |  |
3 |  |  |  |  |  |  |  |

Acquisition Context Templates are described using tables of the following form:

Type:

  (Non-) Extensible

Order:

  (Non-) Significant

Table TID <#>. <Acquisition Context Template Name>

| VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | ---
1 |  |  |  |  |  |
2 |  |  |  |  |  |
3 |  |  |  |  |  |

Protocol Context Templates are described using tables of the following form:

Type:

  (Non-) Extensible

Order:

  (Non-) Significant

Table TID <#>. <Protocol Context Template Name>

| NL | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | ---
1 |  |  |  |  |  |  |
2 |  |  |  |  |  |  |
3 |  |  |  |  |  |  |

The semantics of the fields (columns) of Template tables are defined by subsections of this Section. A row of a Template table specifies either one Content Item or inclusion of another Template that may specify any number of Content Items (see Section 6.2.3 for definition of Included Templates). Each Template table is named by a title, identified by a TID number and further explained by a description such as explanation of Template contents, purpose and use cases.

The following conventions are defined for the form of references to coded concepts, Context Groups and Templates.

Code Meanings are enclosed in quotation marks (for example "cm"). Code Values and Coding Scheme Designators are not enclosed in quotation marks unless a comma occurs in the string.

References to coded concepts take the following form:

- EV or DT (CV, CSD, "CM")

  e.g., an Enumerated Value with only CV, CSD, and CM defined is represented as follows: EV (CV, CSD, "CM"), for example EV (T-04000, SNM3, "Breast").

- MemberOf { BCID or DCID (CID) CNAME } MemberOf selects one term from the specified context group.

If reference to a specific coding scheme version is required, it takes the following form:

- EV or DT (CV, CSD [CSV], "CM")

  e.g., DT (D3-81922, SRT [V1], "Aortic fistula").

References to Context Groups take the following form:

- BCID or DCID (CID) CNAME

  e.g., Defined Context Group 5000 is represented as follows: DCID (5000) Language.

References to Templates take the following form:

- BTID or DTID (TID) TNAME

  e.g., Baseline Template 1000 is represented as follows: BTID (1000) Quotation.

### 6.1.1 Row Number

Each row of a Template Table is denoted by a row number. The first row is numbered 1 and subsequent rows are numbered in ascending order with increments of 1. This number denotes a row for convenient description as well as reference in conditions. The Row Number of a Content Item in a Template may or may not be the same as the ordinal position of the corresponding Sequence Item (representing the Content Item) in a Content Sequence (0040,A730), depending on the number of times the Content Item is repeated.

The Content Item specified in the first row of a Template table may be of any Value Type. Specifically, it is not constrained to be a CONTAINER.

### 6.1.2 Nesting Level (NL)

The nesting level of Content Items is denoted by ">" symbols, one per level of nesting below the initial Source Content Item (of the Template) in a manner similar to the depiction of nested Sequences of Items in Modules Tables in PS3.3. When it is necessary to specify the Target Content Item(s) of a relationship, they are specified in the row(s) immediately following the corresponding Source Content Item. The Nesting Level of a Target Content Item is one greater than the Nesting Level of the corresponding (parent) Source Content Item. The Content Item specified in row 1 of a Template Table is at the top level (i.e., no ">" symbol is ever present in the NL field for the first Content Item in the table).

Acquisition Context templates have no Nesting Level field. Protocol Context and UPS Processing Parameter templates allow a single Nesting Level implemented through the Content Item Modifier Sequence (see PS3.3).

### 6.1.3 Relationship With Source Content Item (Parent)

Relationship Type and Relationship Mode (i.e., By-value or By-reference) constraints, if defined, are specified in this field, as described Table 6.1.3-1.

Relationship Type and Mode are specified for each row that specifies a target content item.

Relationship Type and Mode may also be specified when another Template is included, either "top-down" or "bottom-up" or both (i.e., in the "INCLUDE Template" row of the calling Template, or in all rows of the included Template, or in both places). There shall be no conflict between the Relationship Type and Mode of a row that includes another Template and the Relationship Type and Mode of the rows of the included Template.

### Note

SR IODs specify Enumerated Values for Relationship Types. If a Relationship Type other than one of the Defined Terms for Relationship Type (0040,A010) is specified in a Private SOP Class, there is a significant risk to interoperability. Documentation accompanying Templates for Private SOP Classes should define any Relationship-type extensions in the manner that the Standard Relationship Types are defined in PS3.3.

Acquisition Context and Protocol Context templates have no Relationship field.

Table 6.1.3-1. Syntax of Relationship Constraints

Expression | Definition
--- | ---
RTYPE | Relationship Mode is By-value and Relationship Type is RTYPE. For example, "INFERRED FROM".
R-RTYPE | Relationship Mode is By-reference and Relationship Type is RTYPE. For example, "R-INFERRED FROM".

### 6.1.4 Value Type (VT)

The Value Type field specifies the SR Value Type of the Content Item or conveys the word "INCLUDE" to indicate that another Template is to be included (substituted for the row). See Section 6.2.3 for further description of "Included Templates".

### 6.1.5 Concept Name

Any constraints on Concept Name are specified in this field as defined or enumerated coded entries, or as baseline or defined context groups. Alternatively, when the VT field is "INCLUDE", the Concept Name field specifies the template to be included.

### 6.1.6 Value Multiplicity (VM)

The VM field indicates the number of times that either a Content Item of the specified pattern or an included Template may appear in this position. Table 6.1.6-1 specifies the values that are permitted in this field.

Table 6.1.6-1. Permitted Values for VM

Expression | Definition
--- | ---
i (where 'i' represents an integer) | Exactly i occurrences, where i>=1. E.g., when i=1 there shall be one occurrence of the Content Item in this position.
i-j (where 'i' and 'j' represent integers) | From i to j occurrences, where i and j are >=1 and j>i.
i-n (where 'i' and 'n' represent integers) | i or more occurrences, where i>=1.

### 6.1.7 Requirement Type

The Requirement Type field specifies the requirements on the presence or absence of the Content Item or included Template.

### Note

There is typically no need to specify Requirement Type separately for SCU and SCP of the Basic SR SOP Classes, because the SCP is required to support the entire content of any SR Document it receives. Therefore, for Basic SR SOP Classes, Requirement Type effectively only applies to the SCU.

The following symbols are used:

M

  Mandatory. Shall be present.

MC

  Mandatory Conditional. Shall be present if the specified condition is satisfied.

U

  User Option. May or may not be present.

UC

  User Option Conditional. May not be present. May be present according to the specified condition.

### Note

There is an interaction between the VM and the Requirement Type with respect to the number of times that a content item (or included Template) may actually be present, as follows:

Req Type | VM | Actual number of occurrences in the content tree
--- | --- | ---
M | 1 | 1
M | 1-n | 1 to n
U | 1 | 0 or 1
U | 1-n | 0 to n

### 6.1.8 Condition

The Condition field specifies any conditions upon which presence or absence of the Content Item or its values depends. This field specifies any Concept Name(s) or Values upon which there are dependencies.

References in Condition statements to coded concepts or values, whether to select a content item to test or to specify a value to test against, are of the form (CV, CSD, "CM"). As is always the case for coded entries, the matching is performed against CV and CSD, irrespective of the string value of CM.

References may also be made to row numbers (e.g., to specify exclusive OR conditions that span multiple rows of a Template table).

The following abbreviations are used:

XOR

  Exclusive OR. One and only one row shall be selected from mutually-exclusive options.

  ### Note

  For example, if one of rows 1, 2, 3 or 4 may be included, then for row 2, the abbreviation "XOR rows 1, 3, 4" is specified for the condition.

IF

  Shall be present if the condition is TRUE; may be present otherwise.

IFF

  If and only if. Shall be present if the condition is TRUE; shall not be present otherwise.

### 6.1.9 Value Set Constraint

Value Set Constraints, if any, are specified in this field as defined or enumerated coded entries, or as baseline or defined context groups.

The Value Set Constraint column may specify a default value for the Content Item if the Content Item is not present, either as a fixed value, or by reference to another Content Item, or by reference to an Attribute from the data set other than within the Content Sequence (0040,A730).

#### 6.1.9.1 NUM Units Constraint

Constraints on units of measurement, if any, are specified in the Value Set Constraint field if and only if the Value Type is NUM. The constraints are specified either as defined or enumerated coded entries, or as baseline or defined context groups.

#### 6.1.9.2 CONTAINER Continuation Flag Constraint

The value of the Continuity of Content Flag (0040,A050) may be specified in the Value Set Constraint field if and only if the Value Type is CONTAINER.

### Note

The SR Document Content Module specifies "SEPARATE" and "CONTINUOUS" as the Enumerated Values for Continuity of Content Flag (0040,A050).

#### 6.1.9.3 SCOORD Graphic Type Constraint

Constraints on the value of the Graphic Type(0070,0023) may be specified in the Value Set Constraint field if and only if the Value Type is SCOORD. The constraint may specify a set of allowed values, or a set of disallowed values. For example:

- GRAPHIC TYPE = {POINT}

- GRAPHIC TYPE = {CIRCLE, ELLIPSE}

- GRAPHIC TYPE = not {MULTIPOINT}

## 6.2 Special Conventions for Template Tables

### 6.2.1 Multiple Value Sets Depending On Different Conditions

When a Content Item may have different value sets, each depending on different conditions, the description of each different case begins in a separate row of the Template Table.

### 6.2.2 Target Content Items of Relationships

When it is necessary to specify the Target Content Item(s) of a relationship, they are specified in the row(s) immediately following the Source Content Item. The Nesting level of a Target Content Item (or set of Target Content Items specified indirectly via an 'include Template' macro) is one greater than the Nesting Level of the corresponding Source Content Item, as indicated by an increase in the number of ">" characters in the nesting level.

When a Content Item may be the Source of multiple relationships having different Relationship Types and/or different Relationship Modes and/or different patterns of Target Content Item(s), the description of each different case begins in a separate row of the Template Table.

When the Source Content Item of a relationship has VM of greater than 1, the specified pattern of Target Content Items applies to all instantiations of the Source Content Item.

### Note

For example, if a Template specifies that the VM of a Source Content Item is 1-n and specifies a By-value relationship to two CODE Content Items with particular value set constraints, then each instantiation of the Source Content Item has a By-value relationship to two CODE Content Items with the specified value constraints.

When a Source Content Item that has a Requirement Type of U, UC or MC is not present (is not instantiated), no Target Content Items of that Source Content Item are present, even if one or more of the Target Content Items is designated with a Requirement Type of M or MC.

### Note

In other words, potential children are not present when there is no parent.

### 6.2.3 Inclusion of Templates

A Template may specify another Template to be included by specifying "INCLUDE" in the Value Type field and the identifier of the included Template in the Concept Name field. All of the rows of the specified Template are in included in the invoking Template, effectively substituting the specified template for the row where the inclusion is invoked. Whether or not the inclusion is user optional, mandatory or conditional is specified in the Requirement and Condition fields. The number of times the included Template may be repeated is specified in the VM field.

#### 6.2.3.1 Template Parameters

A Template that is included by another Template may include parameters that are replaced by values defined in the invoking Template. Parameters may be used to specify coded concepts or Context Groups in the Concept Name, Condition, or Value Set Constraint fields of a Template.

An included Template that accepts parameters shall be introduced by a table listing those parameters of the form:

Parameter Name | Parameter Usage
--- | ---
... | ...
... | ...

Parameters are indicated by a name beginning with the character "$".

The invoking Template may specify the value of the parameters in the included Template by name in the Value Set Constraint field of the INCLUDE row. The parameter in the included Template shall be replaced by the specified parameter value. Specification of a parameter value shall be of one of the following forms:

Notation | Definition
--- | ---
$parametername = EV or DT (CV, CSD, "CM") | The parameter passed to the template is the specified coded term.
$parametername = (CV, CSD, "CM") | The parameter passed to the template is the specified coded term, used as a parameter in a Condition field of the included Template.
$parametername = BCID or DCID (CID) CNAME | The parameter passed to the template is the Context Group.
$parametername = MemberOf {BCID or DCID (CID) CNAME} | The parameter passed to the template is a single coded term from the Context Group in curly braces.

The specification of a parameter value is valid only for the directly included template. Therefore, it needs to be explicitly respecified in templates intermediate between the originally specifying Template and the target Template. The intermediate Template may use the same parameter name as used by the Template it invokes; in such a case, the intermediate Template would invoke the subsidiary Template with a specification in the Value Set Constraint field such as:

$parametername = $parametername

### Note

In this case, the left hand instance of $parametername is the name in the subsidiary template, and the right hand instance is the (parametrized) value passed into the current template.

The invoking template is not required to specify all parameters of included templates. If not specified, the value set (term or context group) for that parameter is unconstrained. An unconstrained value in a Condition will cause the condition to fail.

### 6.2.4 Post-coordinated Codes and Has Concept Modifier Relationship

Though it may not be explicitly shown in a particular Template, the use of any coded Concept Name in any Content Item may be defined in a post-coordinated rather than pre-coordinated manner, unless explicitly forbidden by the IOD or the Template.

Accordingly, any such Content Item may have any number of Target Content Items via a "HAS CONCEPT MOD" relationship, even if not explicitly specified in a Template. Each Target Content Item of such a relationship may be more complicated than a single Content Item if the IOD permits (i.e., the post-coordinated concept may potentially be defined by a complex sub-tree).

### 6.2.5 Extension of Templates

An Extensible Template may be extended in an Application generating SOP Instances to include additional Content Items in its definition. Such Content Items shall not duplicate concepts for which an encoding is defined in the Template. I.e., if a method is provided for the encoding of a concept in the Template, that concept shall not be encoded using a different Content Item in an extension to the Template.

### Note

There is no requirement that the included additional Content Items in a Template extension be placed at the end of the Template. The additional Content Items may be included at any semantically appropriate location in the Template, regardless of whether the order of Content Items in the Template is significant.

A Non-extensible Template shall not be modified in an Application by the addition of Content Items to its definition.

### Note

The set of Content Items in either an Extensible or a Non-extensible Template may be changed in subsequent editions of the Standard, in accordance with the procedures of the DICOM Standards Committee.

A Non-Extensible Template may include a Template that is Extensible. In invoking such a Template, the content structure of SOP Instances created from the Non-Extensible Template may vary according to the varying content structure allowed by the extension of the included Template.

### Note

Specification of such extensible content in a Non-Extensible Template may be desirable if the Template defines, e.g., a fixed top level structure into which a variety of lower level structures may be "plugged".

# 7 DCMR Context Group Specifications

Context Groups specify Value Set restrictions for Code Value (0008,0100) and Code Meaning (0008,0104) of Code Sequence Attributes for given functional or operational contexts. This Section specifies the semantics of DCMR Context Group Tables.

## 7.1 Context Group Table Field Definition

Context Groups are described using tables of the following form (optional columns are shown with italic column titles):

Type:

  (Non-) Extensible

Version:

  <yyyymmdd>

Table CID <#>. <Context Group Name>

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning | <Reference Terminology> Equivalent Value
--- | --- | --- | --- | ---
... | ... | ... | ... | ...
... | ... | ... | ... | ...

A row of a Context Group table specifies one coded concept. Each Context Group table is named by a title and identified by a CID number and version.

The columns of the tables consist of:

- Coding Scheme Designator (0008,0102)

- Code Value (0008,0100)

- Code Meaning (0008,0104)

In those cases where it is necessary, Coding Scheme Version (0008,0103) may also be specified. This column may be absent if Coding Scheme Version is not required for any of the coded concepts in the Context Group.

The value specified in the Code Meaning field is an acceptable value for the specified code value, but does not preclude the use of other synonymous text in the same or other language.

### Note

Some coding schemes such as LOINC do not specify the equivalent of a Code Meaning.

If further description of the concept represented by the code is required in the DCMR (rather than referring to an external coding scheme), it is included in a separate table.

An optional column may provide an informative mapping from the coded concepts of the Context Group to a reference terminology specified in the column heading.

A Context Group may alternatively be defined by narrative reference to an externally defined coding scheme.

### Note

See for instance CID 82 “Units of Measurement”.

## 7.2 Special Conventions for Context Group Tables

### 7.2.1 Include Context Group

The 'Include Context Group' macro is a concise mechanism for including (by-reference) all of the rows of a specified Context Group in the invoking Context Group, effectively substituting the specified Context Group for the row where the macro is invoked. If an 'Include Context Group' is specified, it shall be specified in the Concept Name column of a Context Group Table. Table 7.2.1-1 specifies the syntax of the 'Include Context Group macro. Inclusion may be nested, in that included Context Groups may themselves include other Context Groups. This gives rise to the possibility of circular inclusion and multiple inclusion, in which case the Context Group shall consist of the transitive closure of the set of all coded concepts within the included Context Groups.

### Note

For example, it is reasonable to have the following definitions for context groups:

- Context ID 1, includes Context IDs 2 and 3

- Context ID 2, includes Context IDs 4 and 5

- Context ID 3, includes Context IDs 5 and 6

- Context ID 4 contains a, b, c

- Context ID 5 contains e, f, g

- Context ID 6 contains a, h, i

The contents of Context ID 1 will be a, b, c, e, f, g, h, i.

Table 7.2.1-1. Include Context Group Macro

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
… | … | …
Include CID nnn
… | … | …

### 7.2.2 Units of Measurement

Context Group 82 is defined to include all units of measurement relevant to DICOM IODs. In the past it was envisaged that an extensible list of pre-coordinated codes would be included in the mapping resource.

DICOM has now adopted the Unified Codes for Units of Measurement (UCUM) standard for all units of measurement. This coding scheme allows for the "construction" of pre-coordinated codes from atomic components.

The specialization of the UCUM standard as it is used in DICOM involves the following rules:

- the Coding Scheme Designator is specified as "UCUM"

- the version of UCUM from which a code is constructed is not required, as it is not needed to resolve ambiguity in the Code Value or Code Meaning; however, there is no restriction on the version being specified in Coding Scheme Version

- the Code Value will be constructed from UCUM and make use of the "case-sensitive" form of UCUM code (e.g., "ml/s")

- the Code Meaning for other than UCUM unity may be one of the following:

  - the "print" value specified in UCUM (e.g., "mmHg" for Code Value mm[Hg])

  - the same string as sent in the Code Value (e.g., "ml/s")

  - constructed from the "names" of individual components using the Americanized form of name (e.g., "milliliters/second")

  - constructed from the "names" of individual components using the European form of name (e.g., " millilitres /second")

- In the case of UCUM unity ("1", or curly braces expression) it is forbidden to use "1" as a Code Meaning. Annex G provides Code Meanings for a Code Value (0008,0100) of 1. A Template or Context Group may constrain the Code Meaning according to the following rules:

  - UCUM default unit 1 shall use one of the Code Meaning synonyms specified in Annex G

  - ratios of identically dimensioned values may use ({ratio}, UCUM, "ratio")

  - unitless numeric scores may use ({M:N}, UCUM, "range: M:N") to specify the minimum and maximum value, for example, ({0:10}, UCUM, "range: 0:10")

  - counts using UCUM annotation shall always use the text within the curly braces as the Code Meaning, for example, ({masses}, UCUM, "masses")

  - compositions of a curly braces expression with other UCUM values may use a conventional clinical representation, for example, ({H.B.}/min, UCUM, "BPM")

The UCUM standard states that the preferred display values for codes deg (degrees of plane angle) and Cel (degrees Celsius) are "°" and "°C". However, the character ° does not have a representation in the DICOM default character set (ASCII, ISO-IR 6). The Code Meaning specified in this Part therefore uses "deg" and "C". SOP Instances that specify a Specific Character Set that allows the character ° may use Code Meanings "°" and "°C".

### Note

1. Code Meaning "C" formally conflicts with the Code Meaning for Coulomb. In the context of DICOM use, the possibility of confusion to a user based on the display of the Code Meaning is considered remote, as there is little use of Coulomb in imaging, and the context of the displayed item Concept Name would resolve between temperature and electric charge. Automated processing based on the Code Values should not face an issue as the Code Values differ.

2. The character ° has Unicode code point U+00B0, and is represented as 0xB0 in ISO-IR 100 (Latin-1), ISO-IR 101 (Latin-2), ISO-IR 109 (Latin-3), ISO-IR 110 (Latin-4), ISO-IR 126 (Greek), ISO-IR 138 (Hebrew), and ISO-IR 148 (Latin-5). It is not encodable in ISO-IR 13 (Katakana), ISO-IR 144 (Cyrillic), ISO-IR 127 (Arabic), or ISO-IR 166 (Thai).

### 7.2.3 Extension of Context Groups

An Application may extend an Extensible Context Group by adding terms for new concepts. Applications may not substitute other terms of the same concept in the Context Group. Applications may not add a term that means "unspecified" or "missing" or "unknown" similar; if such a concept is intended to be permitted then the Standard will include it in the Context Group already. Such extension may be made without a change in Context Group Identifier, but with the specification of Context Group Extensions (see PS3.3).

Non-extensible Context Groups shall not be modified in an Application.

### Note

The set of concepts in either an Extensible or a Non-extensible Context Group may be changed in subsequent editions of the Standard, in accordance with the procedures of the DICOM Standards Committee.

# 8 Coding Schemes

Table 8-1 lists the coding schemes (and their designators) defined for use in DICOM. Additionally, any coding scheme may be used that has an entry in the HL7 Registry of Coding Schemes (HL7 v2 Table 0396, or the equivalent online registry), in which case the HL7 Symbolic Name shall be used as the value for the Coding Scheme Designator in DICOM, as long as it does not conflict with an entry Table 8-1 and fits within the Value Representation of the DICOM Coding Scheme Designator (0008,0102) attribute. As specified in the HL7 v2 Table 0396, local or private coding schemes shall be identified by an alphanumeric identifier beginning with the characters "99".

### Note

1. An earlier version of this table was formerly contained in Annex D of PS3.3.

2. See Section 8.2 “Coding Scheme Designator and Coding Scheme Version” in PS3.3 for further description.

3. The Coding Scheme UIDs are provided for reference only; the normative specification of UIDs and their associated meaning is the responsibility of the coding scheme developer and/or HL7.

4. The HL7 registration of Coding Schemes is available at http://www.hl7.org/oid/index.cfm.

5. Publication of codes or references to coding schemes within DICOM does not constitute a grant of intellectual property rights to implementers. Use of some Coding Schemes may require a license, or purchase of the relevant coding scheme publication. Implementers should consult the relevant coding scheme publisher; see also Section 2.

Table 8-1. Coding Schemes

Coding Scheme Designator | Coding Scheme UID | Description
--- | --- | ---
ACR | 2.16.840.1.113883.6.76 | ACR Index for Radiological Diagnosis Revised, 3rd Edition 1986
ASTM-sigpurpose | 1.2.840.10065.1.12 | ASTM E 2084 Signature Purpose codes (see Annex A1 of ASTM E 2084), ASTM Subcommittee E 31.20 Data and System Security for Health Information
BARI |  | Bypass Angiography Revascularization Investigation, Alderman, EL and Stadius, M, Coronary Artery Disease 1992, 3:1189-1207; endorsed by ACC/AHA Guidelines for Coronary Angiography, J Am Coll Cardiol 1999, 33:1791
BI |  | ACR Breast Imaging Reporting and Data System (BI-RADS®), Coding Scheme Version (0008,0103) is required; code values are section and paragraph identifiers within the publication where the code meaning is defined (e.g., "I.D.1", where I = Breast Imaging Lexicon, D = Special Cases, 1 = Tubular Density, as the code value for "Tubular Density"). ### Note In the HL7 registry, the abbreviation BI is assigned to a different coding scheme, specifically the Beth Israel problem list.
C4 | 2.16.840.1.113883.6.12 | American Medical Association's Current Procedure Terminology 4 (CPT-4)
C5 | 2.16.840.1.113883.6.82 | American Medical Association's Current Procedure Terminology 5 (CPT-5)
CD2 | 2.16.840.1.113883.6.13 | American Dental Association's (ADA) Current Dental Terminology 2 (CDT-2)
DCM | 1.2.840.10008.2.16.4 | DICOM Controlled Terminology; PS3.16 Content Mapping Resource, Annex D (Note that HL7 also specifies an OID of 2.16.840.1.113883.6.31, but deprecates it in favor of 1.2.840.10008.2.16.4).
DCMUID | 1.2.840.10008.2.6.1 | DICOM UID Registry
FMA | 2.16.840.1.113883.6.119 | Digital Anatomist Foundational Model of Anatomy
HPC | 2.16.840.1.113883.6.14 | Healthcare Financing Administration (HCFA) Common Procedure Coding System (HCPCS)
I10 | 2.16.840.1.113883.6.3 | International Classification of Diseases revision 10 (ICD-10)
I10P | 2.16.840.1.113883.6.4 | ICD-10 Procedure Coding System (ICD 10 PCS)
I9 | 2.16.840.1.113883.6.42 | International Classification of Diseases revision 9 (ICD-9)
I9C | 2.16.840.1.113883.6.2 | International Classification of Diseases revision 9, with Clinical Modifications (ICD-9-CM)
ISO639_1 | 2.16.840.1.113883.6.99 | Two-letter language codes ### Note HL7 uses "ISO639-1" for the symbolic name, with a hyphen rather than an underscore
ISO639_2 | 2.16.840.1.113883.6.100 | Three-letter language codes ### Note HL7 uses "ISO639-2" for the symbolic name, with a hyphen rather than an underscore
ISO3166_1 | 2.16.1 | ISO 3166-1 alpha-2 Country Codes ### Note HL7 uses "ISO3166-1" for the symbolic name, with a hyphen rather than an underscore
ISO5218_1 |  | Representation of Human Sexes (not used - see note)
ISO_OID |  | ISO/IEC 8824-1- Information Technology - Abstract Syntax 1 (ASN.1): Specification of Basic Notation, and ISO/IEC 9834-1 - Information technology - Open Systems Interconnection - Procedures for the operation of OSI Registration Authorities: General procedures and top arcs of the ASN.1 Object Identifier tree
LN | 2.16.840.1.113883.6.1 | Logical Observation Identifier Names and Codes, Regenstrief Institute
MDC | 2.16.840.1.113883.6.24 | ISO/IEEE 11073 Medical Device Nomenclature, including all its subsections (-10101, -10102, etc.), encoded as decimal strings <partition>:<element>
MDNS |  | Universal Medical Device (UMD) Nomenclature System
MSH | 2.16.840.1.113883.6.177 | US National Library of Medicine (NLM) Medical Subject Headings (MeSH) See http://www.nlm.nih.gov/mesh/meshhome.html.
NBD | 2.16.840.1.113883.15.2 | NASPE/BPEG Defibrillator Code Bernstein AD, et al."The NASPE/BPEG Defibrillator Code" PACE, 16:1776-1780, 1993
NBG | 2.16.840.1.113883.15.3 | NASPE/BPEG Generic Pacemaker Code (2000) Bernstein AD, et al."The Revised NASPE/BPEG Generic Code for antibradycardia, adaptive-rate, and multisite pacing" PACE, 25:260-264, 2000
NCDR |  | American College of Cardiology National Cardiovascular Data Registry™ Cath Lab Module Version 1.1, 1997; Version 2.0b, 1999
NCIt | 2.16.840.1.113883.3.26.1.1 | NCI Thesaurus. See http://ncit.nci.nih.gov/
NICIP | 2.16.840.1.113883.2.1.3.2.4.21 | UK National Health Service National Interim Clinical Imaging Procedures (NICIP) Short Code (e.g., "CCHAPC" for CT Thorax abdomen pelvis with contrast)
NPI |  | HCFA National Provider Identifier
POS | 2.16.840.1.113883.6.50 | HCFA Place of Service (POS) Codes for Professional Claims
RADLEX | 2.16.840.1.113883.6.256 | RadLex
RFC3066 | 2.16.840.1.113883.6.121 | RFC 3066, Tags for the Identification of Languages, Internet Engineering Task Force ### Note HL7 uses "IETF3066" for the symbolic name.
99SDM | 2.16.840.1.113883.6.53 | SNOMED DICOM Microglossary (Retired) (see Section 8.1)
SCPECG |  | Standard Communications Protocol for Computer-Assisted Electrocardiography, Draft proposal for ISO Standard, AAMI, Revision 1.3
SNM3 | 2.16.840.1.113883.6.51 | SNOMED International Version 3 (see Section 8.1)
SRT | 2.16.840.1.113883.6.96 | SNOMED-CT, using the "SNOMED-RT style" code values (see Section 8.1) ### Note HL7 uses "SNM" for the symbolic name.
UCUM | 2.16.840.1.113883.6.8 | Unified Code for Units of Measure
UMLS | 2.16.840.1.113883.6.86 | UMLS codes as CUIs making up the values in a coding system
UPC | 2.16.840.1.113883.6.55 | Universal Product Code - Universal Code Council

### Note

Coding Scheme Designator ISO5218_1 was improperly specified in earlier editions of the standard. The codes attributed to that coding scheme have been added to the DICOM Controlled Terminology, and thus all references to coding scheme ISO5218_1 should be considered equivalent to coding scheme DCM.

## 8.1 SNOMED

SNOMED (the Systematized Nomenclature of Medicine) is the preferred coding system within DICOM for anatomy, clinical findings, procedures, pharmaceutical/biologic products (including contrast agents), and other clinical terms.

### Note

SNOMED is a registered trademark of the International Health Terminology Standards Development Organisation .

SNOMED has had various versions, including SNOMED International (Version 3), which was issued in 1993 and revised through 1998, SNOMED Reference Terminology, the successor to SNOMED 3 that was published between 1999 and 2001, and SNOMED Clinical Terms, which has been the name since 2002. The coding scheme is fully backward-compatible across SNOMED 3, SNOMED-RT, and SNOMED CT. SNOMED CT introduced a solely numeric set of codes (ConceptID) in addition to the former alphanumeric codes (SnomedID), but all SNOMED terminology concepts have both a numeric and an alphanumeric code.

In previous editions of the DICOM Standard, the following Coding Scheme Designators were used for SNOMED codes in DICOM:

- "99SDM", denoting the provisional SNOMED DICOM Microglossary

- "SNM3", denoting SNOMED International (Version 3)

- "SRT", originally denoting SNOMED-RT

All uses of SNOMED coded terms in DICOM are now indicated by the Coding Scheme Designator "SRT", identifying them as SNOMED CT terms using the "SNOMED-RT style" alphanumeric code values, with some exceptions:

- The “Nuclear Medicine Image IOD” in PS3.3 and “Positron Emission Tomography Image IOD” in PS3.3 in some code sequences require the Coding Scheme Designator "99SDM" as an Enumerated Value (see PS3.3).

- The Mammography View Codes of CID 4014 “View for Mammography” and CID 4015 “View Modifier for Mammography” may use the Coding Scheme Designator "SNM3" for implementation adherence to regulatory approvals.

Consequently, when a Coding Scheme Designator of "99SDM" or "SNM3" is encountered, it shall be treated as equivalent to "SRT" for the purpose of interpreting the Code Value.

### Note

"SRT" as a coding scheme designator is used only in the DICOM Standard. HL7v2 did not standardize a coding scheme designator for SNOMED-RT.

When interoperating with systems that use SNOMED CT codes obtained from a source other than the DICOM Standard, Application Entities may receive Code Sequences with a Coding Scheme Designator of "SNOMED-CT" and a numeric ConceptID code. It is the responsibility of such Application Entities to convert any such codes to the alphanumeric SnomedID with Coding Scheme Designator "SRT" for use in DICOM objects and services.

### 8.1.1 Use of SNOMED Anatomic Concepts

In general, DICOM uses the anatomic concepts with the term "structure", rather than with the term "entire". This is an important distinction in SNOMED. "Entire" is a child concept to "structure", has a more restricted meaning, and typically is used in conjunction with treatments (e.g., "excision of entireright kidney"). It is used in distinction to other sibling children of the parent concept that may identify parts of the parent anatomic feature. Since imaging typically targets both the anatomic feature and the area around it, or sometimes just part of the anatomic feature, DICOM usually uses "structure" concepts that are more inclusive than the "entire" concepts.

## 8.2 ISO_OID

ISO 8824 and ISO 9834-1 are the standards defined for the generation of object identifiers that are used as DICOM Unique Identifiers (see PS3.5), can also serve as a general mechanism for identifying organizations and objects defined by those organizations.

When the Coding Scheme Designator is ISO_OID, the Code Value shall be the numeric (dot delimited) form of a valid object identifier.

A repository of known existing object identifiers can be found at http://www.oid-info.com/index.htm. For example:

- the ISO 9834-1 assigned numeric object identifier for the country France, is "1.0.3166.2.2.1.250" (since ISO 3166 defines a means for maintaining country codes using object identifiers)

- the object identifier for the RIPEMD-160 cryptographic hash function is "1.0.10118.3.0.49"

- the object identifier for the HL7 V2 table of codes for marital status is "2.16.840.1.113883.12.2"

The re-use of object identifiers for existing concepts that do not have an alternative more appropriate coding scheme compatible with DICOM provides a mechanism to avoid defining new codes. For example, HL7 assigned object identifiers can be found at http://www.hl7.org/oid/index.cfm.

Though the intent of ISO_OID is to define organizational roots for the hierarchical assignment of object identifiers, and not specifically to identify organizations per se, the organizational root values can be construed as identifying the organization. For example, the DICOM Standards Organization itself can be identified by the value "1.2.840.10008". See also CID 5002 “Organizations”.

# A Structured Reporting Templates (Normative)

This Annex specifies the content of Standard Templates that may be used by DICOM SR IODs.

## General Templates

### TID 300 Measurement

This Template provides a general structure for a numeric measurement, together with evaluations of its normality and/or significance, and the inference source(s) for its value. This structure is instantiated by inclusion of this Template with specific contextual parameters from a parent Template.

Table TID 300. Parameters

Parameter Name | Parameter Usage
--- | ---
$Measurement | Coded term or Context Group for Concept Name of measurement
$Units | Units of Measurement
$ModType | Modifier Name for Concept Name of measurement
$ModValue | Modifier Value for Concept Name of measurement
$Method | Value for Measurement Method
$Derivation | Value for Measurement Derivation
$TargetSite | Value for Anatomic Location of measurement
$TargetSiteMod | Modifier Value for Anatomic Location of measurement
$Equation | Coded term or Context Group for the equation or table from which the measurement was derived or computed
$ImagePurpose | Purpose of Reference for an image used as a source of the measurement
$WavePurpose | Purpose of Reference for a waveform used as a source of the measurement
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$DerivationParameter | Coded term or Context Group for Concept Name of a derivation parameter
$DerivationParameterUnits | Units of derivation parameter

Type:

  Extensible

Order:

  Significant

Table TID 300. Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | $Measurement | 1 | M |  | UNITS = $Units
2 | > | HAS CONCEPT MOD | CODE | $ModType | 1-n | U |  | $ModValue
3 | > | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | $Method
4 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | U |  | $Derivation
5 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | U |  | $TargetSite
6 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
7 | >> | HAS CONCEPT MOD | CODE | DT (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | $TargetSiteMod
8 | > | HAS PROPERTIES | INCLUDE | DTID 310 “Measurement Properties” | 1 | U |  | $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority
9 | > | INFERRED FROM | NUM | $DerivationParameter | 1-n | UC | XOR Row 10 | UNITS = $DerivationParameterUnits
10 | > | R-INFERRED FROM | NUM | $DerivationParameter | 1-n | UC | XOR Row 9 | UNITS = $DerivationParameterUnits
11 | > | INFERRED FROM | INCLUDE | DTID 315 “Equation or Table” | 1 | UC | XOR Row 12 | $Equation = $Equation
12 | > | INFERRED FROM | TEXT | DCID 228 “Equation or Table” | 1 | UC | XOR Row 11 |
13 | > |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1-n | U |  | $Purpose = $ImagePurpose
14 | > |  | INCLUDE | DTID 321 “Waveform or Temporal Coordinates” | 1-n | U |  | $Purpose = $WavePurpose
15 | > |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
16 | > | HAS CONCEPT MOD | TEXT | EV (121050, DCM, "Equivalent Meaning of Concept Name") | 1 | U |  |
17 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
18 | > | INFERRED FROM | COMPOSITE | EV (126100, DCM, "Real World Value Map used for measurement") | 1 | U |  | SOP Class UID shall be Real World Value Mapping Storage ("1.2.840.10008.5.1.4.1.1.67")

Content Item Descriptions

Rows 2, 3, 4, 5 | The HAS CONCEPT MOD items allow the explicit definition of terms for post-coordination of the measurement concept name. Additional post-coordinated modifier terms may be included in a SOP Instance based on this template, in accordance with section 6.2.4, or as defined by templates that invoke this template and explicitly define additional post-coordinated modifiers (e.g., TID 5203).
Rows 9, 10 | The INFERRED FROM items allow the specification (by-value or by-reference) of numeric values that were used in the derivation of the numeric measurement of Row 1. The nature of the inference is not explicitly conveyed; it may be implicit in the Concept Names of the measurements. Inference by-reference is valid only in SOP Classes that permit the INFERRED FROM relationship by-reference.
Row 16 | Equivalent Meaning of Concept Name allows the creating application to specify the preferred composed concept name representing the measurement and the associated post-coordinated concept modifiers. The concept modifiers may include those specified in this template, in a template that invokes this template, or at the option of the creating application in accordance with section 6.2.4. This composed concept name may be rendered by a display application.
Row 18 | Row 18 is a reference to an RWV that describes how measurements were made in units that differ from the stored pixel values in the images referenced in Row 13. E.g., for a PET SUVbw measurement, the mapping from activity/concentration units in the referenced image that was used (and which may be reused for measurements in the future) may be encoded in a referenced RWV instance. This reference overrides any reference in an including template (such as for a Measurement Group).

### TID 310 Measurement Properties

This Template provides the properties of a numeric measurement, including evaluations of its normality and/or significance, its relationship to a reference population, and an indication of its selection from a set of measurements.

Table TID 310. Parameters

Parameter Name | Parameter Usage
--- | ---
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement

Type:

  Extensible

Order:

  Significant

Table TID 310. Measurement Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121402, DCM, "Normality") | 1 | U |  | DCID 222 “Normality Codes”
2 |  |  | INCLUDE | DTID 311 “Measurement Statistical Properties” | 1 | U |  | $RefAuthority = $RefAuthority
3 |  |  | INCLUDE | DTID 312 “Normal Range Properties” | 1 | U |  | $RangeAuthority = $RangeAuthority
4 |  |  | CODE | EV (121403, DCM, "Level of Significance") | 1 | U |  | DCID 220 “Level of Significance”
5 |  |  | NUM | DCID 225 “Measurement Uncertainty Concepts” | 1-n | U |  |
6 |  |  | CODE | EV (121404, DCM, "Selection Status") | 1 | U |  | DCID 224 “Selection Method”

### TID 311 Measurement Statistical Properties

This Template provides the statistical properties of a reference population for a numeric measurement, and/or the position of a measurement in such a reference population.

Table TID 311. Parameters

Parameter Name | Parameter Usage
--- | ---
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population

Type:

  Extensible

Order:

  Significant

Table TID 311. Measurement Statistical Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 221 “Measurement Range Concepts” | 1-n | M |  |
2 |  |  | TEXT | EV (121405, DCM, "Population description") | 1 | U |  |
3 |  |  | TEXT | EV (121406, DCM, "Reference Authority") | 1 | UC | XOR row 4 |
4 |  |  | CODE | EV (121406, DCM, "Reference Authority") | 1 | UC | XOR row 3 | $RefAuthority

### TID 312 Normal Range Properties

This Template provides the normal range of values for a numeric measurement.

Table TID 312. Parameters

Parameter Name | Parameter Usage
--- | ---
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement

Type:

  Extensible

Order:

  Significant

Table TID 312. Normal Range Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 223 “Normal Range Values” | 1-n | M |  |
2 |  |  | TEXT | EV (121407, DCM, "Normal Range description") | 1 | U |  |
3 |  |  | TEXT | EV (121408, DCM, "Normal Range Authority") | 1 | UC | XOR row 4 |
4 |  |  | CODE | EV (121408, DCM, "Normal Range Authority") | 1 | UC | XOR row 3 | $RangeAuthority

### TID 315 Equation or Table

Table TID 315. Parameters

Parameter Name | Parameter Usage
--- | ---
$Equation | Coded term or Context Group for the equation or table from which a measurement was derived or computed

Type:

  Extensible

Order:

  Significant

Table TID 315. Equation or Table

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DCID 228 “Equation or Table” | 1 | M |  | $Equation
2 | > | HAS PROPERTIES | NUM |  | 1-n | U |  |
3 | > | R-HAS PROPERTIES | NUM |  | 1-n | U |  |

Content Item Descriptions

Row 2 | The HAS PROPERTIES allows the specification of the numeric values used as input to the equation or table identified in Row 1.
Row 3 | The HAS PROPERTIES allows the specification by-reference of the numeric values used as input to the equation or table. This row is valid only in SOP Classes that permit the HAS PROPERTIES relationship by-reference.

### Note

For example, if Row 1 identifies a specific Body Surface Area equation, Rows 2 and 3 can be used to convey (by-value or by-reference) the Patient Height and Patient Weight numeric measurements used in the BSA computation.

### TID 320 Image or Spatial Coordinates

This Template provides a general structure for inference from an image, either as a whole, or with specific spatial coordinates, as a single included Template in the invoking Template. If allowed by the IOD, the Image Content Item may be included by-reference.

Table TID 320. Parameters

Parameter Name | Parameter Usage
--- | ---
$Purpose | Purpose of Reference for an image used as a source of the measurement

Type:

  Extensible

Order:

  Significant

Table TID 320. Image or Spatial Coordinates

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | INFERRED FROM | IMAGE | $Purpose | 1 | MC | XOR Rows 2, 3 |
2 |  | R-INFERRED FROM | IMAGE |  | 1 | MC | XOR Rows 1, 3 |
3 |  | INFERRED FROM | SCOORD | $Purpose | 1 | MC | XOR Rows 1, 2 |
4 | > | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 5 |
5 | > | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 4 |

### TID 321 Waveform or Temporal Coordinates

This Template provides a general structure for referencing a waveform, either as a whole, or with specific temporal coordinates, as a single included Template in the invoking Template. If allowed by the IOD, the Waveform Content Item may be included by-reference.

Table TID 321. Parameters

Parameter Name | Parameter Usage
--- | ---
$Purpose | Purpose of Reference for a waveform used as a source of the measurement

Type:

  Extensible

Order:

  Significant

Table TID 321. Waveform or Temporal Coordinates

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | INFERRED FROM | WAVEFORM | $Purpose | 1 | MC | XOR Rows 2, 3 |
2 |  | R-INFERRED FROM | WAVEFORM |  | 1 | MC | XOR Rows 1, 3 |
3 |  | INFERRED FROM | TCOORD | $Purpose | 1 | MC | XOR Rows 1, 2 |
4 | > | SELECTED FROM | WAVEFORM |  | 1 | MC | XOR Row 5 |
5 | > | R-SELECTED FROM | WAVEFORM |  | 1 | MC | XOR Row 4 |

### TID 350 References to Supporting Evidence

This template provides references to supporting evidence in the form of DICOM composite objects. This includes references to images, spatial coordinates on images, and other composite objects, such as Structured Reports.

Type:

  Non-Extensible

Order:

  Significant

Table TID 350. References to Supporting Evidence

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | IMAGE | BCID 7003 “Diagnostic Imaging Report Purposes of Reference” | 1-n | U |  |
2 |  |  | SCOORD | BCID 7003 “Diagnostic Imaging Report Purposes of Reference” | 1-n | U |  |
3 | > | SELECTED FROM | IMAGE |  | 1 | M |  |
4 |  |  | COMPOSITE | DT (122073, DCM, "Current procedure evidence") | 1-n | U |  |
5 | > | HAS CONCEPT MOD | CODE | EV (121144, DCM, "Document Title") | 1 | U |  |

### TID 351 Previous Reports

This general template provides a means to reference previous structured reporting composite object instances.

Type:

  Extensible

Order:

  Significant

Table TID 351. Previous Reports

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111549, DCM, "Previous Reports") | 1 | M |  |
2 | > | CONTAINS | COMPOSITE |  | 1-n | M |  |

Content Item Descriptions

Row 2 | Concept Name may be the Root Concept Name (title) of a Structured Report composite object instance.

### TID 1000 Quotation

Unless otherwise specified, content in an SR tree is "directly" observed. When material is quoted (from a source that is either a document or something spoken), then it is necessary to specify:

- the fact that one is quoting

- who is doing the quoting

- the source of the quote

- who is being quoted, and who and what the quote is about

This template establishes a mechanism for quoting by specifying:

- the fact that one is quoting, by the presence of the contents of the template in the tree

- that the "observer context" above the invocation of this template establishes who is doing the quoting

- the source of the quote, by the values of the content items in this template

- who is being quoted, and who and what the quote is about, by the observation context that is established at the start of the quoted material

This template may be invoked recursively, to nest quotes within quotes. In essence, the chain of who is quoting whom can be established by maintaining a "stack" of observer context.

If a dimension of observation context is the same in the quoted material as in the enclosing tree, then the observation context does not need to be respecified (e.g., the quote may be about the same subject or procedure). Typically, the observer context would change (unless one were quoting oneself).

In the case of quoting something that was spoken, the "observer" is the person speaking.

TID 1000 is attached using HAS OBS CONTEXT relationships to the top node of the material that is being quoted. The presence of the Quoted Source concept signals the fact that the material is quoted rather than directly observed.

Type:

  Extensible

Order:

  Significant

Table TID 1000. Quotation

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS OBS CONTEXT | CODE | EV (121001, DCM, "Quotation Mode") | 1 | M |  | EV (121003, DCM, "Document") EV (121004, DCM, "Verbal")
2 |  | HAS OBS CONTEXT | COMPOSITE | EV (121002, DCM, "Quoted Source") | 1 | MC | Required if quoted material source is a DICOM composite object |
3 |  | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |

### TID 1001 Observation Context

Specifies attributes of observation context that may be defined, extended or replaced at any location in the SR tree.

This includes attributes that specify:

- who or what the observation is about ("subject context")

- what procedure the observation is about ("procedure context")

- who or what is making the observation ("observer context")

Establishing context includes two aspects of each dimension: identification and description (e.g., patient name and ID vs. patient's age, height or weight).

Whenever one dimension of context is changed or an attribute is added, all attributes of that dimension of context are "flushed", that is they need to be repeated in their entirety. For example, when the subject is changed from patient (name, id) to fetus (number), then the parameters of the patient are discarded. E.g., the patient's ID does not apply to the fetus.

"Extending" the same class and dimension of observation context isn't feasible, since one cannot "null out" or remove a previously set attribute. Any time a dimension of observation context is "replaced", any attributes that are unspecified remain unspecified (i.e., they are not inherited).

Type:

  Non-Extensible

Order:

  Significant

Table TID 1001. Observation Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | MC | Required if all aspects of observer context are not inherited. | Defaults to the attributes of the Author Observer Sequence (0040,A078), or the Verifying Observer Sequence (0040,A073) if the Author Observer Sequence is not present
2 |  | HAS OBS CONTEXT | INCLUDE | DTID 1005 “Procedure Context” | 1 | MC | Required if all aspects of procedure context are not inherited. |
3 |  | HAS OBS CONTEXT | INCLUDE | DTID 1006 “Subject Context” | 1 | MC | Required if all aspects of observation subject context are not inherited. |

### TID 1002 Observer Context

The observer (person or device) that created the Content Items to which this context applies.

Whenever this template is invoked, all previously inherited attributes of Observer Context are discarded and replaced.

There may be more than one observer, as this template may be invoked with a VM 1-n, and both person and device observers. In such a case, the Content Items of TID 1003 “Person Observer Identifying Attributes” and TID 1004 “Device Observer Identifying Attributes” shall be included in the order in which the values of Observer Type are specified. Since TID 1003 “Person Observer Identifying Attributes” and TID 1004 “Device Observer Identifying Attributes” both include a single mandatory Content Item as their first Content Item, which observer is being described can be determined

Type:

  Non-Extensible

Order:

  Significant

Table TID 1002. Observer Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS OBS CONTEXT | CODE | EV (121005, DCM, "Observer Type") | 1 | MC | IF Observer type is device | DCID 270 “Observer Type” Defaults to (121006, DCM, "Person")
2 |  | HAS OBS CONTEXT | INCLUDE | DTID 1003 “Person Observer Identifying Attributes” | 1 | MC | IFF Row 1 value = (121006, DCM, "Person") or Row 1 is absent |
3 |  | HAS OBS CONTEXT | INCLUDE | DTID 1004 “Device Observer Identifying Attributes” | 1 | MC | IFF Row 1 value = (121007, DCM, "Device") |

### TID 1003 Person Observer Identifying Attributes

This template contains identifying (and optionally descriptive) attributes of persons that are observers.

Type:

  Extensible

Order:

  Significant

Table TID 1003. Person Observer Identifying Attributes

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | PNAME | EV (121008, DCM, "Person Observer Name") | 1 | M |  |
2 |  |  | TEXT | EV (121009, DCM, "Person Observer's Organization Name") | 1 | U |  | Defaults to Institution Name (0008,0080) of the General Equipment Module
3 |  |  | CODE | EV (121010, DCM, "Person Observer's Role in the Organization") | 1 | U |  | BCID 7452 “Organizational Roles”
4 |  |  | CODE | EV (121011, DCM, "Person Observer's Role in this Procedure") | 1 | U |  | BCID 7453 “Performing Roles”

### TID 1004 Device Observer Identifying Attributes

This template (derived from the DICOM General Equipment Module of PS3.3) contains identifying (and optionally descriptive) attributes of devices that are observers.

Type:

  Extensible

Order:

  Significant

Table TID 1004. Device Observer Identifying Attributes

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | UIDREF | EV (121012, DCM, "Device Observer UID") | 1 | M |  |
2 |  |  | TEXT | EV (121013, DCM, "Device Observer Name") | 1 | U |  | Defaults to value of Station Name (0008,1010) in General Equipment Module
3 |  |  | TEXT | EV (121014, DCM, "Device Observer Manufacturer") | 1 | U |  | Defaults to value of Manufacturer (0008,0070) in General Equipment Module
4 |  |  | TEXT | EV (121015, DCM, "Device Observer Model Name") | 1 | U |  | Defaults to value of Manufacturer's Model Name (0008,1090) in General Equipment Module
5 |  |  | TEXT | EV (121016, DCM, "Device Observer Serial Number") | 1 | U |  | Defaults to value of Device Serial Number (0018,1000) in General Equipment Module
6 |  |  | TEXT | EV (121017, DCM, "Device Observer Physical Location During Observation") | 1 | U |  |
7 |  |  | CODE | EV (113876, DCM, "Device Role in Procedure") | 1-n | U |  | BCID 7445 “Device Participating Roles”

Content Item Descriptions

Row 7 | If the device performing the observations has other roles, e.g., as the irradiating device in a dose report, this may be recorded here, if not implicit.

### TID 1005 Procedure Context

This template contains identifying (and optionally descriptive) attributes of the procedure that is the source of evidence being interpreted.

Whenever this template is invoked, all previously inherited attributes of Procedure Context are discarded and replaced.

### Note

If an observed digital image is identified by other than a DICOM UID, a Study Instance UID must be generated for the non-DICOM evidence. The same must be done to document interpretation of hard-copy radiographs generated outside of the scope of the DICOM system.

Type:

  Non-Extensible

Order:

  Significant

Table TID 1005. Procedure Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | UIDREF | EV (121018, DCM, "Procedure Study Instance UID") | 1 | U |  | Defaults to Study Instance UID (0020, 000D) of General Study Module
2 |  |  | UIDREF | EV (121019, DCM, "Procedure Study Component UID") | 1-n | U |  |
3 |  |  | TEXT | EV (121020, DCM, "Placer Number") | 1 | U |  |
4 | > | HAS CONCEPT MOD | TEXT | EV (110190, DCM, "Issuer of Identifier") | 1 | U |  | See note
5 |  |  | TEXT | EV (121021, DCM, "Filler Number") | 1 | U |  |
6 | > | HAS CONCEPT MOD | TEXT | EV (110190, DCM, "Issuer of Identifier") | 1 | U |  | See note
7 |  |  | TEXT | EV (121022, DCM, "Accession Number") | 1 | U |  | Defaults to (0008,0050)
8 | > | HAS CONCEPT MOD | TEXT | EV (110190, DCM, "Issuer of Identifier") | 1 | U |  | See note
9 |  |  | CODE | EV (121023, DCM, "Procedure Code") | 1-n | U |  | Defaults to Procedure Code Sequence (0008,1032) of General Study Module

Content Item Descriptions

Rows 5, 6 | The issuer shall be formatted in accordance with the HL7v2 Hierarchic Designator Data Type. That format is [ Namespace ID]^[ Universal ID^ Universal ID Type], where Namespace IDidentifies an entity within the local namespace or domain, Universal IDis a universal or unique identifier for an entity, and Universal ID Typespecifies the standard format of the Universal ID (see HL7 v2 Section 2.A.33).

### TID 1006 Subject Context

This template contains identifying (and optionally descriptive) attributes of the subject of the observation.

Subject context identifies (and optionally) describes the subject of the observation, whether it be a patient (human or animal), a fetus (human or animal), a specimen, or a device.

Type:

  Non-Extensible

Order:

  Significant

Table TID 1006. Subject Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121024, DCM, "Subject Class") | 1 | MC | IF subject is not the Patient | DCID 271 “Observation Subject Class” Defaults to (121025, DCM, "Patient")
2 |  |  | INCLUDE | DTID 1007 “Subject Context, Patient” | 1 | UC | IFF Row 1 value = (121025, DCM, "Patient") or Row 1 is absent | May be used for human or animal patients
3 |  |  | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | UC | IFF Row 1 value = (121026, DCM, "Fetus") | May be used for human or animal fetuses
4 |  |  | INCLUDE | DTID 1009 “Subject Context, Specimen” | 1 | UC | IFF Row 1 value = (121027, DCM, "Specimen") |
5 |  |  | INCLUDE | DTID 1010 “Subject Context, Device” | 1 | UC | IFF Row 1 value = (121192, DCM, "Device Subject") |

### TID 1007 Subject Context, Patient

Identifies (and optionally describes) a patient who is the subject.

Type:

  Extensible

Order:

  Significant

Table TID 1007. Subject Context, Patient

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | UIDREF | EV (121028, DCM, "Subject UID") | 1 | U |  | E.g., SOP Instance UID of Detached Patient Instance
2 |  |  | PNAME | EV (121029, DCM, "Subject Name") | 1 | MC | Required if not inherited. | Defaults to value of Patient's Name (0010,0010) in Patient Module
3 |  |  | CODE | EV (121030, DCM, "Subject ID") | 1 | MC | Required if not inherited. | Defaults to value of Patient ID (0010,0020) in Patient Module
4 |  |  | DATE | EV (121031, DCM, "Subject Birth Date") | 1 | U |  | Defaults to value of Patient's Birth Date (0010,0030) in Patient Module
5 |  |  | CODE | EV (121032, DCM, "Subject Sex") | 1 | U |  | Defaults to value equivalent to Patient's Sex (0010,0040) in Patient Module DCID 7455 “Sex”
6 |  |  | NUM | EV (121033, DCM, "Subject Age") | 1 | U |  | Defaults to value of Patient's Age (0010,1010) in Patient Study Module UNITS = DCID 7456 “Units of Measure for Age”
7 |  |  | CODE | EV (121034, DCM, "Subject Species") | 1 | MC | Required if not inherited. | DCID 7454 “Species” Defaults to (L-85B00, SRT, "homo sapiens").
8 |  |  | CODE | EV (121035, DCM, "Subject Breed") | 1 | U |  | Defaults to value of Patient Breed Code Sequence (0010,2293) in General Patient Module DCID 7480 “Breed”

### TID 1008 Subject Context, Fetus

Identifies (and optionally describes) a fetus who is the subject.

Type:

  Extensible

Order:

  Significant

Table TID 1008. Subject Context, Fetus

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | PNAME | EV (121036, DCM, "Mother of fetus") | 1 | U |  | Defaults to an observation subject that is a patient prior to replacing the Observation Subject Class with Fetus.
2 |  |  | UIDREF | EV (121028, DCM, "Subject UID") | 1 | U |  | For longitudinal tracking of individual fetuses
3 |  |  | TEXT | EV (121030, DCM, "Subject ID") | 1 | MC | IF row 4 is absent | For longitudinal tracking of individual fetuses (human readable value e.g., "A" or "1")
4 |  |  | TEXT | EV (11951-1, LN, "Fetus ID") | 1 | MC | IF row 3 is absent | For separation of multiple fetuses during this procedure e.g., fetus '1" of '2" … not for longitudinal comparisons.; i.e., the "m" of fetus "m" of "n"
5 |  |  | NUM | EV (11878-6, LN, "Number of Fetuses by US") | 1 | U | XOR Row 6 | UNITS = EV (1, UCUM, "no units")
6 |  |  | NUM | EV (55281-0, LN, "Number of Fetuses") | 1 | UC | XOR Row 5 | UNITS = EV (1, UCUM, "no units")

Content Item Descriptions

Row 5, 6 | The "n" of fetus "m" of "n"; either the code for the ultrasound method (Row 5) or for the non-specific method (Row 6) may be used

### TID 1009 Subject Context, Specimen

Identifies (and optionally describes) a specimen that is the subject.

Type:

  Extensible

Order:

  Significant

Table TID 1009. Subject Context, Specimen

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | UIDREF | EV (121039, DCM, "Specimen UID") | 1 | U |  |
2 |  |  | INCLUDE | DTID 1007 “Subject Context, Patient” | 1 | UC | IFF the source of the specimen is a human or animal patient |
3 |  |  | TEXT | EV (121041, DCM, "Specimen Identifier") | 1 | U |  |
4 |  |  | TEXT | EV (111724, DCM, "Issuer of Specimen Identifier") | 1 | U |  | See content item descriptions
5 |  |  | CODE | EV (R-00254, SRT, "Specimen Type") | 1 | U |  | DCID 8103 “Anatomic Pathology Specimen Types”
6 |  |  | TEXT | EV (111700, DCM, "Specimen Container Identifier") | 1 | U |  |

Content Item Descriptions

Rows 5, 6 | The Issuer of Specimen Identifier shall be formatted in accordance with the HL7 v2 Hierarchic Designator data type (see HL7 v2.6 Section 2.A.33), i.e., [ Namespace ID]^[ Universal ID^Universal ID Type]

### TID 1010 Subject Context, Device

Identifies (and optionally describes) a device that is the subject of observations.

Type:

  Extensible

Order:

  Significant

Table TID 1010. Subject Context, Device

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (121193, DCM, "Device Subject Name") | 1 | M |  |
2 |  |  | UIDREF | EV (121198, DCM, "Device Subject UID") | 1 | U |  |
3 |  |  | TEXT | EV (121194, DCM, "Device Subject Manufacturer") | 1 | U |  |
4 |  |  | TEXT | EV (121195, DCM, "Device Subject Model Name") | 1 | U |  |
5 |  |  | TEXT | EV (121196, DCM, "Device Subject Serial Number") | 1 | U |  |
6 |  |  | TEXT | EV (121197, DCM, "Device Subject Physical Location during observation") | 1 | U |  |

### TID 1020 Person Participant

This template describes a person participating in an activity as other than an observer or subject. E.g., for a dose report documenting an irradiating procedure, participants include the person administering the irradiation and the person authorizing the irradiation.

This Template is included with specific contextual parameters from a parent Template.

Table TID 1020. Parameters

Parameter Name | Parameter Usage
--- | ---
$PersonProcedureRole | Coded term or Context Group for the Concept Name that describes the nature of the person's participation in this procedure.

Type:

  Extensible

Order:

  Significant

Table TID 1020. Person Participant

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | PNAME | EV (113870, DCM, "Person Name") | 1 | M |  |
2 | > | HAS PROPERTIES | CODE | EV (113875, DCM, "Person Role in Procedure") | 1 | M |  | $PersonProcedureRole
3 | > | HAS PROPERTIES | TEXT | EV (113871, DCM, "Person ID") | 1 | U |  |
4 | > | HAS PROPERTIES | TEXT | EV (113872, DCM, "Person ID Issuer") | 1 | U |  |
5 | > | HAS PROPERTIES | TEXT | EV (113873, DCM, "Organization Name") | 1 | U |  |
6 | > | HAS PROPERTIES | CODE | EV (113874, DCM, "Person Role in Organization") | 1 | U |  | BCID 7452 “Organizational Roles”

Content Item Descriptions

Row 1 | The name of the person participating in the role identified in Row 2.
Row 2 | The procedural role played by the person in this procedure.
Row 6 | The organizational role played by the person in the organization.

### TID 1021 Device Participant

This template describes a device participating in an activity as other than an observer or subject. E.g., for a dose report documenting an irradiating procedure, participants include the irradiating device.

This Template is included with specific contextual parameters from a parent Template.

Table TID 1021. Parameters

Parameter Name | Parameter Usage
--- | ---
$DeviceProcedureRole | Coded term or Context Group for the Concept Name that describes the nature of the device's participation in this procedure.

Type:

  Extensible

Order:

  Significant

Table TID 1021. Device Participant

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (113876, DCM, "Device Role in Procedure") | 1 | M |  | $DeviceProcedureRole
2 | > | HAS PROPERTIES | TEXT | EV (113877, DCM, "Device Name") | 1 | U |  |
3 | > | HAS PROPERTIES | TEXT | EV (113878, DCM, "Device Manufacturer") | 1 | M |  |
4 | > | HAS PROPERTIES | TEXT | EV (113879, DCM, "Device Model Name") | 1 | M |  |
5 | > | HAS PROPERTIES | TEXT | EV (113880, DCM, "Device Serial Number") | 1 | M |  |
6 | > | HAS PROPERTIES | UIDREF | EV (121012, DCM, "Device Observer UID") | 1 | M |  |

Content Item Descriptions

Row 1 | If no Device Procedure Role is provided, BCID 7445 “Device Participating Roles” may be used.
Row 2 | This may be used for the name by which the organization manages the device.

### TID 1200 Language Designation

Defines a mechanism for specifying a language, optionally with designation of the country in which that language applies.

### Note

1. For example, the French language could be specified unmodified, or French as written in France or Canada could be distinguished.

2. The language codes specified in CID 5000 “Languages” optionally allow the encoding of the country of language in the code value for the language. Encoding of the country of language in a separate subsidiary Concept Modifier content item is allowed for backward compatibility with previous editions of the Standard.

Type:

  Extensible

Order:

  Significant

Table TID 1200. Language Designation

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121045, DCM, "Language") | 1 | M |  | DCID 5000 “Languages”
2 | > | HAS CONCEPT MOD | CODE | EV (121046, DCM, "Country of Language") | 1 | U |  | DCID 5001 “Countries”

### TID 1201 Language of Value

Defines a mechanism for specifying the language in which the value of the parent content item (only) is written. Does not specify the language of the Concept Name of the parent content item, nor of any other descendants of the parent content item.

### Note

The language codes specified in CID 5000 “Languages” optionally allow the encoding of the country of language in the code value for the language. Encoding of the country of language in a separate subsidiary Concept Modifier content item is allowed for backward compatibility with previous editions of the Standard.

Type:

  Non-Extensible

Order:

  Significant

Table TID 1201. Language of Value

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | CODE | EV (121047, DCM, "Language of Value") | 1 | M |  | DCID 5000 “Languages”
2 | > | HAS CONCEPT MOD | CODE | EV (121046, DCM, "Country of Language") | 1 | U |  | DCID 5001 “Countries”

### TID 1202 Language of Name and Value

Defines a mechanism for specifying the language in which the value and the Concept Name of the parent content item (only) is written. Does not specify the language of any other descendants of the parent content item.

### Note

The language codes specified in CID 5000 “Languages” optionally allow the encoding of the country of language in the code value for the language. Encoding of the country of language in a separate subsidiary Concept Modifier content item is allowed for backward compatibility with previous editions of the Standard.

Type:

  Extensible

Order:

  Significant

Table TID 1202. Language of Name and Value

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | CODE | EV (121048, DCM, "Language of Name and Value") | 1 | M |  | DCID 5000 “Languages”
2 | > | HAS CONCEPT MOD | CODE | EV (121046, DCM, "Country of Language") | 1 | U |  | DCID 5001 “Countries”

### TID 1204 Language of Content Item and Descendants

Defines a mechanism for specifying the language in which the value and the Concept Name of the parent content item and any other descendants of the parent content item is written.

### Note

The language codes specified in CID 5000 “Languages” optionally allow the encoding of the country of language in the code value for the language. Encoding of the country of language in a separate subsidiary Concept Modifier content item is allowed for backward compatibility with previous editions of the Standard.

Type:

  Non-Extensible

Order:

  Significant

Table TID 1204. Language of Content Item and Descendants

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | CODE | EV (121049, DCM, "Language of Content Item and Descendants") | 1 | M |  | DCID 5000 “Languages”
2 | > | HAS CONCEPT MOD | CODE | EV (121046, DCM, "Country of Language") | 1 | U |  | DCID 5001 “Countries”

### TID 1210 Equivalent Meaning(s) of Concept Name

Defines a mechanism for specifying one or more equivalent meanings for the Concept Name of the parent content item.

Type:

  Non-Extensible

Order:

  Significant

Table TID 1210. Equivalent Meaning(s) of Concept Name

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | TEXT | EV (121050, DCM, "Equivalent Meaning of Concept Name") | 1-n | MC | XOR Row 3 | Plain text equivalent of code meaning of the concept name of the content item being modified, in the specified language from the specified country, using the default character set or a character set selected from Specified Character Set
2 | > |  | INCLUDE | DTID 1201 “Language of Value” | 1 | U |  |
3 |  | HAS CONCEPT MOD | CODE | EV (121050, DCM, "Equivalent Meaning of Concept Name") | 1-n | MC | XOR Row 1 |
4 | > |  | INCLUDE | DTID 1201 “Language of Value” | 1 | U |  |

### TID 1211 Equivalent Meaning(s) of Value

Defines a mechanism for specifying one or more equivalent meanings for the Value of the parent content item.

Type:

  Extensible

Order:

  Significant

Table TID 1211. Equivalent Meaning(s) of Value

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | TEXT | EV (121051, DCM, "Equivalent Meaning of Value") | 1-n | MC | XOR Row 3 | Plain text equivalent of code meaning of the value of the content item being modified, in the specified language from the specified country, using the default character set or a character set selected from Specified Character Set
2 | > |  | INCLUDE | DTID 1201 “Language of Value” | 1 | U |  |
3 |  | HAS CONCEPT MOD | CODE | EV (121051, DCM, "Equivalent Meaning of Value") | 1-n | MC | XOR Row 1 |
4 | > |  | INCLUDE | DTID 1201 “Language of Value” | 1 | U |  |

### Note

1. For example, to describe a longer, more meaningful equivalent (in the same language) for a procedure code than is defined in a coding scheme:

   CODE: (121023, DCM, "Procedure Code") = (50291CC, ICD10PCS, "IMAGING:CNS:CT:SELLA:LOWOSMOLAR:IT, U, E:2PLANE3D")

   > HAS CONCEPT MOD TEXT: (121051, DCM, "Equivalent meaning of value") = "imaging study central nervous system of the sella turcica/pituitary gland with low osmolar contrast intrathecal, unenhanced and enhanced, in two planes with 3D reconstructions"

2. For example, to specify a concept name and value in both French and English in Canada:

   CODE:(T-D0005, SRT, "Anatomical structure") = (T-04000, SRT, "Breast")

   > HAS CONCEPT MOD CODE: (121048, DCM, "Language of name and value") = (en-CA, RFC3066, "English, Canada")

   > HAS CONCEPT MOD CODE: (121050, DCM, "Equivalent meaning of concept name") = (T-D0005, SRT, " Structure de l'anatomie ")

   >> HAS CONCEPT MOD CODE: (121047, DCM, " Langue de la valeur ") = (fr-CA, RFC3066, " Français, Canadien ")

   > HAS CONCEPT MOD CODE: (121051, DCM, "Equivalent meaning of value") = (T-04000, SRT, " Sein ")

   >> HAS CONCEPT MOD CODE: (121047, DCM, " Langue de la valeur ") = (fr-CA, RFC3066, " Français, Candie ")

### TID 1350 Negation Modifier, Presence of Finding

Concept Name Modifier for negation of the presence of a finding represented by a post-coordinated concept.

### Note

1. For example, negation modifier applied to "distention" in the post-coordinated structure:

   CODE: "anatomic location" = "bile duct"

   > HAS PROPERTY CODE: "morphology" = "distention"

   >> HAS CONCEPT MOD CODE: "presence of property" = "absent"

   means: "bile duct distention not present"

2. The presence-negation modifier modifies the entire post-coordinated concept, not just the source content item of the HAS CONCEPT MOD relationship. The entire branch of the tree from the content item is included in the post-coordinated structure that is negated.

Type:

  Extensible

Order:

  Significant

Table TID 1350. Negation Modifier, Presence of Finding

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | CODE | EV (121052, DCM, "Presence of property") | 1 | M |  | DCID 240 “Present-Absent”

### TID 1400 Linear Measurement

Type:

  Extensible

Order:

  Significant

Table TID 1400. Linear Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 7470 “Linear Measurements” | 1 | M |  | UNITS = DCID 7460 “Units of Linear Measurement”
2 | > | INFERRED FROM | SCOORD | EV (121055, DCM, "Path") | 1 | UC | XOR Row 5 | GRAPHIC TYPE = {POLYLINE, CIRCLE, ELLIPSE}
3 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 4 |
4 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 3 |
5 | > | INFERRED FROM | SCOORD | EV (121230, DCM, "Path Vertex") | 2-n | UC | XOR Row 2 | GRAPHIC TYPE = {POINT}
6 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 7 |
7 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 6 |
8 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |

Content Item Descriptions

Row 2 | Path shall represent the measured path or a projection of the measured path in the image. The Graphic Type (0070,0023) of the Path SCOORD shall be: an open POLYLINE with two different points (to measure length, diameter, distance, proximity, etc), a CIRCLE or ELLIPSE (to measure circumference) or an open or closed POLYLINE (closed polygon) to measure path length (open) or perimeter (closed).
Row 5 "Path Vertex" | A measured path that traverses two or more images (e.g., the ends of the path are in different cross-sectional plane images) shall be identified by vertices along the path. The Path Vertices shall be ordered by the order of their SCOORD Content Items to identify the measured path. The Graphic Type (0070,0023) of each SCOORD shall be POINT

### TID 1401 Area Measurement

Type:

  Extensible

Order:

  Significant

Table TID 1401. Area Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 7471 “Area Measurements” | 1 | M |  | Value shall be > 0 UNITS = DCID 7461 “Units of Area Measurement”
2 | > | INFERRED FROM | SCOORD | EV (121056, DCM, "Area Outline") | 1 | MC | IF concept name of Row 1 is (G-A16A, SRT, "Area of defined region"), and IFF Row 5 or 6 not present. | GRAPHIC TYPE = not {MULTIPOINT}
3 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 4 |
4 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 3 |
5 | > | INFERRED FROM | IMAGE | EV (121214, DCM, "Referenced Segmentation Frame") | 1 | MC | IF concept name of Row 1 is (G-A16A, SRT, "Area of defined region"), and IFF Row 2 or 6 not present. | Reference shall be to a Segmentation Image, with a single value specified in Referenced Frame Number
6 | > | R- INFERRED FROM | IMAGE |  | 1 | MC | IF concept name of Row 1 is (G-A16A, SRT, "Area of defined region"), and IFF Row 2 or 5 not present. | Reference shall be to a Segmentation Image, with a single value specified in Referenced Frame Number
7 | > | HAS PROPERTIES | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | DCID 7473 “General Area Calculation Methods”
8 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |

Content Item Descriptions

Row 2 "Area Outline" | A Graphic Type of POINT implies that the object is a single pixel and the object's area is the area of the pixel. Otherwise the type shall be a closed POLYLINE (start and end point the same) or a CIRCLE or an ELLIPSE.
Rows 5, 6 | Referenced Frame Number (0008,1160) is an attribute of the IMAGE Content Item. If the Referenced Segmentation SOP Instance has Segmentation Type (0062,0001) value BINARY, it identifies the area of defined (measured) region by pixel values in the referenced frame with value 1. For Segmentation Type value FRACTIONAL, the area is computed by an implementation dependent method. Frame number shall be specified even if the Segmentation SOP Instance has only a single frame.

### TID 1402 Volume Measurement

Type:

  Extensible

Order:

  Significant

Table TID 1402. Volume Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 7472 “Volume Measurements” | 1 | M |  | Value shall be > 0 UNITS = DCID 7462 “Units of Volume Measurement”
2 | > | INFERRED FROM | SCOORD | EV (121057, DCM, "Perimeter Outline") | 1-n | UC | XOR row 5, 6 | GRAPHIC TYPE = not {MULTIPOINT}
3 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 4 |
4 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 3 |
5 | > | INFERRED FROM | IMAGE | EV (121191, DCM, "Referenced Segment") | 1 | UC | XOR row 2, 6 | Reference shall be to a Segmentation Image, with a value specified in Referenced Segment Number
6 | > | R- INFERRED FROM | IMAGE |  | 1 | UC | XOR row 2, 5 | Reference shall be to a Segmentation Image, with a value specified in Referenced Segment Number
7 | > | HAS PROPERTIES | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | DCID 7474 “General Volume Calculation Methods”
8 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |

Content Item Descriptions

Row 2 "Perimeter Outline" | The two dimensional perimeter of the volume's intersection with or projection into the image. A Graphic Type of POINT implies that the volume's intersection or projection in a plane is a single pixel. A single pixel projection perimeter cannot cause a volume calculation to become 0. Otherwise the type shall be a closed POLYLINE (start and end point the same) or a CIRCLE or an ELLIPSE.
Rows 5, 6 | Referenced Segment Number (0062,000B) is an attribute of the IMAGE Content Item. If the Referenced Segmentation SOP Instance has Segmentation Type (0062,0001) value BINARY, it identifies the defined (measured) volume by pixel/voxel values in the frames of the referenced segment with value 1. For Segmentation Type value FRACTIONAL, the volume is computed by an implementation dependent method. Segment number shall be specified even if the Segmentation SOP Instance has only a single segment.

### TID 1404 Numeric Measurement

Type:

  Extensible

Order:

  Significant

Table TID 1404. Numeric Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | No baseline CID | 1 | M |  | UNITS = DCID 82 “Units of Measurement”
2 | > | INFERRED FROM | SCOORD | No baseline CID | 1-n | UC | XOR Row 5, 6 |
3 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 4 |
4 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 3 |
5 | > | R-INFERRED FROM | IMAGE | BCID 7003 “Diagnostic Imaging Report Purposes of Reference” | 1-n | UC | XOR Row 2, 6 |
6 | > | INFERRED FROM | IMAGE | BCID 7003 “Diagnostic Imaging Report Purposes of Reference” | 1-n | UC | XOR Row 2, 5 |
7 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |

Content Item Descriptions

Row 2 | The SCOORD may indicate the points or area where the measurement was taken (e.g., a POINT showing the pixel location of a density measurement, or an open POLYLINE of three points showing the calculation of an angle).
Rows 3, 5 | Valid only in IODs that permit relationships by-reference.

### TID 1406 Three Dimensional Linear Measurement

Type:

  Extensible

Order:

  Significant

Table TID 1406. Three Dimensional Linear Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 7470 “Linear Measurements” | 1 | M |  | UNITS = DCID 7460 “Units of Linear Measurement”
2 | > | INFERRED FROM | SCOORD3D | EV (121055, DCM, "Path") | 1 | M |  | GRAPHIC TYPE = {POLYLINE, ELLIPSE, POLYGON}
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |

Content Item Descriptions

Rows 5, 6 | Path shall represent the measured path in a reference coordinate space. The Graphic Type (0070,0023) of the Path SCOORD3D shall be: an open POLYLINE with two or more different (x,y,z) triplets (to measure length, diameter, distance, proximity, etc.), an ELLIPSE (to measure circumference) or a closed POLYGON to measure perimeter, where the (x,y,z) triplets are coplanar.

### TID 1410 Planar ROI Measurements

This Template provides a general structure to report one or more measurements for some metric, e.g., density, flow, or concentration, over a planar region of interest in an image. The ROI may be specified by an SCOORD on an image, or by a Segmentation Image.

Table TID 1410. Parameters

$Measurement | Coded term or Context Group for Concept Name of measurement
$Units | Units for the measurement
$ModType | Modifier Name for Concept Name of measurement
$ModValue | Modifier Value for Concept Name of measurement
$Method | Value for Measurement Method
$Derivation | Value for Measurement Derivation
$TargetSite | Value for Anatomic Location of measurement
$TargetSiteMod | Modifier Value for Anatomic Location of measurement
$Equation | Coded term or Context Group for the equation or table from which the measurement was derived or computed
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$DerivationParameter | Coded term or Context Group for Concept Name of a derivation parameter
$DerivationParameterUnits | Units of derivation parameter
$QualitativeEvaluations | Evaluations encoded with code or text responses

Type:

  Extensible

Order:

  Significant

Table TID 1410. Planar ROI Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125007, DCM, "Measurement Group") | 1 | M |  |
2 | > | HAS OBS CONTEXT | TEXT | DT (112039, DCM, "Tracking Identifier") | 1 | M |  |
3 | > | HAS OBS CONTEXT | UIDREF | EV (112040, DCM, "Tracking Unique Identifier") | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 1502 “Time Point Context” | 1 | U |  |
5 | > | CONTAINS | SCOORD | EV (111030, DCM, "Image Region") | 1 | MC | XOR Row 7 | GRAPHIC TYPE = not {MULTIPOINT}
6 | >> | SELECTED FROM | IMAGE |  | 1 | M |  |
7 | > | CONTAINS | IMAGE | EV (121214, DCM, "Referenced Segmentation Frame") | 1 | MC | XOR Row 5 | Reference shall be to a Segmentation Image, with a single value specified in Referenced Frame Number, and with a single value specified in Referenced Segment Number
8 | > | CONTAINS | IMAGE | EV (121233, DCM, "Source image for segmentation") | 1 | MC | IFF Row 7 |
9 | > | CONTAINS | IMAGE | EV (121200, DCM, "Illustration of ROI") | 1 | U |  |
10 | > | CONTAINS | COMPOSITE | EV (126100, DCM, "Real World Value Map used for measurement") | 1 | U |  | SOP Class UID shall be Real World Value Mapping Storage ("1.2.840.10008.5.1.4.1.1.67")
11 | > | CONTAINS | INCLUDE | DTID 1419 “ROI Measurements” | 1 | M |  | $Measurement = $Measurement $Units = $Units $ModType = $ModType $ModValue = $ModValue $Method = $Method $Derivation = $Derivation $TargetSite = $TargetSite $TargetSiteMod = $TargetSiteMod $Equation = $Equation $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority $DerivationParameter = $DerivationParameter $DerivationParameterUnits = $DerivationParameterUnits
12 | > | CONTAINS | CODE | $QualitativeEvaluations | 1-n | U |  |
13 | > | CONTAINS | TEXT | $QualitativeEvaluations | 1-n | U |  |

Content Item Descriptions

Rows 2, 3 | The Tracking Identifier and Tracking Unique Identifier are defined as a text label or unique identifier (respectively) used for tracking a finding or feature, potentially across multiple reporting objects, over time. As such, they are distinct from the Observation UID (0040,A171), which is unique identifier of the specific Content Item and its subsidiary Content Items that constitute an individual observation, and would be different for different observations on different occasions of the same finding or feature.
Rows 6, 7 | Referenced Frame Number (0008,1160) is an attribute of the IMAGE Content Item, and shall be present with a single value. If the Referenced Segmentation SOP Instance has Segmentation Type (0062,0001) value BINARY, it identifies the area of defined (measured) region of interest by pixel values in the referenced frame with value 1. For Segmentation Type value FRACTIONAL, the area is computed by an implementation dependent method. Frame number shall be specified even if the Segmentation SOP Instance has only a single frame.
Row 8 | Identifies the source image that was segmented to identify the ROI, and whose properties are described in this container.
Row 9 | This referenced image may contain a "screen shot" illustrating a rendered version of the ROI.
Row 10 | The reference to an RWV in Row 10 allows measurements to be made in units that differ from the stored pixel values in the images referenced elsewhere in the template. E.g., for a PET SUVbw measurement, the mapping from activity/concentration units in the referenced image that was used (and which may be reused for measurements in the future) may be encoded in a referenced RWV instance. This reference applies to any measurements in included templates, unless overridden).
Rows 12, 13 | Allows encoding a flat list of name-value pairs that are coded questions with coded or text answers, for example, to record categorical observations related to the subject of the measurement group.

### TID 1411 Volumetric ROI Measurements

This Template provides a general structure to report one or more measurements for some metric, e.g., density, flow, or concentration, over a volumetric region of interest in a set of images or a Frame of Reference. The volumetric ROI may be specified by a set of SCOORDs on an image set representing a volume, by a volumetric Segmentation Image, by a volume defined in a Surface Segmentation, or by a SCOORD3D.

Table TID 1411. Parameters

$Measurement | Coded term or Context Group for Concept Name of measurement
$Units | Units for the measurement
$ModType | Modifier Name for Concept Name of measurement
$ModValue | Modifier Value for Concept Name of measurement
$Method | Value for Measurement Method
$Derivation | Value for Measurement Derivation
$TargetSite | Value for Anatomic Location of measurement
$TargetSiteMod | Modifier Value for Anatomic Location of measurement
$Equation | Coded term or Context Group for the equation or table from which the measurement was derived or computed
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$DerivationParameter | Coded term or Context Group for Concept Name of a derivation parameter
$DerivationParameterUnits | Units of derivation parameter
$QualitativeEvaluations | Evaluations encoded with code or text responses

Type:

  Extensible

Order:

  Significant

Table TID 1411. Volumetric ROI Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125007, DCM, "Measurement Group") | 1 | M |  |
2 | > | HAS OBS CONTEXT | TEXT | DT (112039, DCM, "Tracking Identifier") | 1 | M |  |
3 | > | HAS OBS CONTEXT | UIDREF | EV (112040, DCM, "Tracking Unique Identifier") | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 1502 “Time Point Context” | 1 | U |  |
5 | > | CONTAINS | SCOORD | EV (111030, DCM, "Image Region") | 1-n | MC | XOR Rows 7, 10 | GRAPHIC TYPE = not {MULTIPOINT}
6 | >> | SELECTED FROM | IMAGE |  | 1 | M |  |
7 | > | CONTAINS | IMAGE | EV (121191, DCM, "Referenced Segment") | 1 | MC | XOR Rows 5, 10 | Reference shall be to a Segmentation Image or Surface Segmentation object, with a single value specified in Referenced Segment Number
8 | > | CONTAINS | IMAGE | EV (121233, DCM, "Source image for segmentation") | 1-n | MC | XOR Row 9 and IFF Row 7 |
9 | > | CONTAINS | UIDREF | EV (121232, DCM, "Source series for image segmentation") | 1 | MC | XOR Row 8 and IFF Row 7 |
10 | > | CONTAINS | SCOORD3D | EV (121231, DCM, "Volume Surface") | 1 | MC | XOR Rows 5, 7 | GRAPHIC TYPE = {ELLIPSOID}
11 | > | CONTAINS | IMAGE | EV (121233, DCM, "Source image for segmentation") | 1-n | MC | XOR Row 12 and IFF Row 10 |
12 | > | CONTAINS | UIDREF | EV (121232, DCM, "Source series for segmentation") | 1 | MC | XOR Row 11 and IFF Row 10 |
13 | > | CONTAINS | IMAGE | EV (121200, DCM, "Illustration of ROI") | 1-n | U |  |
14 | > | CONTAINS | COMPOSITE | EV (126100, DCM, "Real World Value Map used for measurement") | 1 | U |  | SOP Class UID shall be Real World Value Mapping Storage ("1.2.840.10008.5.1.4.1.1.67")
15 | > | CONTAINS | INCLUDE | DTID 1419 “ROI Measurements” | 1 | M |  | $Measurement = $Measurement $Units = $Units $ModType = $ModType $ModValue = $ModValue $Method = $Method $Derivation = $Derivation $TargetSite = $TargetSite $TargetSiteMod = $TargetSiteMod $Equation = $Equation $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority $DerivationParameter = $DerivationParameter $DerivationParameterUnits = $DerivationParameterUnits
16 | > | CONTAINS | CODE | $QualitativeEvaluations | 1-n | U |  |
17 | > | CONTAINS | TEXT | $QualitativeEvaluations | 1-n | U |  |

Content Item Descriptions

Rows 2, 3 | The Tracking Identifier and Tracking Unique Identifier are defined as a text label or unique identifier (respectively) used for tracking a finding or feature, potentially across multiple reporting objects, over time. As such, they are distinct from the Observation UID (0040,A171), which is unique identifier of the specific Content Item and its subsidiary Content Items that constitute an individual observation, and would be different for different observations on different occasions of the same finding or feature.
Rows 6, 7 | Referenced Segment Number (0062,000B) is an attribute of the IMAGE Content Item, and shall be present with a single value. If the Referenced SOP Instance is a Segmentation Image, it shall have a defined Frame of Reference. If it has Segmentation Type (0062,0001) value BINARY, it identifies the volume of defined (measured) region of interest by voxel values in the referenced segment with value 1. If it has Segmentation Type value FRACTIONAL, the volume is defined by an implementation dependent method. If the referenced SOP Instance is a Surface Segmentation, the referenced segment shall constitute a finite volume. It identifies the volume of the defined (measured) region of interest by the interior of the finite volume. Segment number shall be specified even if the Segmentation SOP Instance has only a single segment.
Rows 8, 11 | Identifies the source images that were segmented to identify the ROI, when, for example a subset of images in a series was used.
Rows 9, 12 | Identifies the source series of images that were segmented to identify the ROI, when, for example an entire set of images in a series was used.
Row 13 | These referenced images may contain "screen shot" illustrating rendered versions of the ROI.
Row 14 | The reference to an RWV in Row 14 allows measurements to be made in units that differ from the stored pixel values in the images referenced elsewhere in the template. E.g., for a PET SUVbw measurement, the mapping from activity/concentration units in the referenced image that was used (and which may be reused for measurements in the future) may be encoded in a referenced RWV instance. This reference applies to any measurements in included templates, unless overridden).
Rows 16, 17 | Allows encoding a flat list of name-value pairs that are coded questions with coded or text answers, for example, to record categorical observations related to the subject of the measurement group.

### TID 1419 ROI Measurements

This Template encodes measurements for some metric, e.g., density, flow, or concentration.

Table TID 1419. Parameters

$Measurement | Coded term or Context Group for Concept Name of measurement
$Units | Units for the measurement
$ModType | Modifier Name for Concept Name of measurement
$ModValue | Modifier Value for Concept Name of measurement
$Method | Value for Measurement Method
$Derivation | Value for Measurement Derivation
$TargetSite | Value for Anatomic Location of measurement
$TargetSiteMod | Modifier Value for Anatomic Location of measurement
$Equation | Coded term or Context Group for the equation or table from which the measurement was derived or computed
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$DerivationParameter | Coded term or Context Group for Concept Name of a derivation parameter
$DerivationParameterUnits | Units of derivation parameter

Type:

  Extensible

Order:

  Significant

Table TID 1419. ROI Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | $Method
2 |  | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | U |  | $TargetSite
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
4 | > | HAS CONCEPT MOD | CODE | DT (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | $TargetSiteMod
5 |  |  | NUM | $Measurement | 1-n | M |  | UNITS = $Units
6 | > | HAS CONCEPT MOD | CODE | $ModType | 1-n | U |  | $ModValue
7 | > | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | $Method
8 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | U |  | $Derivation
9 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | U |  | $TargetSite
10 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
11 | >> | HAS CONCEPT MOD | CODE | DT (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | $TargetSiteMod
12 | > | HAS PROPERTIES | INCLUDE | DTID 310 “Measurement Properties” | 1 | U |  | $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority
13 | > | INFERRED FROM | NUM | $DerivationParameter | 1-n | UC | XOR Row 14 | $DerivationParameterUnits
14 | > | R-INFERRED FROM | NUM | $DerivationParameter | 1-n | UC | XOR Row 13 | $DerivationParameterUnits
15 | > | INFERRED FROM | INCLUDE | DTID 315 “Equation or Table” | 1 | UC | XOR Row 16 | $Equation = $Equation
16 | > | INFERRED FROM | TEXT | DCID 228 “Equation or Table” | 1 | UC | XOR Row 15 |
17 | > |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
18 | > | HAS CONCEPT MOD | TEXT | EV (121050, DCM, "Equivalent Meaning of Concept Name") | 1 | U |  |

Content Item Descriptions

Row 5 | Specifies the metric for which measurements are reported, e.g., density, flow, or concentration. This metric is computed at each sample point (e.g., pixel or voxel) in an ROI (defined in the invoking template), but those individual point measurements are not encoded. Instead, just the summary measurements for the ROI are encoded, and the means of computing a single value is defined in Row 8 Derivation (e.g., mean).
Rows 1, 2, 3, 4, 6, 7, 8, 9, 10, 11 | The HAS CONCEPT MOD items allow the explicit definition of terms for post-coordination of the measurement concept name. Additional post-coordinated modifier terms may be included in a SOP Instance based on this template, in accordance with section 6.2.4, or as defined by templates that invoke this template and explicitly define additional post-coordinated modifiers.
Rows 13, 14 | The INFERRED FROM items allow the specification (by-value or by-reference) of numeric values that were used in the derivation of the numeric measurement of Row 1. The nature of the inference is not explicitly conveyed; it may be implicit in the Concept Names of the measurements. Inference by-reference is valid only in SOP Classes that permit the INFERRED FROM relationship by-reference.
Row 18 | Equivalent Meaning of Concept Name allows the creating application to specify the preferred composed concept name representing the measurement and the associated post-coordinated concept modifiers. The concept modifiers may include those specified in this template, in a template that invokes this template, or at the option of the creating application in accordance with section 6.2.4. This composed concept name may be rendered by a display application.

### TID 1420 Measurements Derived From Multiple ROI Measurements

This Template encodes measurements for some metric, e.g., density, flow, or concentration, which are acquired over some defined sampling (e.g., over successive time slots in a dynamic contrast enhanced acquisition).

Table TID 1420. Parameters

$Measurement | Coded term or Context Group for Concept Name of measurement
$MeasurementUnits | Units for the measurement
$ModType | Modifier Name for Concept Name of measurement
$ModValue | Modifier Value for Concept Name of measurement
$Method | Value for Measurement Method
$Derivation | Value for Measurement Derivation
$TargetSite | Value for Anatomic Location of measurement
$TargetSiteMod | Modifier Value for Anatomic Location of measurement
$Equation | Coded term or Context Group for the equation or table from which the measurement was derived or computed
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$StatisticalRefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$StatisticalRangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$DerivationParameter | Coded term or Context Group for Concept Name of a derivation parameter
$DerivationParameterUnits | Units of derivation parameter

Type:

  Extensible

Order:

  Non-Significant

Table TID 1420. Measurements Derived From Multiple ROI Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DCID 7465 “Measurements Derived From Multiple ROI Measurements” | 1-n | M |  |
2 | > | R-INFERRED FROM | INCLUDE | DTID 1410 “Planar ROI Measurements” | 1-n | MC | XOR Row 3 | $Measurement = $Measurement $Units = $MeasurementUnits $ModType = $ModType $ModValue = $ModValue $Method = $Method $Derivation = $Derivation $TargetSite = $TargetSite $TargetSiteMod = $TargetSiteMod $Equation = $Equation $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority $DerivationParameter = $DerivationParameter $DerivationParameterUnits = $DerivationParameterUnits
3 | > | R-INFERRED FROM | INCLUDE | DTID 1411 “Volumetric ROI Measurements” | 1-n | MC | XOR Row 2 | $Measurement = $Measurement $Units = $MeasurementUnits $ModType = $ModType $ModValue = $ModValue $Method = $Method $Derivation = $Derivation $TargetSite = $TargetSite $TargetSiteMod = $TargetSiteMod $Equation = $Equation $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority $DerivationParameter = $DerivationParameter $DerivationParameterUnits = $DerivationParameterUnits
4 | > | HAS PROPERTIES | INCLUDE | DTID 310 “Measurement Properties” | 1 | U |  | $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority

Content Item Descriptions

Row 1 | Specifies the type of derived measurement reported, e.g., the mean of the individual ROI mean density values. Note that the units may be different from the units in the ROI measurements
Rows 2, 3 | The measurement values of each ROI that contributes to the derived measurement, e.g., the mean density within an ROI. These are specified by reference, so as to not have to repeat the ROI information when it contributes to multiple derived measurements (e.g., if both mean and SD of ROI mean density values were specified).

### TID 1500 Measurement Report

This root template that encodes a list of Measurement Groups each containing lists of measurements, together with any derived measurements.

Each Measurement Group is identified by Tracking ID and UIDs.

An image library is available to describe characteristics of the images referenced by the measurements, if any.

Type:

  Extensible

Order:

  Non-Significant

Root:

  Yes

Table TID 1500. Measurement Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DCID 7021 “Measurement Report Document Titles” | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
4 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1-n | M |  | BCID 100 “Quantitative Diagnostic Imaging Procedures”
5 | > | CONTAINS | INCLUDE | DTID 1600 “Image Library” | 1 | M |  |
6 | > | CONTAINS | CONTAINER | EV (126010, DCM, "Imaging Measurements") | 1 | C | IF row 10 and 12 are absent |
7 | >> | CONTAINS | INCLUDE | DTID 1410 “Planar ROI Measurements” | 1-n | U |  | $Measurement = BCID 7469 “Generic Intensity and Size Measurements” $Units = BCID 7181 “Abstract Multi-dimensional Image Model Component Units” $Derivation = BCID 7464 “General Region of Interest Measurement Modifiers” $Method = BCID 6147 “Response Criteria”
8 | >> | CONTAINS | INCLUDE | DTID 1411 “Volumetric ROI Measurements” | 1-n | U |  | $Measurement = BCID 7469 “Generic Intensity and Size Measurements” $Units = BCID 7181 “Abstract Multi-dimensional Image Model Component Units” $Derivation = BCID 7464 “General Region of Interest Measurement Modifiers” $Method = BCID 6147 “Response Criteria”
9 | >> | CONTAINS | INCLUDE | DTID 1501 “Measurement Group” | 1-n | U |  | $Measurement = BCID 7469 “Generic Intensity and Size Measurements” $Units = BCID 7181 “Abstract Multi-dimensional Image Model Component Units” $Derivation = BCID 7464 “General Region of Interest Measurement Modifiers” $Method = BCID 6147 “Response Criteria”
10 | > | CONTAINS | CONTAINER | EV (126011, DCM, "Derived Imaging Measurements") | 1 | C | IF row 6 and 12 are absent |
11 | >> | CONTAINS | INCLUDE | DTID 1420 “Measurements Derived From Multiple ROI Measurements” | 1-n | U |  |
12 | > | CONTAINS | CONTAINER | EV (C0034375, UMLS, "Qualitative Evaluations") | 1 | C | IF row 6 and 10 are absent |
13 | >> | CONTAINS | CODE |  | 1-n | U |  |
14 | >> | CONTAINS | TEXT |  | 1-n | U |  |

Content Item Descriptions

Rows 6, 10, 12 | The conditions require that all except one of the "heading" containers is present, though each may be empty.
Rows 7, 8, 9 | The baseline context groups defined allow for generic intensity and size measurements, regardless of the geometry of the ROI (e.g., linear distance can be measured on volumes, or volume can be estimated from a linear distance), and being baseline, do not constrain the invoker from using other appropriate concepts specific to the application.
Row 9 | A Measurement Group is used to contain one or more individual measurements that are invocations of TID 300, consistent with TIDs 1410 and 1411, which both already have Measurement Group containers as their roots.
Rows 12, 13, 14 | These content items allow encoding a flat list of name-value pairs that are coded questions with coded or text answers, for example, to record categorical observations related to the entire subject of the report rather than specific measurement groups.

### TID 1501 Measurement Group

This template groups Measurements into a Measurement Group.

Each Measurement Group is identified by Tracking ID and UIDs, and may be described as having being made at a particular time point.

Measurement groups may contain various common measurement modifiers that are shared by all measurements in the group, such as method and finding site.

Table TID 1501. Parameters

Parameter Name | Parameter Usage
--- | ---
$Measurement | Coded term or Context Group for Concept Name of measurement
$Units | Units of Measurement
$ModType | Modifier Name for Concept Name of measurement
$ModValue | Modifier Value for Concept Name of measurement
$Method | Value for Measurement Method
$Derivation | Value for Measurement Derivation
$TargetSite | Value for Anatomic Location of measurement
$TargetSiteMod | Modifier Value for Anatomic Location of measurement
$Equation | Coded term or Context Group for the equation or table from which the measurement was derived or computed
$ImagePurpose | Purpose of Reference for an image used as a source of the measurement
$WavePurpose | Purpose of Reference for a waveform used as a source of the measurement
$RefAuthority | Bibliographic reference or authority for statistical properties of a reference population
$RangeAuthority | Bibliographic reference or authority for the normal range of the measurement
$DerivationParameter | Coded term or Context Group for Concept Name of a derivation parameter
$DerivationParameterUnits | Units of derivation parameter
$QualitativeEvaluations | Evaluations encoded with code or text responses

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1501. Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (125007, DCM, "Measurement Group") | 1 | M |  |
2 | > | HAS OBS CONTEXT | TEXT | DT (112039, DCM, "Tracking Identifier") | 1 | M |  |
3 | > | HAS OBS CONTEXT | UIDREF | EV (112040, DCM, "Tracking Unique Identifier") | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 1502 “Time Point Context” | 1 | U |  |
5 | > | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | $Method
6 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | U |  | $TargetSite
7 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
8 | >> | HAS CONCEPT MOD | CODE | DT (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | $TargetSiteMod
9 | > | CONTAINS | COMPOSITE | EV (126100, DCM, "Real World Value Map used for measurement") | 1 | U |  | SOP Class UID shall be Real World Value Mapping Storage ("1.2.840.10008.5.1.4.1.1.67")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = $Measurement $Units = $Units $ModType = $ModType $ModValue = $ModValue $Method = $Method $Derivation = $Derivation $TargetSite = $TargetSite $TargetSiteMod = $TargetSiteMod $Equation = $Equation $RefAuthority = $RefAuthority $RangeAuthority = $RangeAuthority $DerivationParameter = $DerivationParameter $DerivationParameterUnits = $DerivationParameterUnits
11 | > | CONTAINS | CODE | $QualitativeEvaluations | 1-n | U |  |
12 | > | CONTAINS | TEXT | $QualitativeEvaluations | 1-n | U |  |

Content Item Descriptions

Rows 2, 3, 10 | The included TID 300 already contains an optional inclusion of TID 1408 Tracking Identifier, which in turns allows for either or both Tracking Identifier and Tracking Unique Identifier content items; the intent of Rows 2 and 3 is not to send these content items twice, but rather to specialize their use such that their presence is mandatory, consistent with TIDs 1410 and 1411.
Row 9 | The reference to an RWV in Row 9 allows measurements to be made in units that differ from the stored pixel values in the images referenced elsewhere in the template. E.g., for a PET SUVbw measurement, the mapping from activity/concentration units in the referenced image that was used (and which may be reused for measurements in the future) may be encoded in a referenced RWV instance. This reference applies to any measurements in included templates, unless overridden).
Rows 11, 12 | Allows encoding a flat list of name-value pairs that are coded questions with coded or text answers, for example, to record categorical observations related to the subject of the measurement group.

### TID 1502 Time Point Context

This template describes information about the time point, for example, at which a measurement was obtained.

Type:

  Extensible

Order:

  Non-Significant

Root:

  No

Table TID 1502. Time Point Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS OBS CONTEXT | TEXT | EV (126070, DCM, "Subject Time Point Identifier") | 1 | U |  |
2 |  | HAS OBS CONTEXT | TEXT | EV (126071, DCM, "Protocol Time Point Identifier") | 1 | U |  |
3 |  | HAS OBS CONTEXT | TEXT | EV (C2348792, UMLS, "Time Point") | 1 | M |  |
4 |  | HAS OBS CONTEXT | CODE | EV (126072, DCM, "Time Point Type") | 1-n | U |  | BCID 6146 “Time Point Types”
5 |  | HAS OBS CONTEXT | NUM | EV (126073, DCM, "Time Point Order") | 1 | U |  | UNITS = EV (1, UCUM, "no units")

Content Item Descriptions

Row 1 | Usually the same value as the Clinical Trial Time Point ID (0012,0050) attribute in the Clinical Trial Study Module, though not confined to clinical trial use. May or may not be human readable, and not required to be a DICOM UID.
Row 2 | All of the subjects within a treatment protocol that are examined at a particular scheduled time point (e.g., "baseline", "pre-treatment", "first post-treatment") will have the same Protocol Time Point Identifier, but different Subject Time Point Identifiers. However, in different protocols, the Protocol Time Point Identifiers for the same conceptual “time point” will be different. E.g., the “baseline” time point will have different Protocol Time Point Identifiers in different protocols. May or may not be human readable, and not required to be a DICOM UID.
Row 3 | Typically a short pre-defined label that has the same scope as Protocol Time Point Identifier (i.e., same conceptual time point within a treatment protocol) but is human-readable and understandable, e.g., "BASELINE" or "TP0", "TP1", etc. Usually the same value as Clinical Trial Time Point Description (0012,0051) attribute in the Clinical Trial Study Module, though not confined to clinical trial use. The Concept Name is selected as (C2348792, UMLS, "Time Point") (which is (C68568, NCIt, "Time Point"), defined as "a specific point in the time continuum, including those established relative to an event") in order to be compatible with external terminologies.
Row 4 | More than one type is permitted, e.g., a time point may be "posttreatment" as well as "unscheduled" or "nadir", etc.
Row 5 | The order is expected to be monotonically increasing within a particular scope of usage, but is not required to start at 0 or 1, nor required to increase in increments of 1 or even the same increment (e.g., to allow for retrospective insertion of unscheduled time points). In clinical usage, the Time Point Order would be expected to be temporally increasing, but in a clinical trial may be a randomized reading order rather than a temporal order.

### TID 1600 Image Library

The Image Library contains references to images and and selected attributes describing them that facilitate analysis without having to retrieve the entire set of referenced images.

Type:

  Extensible

Order:

  Non-Significant

Table TID 1600. Image Library

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111028, DCM, "Image Library") | 1 | M |  |
2 | > | CONTAINS | CONTAINER | EV (126200, DCM, "Image Library Group") | 1-n | U |  |
3 | >> | HAS ACQ CONTEXT | INCLUDE | DTID 1602 “Image Library Entry Descriptors” | 1 | U |  |
4 | >> | CONTAINS | INCLUDE | DTID 1601 “Image Library Entry” | 1-n | U |  |

### TID 1601 Image Library Entry

Each instance of the Image Library Entry template contains the Image SOP Class and Instance UIDs, and selected attributes for an image that facilitate analysis without having to retrieve the entire set of referenced images.

Type:

  Extensible

Order:

  Non-Significant

Table TID 1601. Image Library Entry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | IMAGE |  | 1 | M |  |
2 | > | HAS ACQ CONTEXT | INCLUDE | DTID 1602 “Image Library Entry Descriptors” | 1 | U |  |

### TID 1602 Image Library Entry Descriptors

This template contains selected attributes for an image or group of images. The descriptive information may be copied from images or derived.

Type:

  Extensible

Order:

  Non-Significant

Table TID 1602. Image Library Entry Descriptors

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | CODE | EV (121139, DCM, "Modality") | 1 | U |  |
2 |  | HAS ACQ CONTEXT | CODE | EV (123014, DCM, "Target Region") | 1 | U |  | DCID 4031 “Common Anatomic Regions”
3 |  | HAS ACQ CONTEXT | CODE | EV (111027, DCM, "Image Laterality") | 1 | U |  |
4 |  | HAS ACQ CONTEXT | DATE | EV (111060, DCM, "Study Date") | 1 | U |  |
5 |  | HAS ACQ CONTEXT | TIME | EV (111061, DCM, "Study Time") | 1 | U |  |
6 |  | HAS ACQ CONTEXT | DATE | EV (111018, DCM, "Content Date") | 1 | U |  |
7 |  | HAS ACQ CONTEXT | TIME | EV (111019, DCM, "Content Time") | 1 | U |  |
8 |  | HAS ACQ CONTEXT | DATE | EV (126201, DCM, "Acquisition Date") | 1 | U |  |
9 |  | HAS ACQ CONTEXT | TIME | EV (126202, DCM, "Acquisition Time") | 1 | U |  |
10 |  | HAS ACQ CONTEXT | UIDREF | EV (112227, DCM, "Frame of Reference UID") | 1 | U |  |
11 |  | HAS ACQ CONTEXT | NUM | EV (110910, DCM, "Pixel Data Rows") | 1 | U |  | UNITS = EV ({pixels}, UCUM, "pixels")
12 |  | HAS ACQ CONTEXT | NUM | EV (110911, DCM, "Pixel Data Columns") | 1 | U |  | UNITS = EV ({pixels}, UCUM, "pixels")
13 |  | HAS ACQ CONTEXT | INCLUDE | DTID 1603 “Image Library Entry Descriptors for Projection Radiography” | 1 | U |  |
14 |  | HAS ACQ CONTEXT | INCLUDE | DTID 1604 “Image Library Entry Descriptors for Cross-Sectional Modalities” | 1 | U |  |
15 |  | HAS ACQ CONTEXT | INCLUDE | DTID 1605 “Image Library Entry Descriptors for CT” | 1 | U |  |
16 |  | HAS ACQ CONTEXT | INCLUDE | DTID 1606 “Image Library Entry Descriptors for MR” | 1 | U |  |
17 |  | HAS ACQ CONTEXT | INCLUDE | DTID 1607 “Image Library Entry Descriptors for PET” | 1 | U |  |

Content Item Descriptions

Target Region | The value of Anatomic Region Sequence (0008,2218) in the Image IOD, or a code derived from Body Part Examined (0018,0015) using the mapping described in Annex L.

### TID 1603 Image Library Entry Descriptors for Projection Radiography

This template contains selected attributes for a projection radiography image or group of such images. The descriptive information may be copied from images or derived.

Type:

  Extensible

Order:

  Non-Significant

Table TID 1603. Image Library Entry Descriptors for Projection Radiography

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | CODE | EV (111031, DCM, "Image View") | 1 | U |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111032, DCM, "Image View Modifier") | 1-n | U |  |
3 |  | HAS ACQ CONTEXT | TEXT | EV (111044, DCM, "Patient Orientation Row") | 1 | U |  |
4 |  | HAS ACQ CONTEXT | TEXT | EV (111043, DCM, "Patient Orientation Column") | 1 | U |  |
5 |  | HAS ACQ CONTEXT | NUM | EV (111026, DCM, "Horizontal Pixel Spacing") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
6 |  | HAS ACQ CONTEXT | NUM | EV (111066, DCM, "Vertical Pixel Spacing") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
7 |  | HAS ACQ CONTEXT | NUM | EV (112011, DCM, "Positioner Primary Angle") | 1 | U |  |
8 |  | HAS ACQ CONTEXT | NUM | EV (112012, DCM, "Positioner Secondary Angle") | 1 | U |  |

Content Item Descriptions

Patient Orientation Row | First (row) and second (column) components of Patient Orientation (0020,0020) in the Image IOD. See Section C.7.6.1.1.1 “Patient Orientation” in PS3.3 .
Patient Orientation Column
Horizontal Imager Pixel Spacing | The row (first) component of Imager Pixel Spacing (0018,1164) in the Image IOD. See Section C.8.11.4 “DX Detector Module” in PS3.3 .
Vertical Imager Pixel Spacing | The column (second) component of Imager Pixel Spacing (0018,1164) in the Image IOD. See Section C.8.11.4 “DX Detector Module” in PS3.3 .

### TID 1604 Image Library Entry Descriptors for Cross-Sectional Modalities

This template contains selected attributes for a cross-sectional image or group of such images. The descriptive information may be copied from images or derived.

Type:

  Extensible

Order:

  Non-Significant

Table TID 1604. Image Library Entry Descriptors for Cross-Sectional Modalities

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | NUM | EV (111026, DCM, "Horizontal Pixel Spacing") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
2 |  | HAS ACQ CONTEXT | NUM | EV (111066, DCM, "Vertical Pixel Spacing") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
3 |  | HAS ACQ CONTEXT | NUM | EV (112226, DCM, "Spacing between slices") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
4 |  | HAS ACQ CONTEXT | NUM | EV (112225, DCM, "Slice Thickness") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
5 |  | HAS ACQ CONTEXT | NUM | EV (110901, DCM, "Image Position (Patient) X") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
6 |  | HAS ACQ CONTEXT | NUM | EV (110902, DCM, "Image Position (Patient) Y") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
7 |  | HAS ACQ CONTEXT | NUM | EV (110903, DCM, "Image Position (Patient) Z") | 1 | U |  | UNITS = EV (mm, UCUM, "millimeter")
8 |  | HAS ACQ CONTEXT | NUM | EV (110904, DCM, "Image Orientation (Patient) Row X") | 1 | U |  | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
9 |  | HAS ACQ CONTEXT | NUM | EV (110905, DCM, "Image Orientation (Patient) Row Y") | 1 | U |  | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
10 |  | HAS ACQ CONTEXT | NUM | EV (110906, DCM, "Image Orientation (Patient) Row Z") | 1 | U |  | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
11 |  | HAS ACQ CONTEXT | NUM | EV (110907, DCM, "Image Orientation (Patient) Column X") | 1 | U |  | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
12 |  | HAS ACQ CONTEXT | NUM | EV (110908, DCM, "Image Orientation (Patient) Column Y") | 1 | U |  | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
13 |  | HAS ACQ CONTEXT | NUM | EV (110909, DCM, "Image Orientation (Patient) Column Z") | 1 | U |  | UNITS = EV ({-1:1}, UCUM, "{-1:1}")

Content Item Descriptions

Horizontal Imager Pixel Spacing | The row (first) component of Pixel Spacing (0028,0030) in the Image IOD. See Section 10.7.1.1 “Pixel Spacing” in PS3.3 and Section C.7.6.2 “Image Plane Module” in PS3.3 .
Vertical Imager Pixel Spacing | The column (second) component of Pixel Spacing (0028,0030) in the Image IOD. See Section 10.7.1.1 “Pixel Spacing” in PS3.3 and Section C.7.6.2 “Image Plane Module” in PS3.3 .

### TID 1605 Image Library Entry Descriptors for CT

This template contains selected attributes for a CT image or group of such images. The descriptive information may be copied from images or derived.

Type:

  Extensible

Order:

  Non-Significant

Table TID 1605. Image Library Entry Descriptors for CT

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | CODE | EV (113820, DCM, "CT Acquisition Type") | 1 | U |  | DCID 10013 “CT Acquisition Type”
2 |  | HAS ACQ CONTEXT | CODE | EV (113961, DCM, "Reconstruction Algorithm") | 1 | U |  | DCID 10033 “CT Reconstruction Algorithm”

Content Item Descriptions

CT Acquisition Type | A code derived from the value of Acquisition Type (0018,9302) in the Image IOD. See Section C.8.15.3.2 “CT Acquisition Type Macro” in PS3.3 .
Reconstruction Algorithm | A code derived from the value of Reconstruction Algorithm (0018,9315) in the Image IOD. See Section C.8.15.3.7 “CT Reconstruction Macro” in PS3.3 .

### TID 1606 Image Library Entry Descriptors for MR

This template contains selected attributes for a MR image or group of such images. The descriptive information may be copied from images or derived.

Type:

  Extensible

Order:

  Non-Significant

Table TID 1606. Image Library Entry Descriptors for MR

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | TEXT | EV (110909, DCM, "Pulse Sequence Name") | 1 | U |  |

Content Item Descriptions

Pulse Sequence Name | The value of Pulse Sequence Name (0018,9005) or Sequence Name (0018,0024) in the Image IOD. See Section C.8.13.4 “MR Pulse Sequence Module” in PS3.3 .

### TID 1607 Image Library Entry Descriptors for PET

This template contains selected attributes for a PET image or group of such images. The descriptive information may be copied from images or derived.

### Note

The content of this template is similar to that in TID 15101 NM/PET Protocol Context, but is in the form of an SR template rather than a Protocol Context Template, and the content items are not nested as modifiers. There is also some similarity to TID 3307 NM/PET Perfusion Measurement Group.

Type:

  Extensible

Order:

  Non-Significant

Table TID 1607. Image Library Entry Descriptors for PET

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | CODE | EV (C-10072, SRT, "Radionuclide") | 1 | U |  | DCID 4020 “PET Radionuclide”
2 |  | HAS ACQ CONTEXT | CODE | EV (F-61FDB, SRT, "Radiopharmaceutical agent") | 1 | U |  | DCID 4021 “PET Radiopharmaceutical”
3 |  | HAS ACQ CONTEXT | NUM | EV (R-42806, SRT, "Half-life of radiopharmaceutical") | 1 | U |  | UNITS = (s, UCUM, "s")
4 |  | HAS ACQ CONTEXT | DATETIME | EV (123003, DCM, "Radiopharmaceutical Start Date Time") | 1 | U |  |
4 |  | HAS ACQ CONTEXT | DATETIME | EV (123004, DCM, "Radiopharmaceutical Stop Time") | 1 | U |  |
5 |  | HAS ACQ CONTEXT | NUM | EV (123005, DCM, "Radiopharmaceutical Volume") | 1 | U |  | UNITS = DT (cm3, UCUM, "cm3")
6 |  | HAS ACQ CONTEXT | NUM | EV (123006, DCM, "Radionuclide Total Dose") | 1 | U |  | UNITS = DT (Bq, UCUM, "Bq")
7 |  | HAS ACQ CONTEXT | NUM | EV (123007, DCM, "Radiopharmaceutical Specific Activity") | 1 | U |  | UNITS = DT (Bq/mol, UCUM, "Bq/mol")
8 |  | HAS ACQ CONTEXT | CODE | EV (G-C340, SRT, "Route of Administration") | 1 | U |  | BCID 11 “Route of Administration”
9 |  | HAS ACQ CONTEXT | NUM | EV (123009, DCM, "Radionuclide Syringe Counts") | 1 | U |  | UNITS = DT ({counts}/s, UCUM "counts/s")
10 |  | HAS ACQ CONTEXT | NUM | EV (123010, DCM, "Radionuclide Residual Syringe Counts") | 1 | U |  | UNITS = DT ({counts}/s, UCUM "counts/s")
10 |  | HAS ACQ CONTEXT | NUM | EV (126203, DCM, "PET Radionuclide Incubation Time") | 1 | U |  | UNITS = EV (min, UCUM, "min")
12 |  | HAS ACQ CONTEXT | NUM | EV (14749-6, LN, "Glucose") | 1 | U |  | UNITS = EV (mmol/l, UCUM, "mmol/l")
13 |  | HAS ACQ CONTEXT | DATE | EV (109081, DCM, "Glucose Measurement Date") | 1 | M |  |
14 |  | HAS ACQ CONTEXT | TIME | EV (109082, DCM, "Glucose Measurement Time") | 1 | M |  |

Content Item Descriptions

Half-life of radiopharmaceutical | The units for half life are chosen to be seconds, to match the units used for Radionuclide Half Life (0018,1075). See Section C.8.9.2 “PET Isotope Module” in PS3.3 .

### TID 2000 Basic Diagnostic Imaging Report

Basic report template for general diagnostic imaging interpretation reports.

Can only be instantiated at the root node and cannot be included in other templates.

Type:

  Non-Extensible

Order:

  Significant

Root:

  Yes

Table TID 2000. Basic Diagnostic Imaging Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | BCID 7000 “Diagnostic Imaging Report Document Titles” | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1-n | U |  |
3 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
4 | > | HAS CONCEPT MOD | INCLUDE | DTID 1210 “Equivalent Meaning(s) of Concept Name” | 1-n | U |  |
5 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
6 | > | CONTAINS | CONTAINER | BCID 7001 “Diagnostic Imaging Report Headings” | 1-n | U |  |
7 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
8 | >> |  | INCLUDE | DTID 2002 “Report Narrative” | 1 | M |  |

No content items other than those defined in Observation Context TID 1001 “Observation Context” may be the target of a HAS OBS CONTEXT relationship when TID 2000 “Basic Diagnostic Imaging Report” is invoked.

### TID 2001 Basic Diagnostic Imaging Report Observations

Individual numeric or image observations that may be useful for inclusion as individual findings or as the source of inferences in a report.

Type:

  Non-Extensible

Order:

  Significant

Table TID 2001. Basic Diagnostic Imaging Report Observations

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | IMAGE | BCID 7003 “Diagnostic Imaging Report Purposes of Reference” | 1 | MC | XOR Rows 2, 3, 4, 5. |
2 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1 | MC | XOR Rows 1, 3, 4, 5. Shall not be present if the NUM value type is not supported by the IOD. |
3 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1 | MC | XOR Rows 1, 2, 4, 5. Shall not be present if the NUM value type is not supported by the IOD. |
4 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1 | MC | XOR Rows 1, 2, 3, 5. Shall not be present if the NUM value type is not supported by the IOD. |
5 |  |  | INCLUDE | DTID 1404 “Numeric Measurement” | 1 | MC | XOR Rows 1, 2, 3, 4. Shall not be present if the NUM value type is not supported by the IOD. |

### TID 2002 Report Narrative

The Report Narrative allows recording of text, code, and numeric observations. The order of Content Items in the template is not significant; the order of Content Items in a SOP Instance may be significant to the narrative flow of the report.

Type:

  Non-Extensible

Order:

  Non-Significant

Table TID 2002. Report Narrative

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CODE | BCID 7002 “Diagnostic Imaging Report Elements” | 1-n | U |  |
2 | > | INFERRED FROM | INCLUDE | DTID 2001 “Basic Diagnostic Imaging Report Observations” | 1-n | U |  |
3 |  | CONTAINS | TEXT | BCID 7002 “Diagnostic Imaging Report Elements” | 1-n | U |  |
4 | > | INFERRED FROM | INCLUDE | DTID 2001 “Basic Diagnostic Imaging Report Observations” | 1-n | U |  |
5 |  | CONTAINS | INCLUDE | DTID 2001 “Basic Diagnostic Imaging Report Observations” | 1-n | U |  |

### TID 2005 Transcribed Diagnostic Imaging Report

Basic report template for general diagnostic imaging interpretation reports produced in a dictation/transcription workflow. SR documents encoded using this template are intended to be transformable to HL7 Clinical Document Architecture format (see Section X.3 “Transcribed Diagnostic Imaging CDA Instance Content” in PS3.17 ).

This template can be instantiated only at the root node, and cannot be included in other templates.

Observation Context shall be inherited from outside the SR Content tree, and shall not be changed within the Content tree. To satisfy the requirement that Observer Context is inherited, either or both the Author Observer Sequence (0040,A078) or the Verifying Observer Sequence (0040,A073) from the SR Document Module must be present in the SOP Instance.

### Note

See Section C.17.5 “Observation Context Encoding” in PS3.3 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  Yes

Table TID 2005. Transcribed Diagnostic Imaging Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | BCID 7000 “Diagnostic Imaging Report Document Titles” | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121049, DCM, "Language of Content Item and Descendants") | 1 | M |  | DCID 5000 “Languages”
3 | > | CONTAINS | CONTAINER | BCID 7001 “Diagnostic Imaging Report Headings” | 1-n | M |  |
4 | >> | CONTAINS | TEXT | BCID 7002 “Diagnostic Imaging Report Elements” | 1 | U |  |
5 | > | CONTAINS | CONTAINER | EV (55113-5, LN, "Key Images") | 1-n | U |  |
6 | >> | CONTAINS | TEXT | EV (113012, DCM, "Key Object Description") | 1 | U |  |
7 | >> | CONTAINS | IMAGE | Purpose of Reference is not used | 1-n | M |  |

Content Item Descriptions

Row 3 | CONTAINER Concept Name may be absent.
Row 7 | IMAGE Concept Name shall be absent

### TID 2006 Imaging Report With Conditional Radiation Exposure and Protection Information

This template is used for general imaging reports for both radiation producing and non-radiation producing modalities.

For radiation producing modalities, radiation exposure and protection information is required, such as to support nationally-specific legal or standard requirements.

It contains mandatory sections, each of which may appear only once in objects instantiated from the template, including the medical content of the report that comprises relevant medical history data, information on the current request (i.e., clinical question that is expected to be answered by the requested procedure), impressions on the current imaging procedure that has been performed, and radiation exposure and protection information.

This template is a specialization of TID 2000 “Basic Diagnostic Imaging Report”, in that it uses the same structure of headings and content, but mandates the presence and order of specific headings, and extends the subordinate content with specific content items.

Type:

  Non-Extensible

Order:

  Non-Significant

Root:

  Yes

Table TID 2006. Imaging Report With Conditional Radiation Exposure and Protection Information

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | BCID 7000 “Diagnostic Imaging Report Document Titles” | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1-n | U |  |
3 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
4 | > | HAS CONCEPT MOD | INCLUDE | DTID 1210 “Equivalent Meaning(s) of Concept Name” | 1-n | U |  |
5 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
6 | > | CONTAINS | CONTAINER | EV (55111-9, LN, "Current Procedure Descriptions") | 1 | M |  |
7 | >> |  | INCLUDE | DTID 2007 “Imaging Procedure Description” | 1 | M |  |
8 | > | CONTAINS | CONTAINER | EV (55114-3, LN, "Prior Procedure Descriptions") | 1-n | MC | IF relevant prior procedures have been performed. |
9 | >> |  | INCLUDE | DTID 2007 “Imaging Procedure Description” | 1 | M |  |
10 | > | CONTAINS | CONTAINER | EV (11329-0, LN, "History") | 1 | M |  |
11 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
12 | >> |  | INCLUDE | DTID 2002 “Report Narrative” | 1 | M |  |
13 | > | CONTAINS | CONTAINER | EV (55115-0, LN, "Request") | 1 | M |  |
14 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
15 | >> |  | INCLUDE | DTID 2002 “Report Narrative” | 1 | M |  |
16 | > | CONTAINS | CONTAINER | EV (19005-8, LN, "Impressions") | 1 | M |  |
17 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
18 | >> |  | INCLUDE | DTID 2002 “Report Narrative” | 1 | M |  |
19 | > | CONTAINS | INCLUDE | DTID 2008 “Radiation Exposure and Protection Information” | 1 | MC | IF the current procedure exposes the patient to ionizing radiation. |
20 | > | CONTAINS | CONTAINER | BCID 7001 “Diagnostic Imaging Report Headings” | 1-n | U |  |
21 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
22 | >> |  | INCLUDE | DTID 2002 “Report Narrative” | 1 | M |  |

No content items other than those defined in TID 1001 “Observation Context” may be the target of a HAS OBS CONTEXT relationship when TID 2006 “Imaging Report With Conditional Radiation Exposure and Protection Information” is invoked.

Content Item Descriptions

Row 2 | Even though this information is related to the content of Row 6 in TID 2007 “Imaging Procedure Description”, it is present here for consistency with other report templates.
Rows 5, 6 | Information on at least one of the following person observers is mandatory: 1) "Performing Physician" 2) "Performing Technologist". (For those person observers, requirement types as specified in TID 1003 “Person Observer Identifying Attributes” apply. That means that "Person Observer Name" is the only mandatory attribute).
Row 20 | Each heading (concept code from CID 7001 “Diagnostic Imaging Report Headings”) may appear only once, and may not repeat the headings (concept codes) used when instantiating any other rows of this template.

### TID 2007 Imaging Procedure Description

Contains information related to the procedure.

Type:

  Extensible

Order:

  Non-Significant

Table TID 2007. Imaging Procedure Description

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
2 |  | CONTAINS | TEXT | EV (123014, DCM, "Target Region") | 1 | MC | XOR with Row 3 |
3 |  | CONTAINS | CODE | EV (123014, DCM, "Target Region") | 1 | MC | XOR with Row 2 | DCID 4028 “Craniofacial Anatomic Regions” , DCID 4030 “CT, MR and PET Anatomy Imaged” , DCID 4031 “Common Anatomic Regions”
4 |  | CONTAINS | TEXT | EV (121065, DCM, "Procedure Description") | 1 | M |  |
5 |  | CONTAINS | DATE | EV (111060, DCM, "Study Date") | 1 | M |  | Shall be equal to the Study Date (0020,0020) in the General Study Module in the images to which this report applies.
6 |  | CONTAINS | TIME | EV (111061, DCM, "Study Time") | 1 | U |  | If present, shall be equal to the Study Time (0020,0030) in the General Study Module in the images to which this report applies.
7 |  | CONTAINS | COMPOSITE | EV (113701, DCM, "X-Ray Radiation Dose Report") | 1-n | U |  |

### TID 2008 Radiation Exposure and Protection Information

Contains information related to the radiation exposure and protection of the patient, as is required by national legal requirements or standards.

Other information about the current procedure is described in TID 2006 “Imaging Report With Conditional Radiation Exposure and Protection Information” and not repeated here.

Type:

  Extensible

Order:

  Non-Significant

Table TID 2008. Radiation Exposure and Protection Information

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (73569-6, LN, "Radiation Exposure and Protection Information") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | U |  |
3 | > | CONTAINS | CODE | EV (111532, DCM, "Pregnancy Status") | 1 | MC | IF female patient of child-bearing age | DCID 6096 “Pregnancy Status” Pregnancy Status
4 | > | CONTAINS | TEXT | EV (18785-6, LN, "Indications for Procedure") | 1 | M |  |
5 | > | CONTAINS | PNAME | EV (113850, DCM, "Irradiation Authorizing ") | 1 | M |  |
6 | > | CONTAINS | TEXT | EV (113921, DCM, "Radiation Exposure") | 1 | MC | IFF ionizing radiation is applied in the context of the current procedure |
7 | > | CONTAINS | TEXT | EV (113922, DCM, "Radioactive Substance Administered") | 1 | MC | IFF radioactive substance is administered in the context of the current procedure |

Content Item Descriptions

Row 5 | The clinician responsible for determining that the irradiating procedure was appropriate for the indications.
Row 6 | A textual, human-readable description of the radiation exposure is all that is required by this template (such as is sufficient to comply with, for example, German law). Detailed specification of exposure is out of the scope of this template. Such information may be given in a separate SR instances such as described in TID 10001 “Projection X-Ray Radiation Dose” or TID 10011 “CT Radiation Dose”, and referenced from TID 2007 “Imaging Procedure Description”.

### TID 2010 Key Object Selection

The Key Object Selection template is intended for flagging one or more significant images, waveforms, or other composite SOP Instances. Key Object Selection contains:

- coded document title stating the reason for significance of the referenced objects in the Key Object Selection,

- optional free form text comment in an explicitly identified language, and

- optional identification of the observer (device or person) that created the Key Object Selection.

### Note

1. For instance, when this template is used to identify images rejected for quality reasons, the device or person performing the quality assessment is identified in observation context items (invoked through TID 1002 “Observer Context”). The reason for rejection can be included both as a code used as a concept modifier for the document title, and as text description.

2. The order of object references may be significant, e.g., when the title concept is "For Conference".

3. Instances referenced in a Key Object Selection Document may be securely referenced by Digital Signature or MAC mechanisms within the SR Document General Module (see PS3.3).

The Template can only be instantiated at the root node and cannot be included in other templates. The Template is not extensible; that is, no other content items may be added to this template, or the templates that are included, recursively.

Type:

  Non-Extensible

Order:

  Non-Significant

Root:

  Yes

Table TID 2010. Key Object Selection

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DCID 7010 “Key Object Selection Document Title” | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (113011, DCM, "Document Title Modifier") | 1-n | U |  |
3 | > | HAS CONCEPT MOD | CODE | EV (113011, DCM, "Document Title Modifier") | 1 | UC | IF Row 1 Concept Name = (113001, DCM, "Rejected for Quality Reasons") or (113010, DCM, "Quality Issue") | DCID 7011 “Rejected for Quality Reasons”
4 | > | HAS CONCEPT MOD | CODE | EV (113011, DCM, "Document Title Modifier") | 1 | MC | IF Row 1 Concept Name = (113013, DCM, "Best In Set") | DCID 7012 “Best in Set”
5 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
6 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
7 | > | CONTAINS | TEXT | EV (113012, DCM, "Key Object Description") | 1 | U |  |
8 | > | CONTAINS | IMAGE | Purpose of Reference shall not be present | 1-n | MC | At least one of Rows 8, 9 and 10 shall be present |
9 | > | CONTAINS | WAVEFORM | Purpose of Reference shall not be present | 1-n | MC | At least one of Rows 8, 9 and 10 shall be present |
10 | > | CONTAINS | COMPOSITE | Purpose of Reference shall not be present | 1-n | MC | At least one of Rows 8, 9 and 10 shall be present |

### TID 2020 Spectacle Prescription Report

The Spectacle Prescription Report is a structured report used to represent the prescription for a patient. Usually a prescription is for both eyes, but sometimes just one. The Spectacle Prescription Report defines a refractive correction relative to which visual acuity may be measured subjectively, and thus may be referenced by a Visual Acuity Measurements Storage SOP Instance.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 2020. Spectacle Prescription Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111671, DCM, "Spectacle Prescription Report") | 1 | M |  | Root node
2 | > | CONTAINS | CONTAINER | EV (111688, DCM, "Right Eye Rx") | 1 | UC | IF Right Spectacle Lens is prescribed |
3 | >> | CONTAINS | INCLUDE | DTID 2021 “Template for Spectacle Prescription Details” | 1 | M |  |
4 | > | CONTAINS | CONTAINER | EV (111689, DCM, "Left Eye Rx") | 1 | UC | IF Left Spectacle Lens is prescribed |
5 | >> | CONTAINS | INCLUDE | DTID 2021 “Template for Spectacle Prescription Details” | 1 | M |  |
6 | > | CONTAINS | NUM | EV (111679, DCM, "Distance Pupillary Distance") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
7 | > | CONTAINS | NUM | EV (111680, DCM, "Near Pupillary Distance") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
8 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

### TID 2021 Template for Spectacle Prescription Details

Type:

  Extensible

Order:

  Significant

Table TID 2021. Spectacle Prescription Details

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | NUM | EV (F-02FB4, SRT, "Sphere") | 1 | M |  | UNITS = EV ([diop], UCUM, "diopters")
2 |  | CONTAINS | NUM | EV (F-A2143, SRT, "Cylinder Power") | 1 | UC | IF Cylinder is prescribed | UNITS = EV ([diop], UCUM, "diopters")
3 |  | CONTAINS | NUM | EV (F-02FB7, SRT, "Axis") | 1 | MC | IF ROW 2 is Present | UNITS = EV (deg, UCUM, "degrees")
4 |  | CONTAINS | NUM | EV (111672, DCM, "Add Near") | 1 | UC | IF Add Near is prescribed | UNITS = EV ([diop], UCUM, "diopters")
5 |  | CONTAINS | NUM | EV (111673, DCM, "Add Intermediate") | 1 | UC | IF Add Intermediate is prescribed | UNITS = EV ([diop], UCUM, "diopters")
6 |  | CONTAINS | NUM | EV (111674, DCM, "Add Other") | 1 | UC | IF Add Other is prescribed | UNITS = EV ([diop], UCUM, "diopters")
7 |  | CONTAINS | NUM | EV (111675, DCM, "Horizontal Prism Power") | 1 | UC | IF Horizontal Prism is prescribed | UNITS = EV ([p'diop], UCUM, "prism diopters")
8 |  | CONTAINS | CODE | EV (111676, DCM, "Horizontal Prism Base") | 1 | MC | IF ROW 7 is present | DCID 4214 “Ophthalmic Horizontal Directions”
9 |  | CONTAINS | NUM | EV (111677, DCM, "Vertical Prism Power") | 1 | UC | IF Vertical Prism is prescribed | UNITS = EV ([p'diop], UCUM, "prism diopters")
10 |  | CONTAINS | CODE | EV (111678, DCM, "Vertical Prism Base") | 1 | MC | IF Row 9 is present | DCID 4215 “Ophthalmic Vertical Directions”

### TID 2100 Macular Grid Thickness and Volume Report

The Macular Grid Thickness and Volume Report is a structured report encoding the macular grid thickness and volume values derived from ophthalmic images, such as ophthalmic OPT images. This may encode measurements of either or both eyes.

The macular grid conveyed by this report is based upon the grid employed by the Early Treatment of Diabetic Retinopathy Study (ETDRS) to measure area and proximity of macular edema to the anatomic center (fovea) of the macula. See ETDRS Report Number 10.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 2100. Macular Grid Thickness and Volume Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111690, DCM, "Macular Grid Thickness and Volume Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 2101 “Macular Grid Thickness and Volume Measurement” | 1 | MC | IF Row 5 is absent. | $Laterality = EV (G-A100, SRT, "Right")
5 | > | CONTAINS | INCLUDE | DTID 2101 “Macular Grid Thickness and Volume Measurement” | 1 | MC | IF Row 4 is absent. | $Laterality = EV (G-A101, SRT, "Left")

### TID 2101 Macular Grid Thickness and Volume Measurement

This Template encodes the macular grid thickness and volume measurements for a single eye.

Table TID 2101. Parameters

Parameter Name | Parameter Usage
--- | ---
$Laterality | Which eye

Type:

  Extensible

Order:

  Significant

Table TID 2101. Macular Grid Thickness and Volume Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | EV (T-AA000, SRT, "Eye")
3 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | $Laterality
4 | > | CONTAINS | NUM | EV (57108-3, LN, "Macular Grid.Center Point Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
5 | > | CONTAINS | NUM | EV (57109-1, LN, "Macular Grid.Center Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
6 | > | CONTAINS | NUM | EV (57110-9, LN, "Macular Grid.Inner Superior Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
7 | > | CONTAINS | NUM | EV (57111-7, LN, "Macular Grid.Inner Nasal Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
8 | > | CONTAINS | NUM | EV (57112-5, LN, "Macular Grid.Inner Inferior Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
9 | > | CONTAINS | NUM | EV (57113-3, LN, "Macular Grid.Inner Temporal Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
10 | > | CONTAINS | NUM | EV (57114-1, LN, "Macular Grid.Outer Superior Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
11 | > | CONTAINS | NUM | EV (57115-8, LN, "Macular Grid.Outer Nasal Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
12 | > | CONTAINS | NUM | EV (57116-6, LN, "Macular Grid.Outer Inferior Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
13 | > | CONTAINS | NUM | EV (57117-4, LN, "Macular Grid.Outer Temporal Subfield Thickness") | 1 | M |  | UNITS = EV (um, UCUM, "micrometer")
14 | > | CONTAINS | NUM | EV (57118-2, LN, "Macular Grid.Total Volume") | 1 | M |  | UNITS = EV (mm3, UCUM, "mm3")
15 | > | CONTAINS | NUM | EV (111691, DCM, "Number of Images Used for Macular Measurements") | 1 | M |  | UNITS = EV ({images}, UCUM, "images")
16 | > | CONTAINS | NUM | EV (111692, DCM, "Number of Samples Used per Image") | 1 | M |  | UNITS = EV ({samples}, UCUM, "samples")
17 | > | CONTAINS | NUM | EV (111693, DCM, "Analysis Quality Rating") | 1 | M |  | UNITS = EV ({0:100}, UCUM, "range:0:100") Value = 0 - 100
18 | >> | HAS OBS CONTEXT | INCLUDE | DTID 2102 “Quality Rating Identification” | 1 | M |  |
19 | > | CONTAINS | NUM | EV (111694, DCM, "Image Set Quality Rating") | 1 | M |  | UNITS = EV ({0:100}, UCUM, "range:0:100") Value = 0 - 100
20 | >> | HAS OBS CONTEXT | INCLUDE | DTID 2102 “Quality Rating Identification” | 1 | M |  |
21 | > | CONTAINS | NUM | EV (111029, DCM, "Image Quality Rating") | 1-n | U |  | UNITS = EV ({0:100}, UCUM, "range:0:100") Value = 0 - 100
22 | >> | INFERRED FROM | IMAGE | No purpose of reference | 1 | M |  |
23 | >> | HAS OBS CONTEXT | INCLUDE | DTID 2102 “Quality Rating Identification” | 1 | M |  |
24 | > | CONTAINS | CODE | EV (111696, DCM, "Visual Fixation Quality During Acquisition") | 1 | U |  | DCID 4220 “Visual Fixation Quality During Acquisition”
25 | >> | HAS CONCEPT MOD | CODE | EV (111697, DCM, "Visual Fixation Quality Problem") | 1-n | U |  | DCID 4221 “Visual Fixation Quality Problem”
26 | > | CONTAINS | CODE | EV (111698, DCM, "Ophthalmic Macular Grid Problem") | 1-n | U |  | DCID 4222 “Ophthalmic Macular Grid Problem”
27 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

### TID 2102 Quality Rating Identification

This template specifies the algorithm (and parameters) used to create a quality rating for an image or image set.

It is expected that the identified algorithm will create a consistent quality rating when analyzing a given image. If the algorithm allows change to its parameters that would alter the quality rating created, the specific parameters used should be specified.

Type:

  Non-Extensible

Order:

  Significant

Table TID 2102. Quality Rating Identification

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (111001, DCM, "Algorithm Name") | 1 | M |  |
2 |  |  | TEXT | EV (111003, DCM, "Algorithm Version") | 1 | M |  |
3 |  |  | TEXT | EV (122405, DCM, "Algorithm Manufacturer") | 1 | M |  |
4 |  |  | TEXT | EV (111002, DCM, "Algorithm Parameters") | 1-n | U |  |

## Procedure Log IOD Templates

### TID 3001 Procedure Log

The Procedure Log template is intended for the representation of reports or logs of time-stamped events occurring during an image-guided interventional or other procedure.

This Template does not require a particular ordering of the subsidiary Content Items.

### Note

1. The Procedure Log IOD (PS3.3) requires ordering by Observation DateTime; thus log entries of different types (i.e., specified by different Rows in the Template) may appear in any order.

2. While this Template is extensible, the Procedure Log IOD forbids Container Content Items subsidiary to the top level Container.

Type:

  Extensible

Order:

  Non-Significant

Root:

  Yes

Table TID 3001. Procedure Log

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DCID 3400 “Procedure Log Titles” | 1 | M |  | Root node
2 | > |  | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
3 | > |  | INCLUDE | DTID 3601 “Procedure Context” | 1 | M |  |
4 | > | HAS ACQ CONTEXT | TEXT | EV (121121, DCM, "Room identification") | 1 | U |  |
5 | > | HAS ACQ CONTEXT | TEXT | EV (121122, DCM, "Equipment identification") | 1-n | U |  |
6 | > | CONTAINS | TEXT | DCID 3401 “Types of Log Notes” | 1-n | U |  |
7 | >> |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
8 | > | CONTAINS | CODE | EV (121123, DCM, "Patient Status or Event") | 1-n | U |  | DCID 3402 “Patient Status and Events”
9 | >> |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
10 | > | CONTAINS | PNAME | DCID 3404 “Staff Actions” | 1-n | U |  |
11 | >> |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
12 | > | CONTAINS | TEXT | DCID 3427 “Equipment Events” | 1-n | U |  | Equipment identifier
13 | >> |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
14 | > | CONTAINS | INCLUDE | DTID 3100 “Procedure Action” | 1-n | U |  |
15 | > | CONTAINS | INCLUDE | DTID 3101 “Image Acquisition” | 1-n | U |  |
16 | > | CONTAINS | INCLUDE | DTID 3102 “Waveform Acquisition” | 1-n | U |  |
17 | > | CONTAINS | INCLUDE | DTID 3103 “Referenced Object” | 1-n | U |  |
18 | > | CONTAINS | INCLUDE | DTID 3104 “Consumables” | 1-n | U |  |
19 | > | CONTAINS | INCLUDE | DTID 3105 “Lesion Identification and Properties” | 1-n | U |  |
20 | > | CONTAINS | INCLUDE | DTID 3106 “Drugs/Contrast Administered” | 1-n | U |  |
21 | > | CONTAINS | INCLUDE | DTID 3107 “Device Used” | 1-n | U |  |
22 | > | CONTAINS | INCLUDE | DTID 3108 “Intervention” | 1-n | U |  |
23 | > | CONTAINS | CODE | EV (DD-60002, SRT, "Complication of Procedure") | 1-n | U |  | DCID 3413 “Adverse Outcomes”
24 | >> |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
25 | > | CONTAINS | INCLUDE | DTID 3109 “Measurements” | 1-n | U |  |
26 | > | CONTAINS | INCLUDE | DTID 3110 “Impressions or Findings” | 1-n | U |  |
27 | > | CONTAINS | INCLUDE | DTID 3111 “Percutaneous Entry” | 1-n | U |  |
28 | > | CONTAINS | INCLUDE | DTID 3112 “Specimen Obtained” | 1-n | U |  |
29 | > | CONTAINS | INCLUDE | DTID 3113 “Patient Support” | 1-n | U |  |
30 | > | CONTAINS | INCLUDE | DTID 3114 “Patient Assessment” | 1-n | U |  |
31 | > | CONTAINS | INCLUDE | DTID 3115 “ECG ST Assessment” | 1-n | U |  |

Content Item Descriptions

Row 2 | Includes TID 1002 “Observer Context”, which shall be used to record the identity of the person responsible for recording the log, as well as all other participants in the procedure, even though these personnel may not technically be "observers" of the Procedure Log. As participants in the procedure, they are potential sources for events and observations recorded in the Log. TID 1002 “Observer Context” allows the specification of the person's role in the organization (e.g., physician, nurse), as well as the role in the procedure (e.g., circulating, performing, etc.).
Row 5 | Shall be used to record the identity of the major equipment used in the procedure.
Row 6 | May be used to record any event not covered by a specific log entry template.

### TID 3010 Log Entry Qualifiers

The Log Entry Qualifiers Template provides a common means for adding additional description to a procedure log content item. It allows identification of a source for the procedure log entry (other than the recording observer for the log as a whole), a free text comment, a link to a particular Procedure Action item, a link to a particular lesion, or the date/time of recording (if different than the time of the event occurrence recorded in the Observation DateTime of the parent content item).

Type:

  Extensible

Order:

  Significant

Table TID 3010. Log Entry Qualifiers

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
2 |  | HAS PROPERTIES | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
3 |  | HAS OBS CONTEXT | TEXT | EV (121124, DCM, "Procedure Action ID") | 1-n | U |  |
4 |  | HAS OBS CONTEXT | TEXT | EV (121151, DCM, "Lesion Identifier") | 1-n | U |  | Up to 3 numeric characters
5 |  | HAS OBS CONTEXT | DATETIME | EV (121125, DCM, "DateTime of Recording of Log Entry") | 1 | U |  |
6 |  | INFERRED FROM | IMAGE |  | 1-n | U |  |
7 |  | INFERRED FROM | WAVEFORM |  | 1-n | U |  |
8 |  | INFERRED FROM | COMPOSITE |  | 1-n | U |  |
9 |  | HAS OBS CONTEXT | CODE | EV (121135, DCM, "Observation DateTime Qualifier") | 1 | U |  | DCID 3430 “DateTime Qualifiers”

Content Item Descriptions

Row 3 | Procedure Action ID allows linking recorded events to a particular action, step, or phase of a procedure. See description for TID 3100 “Procedure Action”.
Row 4 | Lesion Identifier is specified as a numeric text string, and allows linking recorded events to the diagnosis or therapy of particular lesion. See description for TID 3105 “Lesion Identification and Properties”.

### TID 3100 Procedure Action

The Procedure Action Template is intended for the recording of the beginning or end of procedure steps or action items in a procedure. The level of granularity of the recorded events is not specified, and may vary between institutions, or even be at multiple levels within a single procedure log. There is no requirement for the real-world procedure step or action item recorded with this template to end before another one begins; there may be overlapping or simultaneous procedure steps or action items.

This log entry template may be used to record the start or stop of timers.

Other recorded events in the procedure may be linked to a particular step or action item by Procedure Action ID (see TID 3010 “Log Entry Qualifiers”).

Type:

  Extensible

Order:

  Significant

Table TID 3100. Procedure Action

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DCID 3421 “Procedure Action” | 1 | M |  | BCID 3405 “Procedure Action Values”
2 | > | HAS PROPERTIES | TEXT | EV (121124, DCM, "Procedure Action ID") | 1 | M |  |
3 | > | HAS PROPERTIES | PNAME | BCID 7453 “Performing Roles” | 1-n | U |  |
4 | > | HAS PROPERTIES | NUM | EV (121128, DCM, "Procedure Action Duration") | 1 | U |  |
5 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
6 | > | HAS PROPERTIES | UIDREF | EV (121126, DCM, "Performed Procedure Step SOP Instance UID") | 1 | MC | IFF a Performed Procedure Step SOP Class is used to provide status of the Procedure Step |
7 | > | HAS PROPERTIES | UIDREF | EV (121127, DCM, "Performed Procedure Step SOP Class UID") | 1 | MC | IFF a Performed Procedure Step SOP Class is used to provide status of the Procedure Step |

Content Item Descriptions

Row 2 | The value of the Procedure Action ID shall be uniquely associated with the step or action within the context of the Study, and may be used to associate various Procedure Log entries with the step or action.
Row 3 | May be used to record the identity of staff roles for the purpose of this Procedure Action, which may differ from their roles in the procedure as a whole.

### TID 3101 Image Acquisition

The Image Acquisition Template allows recording of the essential parameters of a digital image acquired during the procedure.

Type:

  Extensible

Order:

  Significant

Table TID 3101. Image Acquisition

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | IMAGE | EV (121138, DCM, "Image Acquired") | 1 | M |  |
2 | > | HAS ACQ CONTEXT | UIDREF | EV (112002, DCM, "Series Instance UID") | 1 | M |  |
3 | > | HAS ACQ CONTEXT | CODE | EV (121139, DCM, "Modality") | 1 | M |  | DCID 29 “Acquisition Modality” Derived from referenced image SOP Instance attribute (0008,0060)
4 | > | HAS PROPERTIES | NUM | EV (121140, DCM, "Number of Frames") | 1 | U |  |
5 | > | HAS PROPERTIES | TEXT | EV (121141, DCM, "Image Type") | 1 | U |  | From referenced image SOP Instance attribute (0008,0008)
6 | > | HAS ACQ CONTEXT | NUM | EV (112011, DCM, "Positioner Primary Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
7 | > | HAS ACQ CONTEXT | NUM | EV (112012, DCM, "Positioner Secondary Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
8 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3102 Waveform Acquisition

The Waveform Acquisition Template allows recording of the essential parameters of a digital waveform acquired during the procedure.

Type:

  Extensible

Order:

  Significant

Table TID 3102. Waveform Acquisition

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | WAVEFORM | EV (121143, DCM, "Waveform Acquired") | 1 | M |  |
2 | > | HAS ACQ CONTEXT | CODE | EV (121139, DCM, "Modality") | 1 | M |  | DCID 29 “Acquisition Modality” Derived from referenced waveform SOP Instance attribute (0008,0060)
3 | > | HAS ACQ CONTEXT | NUM | EV (121142, DCM, "Acquisition Duration") | 1 | U |  |
4 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3103 Referenced Object

The Referenced Object Template allows reference to measurement or report objects, such as prior medical reports, laboratory results, hemodynamic measurement reports, or quantitative analysis reports.

Type:

  Extensible

Order:

  Significant

Table TID 3103. Referenced Object

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | COMPOSITE | BCID 3407 “Purpose of Reference to Object” | 1 | M |  |
2 | > | HAS PROPERTIES | CODE | EV (121144, DCM, "Document Title") | 1 | MC | IFF Row 1 references an SR object | Root node concept of referenced SR object
3 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3104 Consumables

The Consumables Template allows recording of devices (e.g., catheters or stents), drugs, or contrast agents accessed in a procedure. This content item is directed towards inventory control and billing. The actual clinical use of the particular consumable is recorded using TID 3106 “Drugs/Contrast Administered” or TID 3107 “Device Used”.

This template allows recording both consumable retrieval from, and return to, inventory or stock, and disposal of used material. The quantity involved in each recorded transaction may be specified.

Type:

  Extensible

Order:

  Significant

Table TID 3104. Consumables

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DCID 3408 “Actions With Consumables” | 1 | M |  | Vendor or local bar coded values
2 | > | HAS PROPERTIES | TEXT | DCID 3426 “Consumables Parameters” | 1-n | U |  |
3 | > | HAS PROPERTIES | NUM | EV (121146, DCM, "Quantity of Material") | 1 | U |  |
4 | > | HAS PROPERTIES | CODE | EV (121147, DCM, "Billing Code") | 1 | U |  | local billing codes
5 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3105 Lesion Identification and Properties

The Lesion Identification and Properties Template allows recording the identification of each lesion addressed in a procedure. The lesion identifier may be used to relate diagnostic or therapeutic actions with their target lesion (see Row 4 in TID 3010 “Log Entry Qualifiers”). This content item may include the initial visually estimated measurements of stenosis or TIMI flow; measured values from a quantitative measurement report may be referenced indirectly (through TID 3103 “Referenced Object”), or by quotation (TID 3109 “Measurements”). Subsequent (e.g., post-intervention) stenosis measurements may be encoded using TID 3109 “Measurements”, with the Lesion Identifier conveyed through its subsidiary TID 3010 “Log Entry Qualifiers” template.

Type:

  Extensible

Order:

  Significant

Table TID 3105. Lesion Identification and Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (121151, DCM, "Lesion Identifier") | 1 | M |  | Up to 3 numeric characters
2 | > | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3604 “Arterial Lesion Locations”
3 | >> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
4 | > | HAS PROPERTIES | CODE | EV (121153, DCM, "Lesion Risk") | 1 | U |  | DCID 3418 “Lesion Risk”
5 | > | HAS PROPERTIES | NUM | EV (R-101BB, SRT, "Lumen Diameter Stenosis") | 1 | U |  | UNITS = EV (%, UCUM, "%")
6 | >> | HAS CONCEPT MOD | CODE | EV (G-72BB, SRT, "Catheterization Procedure Phase") | 1 | M |  | EV (G-7293, SRT, "Baseline Phase")
7 | >> | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | U |  | DCID 3745 “Calculation Method”
8 | > | HAS PROPERTIES | CODE | EV (122109, DCM, "Baseline TIMI Flow") | 1 | UC | IFF Row 2 specifies a coronary artery | DCID 3713 “TIMI Flow Characteristics”
9 | > | HAS PROPERTIES | CODE | EV (122131, DCM, "Degree of Thrombus") | 1 | U |  | DCID 3714 “Thrombus”
10 | > | HAS PROPERTIES | CODE | EV (F-01740, SRT, "Lesion Margin Characteristics") | 1 | U |  | DCID 3715 “Lesion Margin”
11 | > | HAS PROPERTIES | CODE | EV (122134, DCM, "Vessel Morphology") | 1-n | U |  | DCID 3712 “Vessel Descriptors”
12 | > | HAS PROPERTIES | CODE | EV (122132, DCM, "Severity of Calcification") | 1 | U |  | DCID 3716 “Severity”
13 | > | HAS PROPERTIES | IMAGE | DT (121080, DCM, "Best illustration of finding") | 1 | U |  |
14 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

Content Item Descriptions

Row 1 | Lesion Identifier is specified as a numeric text string in order to facilitate transcoding to DICOM Attribute (0018,3105) Lesion Number and to formats for outcomes registries, such as the ACC National Cardiovascular Data Registry™.

### TID 3106 Drugs/Contrast Administered

The Drugs/Contrast Administered Template allows the recording of the start or end of that type of event, together with its parameters. If start and end are represented by a single log entry (e.g., for an injection), the concept name "Drug/contrast administered" shall be used.

Type:

  Extensible

Order:

  Significant

Table TID 3106. Drugs/Contrast Administered

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DCID 3409 “Administration of Drugs/Contrast” | 1 | M |  | BCID 10 “Interventional Drug” or BCID 12 “Radiographic Contrast Agent”
2 | > | HAS PROPERTIES | TEXT | EV (121145, DCM, "Description of Material") | 1 | U |  |
3 | > | HAS PROPERTIES | CODE | EV (G-C340, SRT, "Route of administration") | 1 | U |  | BCID 11 “Route of Administration”
4 | > | HAS PROPERTIES | NUM | DCID 3410 “Numeric Parameters of Drugs/Contrast” | 1-n | U |  |
5 | > | HAS PROPERTIES | PNAME | EV (121152, DCM, "Person administering drug/contrast") | 1 | U |  |
6 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3107 Device Used

The Device Used Template allows recording of the use of interventional diagnostic and therapeutic devices.

The identification of one device used to deploy another device (e.g., a balloon catheter to deploy a stent) may be described with two entries, with one identified as a deployment device in the Concept Modifier of Row 6 of this template, and linked by the same Procedure Action ID in the Log Entry Qualifiers of the included TID 3010 “Log Entry Qualifiers”.

Type:

  Extensible

Order:

  Significant

Table TID 3107. Device Used

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DCID 3422 “Device Use Actions” | 1 | M |  | BCID 3429 “Catheterization Devices”
2 | > | HAS PROPERTIES | CODE | EV (121150, DCM, "Device Code") | 1-n | U |  | Vendor or local bar coded values
3 | > | HAS PROPERTIES | TEXT | EV (121145, DCM, "Description of Material") | 1 | U |  |
4 | > | HAS PROPERTIES | NUM | DCID 3423 “Numeric Device Characteristics” | 1-n | U |  |
5 | > | HAS PROPERTIES | CODE | EV (G-C0E9, SRT, "Procedure site") | 1 | U |  | BCID 3630 “Cardiovascular Anatomic Locations”
6 | > | HAS CONCEPT MOD | CODE | EV (G-C0E8, SRT, "Has Intent") | 1 | U |  | DT (121155, DCM, "Deployment")
7 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3108 Intervention

The Intervention Template allows recording of interventions, including atherectomy, angioplasty, stent placement, brachytherapy, etc. The record may include reference to an image that documents the intervention.

Type:

  Extensible

Order:

  Significant

Table TID 3108. Intervention

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (122090, DCM, "Intervention Action") | 1 | M |  | DCID 3412 “Intervention Actions and Status”
2 | > | HAS PROPERTIES | CODE | EV (G-C0E9, SRT, "Procedure site") | 1 | M |  | DCID 3604 “Arterial Lesion Locations”
3 | >> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
4 | > | HAS PROPERTIES | TEXT | EV (121154, DCM, "Intervention attempt identifier") | 1 | M |  | Up to 3 numeric characters
5 | > | HAS PROPERTIES | CODE | EV (G-C50A, SRT, "Uses Equipment") | 1-n | U |  | BCID 3411 “Intracoronary Devices”
6 | >> | HAS CONCEPT MOD | CODE | EV (122111, DCM, "Primary Intervention Device") | 1 | MC | IF Device is Primary for this Lesion | DCID 230 “Yes-No”
7 | > | HAS PROPERTIES | NUM | DCID 3425 “Intervention Parameters” | 1-n | U |  |
8 | > | HAS PROPERTIES | IMAGE | BCID 7003 “Diagnostic Imaging Report Purposes of Reference” | 1 | U |  |
9 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

Content Item Descriptions

Row 4 | Intervention attempt Identifier is specified as a numeric text string, and shall be treated as the ordinal of the recorded attempted intervention within this procedure (i.e., "1" for the first attempted intervention, "2" for the second, etc.).

### TID 3109 Measurements

The Measurements Template allows recording of significant measurements, such as vital signs, laboratory results, hemodynamic measurements, or quantitative analysis measurements. These measurements are often quoted from another source, which would be documented in the included TID 3010 “Log Entry Qualifiers”.

Type:

  Extensible

Order:

  Significant

Table TID 3109. Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | No BCID | 1 | U |  |
2 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
3 | > | HAS PROPERTIES | INCLUDE | DTID 310 “Measurement Properties” | 1 | U |  |
4 |  |  | CODE | No BCID | 1 | U |  |
5 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3110 Impressions or Findings

The Impressions or Findings Template allows the recording of unconfirmed (provisional) impressions or findings noted during the procedure. It is not intended to convey the Cath Lab Clinical Report (the formal report from the performing physician), although it may be used (like any Procedure Log entry) for the subsequent construction of the Cath Lab Clinical Report.

A finding that is supported by a specific image frame may reference that image in the INFERRED FROM / IMAGE row of the included TID 3010 “Log Entry Qualifiers” template.

Type:

  Extensible

Order:

  Significant

Table TID 3110. Impressions or Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121071, DCM, "Finding") | 1 | U |  | BCID 3728 “Cath Findings”
2 | > | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
3 | > | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | U |  |
4 | >> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
5 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
6 |  |  | TEXT | BCID 3419 “Findings Titles” | 1 | U |  |
7 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

Content Item Descriptions

Row 3 | Finding Site has no Baseline Context ID specified. Typically terms would be drawn from coronary segments, other arterial segments, myocardial segments, etc.

### TID 3111 Percutaneous Entry

The Percutaneous Entry Template allows recording of the opening or closing of invasive access ports.

Type:

  Extensible

Order:

  Significant

Table TID 3111. Percutaneous Entry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121156, DCM, "Percutaneous Entry Action") | 1 | M |  | DCID 3403 “Percutaneous Entry”
2 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
3 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3112 Specimen Obtained

The Specimen Obtained Template allows recording of obtaining a specimen, and the identifiers for that specimen. This is particularly designed for blood samples that will be analyzed for blood oxygen-related measurements. The analysis of the sample may be recorded in one or more log entries using TID 3109 “Measurements”, or in a separate Structured Report SOP Instance referenced by a log entry using TID 3103 “Referenced Object”.

Type:

  Extensible

Order:

  Significant

Table TID 3112. Specimen Obtained

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121123, DCM, "Patient Status or Event") | 1 | M |  | DCID 3515 “Specimen Collection”
2 | > | HAS ACQ CONTEXT | CODE | EV (R-00254, SRT, "Specimen Type") | 1 | UC | IFF specimen is blood sample | DCID 3520 “Blood Source Type”
3 | > | HAS ACQ CONTEXT | CODE | EV (G-C0E9, SRT, "Procedure site") | 1 | U |  | BCID 3630 “Cardiovascular Anatomic Locations”
4 | > | HAS PROPERTIES | INCLUDE | DTID 1009 “Subject Context, Specimen” | 1 | U |  |

### TID 3113 Patient Support

The Patient Support Template allows recording of the use of various support technologies, including oxygen, ventilation, pacing, etc.

Type:

  Extensible

Order:

  Significant

Table TID 3113. Patient Support

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DCID 3530 “Oxygen Administration Actions” | 1 | U |  | DCID 3531 “Oxygen Administration”
2 | > | HAS PROPERTIES | NUM | EV (121160, DCM, "Oxygen Administration Rate") | 1 | MC | IFF Row 1 Concept is (121161, DCM, "Begin Oxygen Administration") | UNITS = DT (l/min, UCUM, "l/min")
3 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
4 |  |  | CODE | DCID 3550 “Circulatory Support Actions” | 1 | U |  | DCID 3553 “Circulatory Support”
5 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
6 |  |  | CODE | DCID 3551 “Ventilation Actions” | 1 | U |  | DCID 3554 “Ventilation”
7 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |
8 |  |  | CODE | DCID 3552 “Pacing Actions” | 1 | U |  | DCID 3555 “Pacing”
9 | > |  | INCLUDE | DTID 3010 “Log Entry Qualifiers” | 1 | U |  |

### TID 3114 Patient Assessment

The Patient Assessment Template allows recording of the assessment of the patient's cardiovascular, neurological, and/or respiratory condition. A particular use of this template is for "vital signs", which are a specific subset of mandatory patient assessment measurements.

Type:

  Extensible

Order:

  Significant

Table TID 3114. Patient Assessment

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121123, DCM, "Patient Status or Event") | 1 | M |  | DT (121165, DCM, "Patient Assessment Performed") DT (PA-00500, SRT, "Observation of Vital Signs")
2 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (F-008EC, SRT, "Systolic blood pressure") $Units = DCID 3500 “Pressure Units” $Method = BCID 3560 “Blood Pressure Methods”
3 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (F-008ED, SRT, "Diastolic blood pressure") $Units = DCID 3500 “Pressure Units”
4 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (8867-4, LN, "Heart rate") $Units = EV ({H.B.}/min, UCUM, "BPM")
5 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (8310-5, LN, "Body temperature") $Units = EV (Cel, UCUM, "C")
6 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = DCID 3526 “Blood Gas Saturation” $Units = EV (%, UCUM, "%")
7 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (F-043E7, SRT, "Respiration rate") $Units = EV (/min, UCUM, "breaths/min")
8 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1-n | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (122195, DCM, "Pulse Strength") $Method = BCID 3442 “Peripheral Pulse Methods” $TargetSite = BCID 3440 “Peripheral Pulse Locations” $Units = DT ({0:4}, UCUM, "range 0:4")
9 | > | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | MC | IF Row 1 value = (PA-00500, SRT, "Observation of Vital Signs") | $Measurement = EV (F-009EA, SRT, "Pain Score") $Units = DT ({1:10}, UCUM, "range 1:10")
10 | > | HAS PROPERTIES | CODE | DT (8884-9, LN, "Cardiac Rhythm") | 1 | U |  | BCID 3415 “Cardiac Rhythms”
11 | > | HAS PROPERTIES | CODE | DT (9304-7, LN, "Respiration Rhythm") | 1 | U |  | BCID 3416 “Respiration Rhythms”
12 | > | HAS PROPERTIES | CODE | DT (F-043E6, SRT, "Respiration Assessment") | 1 | U |  | BCID 3448 “Airway Assessment”
13 | > | HAS PROPERTIES | CODE | DT (F-046D8, SRT, "Skin condition") | 1-n | U |  | BCID 3446 “Skin Condition”
14 | > | HAS PROPERTIES | CODE | DT (F-04317, SRT, "Patient mental state assessment") | 1 | U |  | No BCID
15 | > | HAS PROPERTIES | TEXT | BCID 3441 “Patient Assessments” | 1-n | U |  |

Content Item Descriptions

Row 8 | Pulse Strength allows the assessment of the patient's pulse at multiple locations using the Topographical concept modifier. It may also be used for a single pulse strength measurement from an unspecified location, as is typical of vital signs assessments.
Row 16 | Allows free text description of patient assessments that are not expressible by coded entries of Rows 10 to 14.

### TID 3115 ECG ST Assessment

The ECG ST Assessment Template allows recording of the assessment of changes in the patient ECG relative to baseline.

Type:

  Extensible

Order:

  Significant

Table TID 3115. ECG ST Assessment

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121123, DCM, "Patient Status or Event") | 1 | M |  | DT (R-41D8B, SRT, "ECG Analysis")
2 | > | HAS PROPERTIES | NUM | DT (122099, DCM, "ST change from baseline") | 1-n | M |  | UNITS = EV (uV, UCUM, "uV")
3 | >> | HAS CONCEPT MOD | CODE | DT (122148, DCM, "Lead ID") | 1 | M |  | BCID 3001 “ECG Leads”

## Quantitative Ventricular Analysis Report SR IOD Templates

The templates that comprise the Quantitative Ventricular Analysis SR are interconnected as in Figure A-3:

Figure A-3. Quantitative Ventricular Analysis Report SR IOD Template Structure

### TID 3202 Ventricular Analysis

The Ventricular Analysis Template provides a CONTAINER with a structure for reporting the result of the ventricular analysis.

Type:

  Extensible

Order:

  Significant

Table TID 3202. Ventricular Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122292, DCM, "Quantitative Ventriculography Report") | 1 | M |  |
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
4 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | U |  |
5 | > | CONTAINS | CONTAINER | EV (122144, DCM, "Quantitative Analysis") | 1-n | M |  |
6 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
7 | >> | HAS OBS CONTEXT | TEXT | EV (111001, DCM, "Algorithm Name") | 1 | M |  |
8 | >> | HAS OBS CONTEXT | TEXT | EV (111003, DCM, "Algorithm Version") | 1 | M |  |
9 | >> | HAS OBS CONTEXT | TEXT | EV (122405, DCM, "Algorithm Manufacturer") | 1 | M |  |
10 | >> | CONTAINS | IMAGE | EV (121112, DCM, "Source of Measurement") | 1-n | M |  |
11 | >>> | HAS CONCEPT MOD | CODE | EV (G-A60B, SRT, "Cardiac Phase") | 1 | M |  | DCID 12233 “Cardiac Phase”
12 | >>> | HAS CONCEPT MOD | CODE | EV (111031, DCM, "Image View") | 1 | MC | If Biplane Analysis | DCID 3466 “Plane Identification”
13 | >> | HAS ACQ CONTEXT | INCLUDE | DTID 3205 “Calibration” | 1-2 | U | VM = 1: Single plane analysis, VM = 2: Biplane analysis | $CalibrationPlane = DCID 3466 “Plane Identification”
14 | >> | HAS ACQ CONTEXT | INCLUDE | DTID 3520 “Hemodynamic Clinical Context” | 1 | U |  |
15 | >> | CONTAINS | INCLUDE | DTID 3206 “VA Main Results” | 1 | M |  |
16 | >> | CONTAINS | INCLUDE | DTID 3207 “AA Main Results” | 1 | U |  |
17 | >> | CONTAINS | INCLUDE | DTID 3208 “Frame-to-Frame Results” | 1 | U |  |
18 | >> | CONTAINS | INCLUDE | DTID 3209 “Centerline Wall Motion” | 1-2 | U | VM = 1: Single plane analysis, VM = 2: Biplane analysis |
19 | >> | CONTAINS | INCLUDE | DTID 3210 “Radial Based Wall Motion” | 1-2 | U | VM = 1: Single plane analysis, VM = 2: Biplane analysis |
20 | >> | CONTAINS | INCLUDE | DTID 3211 “Landmark Based Wall Motion” | 1-2 | U | VM = 1: Single plane analysis, VM = 2: Biplane analysis |

Content Item Descriptions

Row 7 | Identifies the Ventricular Analysis program
Row 8 | Identifies the Ventricular Analysis program version
Row 9 | Identifies the Ventricular Analysis program manufacturer
Row 10 | Identifies the ES and ED images on which the analysis is based, for frame by frame analysis the analyzed image are specified in the frame by frame results (3208) template

### TID 3205 Calibration

The Calibration Template consists of a CONTAINER, with a structure for reporting of the calibration of images used in the analysis.

Table TID 3205. Parameters

Parameter Name | Parameter Usage
--- | ---
$CalibrationPlane | XA Imaging plane

Type:

  Extensible

Order:

  Significant

Table TID 3205. Calibration

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122505, DCM, "Calibration") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111031, DCM, "Image View") | 1 | U |  | $CalibrationPlane
3 | > | HAS OBS CONTEXT | TEXT | EV (111001, DCM, "Algorithm Name") | 1 | MC | IF different from Analysis program specified in the invoking template |
4 | > | HAS OBS CONTEXT | TEXT | EV (111003, DCM, "Algorithm Version") | 1 | MC | IF different from Analysis program specified in the invoking template |
5 | > | HAS OBS CONTEXT | TEXT | EV (122405, DCM, "Algorithm Manufacturer") | 1 | MC | IF different from Analysis program specified in the invoking template |
6 | > | CONTAINS | CODE | EV (122422, DCM, "Calibration Method") | 1 | M |  | DCID 3452 “Calibration Methods”
7 | > | CONTAINS | CODE | EV (122421, DCM, "Calibration Object") | 1 | MC | If row 6 value specifies Calibration Object Used | DCID 3451 “Calibration Objects”
8 | > | CONTAINS | NUM | EV (122423, DCM, "Calibration Object Size") | 1 | MC | If row 6 value specifies Calibration Object Used | DCID 3510 “Catheter Size Units”
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (111026, DCM, "Horizontal Pixel Spacing") $Unit = DT (mm/{pixel}, UCUM, "mm/pixel") $ImagePurpose = EV (121112, DCM, "Source of Measurement")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (111066, DCM, "Vertical Pixel Spacing") $Unit = DT (mm/{pixel}, UCUM, "mm/pixel") $ImagePurpose = EV (121112, DCM, "Source of Measurement")
11 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 3 | Identifies the Calibration program
Row 4 | Identifies the Calibration program version
Row 5 | Identifies the Calibration program manufacturer
Row 7 | Besides a Sphere and a Catheter, a Distance can be identified as a Calibration Object. In this case a distance measurement of a known dimension of the object is used to calculate the pixel size.
Row 8 | The catheter size units is also used to specify the size of other calibration objects (e.g., sphere)
Row 9, 10 | Spacing in the patient body. Point to a single frame containing the image used for calibration if applicable, the actual measurements may be indicated by a SCOORD (see TID 320 “Image or Spatial Coordinates”, row 3)
Row 11 | Secondary Capture image with calibration position

### TID 3206 VA Main Results

The VA Main Results Template consists of a CONTAINER with a structure for reporting the main ventricular analysis measurements.

Type:

  Extensible

Order:

  Significant

Table TID 3206. VA Main Results

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3462 “Chamber Identification”
3 | > | CONTAINS | CODE | EV (122429, DCM, "Volume Method") | 1 | M |  | DCID 3453 “Cardiac Volume Methods”
4 | > | CONTAINS | NUM | EV (122435, DCM, "Regression Volume Exponent") | 1 | U |  | Unit = DT (1, UCUM, "no units")
5 | > | CONTAINS | NUM | EV (122431, DCM, "Regression Slope ED") | 1 | U |  | Unit = DT ({ratio}, UCUM, "ratio")
6 | > | CONTAINS | NUM | EV (122432, DCM, "Regression Offset ED") | 1 | U |  | Unit = DT (ml, UCUM, "ml")
7 | > | CONTAINS | NUM | EV (122433, DCM, "Regression Slope ES") | 1 | U |  | Unit = DT ({ratio}, UCUM, "ratio")
8 | > | CONTAINS | NUM | EV (122434, DCM, "Regression Offset ES") | 1 | U |  | Unit = DT (ml, UCUM, "ml")
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = DCID 3467 “Ejection Fraction” $Unit = DT (%, UCUM, "%")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3468 “ED Volume” $Unit = DT (ml, UCUM, "ml")
11 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3469 “ES Volume” $Unit = DT (ml, UCUM, "ml")
12 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $Unit = DT (ml, UCUM, "ml")
13 | > | CONTAINS | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  | Unit = DT ({H.B.}/min, UCUM, "BPM")
14 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3468 “ED Volume” $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (ml/m2, UCUM, "ml/m2")
15 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3468 “ED Volume” $ModType = EV (121425, DCM, "Index") $ModValue = EV (29463-7, LN, "Patient Weight") $Unit = DT (ml/kg, UCUM, "ml/kg")
16 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3469 “ES Volume” $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (ml/m2, UCUM, "ml/m2")
17 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3469 “ES Volume” $ModType = EV (121425, DCM, "Index") $ModValue = EV (29463-7, LN, "Patient Weight") $Unit = DT (ml/kg, UCUM, "ml/kg")
18 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (ml/m2, UCUM, "ml/m2")
19 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $ModType = EV (121425, DCM, "Index") $ModValue = EV (29463-7, LN, "Patient Weight") $Unit = DT (ml/kg, UCUM, "ml/kg")
20 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32100, SRT, "Cardiac Output") $Unit = DT (l/min, UCUM, "l/min")
21 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32110, SRT, "Cardiac Index") $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (l/min/m2, UCUM, "l/min/m2")
22 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122445, DCM, "Wall Thickness") $Unit = DT (mm, UCUM, "mm")
23 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122446, DCM, "Wall Volume") $Unit = DT (ml, UCUM, "ml")
24 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $Unit = DT (g, UCUM, "gram")
25 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $ModType = EV (121425, DCM, "Index") $ModValue = DCID 3455 “Index Methods” $Unit = DT (g/m2, UCUM, "gram/m2")
26 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $ModType = EV (121425, DCM, "Index") $ModValue = EV (29463-7, LN, "Patient Weight") $Unit = DT (g/kg, UCUM, "gram/kg")
27 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122448, DCM, "Wall Stress") $Unit = DT (dyn/cm2, UCUM, "dynes/cm2")
28 | > | CONTAINS | IMAGE | No purpose of reference | 1-n | U |  |

Content Item Descriptions

Row 28 | Secondary Capture image with ED and/or ES contours

### TID 3207 AA Main Results

The AA Main Results Template consists of a CONTAINER with a structure for reporting the main atrial analysis measurements.

Type:

  Extensible

Order:

  Significant

Table TID 3207. AA Main Results

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3462 “Chamber Identification”
3 | > | CONTAINS | CODE | EV (122429, DCM, "Volume Method") | 1 | M |  | DCID 3453 “Cardiac Volume Methods”
4 | > | CONTAINS | NUM | EV (122435, DCM, "Regression Volume Exponent") | 1 | U |  | Unit = DT (1, UCUM, "no units")
5 | > | CONTAINS | NUM | EV (122431, DCM, "Regression Slope ED") | 1 | U |  | Unit = DT ({ratio}, UCUM, "ratio")
6 | > | CONTAINS | NUM | EV (122432, DCM, "Regression Offset ED") | 1 | U |  | Unit = DT (ml, UCUM, "ml")
7 | > | CONTAINS | NUM | EV (122433, DCM, "Regression Slope ES") | 1 | U |  | Unit = DT ({ratio}, UCUM, "ratio")
8 | > | CONTAINS | NUM | EV (122434, DCM, "Regression Offset ES") | 1 | U |  | Unit = DT (ml, UCUM, "ml")
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3468 “ED Volume” $Unit = DT (ml, UCUM, "ml")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3469 “ES Volume” $Unit = DT (ml, UCUM, "ml")
11 | > | CONTAINS | IMAGE | No purpose of reference | 1-n | U |  |

Content Item Descriptions

Row 11 | Secondary Capture image with ED and/or ES contours

### TID 3208 Frame-to-Frame Results

The Frame-to-Frame Result Template consists of a CONTAINER providing measurements derived from the angiographic images on frame-by-frame basis.

Type:

  Extensible

Order:

  Significant

Table TID 3208. Frame-To-Frame Result

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122499, DCM, "Frame to Frame Analysis")
3 | > | CONTAINS | IMAGE | EV (121112, DCM, "Source of Measurement") | 1-2 | M | VM = 1: Single plane analysis, VM = 2: Biplane analysis |
4 | > | CONTAINS | CODE | EV (122429, DCM, "Volume Method") | 1 | M |  | DCID 3453 “Cardiac Volume Methods”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = DCID 3471 “Estimated Volumes” $TargetSite = DCID 3462 “Chamber Identification” $Unit = DT (ml, UCUM, "ml")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (122445, DCM, "Wall Thickness") $Unit = DT (mm, UCUM, "mm")
7 | > | CONTAINS | IMAGE | No purpose of reference | 1-n | U |  |

Content Item Descriptions

Row 3 | Identifies each frame analyzed, using the multi-valued Referenced Frame Number (0008,1160) attribute of the IMAGE content item.
Row 5, 6 | Includes one measurement for each frame referenced in Row 3.
Row 7 | Secondary Capture image with ventricular contours

### TID 3209 Centerline Wall Motion

The Centerline Wall Motion Template consists of a CONTAINER providing measurements of the centerline wall motion.

Type:

  Extensible

Order:

  Significant

Table TID 3209. Centerline Wall Motion

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122449, DCM, "Centerline Wall Motion Analysis")
3 | >> | HAS CONCEPT MOD | CODE | EV (122410, DCM, "Contour Realignment") | 1 | M |  | DCID 3458 “Contour Realignment”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 100 | M |  | $Measurement = EV (122450, DCM, "Normalized Chord Length") $Unit = DT (%, UCUM, "%")
5 | > | CONTAINS | NUM | EV (122411, DCM, "Threshold Value") | 1 | M |  | Values shall be 1, 2 or 3 UNITS = EV ({sd}, UCUM, "Standard Deviations")
6 | > | CONTAINS | CONTAINER | EV (122451, DCM, "Abnormal Region") | 1-6 | U |  |
7 | >> | CONTAINS | CODE | EV (F-32050, SRT, "Cardiac Wall Motion") | 1 | M |  | DCID 3703 “Wall Motion”
8 | >> | CONTAINS | CODE | EV (R-404F0, SRT, "Circumferential Extent") | 1 | U |  | DCID 3460 “Circumferential Extent”
9 | >> | CONTAINS | NUM | EV (122452, DCM, "First Chord of Abnormal Region") | 1 | M |  | Unit = DT (1, UCUM, "no unit")
10 | >> | CONTAINS | NUM | EV (122453, DCM, "Last Chord of Abnormal Region") | 1 | M |  | Unit = DT (1, UCUM, "no unit")
11 | > | CONTAINS | CONTAINER | EV (122417, DCM, "Regional Abnormal Wall Motion ") | 1-4 | U |  |
12 | >> | HAS CONCEPT MOD | CODE | EV (G-C03E, SRT, "Finding Site") | 1 | M |  | DCID 3461 “Regional Extent”
13 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122459, DCM, "Territory Region Severity") $ModType = EV (F-32050, SRT, "Cardiac Wall Motion") $ModValue = DCID 3703 “Wall Motion” $Unit = DT ({sd}, UCUM, "Standard Deviations")
14 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122461, DCM, "Opposite Region Severity") $ModType = EV (F-32050, SRT, "Cardiac Wall Motion") $ModValue = DCID 3703 “Wall Motion” $Unit = DT ({sd}, UCUM, "Standard Deviations")
15 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 4 | Normalized lengths of the chords determined between ED and ES contour. The measurement template allows the specification of the statistical properties of the normal population and of the chord measurement relative to the population.
Row 8 | If the Circumferential Extent is not specified no limitations to the boundaries for regions are assumed.
Row 11 | The Regional Abnormal Wall Motion container holds the information on the severity of the decreased or increased wall motion of the 4 predefined regions as described in [Sheehan, 1986].
Row 12 | The name of the region with an abnormal ventricular wall motion as described in [Sheehan, 1986].
Row 13 | The severity of the wall motion abnormality expressed in Standard Deviations above or below normal in the territory region as described in [Sheehan, 1986].
Row 14 | The severity of the wall motion abnormality expressed in Standard Deviations above or below normal in the opposite region as described in [Sheehan, 1986].
Row 15 | Secondary Capture image with centerline analysis result

### TID 3210 Radial Based Wall Motion

The Radial Based Wall Motion Template consists of a CONTAINER providing measurements of the radial based wall motion.

Type:

  Extensible

Order:

  Significant

Table TID 3210. Radial Based Wall Motion

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122493, DCM, "Radial Based Wall Motion Analysis")
3 | >> | HAS CONCEPT MOD | CODE | EV (122410, DCM, "Contour Realignment") | 1 | M |  | DCID 3458 “Contour Realignment”
4 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | M |  |
5 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3718 “Myocardial Wall Segments in Projection”
6 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122495, DCM, "Regional Contribution to Ejection Fraction") $Unit = DT (%, UCUM, "%")
7 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (122496, DCM, "Radial Shortening") $Unit = DT (%, UCUM, "%")
8 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 6 | The CREF values of the 6 regions determined for the radial based wall motion
Row 7 | The shortening of the measured radials within the region
Row 8 | Secondary Capture image with radial based analysis result

### TID 3211 Landmark Based Wall Motion

The Landmark Based Wall Motion Template consists of a CONTAINER providing measurements of the landmark based wall motion.

Type:

  Extensible

Order:

  Significant

Table TID 3211. Landmark Based Wall Motion

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122497, DCM, "Landmark Based Wall Motion Analysis")
3 | >> | HAS CONCEPT MOD | CODE | EV (122410, DCM, "Contour Realignment") | 1 | M |  | DCID 3458 “Contour Realignment”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = EV (122498, DCM, "Slice Contribution to Ejection Fraction") $Unit = DT (%, UCUM, "%")
5 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 5 | M |  |
6 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Sites") | 1 | M |  | DCID 3718 “Myocardial Wall Segments in Projection”
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122495, DCM, "Regional Contribution to Ejection Fraction") $Unit = DT (%, UCUM, "%")
8 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 8 | Secondary Capture image with Landmark Based Analysis result

## Quantitative Arterial Analysis Report SR IOD Templates

The templates that comprise the Quantitative Arterial Analysis SR are interconnected as in Figure A-4:

Figure A-4. Quantitative Arterial Analysis Report SR IOD Template Structure

### TID 3213 Quantitative Arterial Analysis

The Quantitative Arterial Analysis Template consists of a CONTAINER with a structure for reporting the result of the quantitative arterial analysis process.

Type:

  Extensible

Order:

  Significant

Table TID 3213. Quantitative Arterial Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122291, DCM, "Quantitative Arteriography Report") | 1 | M |  |
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
4 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | U |  |
5 | > | HAS OBS CONTEXT | TEXT | EV (111001, DCM, "Algorithm Name") | 1 | M |  |
6 | > | HAS OBS CONTEXT | TEXT | EV (111003, DCM, "Algorithm Version") | 1 | M |  |
7 | > | HAS OBS CONTEXT | TEXT | EV (122405, DCM, "Algorithm Manufacturer") | 1 | M |  |
8 | > | CONTAINS | INCLUDE | DTID 3214 “Analyzed Segment” | 1-n | M |  |

Content Item Descriptions

Row 5 | Identifies the Arterial Analysis program
Row 6 | Identifies the Arterial Analysis program version
Row 7 | Identifies the Arterial Analysis program manufacturer

### TID 3214 Analyzed Segment

The Analyzed Segment Template consists of a CONTAINER providing quantitative arterial analysis measurements derived from the angiographic images.

Type:

  Extensible

Order:

  Significant

Table TID 3214. Analyzed Segment

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3604 “Arterial Lesion Locations”
3 | > | CONTAINS | IMAGE | EV (121112, DCM, "Source of Measurement") | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 3205 “Calibration” | 1 | M |  |
5 | > | HAS ACQ CONTEXT | INCLUDE | DTID 3520 “Hemodynamic Clinical Context” | 1 | U |  |
6 | > | HAS ACQ CONTEXT | CODE | EV (G-72BB, SRT, "Catheterization Procedure Phase") | 1 | U |  | DCID 3651 “Hemodynamic Measurement Phase”
7 | > | CONTAINS | SCOORD | EV (122507, DCM, "Left Contour") | 1 | M |  | GRAPHIC TYPE = POLYLINE
8 | >> | R-SELECTED FROM | IMAGE |  | 1 | M |  | Must reference Row 3
9 | > | CONTAINS | SCOORD | EV (122508, DCM, "Right Contour") | 1 | M |  | GRAPHIC TYPE = POLYLINE
10 | >> | R-SELECTED FROM | IMAGE |  | 1 | M |  | Must reference Row 3
11 | > | CONTAINS | INCLUDE | DTID 3219 “Segment Values” | 1 | M |  |
12 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-404FB, SRT, "Minimum") $Unit = DT (mm, UCUM, "mm")
13 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (G-A437, SRT, "Maximum") $Unit = DT (mm, UCUM, "mm")
14 | > | CONTAINS | CONTAINER | EV (122509, DCM, "Diameter Graph") | 1 | U |  |
15 | >> | CONTAINS | NUM | EV (122511, DCM, "Graph Increment") | 1 | M |  | Value = 1 UNITS = DT ({pixels}, UCUM, "pixels")
16 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Unit = DT (mm, UCUM, "mm")
17 | > | CONTAINS | NUM | EV (122382, DCM, "Site of Luminal Minimum ") | 1 | U |  | UNITS = DT ({pixels}, UCUM, "pixels")
18 | > | CONTAINS | NUM | EV (122516, DCM, "Site of Luminal Maximum") | 1 | U |  | UNITS = DT ({pixels}, UCUM, "pixels")
19 | > | CONTAINS | INCLUDE | DTID 3215 “Angiographic Lesion Analysis” | 1-n | U |  |
20 | > | CONTAINS | INCLUDE | DTID 3217 “Sub-segmental Data” | 1-n | U |  |
21 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 1 | Observation DateTime (0040,A032) of container needs to be flagged with the time of the analysis
Row 6 | Numeric coordinates (x,y) identifying the contour points from proximal to distal of left contour. Left is relative to the direction of the blood flow.
Row 9 | Numeric coordinates (x,y) identifying the contour points from proximal to distal of right contour. Right is relative to the direction of the blood flow.
Row 12, 13 | Positions are relative to the midpoint between the first left and right contour points and measured along the midline between the left and right contour.
Row 14 | The X-axis represents the pixel points of the midline of the vessel from proximal to distal. The points on the midline are not necessarily equidistant.
Row 16 | For each point of the midline of the vessel a measurement value for the diameter is calculated.
Row 17, 18 | The positions in the graph are related to the points on the midline of the vessel.
Row 21 | Secondary Capture image with Arterial Analysis contour.

Definition of Left and Right defined by the direction of the blood flow as in Figure A-4b:

Figure A-4b. Direction of Blood Flow

### TID 3215 Angiographic Lesion Analysis

The Angiographic Lesion Analysis Template consists of a CONTAINER providing quantitative arterial analysis measurements derived for an obstruction in a total analyzed segment.

Type:

  Extensible

Order:

  Significant

Table TID 3215. Angiographic Lesion Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (F-00585, SRT, "Lesion Finding") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121151, DCM, "Lesion Identifier") | 1 | M |  |
3 | >> | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3604 “Arterial Lesion Locations”
4 | >>> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-404FB, SRT, "Minimum") $Unit = DT (mm, UCUM, "mm")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Method = DCID 3470 “Vessel Lumen Cross-sectional Area Calculation Methods” $Derivation = EV (R-404FB, SRT, "Minimum") $Unit = DT (mm2, UCUM, "mm2")
7 | > | CONTAINS | CODE | EV (122430, DCM, "Reference Method") | 1 | M |  | DCID 3465 “QA Reference Methods”
8 | > | CONTAINS | CONTAINER | EV (122438, DCM, "Reference Points") | 1 | U |  |
9 | >> | CONTAINS | NUM | EV (122337, DCM, "Relative Position") | 1-n | M |  | UNITS = DT (mm, UCUM, "mm")
10 | >>> | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Unit = DT (mm, UCUM, "mm")
11 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $TargetSite = (122382, DCM, "Site of Luminal Minimum") $Unit = DT (mm, UCUM, "mm")
12 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Derivation = EV (122404, DCM, "Reconstructed") $TargetSite = (122382, DCM, "Site of Luminal Minimum") $Unit = DT (mm2, UCUM, "mm2")
13 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-41D2D, SRT, "Calculated") $TargetSite = EV (122481, DCM, "Contour Start") $Unit = DT (mm, UCUM, "mm")
14 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-41D2D, SRT, "Calculated") $TargetSite = EV (122482, DCM, "Contour End") $Unit = DT (mm, UCUM, "mm")
15 | > | CONTAINS | INCLUDE | DTID 3218 “Position in Arterial Segment” | 1 | M |  |
16 | > | CONTAINS | CONTAINER | EV (122517, DCM, "Densitometric Luminal Cross-sectional Area Graph") | 1 | U |  |
17 | >> | CONTAINS | NUM | EV (122511, DCM, "Graph Increment") | 1 | M |  | Value = 1 UNITS = DT ({pixels}, UCUM, "pixels")
18 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Unit = (mm2, UCUM, "mm2")
19 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Derivation = EV (R-41D2D, SRT, "Calculated") $Method = EV (122474, DCM, "Densitometric method") $TargetSite = EV (122481, DCM, "Contour Start") $Unit = (mm2, UCUM, "mm2")
20 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Derivation = EV (R-41D2D, SRT, "Calculated") $Method = EV (122474, DCM, "Densitometric method") $TargetSite = EV (122482, DCM, "Contour End") $Unit = (mm2, UCUM, "mm2")
21 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (R-101BC, SRT, "Stenotic Lesion Length") $Unit = DT (mm, UCUM, "mm")
22 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (R-101BB, SRT, "Lumen Diameter Stenosis") $Unit = DT (%, UCUM, "%")
23 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (R-101BA, SRT, "Lumen Area Stenosis") $Method = DCID 3470 “Vessel Lumen Cross-sectional Area Calculation Methods” $Unit = DT (%, UCUM, "%")
24 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (122372, DCM, "Lumen Volume") $Method = DCID 3470 “Vessel Lumen Cross-sectional Area Calculation Methods” $Unit = DT (mm3, UCUM, "mm3")
25 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122542, DCM, "Plaque Area") $Unit = DT (mm2, UCUM, "mm2")
26 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122376, DCM, "Total Plaque Volume") $Unit = DT (mm3, UCUM, "mm3")
27 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122544, DCM, "Diameter Symmetry") $Unit = DT ({ratio}, UCUM, "ratio")
28 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122545, DCM, "Area Symmetry") $Unit = DT ({ratio}, UCUM, "ratio")
29 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122546, DCM, "Inflow Angle") $Unit = DT (deg, UCUM, "deg")
30 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122547, DCM, "Outflow Angle") $Unit = DT (deg, UCUM, "deg")
31 | > | CONTAINS | INCLUDE | DTID 3216 “Stenotic Flow Reserve” | 1 | U |  |
32 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 8 | Set of user defined reference position for method that requires local reference position.
Row 9 | Distance of local reference position from an arbitrary landmark.
Row 10 | Diameter at a local reference position.
Row 11 | The reference diameter for the arterial lesion calculated with the applicable reference method
Row 12 | The reference area for the arterial lesion calculated with the applicable reference method
Row 13 | The diameter measurement at the start of the reconstruction line in the diameter graph (TID 3214 “Analyzed Segment” Row 14)
Row 14 | The diameter measurement at the end of the reconstruction line in the diameter graph (TID 3214 “Analyzed Segment” Row 14)
Row 15 | The positions of the lesion, borders of the lesion, etc.
Row 16 | The graph with the calculated cross sectional area results based on the densitometric method
Row 18 | The cross sectional area measurements calculated based on the densitometric method
Row 19 | The cross sectional area measurement at the start of the reconstruction line in the area graph
Row 20 | The cross sectional area measurement at the end of the reconstruction line in the area graph
Row 21 | Measured along the midline of the left and right contour
Row 22 | The diameter stenosis is calculated as follows: (Reference Luminal Diameter - Minimum Luminal Diameter / Reference Luminal Diameter) * 100%
Row 23 | The circular and the densitometric area stenosis are calculated respectively as: (Reference Vessel Lumen Cross-Sectional Area - Minimum Luminal Circular Area / Reference Vessel Lumen Cross-Sectional Area) * 100%(Reference Vessel Lumen Cross-Sectional Area - Minimum Luminal Densitometric Area / Reference Vessel Lumen Cross-Sectional Area) * 100%
Row 24 | Estimated lumen volume between proximal border and distal border of lesion (TID 3218 “Position in Arterial Segment”, row 1 and 2)
Row 32 | Secondary Capture image with obstruction analysis contour

### TID 3216 Stenotic Flow Reserve

The Stenotic Flow Reserve Template consists of a CONTAINER providing quantitative arterial analysis measurements derived for an obstruction in a total analyzed segment.

Type:

  Extensible

Order:

  Significant

Table TID 3216. Stenotic Flow Reserve

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122548, DCM, "Stenotic Flow Reserve") $Unit = DT ({ratio}, UCUM, "ratio")
2 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122549, DCM, "Poiseuille Resistance") $Unit = DT (mm[Hg].s/cm, UCUM, "mmHG.s/cm")
3 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122550, DCM, "Turbulence Resistance") $Unit = DT (mm[Hg].s2/cm2, UCUM, "mmHG.s2/cm2")
4 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122555, DCM, "Estimated Normal Flow") $Unit = DT (ml/s, UCUM, "ml/s")
5 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122551, DCM, "Pressure Drop at SFR") $Unit = DT (mm[Hg], UCUM, "mmHg")
6 |  | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 6 | Secondary Capture image with SFR analysis contour

### TID 3217 Sub-segmental Data

The Sub-segmental Data Template consists of a CONTAINER providing quantitative arterial analysis measurements derived for a sub-segment in a total analyzed segment.

Type:

  Extensible

Order:

  Significant

Table TID 3217. Sub-Segmental Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3604 “Arterial Lesion Locations”
3 | >> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
4 | > | CONTAINS | CODE | EV (122554, DCM, "Segmentation Method") | 1 | M |  | DCID 3456 “Sub-segment Methods”
5 | > | CONTAINS | INCLUDE | DTID 3219 “Segment Values” | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3218 “Position in Arterial Segment” | 1 | M |  |
7 | > | CONTAINS | IMAGE | No purpose of reference | 1 | U |  |

Content Item Descriptions

Row 7 | Secondary Capture image with obstruction analysis contour

### TID 3218 Position in Arterial Segment

The Position in Arterial Segment Template consists of the position content items common for the Angiographic Lesion Analysis and Sub-Segmental Data.

Type:

  Extensible

Order:

  Significant

Table TID 3218. Position in Arterial Segment

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122528, DCM, "Position of Proximal Border") $Unit = DT (mm, UCUM, "mm")
2 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122529, DCM, "Position of Distal Border") $Unit = DT (mm, UCUM, "mm")
3 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122382, DCM, "Site of Luminal Minimum") $Unit = DT (mm, UCUM, "mm")
4 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122516, DCM, "Site of Luminal Maximum") $Unit = DT (mm, UCUM, "mm")
5 |  | CONTAINS | NUM | EV (122528, DCM, "Position of Proximal Border") | 1 | UC | IFF TID 3214 “Analyzed Segment” Row 14 is present | UNITS = DT ({pixels}, UCUM, "pixels")
6 |  | CONTAINS | NUM | EV (122529, DCM, "Position of Distal Border") | 1 | UC | IFF TID 3214 “Analyzed Segment” Row 14 is present | UNITS = DT ({pixels}, UCUM, "pixels")
7 |  | CONTAINS | NUM | EV (122382, DCM, "Site of Luminal Minimum") | 1 | UC | IFF TID 3214 “Analyzed Segment” Row 14 is present | UNITS = DT ({pixels}, UCUM, "pixels")
8 |  | CONTAINS | NUM | EV (122516, DCM, "Site of Luminal Maximum") | 1 | UC | IFF TID 3214 “Analyzed Segment” Row 14 is present | UNITS = DT ({pixels}, UCUM, "pixels")

Content Item Descriptions

Row 1, 2, 3, 4 | Positions are relative to the midpoint of the first left and right contour points and measured along the midline of the left and right contour
Row 5, 6, 7, 8 | The positions are relative to the measurement locations of the Diameter Graph of TID 3214 “Analyzed Segment” row 14.

### TID 3219 Segment Values

The Segment Values Template consists of content items providing quantitative arterial analysis measurements for a total analyzed segment or sub segment.

Type:

  Extensible

Order:

  Significant

Table TID 3219. Segment Values

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122510, DCM, "Length Luminal Segment") $Unit = DT (mm, UCUM, "mm")
2 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-404FB, SRT, "Minimum") $Unit = DT (mm, UCUM, "mm")
3 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (G-A437, SRT, "Maximum") $Unit = DT (mm, UCUM, "mm")
4 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-00317, SRT, "Mean") $Unit = DT (mm, UCUM, "mm")
5 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-0364, SRT, "Vessel Luminal Diameter") $Derivation = EV (R-10047, SRT, "Standard Deviation") $Unit = DT (mm, UCUM, "mm")

Content Item Descriptions

Row 1 | Measured along the midline of the left and right contour.

## IVUS Report Templates

The templates that comprise the IVUS Report within the Evidence Report IOD are interconnected as shown in Figure A-5.

Figure A-5. IVUS Report Template Hierarchy

### TID 3250 IVUS Report

The IVUS Report template is the root structure for the representation of IVUS measurements acquired during a catheterization procedure.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 3250. IVUS Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122325, DCM, "IVUS Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
4 | > |  | INCLUDE | DTID 3601 “Procedure Context” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | U |  |
6 | > | CONTAINS | CONTAINER | EV (111028, DCM, "Image Library") | 1 | U |  |
7 | >> | CONTAINS | IMAGE | No purpose of reference | 1-n | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3251 “IVUS Vessel” | 1-n | M |  |

### TID 3251 IVUS Vessel

The IVUS Vessel template provides a structure for grouping one or more lesions analyzed and/or treated during a single phase of a catheterization procedure, according to vessel (or arterial location).

Type:

  Extensible

Order:

  Significant

Table TID 3251. IVUS Vessel

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | U |  | DCID 3604 “Arterial Lesion Locations”
3 | >> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
4 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | UC | IFF anatomy has laterality | DCID 244 “Laterality”
5 | > | HAS ACQ CONTEXT | CODE | EV (G-72BB, SRT, "Catheterization Procedure Phase") | 1 | U |  | DCID 3480 “IVUS Procedure Phases”
6 | > | CONTAINS | CODE | EV (122134, DCM, "Vessel Morphology") | 1-n | U |  | CID 3712 “Vessel Descriptors”
7 | > | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | U |  |
8 | > | CONTAINS | CODE | EV (115, NCDR [2.0b], "Dissection in segment") | 1 | U |  | DCID 230 “Yes-No”
9 | > | CONTAINS | INCLUDE | DTID 3252 “IVUS Lesion” | 1-n | U |  |

### TID 3252 IVUS Lesion

The IVUS Lesion template provides a structure for grouping measurements and observations made on a single lesion during an Intravascular Ultrasound Procedure.

Type:

  Extensible

Order:

  Significant

Table TID 3252. IVUS Lesion

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (F-00585, SRT, "Lesion Finding") | 1 | M |  |
2 | > | HAS OBS CONTEXT | TEXT | EV (121151, DCM, "Lesion Identifier") | 1 | M |  | Up to 3 numeric characters
3 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | U |  | DCID 3604 “Arterial Lesion Locations”
4 | >>> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
5 | > | HAS ACQ CONTEXT | INCLUDE | DTID 3107 “Device Used” | 1-n | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3253 “IVUS Measurements” | 1 | MC | One or both of rows 6 & 7 must be present |
7 | > | CONTAINS | INCLUDE | DTID 3254 “IVUS Qualitative Assessments” | 1 | MC | One or both of rows 6 & 7 must be present |
8 | > | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | U |  |

Content Item Descriptions

Row 2 | Lesion Identifier is specified as a numeric text string in order to facilitate trans-coding to DICOM Attribute (0018,3105) Lesion Number and to formats for outcomes registries, such as the ACC National Cardiovascular Data Registry™. ### Note Also see TID 3105 “Lesion Identification and Properties”.
Row 3 | Finding site may span multiple segments with the proximal and distal extent specified by separate items. These may not be totally contained with the segment specified at the Vessel level.

### TID 3253 IVUS Measurements

The IVUS measurements template groups together simple distance, area and angle measurements, along with derived measurements that made during an IVUS procedure. Refer to the " ACC Clinical Expert Consensus Document on Standards for Acquisition, measurement and Reporting of Intravascular Ultrasound Studies (IVUS) "for more information.

Type:

  Extensible

Order:

  Significant

Table TID 3253. IVUS Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3481 “IVUS Distance Measurements” $Units = EV (mm, UCUM, "mm") $Derivation = DCID 3488 “Min/Max/Mean” $TargetSite = BCID 3486 “Vascular Measurement Sites”
2 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3482 “IVUS Area Measurements” $Units = EV (mm2, UCUM, "mm2") $Derivation = DCID 3488 “Min/Max/Mean” $TargetSite = BCID 3486 “Vascular Measurement Sites”
3 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3483 “IVUS Longitudinal Measurements” $Units = EV (mm, UCUM, "mm")
4 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (122355, DCM, "Arc of Calcium") $Units = EV (deg, UCUM, "degrees") $TargetSite = BCID 3486 “Vascular Measurement Sites”
5 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (R-101BA, SRT, "Lumen Area Stenosis") $Units = EV (%, UCUM, "%")
6 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122354, DCM, "Plaque Burden") $Units = EV (%, UCUM, "%") $TargetSite = BCID 3486 “Vascular Measurement Sites”
7 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3484 “IVUS Indices and Ratios” $Units = EV ({ratio}, UCUM, "ratio") $TargetSite = BCID 3486 “Vascular Measurement Sites”
8 |  |  | INCLUDE | DTID 3255 “IVUS Volume Measurement” | 1-n | U |  |
9 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122339, DCM, "Stent Volume Obstruction") $Units = EV (%, UCUM, "%")

### TID 3254 IVUS Qualitative Assessments

The IVUS Qualitative Assessments template groups together the qualitative properties of a lesion that are observed during an IVUS procedure. Refer to the "ACC Clinical Expert Consensus Document on Standards for Acquisition, measurement and Reporting of Intravascular Ultrasound Studies (IVUS) " for more information.

Type:

  Extensible

Order:

  Significant

Table TID 3254. IVUS Qualitative Assessments

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (122133, DCM, "Lesion Morphology") | 1-n | U |  | DCID 3491 “IVUS Lesion Morphologies”
2 |  |  | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3494 “IVUS Non Morphological Findings”
3 | > |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
4 |  |  | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (D3-81310, SRT, "Arterial Dissection")
5 | > | HAS CONCEPT MOD | CODE | EV (122387, DCM, "Dissection Classification") | 1 | U |  | DCID 3492 “Vascular Dissection Classifications”
6 | > |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
7 |  |  | CODE | EV (122391, DCM, "Relative Stenosis Severity") | 1 | U |  | DCID 3493 “IVUS Relative Stenosis Severities”
8 |  |  | CODE | EV (108, NCDR [2.0b], "Previously Dilated Lesion ") | 1 | U |  | DCID 3750 “Previously Dilated Lesion”
9 |  |  | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (122393, DCM, "Restenotic Lesion")
10 | > |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
11 |  |  | CODE | EV (111009, DCM, "Calcification Type") | 1 | U |  | DCID 3489 “Calcium Distribution”

### TID 3255 IVUS Volume Measurement

The IVUS Volume Measurement Template contains information describing an IVUS Volumetric measurement

Type:

  Extensible

Order:

  Significant

Table TID 3255. IVUS Volume Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = DCID 3485 “IVUS Volume Measurements” $Units = EV (mm3, UCUM, "mm3") $TargetSite = BCID 3487 “Intravascular Volumetric Regions”
2 | > | HAS PROPERTIES | NUM | EV (122336, DCM, "Vascular Volume measurement length") | 1 | U |  | UNITS = DT (mm, UCUM, "mm")
3 | > | HAS PROPERTIES | NUM | EV (122337, DCM, "Relative position") | 1 | U |  | UNITS = DT (mm, UCUM, "mm")
4 | >> | HAS CONCEPT MOD | CODE | EV (122340, DCM, "Fiducial feature") | 1 | M |  | DCID 3496 “IVUS Fiducial Points”

## Stress Testing Report Templates

### TID 3300 Stress Testing Report

The Stress Testing Report template is the root structure for the representation of measurements and findings of a stress testing procedure.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 3300. Stress Testing Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (18752-6, LN, "Stress Testing Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DCID 3200 “Stress Test Procedure”
3 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
5 | > | CONTAINS | CONTAINER | EV (121109, DCM, "Indications for Procedure") | 1 | U |  |
6 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3201 “Indications for Stress Test”
7 | >> | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1 | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | M |  |
9 | > | CONTAINS | INCLUDE | DTID 3802 “Cardiovascular Patient History” | 1 | U |  |
10 | > | CONTAINS | INCLUDE | DTID 3301 “Stress Test Procedure Description” | 1 | M |  |
11 | > | CONTAINS | INCLUDE | DTID 3303 “Stress Test Phase Data” | 1-n | M |  |
12 | > | CONTAINS | INCLUDE | DTID 3311 “Stress Test Summary” | 1 | U |  |
13 | > | CONTAINS | INCLUDE | DTID 3318 “Comparison to Prior Stress Exam” | 1 | U |  |
14 | > | CONTAINS | INCLUDE | DTID 3320 “Conclusions and Recommendations” | 1 | U |  |

### TID 3301 Stress Test Procedure Description

Type:

  Extensible

Order:

  Significant

Table TID 3301. Stress Test Procedure Description

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121064, DCM, "Current Procedure Descriptions") | 1 | M |  |
2 | > | CONTAINS | CODE | DT (109056, DCM, "Stress Protocol") | 1 | U |  | BCID 3261 “Stress Protocols”
3 | > | CONTAINS | TEXT | DT (109056, DCM, "Stress Protocol") | 1 | U |  |
4 | > | CONTAINS | CODE | DT (10:11345, MDC, "Lead System") | 1 | U |  | BCID 3263 “Electrode Placement Values”
5 | > | CONTAINS | CODE | DT (A-17200, SRT, "Exerciser Device") | 1 | U |  | BCID 3203 “Exerciser Device”
6 | > | CONTAINS | CODE | DT (G-C11C, SRT, "Pharmacological Stress Agent") | 1 | MC | IFF Pharmacological Stress used | BCID 3204 “Stress Agents”
7 | > | CONTAINS | CONTAINER | EV (122700, DCM, "Indications for Pharmacological Stress") | 1 | MC | IFF Pharmacological Stress used |
8 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3205 “Indications for Pharmacological Stress Test”
9 | > | CONTAINS | CODE | DT (P0-0099A, SRT, "Imaging procedure") | 1 | MC | IFF imaging used in procedure | DCID 3206 “Non-invasive Cardiac Imaging Procedures”
10 | > | CONTAINS | CODE | DT (125203, DCM, "Acquisition Protocol") | 1 | UC | IFF Nuclear imaging | DCID 3110 “Nuclear Cardiology Protocols”
11 | > | CONTAINS | CODE | DT (125203, DCM, "Acquisition Protocol") | 1 | UC | IFF PET imaging | DCID 3106 “PET Cardiology Protocols”
12 | > | CONTAINS | TEXT | DT (121141, DCM, "Image Type") | 1 | UC | IFF Nuclear or PET imaging | STATIC, DYNAMIC, or GATED. See note.
13 | > | CONTAINS | CODE | DT (RID11248, RADLEX, "Cardiac Gating") | 1 | U |  | DCID 3104 “Cardiac Synchronization Technique”
14 | > | CONTAINS | CODE | DT (125203, DCM, "Acquisition Protocol") | 1 | UC | IFF Contrast echocardiography | DT (P5-B3008, SRT, "Contrast echocardiography")
15 | > | CONTAINS | CODE | DT (113743, DCM, "Patient Orientation") | 1 | U |  | DCID 19 “Patient Orientation”
16 | >> | HAS CONCEPT MOD | CODE | EV (113744, DCM, "Patient Orientation Modifier") | 1 | U |  | DCID 20 “Patient Orientation Modifier”
17 | > | CONTAINS | TEXT | DT (121065, DCM, "Procedure Description") | 1 | U |  |
18 | > | CONTAINS | DATETIME | DT (122701, DCM, "Procedure Time Base") | 1 | U |  | See note.

Content Item Descriptions

Row 12 | Image Type may be copied from the NM Image SOP Instance attribute Image Type (0008,0008) value 3, or from the PET Image SOP Instance attribute Series Type (0054,1000).
Row 18 | The Procedure Time Base is the time from which elapsed times are measured. The Study Time (0008,0030) may include the patient prep period, while this Procedure Time Base is typically established when baseline data collection begins.

### TID 3303 Stress Test Phase Data

The Stress Test Phase Data template provides a structure for measurements acquired during a single procedure phase.

Type:

  Extensible

Order:

  Significant

Table TID 3303. Stress Test Phase Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  | See note.
2 | > | HAS ACQ CONTEXT | CODE | EV (G-7292, SRT, "Procedure phase") | 1 | MC | XOR row 3 | BCID 3207 “Stress Test Procedure Phases”
3 | > | HAS ACQ CONTEXT | CODE | EV (G-7292, SRT, "Procedure phase") | 1 | MC | IFF Nuclear Imaging; XOR row 2 | DCID 3101 “Cardiac Procedural State Values”
4 | > | HAS ACQ CONTEXT | TEXT | EV (G-7292, SRT, "Procedure phase") | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 3301 “Stress Test Procedure Description” | 1 | MC | IFF protocol changed from initial specification |
6 | > | HAS ACQ CONTEXT | NUM | EV (109055, DCM, "Protocol Stage") | 1 | U |  | UNITS = DT ({stage}, UCUM, "stage")
7 | > | CONTAINS | INCLUDE | DTID 3304 “Stress Test Measurement Group” | 1-n | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3307 “NM/PET Perfusion Measurement Group” | 1 | MC | IFF Nuclear or PET Imaging |
9 | > | CONTAINS | INCLUDE | DTID 3309 “Stress Echo Measurement Group” | 1 | UC | IFF Echocardiography Imaging |

Content Item Descriptions

Row 1 | The Container shall have a specific Content Item Observation DateTime (0040,A032) attribute to indicate the time at which the phase began.

### TID 3304 Stress Test Measurement Group

Each instance of the Stress Test Measurement Group represents a group of data elements acquired at approximately the same instant, and conventionally rendered as row in a tabular presentation. It is typically generated during the Stress exam whenever a time interval elapses (for example, every minute of the phase), when a technician observes data worth capturing, or when measurements exceed a given range.

Type:

  Extensible

Order:

  Significant

Table TID 3304. Stress Test Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  | See note.
2 | > | CONTAINS | NUM | DT (F-031F9, SRT, "Time since start of exam") | 1 | M |  | UNITS = DT (min, UCUM, "min")
3 | > | CONTAINS | NUM | DT (122710, DCM, "Time since start of stage") | 1 | M |  | UNITS = DT (min, UCUM, "min")
4 | > | CONTAINS | NUM | DT (122702, DCM, "Treadmill speed") | 1 | U |  | UNITS = DCID 3212 “Treadmill Speed”
5 | > | CONTAINS | NUM | DT (122703, DCM, "Treadmill gradient") | 1 | U |  | UNITS = EV (%, UCUM, "%")
6 | > | CONTAINS | NUM | DT (122704, DCM, "Ergometer power") | 1 | U |  | UNITS = EV (W, UCUM, "Watts")
7 | > | CONTAINS | NUM | DT (122709, DCM, "Activity workload") | 1 | U |  | UNITS = DT ([MET], UCUM, "METS")
8 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DT (122706, DCM, "Rating of Perceived Exertion") $Method = BCID 3239 “Perceived Exertion Scales”
9 | > | CONTAINS | NUM | DT (122705, DCM, "Pharmacological Stress Agent Dose Rate") | 1 | MC | IFF Pharmacological Stress used | UNITS = EV (ug/kg/min, UCUM, "ug/kg/min")
10 | > | CONTAINS | INCLUDE | DTID 3106 “Drugs/Contrast Administered” | 1 | U |  | See note.
11 | > | CONTAINS | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  | UNITS = EV ({H.B.}/min, UCUM, "BPM")
12 | > | CONTAINS | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
13 | > | CONTAINS | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
14 | > | CONTAINS | NUM | DT (122707, DCM, "Number of Ectopic Beats") | 1 | U |  | UNITS = EV ({beats}, UCUM, "beats")
15 | >> | HAS PROPERTIES | NUM | DT (R-40861, SRT, "Period of collection") | 1 | M |  | UNITS = DT (min, UCUM, "min")
16 | >> | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | U |  | BCID 3234 “Ectopic Beat Morphology”
17 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DT (F-03204, SRT, "ST Elevation") $Units = DT (mV, UCUM, "mV") $TargetSite = DCID 3001 “ECG Leads”
18 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DT (F-38279, SRT, "ST Depression") $Units = DT (mV, UCUM, "mV") $TargetSite = DCID 3001 “ECG Leads”
19 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3228 “ECG Timing Measurements” $Units = DT (ms, UCUM, "ms") $TargetSite = DCID 3001 “ECG Leads” See note.
20 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3227 “QTc Measurements” $Units = DT (ms, UCUM, "ms") $TargetSite = DCID 3001 “ECG Leads” $Equation = DCID 3678 “Qt Correction Algorithms” See note.
21 | >> | INFERRED FROM | NUM | DT (2:16000, MDC, "RR Interval for QTc") | 1 | U |  | UNITS = DT (ms, UCUM, "ms") See note.
22 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3229 “ECG Axis Measurements” $Units = DT (deg, UCUM, "deg") See note.
23 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3526 “Blood Gas Saturation” $Units = EV (%, UCUM, "%")
24 | > | CONTAINS | NUM | DT (122708, DCM, "Double Product") | 1 | U |  | UNITS = DT (mm[Hg].{H.B.}/min, UCUM, "mmHg.BPM")
25 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3220 “Stress Symptoms”
26 | > | CONTAINS | CODE | EV (F-00033, SRT, "ECG Finding") | 1-n | U |  | DCID 3230 “ECG Findings”
27 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

Content Item Descriptions

Row 1 | The Container shall have a specific Content Item Observation DateTime (0040,A032) attribute to indicate the time at which the measurements were made.
Row 10 | Included TID 3106 “Drugs/Contrast Administered” allows the recording of test medications other than the Pharmacological Stress Agent identified in TID 3301 “Stress Test Procedure Description”.
Rows 17, 18 | ECG ST elevation/depression is measured in units of mV, but is conventionally reported in units of mm, based on strip recordings with scaling of 100 uV/mm. The display application should render these measurements in units meaningful to the user.
Row 19 | Note that the MDC codes for "per lead" measurements specified in CID 3228 “ECG Timing Measurements” are base codes for post-coordination with lead identifiers conveyed in the Target Site modifier in TID 300 “Measurement”. MDC also defines pre-coordinated codes that include both the measurement and the lead, which may be used in this row.
Row 20 | Note that the MDC code for "QTc interval per lead" specified in CID 3227 “QTc Measurements” is a base code for post-coordination with lead identifiers conveyed in the Target Site modifier in TID 300 “Measurement”. MDC also defines pre-coordinated codes that include both the measurement and the lead, which may be used in this row. Note that TID 300 “Measurement” enables the encoding of a non-standard correction algorithm, either as a local code, or as a TEXT Method Citation (see TID 300 “Measurement” row 12).
Row 21 | R-R interval used for QT correction algorithm
Row 22 | Recommended range for ECG axis measurements is -90° to +270°

### TID 3307 NM/PET Perfusion Measurement Group

Type:

  Extensible

Order:

  Significant

Table TID 3307. NM/PET Perfusion Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DCID 3108 “NM/PET Procedures”
3 | > | CONTAINS | CODE | EV (F-61FDB, SRT, "Radiopharmaceutical agent") | 1 | M |  | DCID 3111 “Nuclear Cardiology Radiopharmaceuticals”
4 | > | CONTAINS | NUM | EV (123006, DCM, "Radionuclide Total Dose") | 1 | M |  | DCID 3083 “Units of Radioactivity”
5 | > | CONTAINS | DATETIME | EV (123003, DCM, "Radiopharmaceutical Start Time") | 1 | M |  |
6 | > | CONTAINS | NUM | DT (122711, DCM, "Exercise duration after stress agent injection") | 1 | U |  | UNITS = DT (min, UCUM, "min")
7 | > | CONTAINS | DATETIME | EV (122712, DCM, "Imaging Start Time") | 1 | M |  |
8 | > | CONTAINS | CODE | EV (122713, DCM, "Attenuation correction") | 1 | U |  | BCID 3112 “Attenuation Correction”
9 | >> | HAS PROPERTIES | CODE | EV (111001, DCM, "Algorithm Name") | 1 | U |  | BCID 3117 “Attenuation Correction Methods”
10 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | BCID 3113 “Types of Perfusion Defects”
11 | >> | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | BCID 3717 “Myocardial Wall Segments”
12 | >> | HAS PROPERTIES | CODE | EV (112025, DCM, "Size Descriptor") | 1 | M |  | BCID 252 “S-M-L Size Descriptor”
13 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | M |  | BCID 3716 “Severity”
14 | > | CONTAINS | CODE | EV (F-02220, SRT, "Left Ventricular Function") | 1 | U |  | BCID 3119 “LV Function”
15 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | BCID 3716 “Severity”
16 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (10230-1, LN, "LV Ejection Fraction") $Units = EV (%, UCUM, "%") $Derivation = DT (R-41D2D, SRT, "Calculated")
17 | > | CONTAINS | CODE | EV (F-02236, SRT, "Left Ventricular Size") | 1 | U |  | BCID 3122 “Ventricular Enlargement”
18 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8821-1, LN, "Left Ventricular ED Volume") $Units = EV (ml, UCUM, "ml")
19 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8823-7, LN, "Left Ventricular ES Volume") $Units = EV (ml, UCUM, "ml")
20 | > | CONTAINS | INCLUDE | DTID 5204 “Wall Motion Analysis” | 1 | U |  | $Procedure = DCID 3108 “NM/PET Procedures”

### TID 3309 Stress Echo Measurement Group

Type:

  Extensible

Order:

  Significant

Table TID 3309. Stress Echo Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (P5-B3000, SRT, "Echocardiography")
3 | > | CONTAINS | INCLUDE | DTID 5203 “Echo Measurement” | 1-n | U |  | $Measurement = DCID 12200 “Echocardiography Left Ventricle” $Method = CID 12227 “Echocardiography Measurement Method”
4 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | IF Row 3 measurement concept is in CID 12222 “Orifice Flow Properties” | EV (T-32600, SRT, "Left Ventricle")
5 | > | CONTAINS | INCLUDE | DTID 5203 “Echo Measurement” | 1-n | U |  | $Measurement = DCID 12211 “Echocardiography Aortic Valve” $Method = CID 12227 “Echocardiography Measurement Method”
6 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | IF Row 5 measurement concept is in CID 12222 “Orifice Flow Properties” | EV (T-35400, SRT, "Aortic Valve")
7 | > | CONTAINS | INCLUDE | DTID 5203 “Echo Measurement” | 1-n | U |  | $Measurement = DCID 12207 “Echocardiography Mitral Valve” $Method = CID 12227 “Echocardiography Measurement Method”
8 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | IF Row 7 measurement concept is in CID 12222 “Orifice Flow Properties” | EV (T-35300, SRT, "Mitral Valve")
9 | > | CONTAINS | INCLUDE | DTID 5203 “Echo Measurement” | 1-n | U |  | $Measurement = DCID 12208 “Echocardiography Tricuspid Valve” $Method = CID 12227 “Echocardiography Measurement Method”
10 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | IF Row 9 measurement concept is in CID 12222 “Orifice Flow Properties” | EV (T-35100, SRT, "Tricuspid Valve")
11 | > | CONTAINS | INCLUDE | DTID 5204 “Wall Motion Analysis” | 1 | U |  | $Procedure = DT (P5-B3121, SRT, "Echocardiography for Determining Ventricular Contraction")

Content Item Descriptions

Rows 3-10 | These invocations of TID 5203 “Echo Measurement” do not include an inherited Findings Site concept, for example as in the invocations of TID 5203 “Echo Measurement” from TID 5202 “Echo Section”. Echo measurements that do not have the associated Finding Site pre-coordinated in the measurement concept (i.e., the orifice flow measurements of CID 12222 “Orifice Flow Properties”), shall have the Finding Site explicitly post-coordinated with a Concept Modifier (Rows 4, 6, 8, and 10). This template does not include the concept of an Image Library, for example as used in TID 5200 “Echocardiography Procedure Report”. Image Content Items in the Echo Measurement template shall be included with by-value relationships, not with by-reference relationships.

### TID 3311 Stress Test Summary

Type:

  Extensible

Order:

  Significant

Table TID 3311. Stress Test Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121111, DCM, "Summary") | 1 | U |  |
3 | > | CONTAINS | INCLUDE | DTID 3312 “Physiological Summary” | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 3313 “Stress ECG Summary” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 3317 “Stress Imaging Summary” | 1 | U |  |
6 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (D3-13025, SRT, "Exercise-induced angina") See note.
7 | >> |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | M |  |
8 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3220 “Stress Symptoms”
9 | > | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1 | U |  |
10 | > | CONTAINS | CODE | DT (G-0180, SRT, "Reason for stopping test") | 1 | U |  | DCID 3221 “Stress Test Termination Reasons”
11 | > | CONTAINS | NUM | DT (122715, DCM, "Pharmacological Stress Agent Dose") | 1 | U |  | DT (mg/kg, UCUM, "mg/kg")

Content Item Descriptions

Rows 6-7 | These rows allow an explicit finding of presence or absence of exercise-induced angina through the TID 1350 “Negation Modifier, Presence of Finding” Concept Modifier "Presence of property"

### TID 3312 Physiological Summary

Type:

  Extensible

Order:

  Significant

Table TID 3312. Physiological Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | DT (40443-4, LN, "Resting Heart Rate") | 1 | M |  | UNITS = DT ({H.B.}/min, UCUM, "BPM")
2 |  |  | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | M |  | UNITS = DCID 3500 “Pressure Units”
3 | > | HAS CONCEPT MOD | CODE | EV (109054, DCM, "Patient State") | 1 | M |  | DT (F-01604, SRT, "Resting State")
4 |  |  | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | M |  | UNITS = DCID 3500 “Pressure Units”
5 | > | HAS CONCEPT MOD | CODE | EV (109054, DCM, "Patient State") | 1 | M |  | DT (F-01604, SRT, "Resting State")
6 |  |  | NUM | DT (F-04F92, SRT, "Target HR") | 1 | M |  | UNITS = DT ({H.B.}/min, UCUM, "BPM")
7 |  |  | NUM | DT (F-04FA6, SRT, "Maximum HR Achieved") | 1 | M |  | UNITS = DT ({H.B.}/min, UCUM, "BPM")
8 |  |  | NUM | DT (F-04FA6, SRT, "Maximum HR Achieved") | 1 | M |  | UNITS = EV (%, UCUM, "%")
9 | > | HAS CONCEPT MOD | CODE | EV (121425, DCM, "Index") | 1 | M |  | DT (F-04F92, SRT, "Target HR")
10 |  |  | NUM | DT (122716, DCM, "Maximum Power Output Achieved") | 1 | U |  | UNITS = DT (W, UCUM, "Watts")
11 |  |  | NUM | DT (122717, DCM, "Peak activity workload") | 1 | U |  | UNITS = DT ([MET], UCUM, "METS")
12 |  |  | CODE | DT (F-04F9F, SRT, "HR Response") | 1 | U |  | DCID 3210 “Speed of Response”
13 |  |  | NUM | DT (F-00E11, SRT, "Maximum systolic blood pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
14 |  |  | NUM | DT (F-00E21, SRT, "Maximum diastolic blood pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
15 |  |  | CODE | DT (F-04F74, SRT, "BP Response") | 1 | U |  | DCID 3210 “Speed of Response”
16 |  |  | NUM | DT (122718, DCM, "Peak Double Product") | 1 | U |  | UNITS = DT (mm[Hg].{H.B.}/min, UCUM, "mmHg.BPM")
17 |  |  | NUM | DT (F-031F8, SRT, "Total Exercise duration") | 1 | U |  | UNITS = DT (min, UCUM, "min")
18 |  |  | NUM | DT (F-031F7, SRT, "Total test duration") | 1 | U |  | UNITS = DT (min, UCUM, "min")
19 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DT (F-04FCC, SRT, "Functional capacity") See note.
20 |  |  | TEXT | DT (F-04FCC, SRT, "Functional capacity") | 1 | U |  |
21 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (122760, DCM, "Stress test score") $Method = BCID 3238 “Stress Scoring Scales”
22 |  |  | NUM | DT (F-04FCA, SRT, "Heart rate recovery time") | 1 | U |  | UNITS = DT (s, UCUM, "s")
23 |  |  | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | BCID 3213 “Stress Hemodynamic Findings”
24 |  |  | CODE | EV (F-00F4E, SRT, "Cardiovascular event risk") | 1 | U |  | BCID 3118 “Level of Risk”

Content Item Descriptions

Row 22 | Numerical scoring of a patient's functional capacity shall include the range of the scoring system in the Units of Measurement (see Section 7.2.2), and may include a coded identifier for the scoring system in the Method concept modifier of TID 300 “Measurement”.

### TID 3313 Stress ECG Summary

Type:

  Extensible

Order:

  Significant

Table TID 3313. Stress ECG Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (F-00033, SRT, "ECG Finding") | 1 | U |  | Device Generated Test Summary
2 | > | HAS OBS CONTEXT | CODE | EV (121005, DCM, "Observer Type") | 1 | M |  | EV (121007, DCM, "Device")
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1004 “Device Observer Identifying Attributes” | 1 | M |  |
4 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (F-03204, SRT, "ST Elevation") $Units = DT (mV, UCUM, "mV") $Derivation = EV (G-A437, SRT, "Maximum") $TargetSite = DCID 3001 “ECG Leads”
5 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (F-38279, SRT, "ST Depression") $Units = DT (mV, UCUM, "mV") $Derivation = EV (G-A437, SRT, "Maximum") $TargetSite = DCID 3001 “ECG Leads”
6 |  |  | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (F-38287, SRT, "T wave alternans") $Units = DT (uV, UCUM, "uV") $Derivation = EV (G-A437, SRT, "Maximum") $TargetSite = DCID 3001 “ECG Leads”
7 |  |  | CODE | EV (F-38035, SRT, "ST Segment Finding") | 1 | U |  | BCID 3231 “ST Segment Findings”
8 | > | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | U |  | BCID 3232 “ST Segment Location”
9 | > | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | U |  | BCID 3233 “ST Segment Morphology”
10 |  |  | NUM | DT (122707, DCM, "Number of Ectopic Beats") | 1 | U |  | UNITS = EV ({beats}, UCUM, "beats")
11 | > | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | U |  | BCID 3234 “Ectopic Beat Morphology”
12 |  |  | CODE | DT (8884-9, LN, "Cardiac Rhythm") | 1-2 | U |  | BCID 3415 “Cardiac Rhythms” See note.
13 | > | HAS CONCEPT MOD | CODE | EV (109054, DCM, "Patient State") | 1 | M |  | DCID 3102 “Rest-Stress”
14 |  |  | CODE | EV (F-00033, SRT, "ECG Finding") | 1-n | U |  | BCID 3230 “ECG Findings”
15 | > | HAS CONCEPT MOD | CODE | EV (109054, DCM, "Patient State") | 1 | U |  | BCID 3262 “ECG Patient State Values”

Content Item Descriptions

Rows 4-14 | Each observation (measurement or finding) may have a specific Content Item Observation DateTime attribute to indicate the time in the procedure at which the observation was made (e.g., time of maximum heart rate, or time of occurrence of an arrhythmia).
Row 12-13 | This Concept and the associated Concept Modifier may be instantiated twice, once for resting state measurements, once for stress.

### TID 3317 Stress Imaging Summary

Type:

  Extensible

Order:

  Significant

Table TID 3317. Stress Imaging Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | DT (122739, DCM, "Overall study quality") | 1 | M |  | BCID 3114 “Study Quality”
2 |  |  | CODE | DT (113010, DCM, "Quality Issue") | 1 | U |  | BCID 3115 “Stress Imaging Quality Issues”
3 |  |  | CODE | EV (121071, DCM, "Finding") | 1 | U |  | BCID 3116 “NM Extracardiac Findings”
4 |  |  | INCLUDE | DTID 300 “Measurement” | 1-2 | U |  | $Measurement = DT (F-04F76, SRT, "Perfusion defect extent") $ModType = EV (109054, DCM, "Patient State") $ModValue = DCID 3102 “Rest-Stress” $Units = EV (%, UCUM, "%") See note.
5 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DT (F-04FCD, SRT, "Stress ischemia extent") $Units = EV (%, UCUM, "%")
6 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DT (122762, DCM, "Number of diseased vessel territories") $Units = EV ({territories}, UCUM, "territories")
7 |  |  | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (D3-13040, SRT, "Coronary artery disease")
8 | > | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | M |  | BCID 3016 “Major Coronary Arteries”
9 |  |  | CODE | EV (F-0238D, SRT, "Myocardial perfusion") | 1-2 | U |  | BCID 3120 “Perfusion Findings”
10 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | BCID 3463 “Ventricle Identification”
11 | > | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | U |  | BCID 3121 “Perfusion Morphology”
12 |  |  | CODE | EV (121071, DCM, "Finding") | 1 | U |  | DT (D4-31124, SRT, "Transient cavitary dilatation")
13 | > |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
14 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DT (F-04FB4, SRT, "Transient cavitary dilatation ratio") $Units = EV ({ratio}, UCUM, "ratio")
15 |  |  | INCLUDE | DTID 300 “Measurement” | 1-2 | U |  | $Measurement = EV (10230-1, LN, "LV Ejection Fraction") $ModType = EV (109054, DCM, "Patient State") $ModValue = DCID 3102 “Rest-Stress” $Units = EV (%, UCUM, "%") See note.

Content Item Descriptions

Row 4 | This row may be instantiated twice, once for resting state measurements, once for stress.
Row 15 | The LVEF code specified in this row is defined in LOINC with method "imaging". LVEF measurement by ultrasound may also be encoded elsewhere in the Content Tree (e.g., in TID 3309 “Stress Echo Measurement Group”) with LOINC code 18043-0, which has method "ultrasound". It is recommended that such findings from the per-phase measurements be summarized here with the generic "LVEF by Imaging" concept code.

### TID 3318 Comparison to Prior Stress Exam

This template describes changes in findings from a prior stress exam. Comparison is to only one prior exam, even though the generic concept name for the template uses the plural "exams".

Type:

  Extensible

Order:

  Significant

Table TID 3318. Comparison to Prior Stress Exam

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111424, DCM, "Comparison to previous exams") | 1 | M |  |
2 | > | CONTAINS | CODE | DT (121058, DCM, "Procedure Reported") | 1 | U |  | DCID 3200 “Stress Test Procedure”
3 | >> | HAS PROPERTIES | DATETIME | EV (122146, DCM, "Procedure DateTime") | 1 | U |  |
4 | >> | HAS PROPERTIES | UIDREF | EV (121018, DCM, "Procedure Study Instance UID") | 1 | U |  |
5 | >> | HAS PROPERTIES | COMPOSITE | EV (122075, DCM, "Prior report for current patient") | 1-n | U |  |
6 | > | CONTAINS | CODE | DT (F-03D1D, SRT, "Exercise tolerance") | 1 | U |  | BCID 3236 “Tolerance Comparison Findings”
7 | > | CONTAINS | CODE | DT (F-0238D, SRT, "Myocardial Perfusion") | 1 | U |  | BCID 3235 “Perfusion Comparison Findings”
8 | >> | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1-n | U |  | BCID 3215 “Perfusion Finding Method”
9 | > | CONTAINS | CODE | DT (F-02225, SRT, "LV Wall motion") | 1-n | U |  | BCID 3237 “Wall Motion Comparison Findings”
10 | >> | HAS CONCEPT MOD | CODE | EV (109054, DCM, "Patient State") | 1 | M |  | DCID 3102 “Rest-Stress”
11 | > | CONTAINS | CODE | EV (111424, DCM, "Comparison to previous exams") | 1 | U |  | BCID 3217 “Comparison Finding”
12 | > | CONTAINS | NUM | DT (122768, DCM, "Difference in Ejection Fraction") | 1 | U |  | UNITS = EV (%, UCUM, "%")
13 | > | CONTAINS | NUM | DT (122769, DCM, "Difference in ED LV Volume") | 1 | U |  | UNITS = EV (ml, UCUM, "ml")
14 | > | CONTAINS | NUM | DT (122769, DCM, "Difference in ED LV Volume") | 1 | U |  | UNITS = EV (ml/m2, UCUM, "ml/m2")
15 | >> | HAS CONCEPT MOD | CODE | EV (121425, DCM, "Index") | 1 | M |  | DT (8277-6, LN, "BSA")

### TID 3320 Conclusions and Recommendations

Type:

  Extensible

Order:

  Non-Significant

Table TID 3320. Conclusions and Recommendations

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121076, DCM, "Conclusions") | 1 | MC | IF Completion Flag (0040,A491) = COMPLETE |
2 | > | CONTAINS | TEXT | EV (121077, DCM, "Conclusion") | 1 | U |  |
3 | > | CONTAINS | CODE | EV (F-00033, SRT, "ECG Finding") | 1 | M |  | DCID 3208 “Summary Codes Exercise ECG”
4 | > | CONTAINS | CODE | EV (F-01969, SRT, "Imaging Finding") | 1 | M |  | DCID 3209 “Summary Codes Stress Imaging”
5 |  |  | CONTAINER | EV (121074, DCM, "Recommendations") | 1 | U |  |
6 | > | CONTAINS | TEXT | EV (121075, DCM, "Recommendation") | 1 | U |  |

## Hemodynamics Report Templates

The templates that comprise the Hemodynamic Report are interconnected as shown in Figure A-6.

Figure A-6. Hemodynamic Report Template Hierarchy

Figure A-6 shows only the use of templates specific to the Hemodynamic Report; it does not show common structural templates such as TID 300 “Measurement”.

### Note

Figure A-6 shows the relationship of templates; it does not show the structural hierarchy of Content Items in the IOD. See Figure L-1 “Hemodynamics Report Structure” in PS3.17 .

### TID 3500 Hemodynamics Report

The Hemodynamic Report template is the root structure for the representation of measurements acquired during a procedure in a cardiac catheterization lab.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 3500. Hemodynamics Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122120, DCM, "Hemodynamics Report") | 1 | M |  | Root node
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
3 | > |  | INCLUDE | DTID 3601 “Procedure Context” | 1 | M |  |
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | M |  |
5 | > | HAS ACQ CONTEXT | INCLUDE | DTID 3603 “Procedure Environmental Characteristics” | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3501 “Hemodynamics Measurement Group” | 1-n | M |  |
7 | > | CONTAINS | INCLUDE | DTID 3570 “Summary, Hemodynamics” | 1 | U |  |

### TID 3501 Hemodynamics Measurement Group

The Hemodynamic Measurement Group template provides a structure for measurements acquired during a single procedure phase in a cardiac catheterization lab.

Type:

  Extensible

Order:

  Significant

Table TID 3501. Hemodynamics Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS ACQ CONTEXT | CODE | EV (G-72BB, SRT, "Catheterization Procedure Phase") | 1 | M |  | DCID 3651 “Hemodynamic Measurement Phase”
3 | > | HAS ACQ CONTEXT | INCLUDE | DTID 3520 “Hemodynamic Clinical Context” | 1 | U |  |
4 | > | HAS ACQ CONTEXT | TEXT | EV (121124, DCM, "Procedure Action ID") | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 3510 “Vital Signs” | 1-n | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3504 “Arterial Pressure Measurement” | 1-n | U |  |
7 | > | CONTAINS | INCLUDE | DTID 3505 “Atrial Pressure Measurement” | 1-n | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3506 “Venous Pressure Measurement” | 1-n | U |  |
9 | > | CONTAINS | INCLUDE | DTID 3507 “Ventricular Pressure Measurement” | 1-n | U |  |
10 | > | CONTAINS | INCLUDE | DTID 3508 “Gradient Measurement” | 1-n | U |  |
11 | > | CONTAINS | INCLUDE | DTID 3509 “Blood Velocity Measurement” | 1-n | U |  |
12 | > | CONTAINS | INCLUDE | DTID 3515 “Cardiac Output Measurement by Indicator Dilution” | 1-n | U |  |
13 | > | CONTAINS | INCLUDE | DTID 3516 “Blood Lab Measurements” | 1-n | U |  |
14 | > | CONTAINS | INCLUDE | DTID 3560 “Derived Hemodynamic Measurements” | 1-n | U |  |
15 | > | CONTAINS | INCLUDE | DTID 3714 “ECG Lead Measurements” | 1-n | U |  |

Content Item Descriptions

Row 4 | Procedure Action ID allows linkage between the hemodynamic measurements recorded in this Template and a procedure step or phase recorded in the Procedure Log, e.g., using TID 3100 “Procedure Action”.

### TID 3504 Arterial Pressure Measurement

The Arterial Pressure Measurement Template consists of a CONTAINER, with an acquisition context of the measurement anatomic site, and containing systolic, diastolic, and mean measurements. This implies that the name of the measurement is effectively post-coordinated from the measurement name, the Hemodynamic Measurement Group container (procedure phase) name, and the acquisition context (finding site) value.

Type:

  Extensible

Order:

  Significant

Table TID 3504. Arterial Pressure Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (P2-36102, SRT, "Arterial pressure measurements") | 1 | M |  |
2 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | M |  | $LocationName = EV (G-C0E3, SRT, "Finding Site") $LocationValue = DCID 3606 “Arterial Source Locations”
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (8480-6, LN, "Intravascular arterial Systolic pressure") $Units = DCID 3500 “Pressure Units”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (8462-4, LN, "Intravascular arterial Diastolic pressure") $Units = DCID 3500 “Pressure Units”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (8478-0, LN, "Intravascular arterial mean pressure") $Units = DCID 3500 “Pressure Units”
6 | > | CONTAINS | INCLUDE | DTID 3550 “Pressure Waveform Measurements” | 1-n | U |  |

### TID 3505 Atrial Pressure Measurement

The Atrial Pressure Measurement Template consists of a CONTAINER, with an acquisition context of the measurement anatomic site, and containing a-wave, v-wave, and mean measurements. This implies that the name of the measurement is effectively post-coordinated from the measurement name, the Hemodynamic Measurement Group container (procedure phase) name, and the acquisition context (finding site) value.

Type:

  Extensible

Order:

  Significant

Table TID 3505. Atrial Pressure Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122121, DCM, "Atrial pressure measurements") | 1 | M |  |
2 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | M |  | $LocationName = EV (G-C0E3, SRT, "Finding Site") $LocationValue = DCID 3608 “Atrial Source Locations”
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (109016, DCM, "A-wave peak pressure") $Units = DCID 3500 “Pressure Units”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (109034, DCM, "V-wave peak pressure") $Units = DCID 3500 “Pressure Units”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (F-31150, SRT, "Mean blood pressure") $Units = DCID 3500 “Pressure Units”
6 | > | CONTAINS | INCLUDE | DTID 3550 “Pressure Waveform Measurements” | 1-n | U |  |

### TID 3506 Venous Pressure Measurement

The Venous Pressure Measurement Template consists of a CONTAINER, with an acquisition context of the measurement anatomic site, and containing a mean measurement. This implies that the name of the measurement is effectively post-coordinated from the measurement name, the Hemodynamic Measurement Group container (procedure phase) name, and the acquisition context (finding site) value.

Type:

  Extensible

Order:

  Significant

Table TID 3506. Venous Pressure Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (P2-36110, SRT, "Venous pressure measurements") | 1 | M |  |
2 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | M |  | $LocationName = EV (G-C0E3, SRT, "Finding Site") $LocationValue = DCID 3607 “Venous Source Locations”
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (F-31150, SRT, "Mean blood pressure") $Units = DCID 3500 “Pressure Units”
4 | > | CONTAINS | INCLUDE | DTID 3550 “Pressure Waveform Measurements” | 1-n | U |  |

### TID 3507 Ventricular Pressure Measurement

The Ventricular Pressure Measurement Template consists of a CONTAINER, with an acquisition context of the measurement anatomic site, and containing systolic and end-diastolic measurements. This implies that the name of the measurement is effectively post-coordinated from the measurement name, the Hemodynamic Measurement Group container (procedure phase) name, and the acquisition context (finding site) value.

Type:

  Extensible

Order:

  Significant

Table TID 3507. Ventricular Pressure Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122122, DCM, "Ventricular pressure measurements") | 1 | M |  |
2 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | M |  | $LocationName = EV (G-C0E3, SRT, "Finding Site") $LocationValue = DCID 3609 “Ventricular Source Locations”
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | IFF $LocationValue selected in row 2 is (T-32600, SRT, "Left Ventricle") or subsite thereof | $Measurement = EV (F-03E0D, SRT, "Left Ventricular Systolic blood pressure") $Units = DCID 3500 “Pressure Units”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | IFF $LocationValue selected in row 2 is (T-32600, SRT, "Left Ventricle") or subsite thereof | $Measurement = EV (F-03E0E, SRT, "Left Ventricular End Diastolic pressure") $Units = DCID 3500 “Pressure Units”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | IFF $LocationValue selected in row 2 is (T-32500, SRT, "Right Ventricle") or subsite thereof | $Measurement = EV (F-03DFE, SRT, "Right Ventricular Systolic blood pressure") $Units = DCID 3500 “Pressure Units”
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | IFF $LocationValue selected in row 2 is (T-32500, SRT, "Right Ventricle") or subsite thereof | $Measurement = EV (F-03E02, SRT, "Right Ventricular End Diastolic pressure") $Units = DCID 3500 “Pressure Units”
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | IFF $LocationValue selected in row 2 is (T-32400, SRT, "Common Ventricle") | $Measurement = EV (122194, DCM, "Ventricular Systolic blood pressure") $Units = DCID 3500 “Pressure Units”
8 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | IFF $LocationValue selected in row 2 is (T-32400, SRT, "Common Ventricle") | $Measurement = EV (122191, DCM, "Ventricular End Diastolic pressure") $Units = DCID 3500 “Pressure Units”
9 | > | CONTAINS | INCLUDE | DTID 3550 “Pressure Waveform Measurements” | 1-n | U |  |

### TID 3508 Gradient Measurement

The Gradient Measurement Template consists of a CONTAINER, with an acquisition context of the measurement anatomic site, and containing the gradient measurement. This implies that the name of the measurement is effectively post-coordinated from the measurement name, the Hemodynamic Measurement Group container (procedure phase) name, and the acquisition context (finding site) value.

Type:

  Extensible

Order:

  Significant

Table TID 3508. Gradient Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122123, DCM, "Gradient assessment") | 1 | M |  |
2 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | MC | XOR with Rows 3 & 4 IFF single location is appropriate | $LocationName = EV (G-C0E3, SRT, "Finding Site") $LocationValue = DCID 3610 “Gradient Source Locations”
3 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | MC | XOR with Row 2 IFF a dual location is appropriate | $LocationName = EV (121116, DCM, "Proximal Finding Site") $LocationValue = DCID 3630 “Cardiovascular Anatomic Locations”
4 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | MC | XOR with Row 2 IFF a dual location is appropriate | $LocationName = EV (121117, DCM, "Distal Finding Site") $LocationValue = DCID 3630 “Cardiovascular Anatomic Locations”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = EV (F-023F7, SRT, "Pressure Gradient") $Units = DCID 3500 “Pressure Units” $Derivation = DCID 3627 “Measurement Type”
6 | > | CONTAINS | INCLUDE | DTID 3550 “Pressure Waveform Measurements” | 1-n | U |  |

Content Item Descriptions

Row 5 | Is used to encode the gradient measurements. Uses TID 300 “Measurement” for the content item structure of the measurements. That template allows an INFERRED FROM relationship to other numeric measurements used in the computation or derivation of the current measurement. In the case of a gradient measurement, the pressure measurements at the distal and proximal locations may thus be explicitly conveyed.

### TID 3509 Blood Velocity Measurement

The Blood Velocity Measurement Template consists of a CONTAINER, with an acquisition context of the measurement anatomic site, and containing primary (not derived) velocity measurements, e.g., from a Doppler catheter. Derived velocity measurements may be recorded using TID 3560 “Derived Hemodynamic Measurements”.

Type:

  Extensible

Order:

  Significant

Table TID 3509. Blood Velocity Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122124, DCM, "Blood velocity measurements") | 1 | M |  |
2 | > |  | INCLUDE | DTID 3530 “Hemodynamic Acquisition Context” | 1 | M |  | $LocationName = EV (G-C0E9, SRT, "Procedure site") $LocationValue = BCID 3606 “Arterial Source Locations”
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = DCID 3612 “Blood Velocity Measurements” $Units = EV (mm/s, UCUM, "mm/s")
4 | > | CONTAINS | INCLUDE | DTID 3550 “Pressure Waveform Measurements” | 1-n | U |  |

### TID 3510 Vital Signs

The Vital Signs template consists of a CONTAINER containing the various vital signs measurements. These measurements may be acquired automatically from patient monitoring equipment, or may be entered based on manual measurements.

Type:

  Extensible

Order:

  Significant

Table TID 3510. Vital Signs

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (8716-3, LN, "Vital Signs") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-008EC, SRT, "Systolic blood pressure") $Units = DCID 3500 “Pressure Units” $Method = BCID 3560 “Blood Pressure Methods”
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-008ED, SRT, "Diastolic blood pressure") $Units = DCID 3500 “Pressure Units”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8867-4, LN, "Heart rate") $Units = EV ({H.B.}/min, UCUM, "BPM")
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8310-5, LN, "Body temperature") $Units = EV (Cel, UCUM, "C")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = DCID 3526 “Blood Gas Saturation” $Units = EV (%, UCUM, "%")
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-043E7, SRT, "Respiration rate") $Units = EV (/min, UCUM, "breaths/min")
8 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122195, DCM, "Pulse Strength") $Units = DT ({0:4}, UCUM, "range 0:4")
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-009EA, SRT, "Pain Score") $Units = DT ({1:10}, UCUM, "range 1:10")
10 | > | CONTAINS | CODE | DT (8884-9, LN, "Cardiac Rhythm") | 1 | U |  | BCID 3415 “Cardiac Rhythms”
11 | > | CONTAINS | CODE | DT (9304-7, LN, "Respiration Rhythm") | 1 | U |  | BCID 3416 “Respiration Rhythms”

### TID 3515 Cardiac Output Measurement by Indicator Dilution

The Cardiac Output Measurement by Indicator Dilution template consists of a CONTAINER containing the measurement and significant parameters of the technical method. If the measurement is based on a DICOM Hemodynamic Waveform SOP Instance, that object may also be referenced.

### Note

Fick Cardiac Output is encoded in TID 3560 “Derived Hemodynamic Measurements”.

Type:

  Extensible

Order:

  Significant

Table TID 3515. Cardiac Output Measurement By Indicator Dilution

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (P2-34201, SRT, "Cardiac Output measurement") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (8737-9, LN, "Cardiac Output by Indicator Dilution") $Units = EV (l/min, UCUM, "l/min") $Method = DCID 3628 “Cardiac Output Methods” $WavePurpose = DT (121112, DCM, "Source of Measurement")
3 | > | HAS ACQ CONTEXT | NUM | EV (122319, DCM, "Catheter Size") | 1 | MC | IFF Row 2 indicates a thermal method | UNITS = DCID 3510 “Catheter Size Units”
4 | > | HAS ACQ CONTEXT | NUM | EV (122320, DCM, "Injectate Temperature") | 1 | MC | IFF Row 2 indicates a thermal method | UNITS = EV (Cel, UCUM, "C")
5 | > | HAS ACQ CONTEXT | NUM | EV (122321, DCM, "Injectate Volume") | 1 | M |  | UNITS = DT (ml, UCUM, "ml")
6 | > | HAS ACQ CONTEXT | NUM | EV (122322, DCM, "Calibration Factor") | 1 | M |  | UNITS = DT (1, UCUM, "no units")

### TID 3516 Blood Lab Measurements

The Blood Lab Measurements template provides for the recording of measurements made on blood samples obtained during a catheterization procedure. The type and anatomic source of the blood is recorded as acquisition context. The results from the blood chemistry measurement system are quoted; the measurement names may be pre-coordinated with the type or source of the blood, or generic measurement names may be reported. In the latter case, the full measurement concept name may be effectively post-coordinated using the recorded acquisition context.

Type:

  Extensible

Order:

  Significant

Table TID 3516. Blood Lab Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122125, DCM, "Blood lab measurements") | 1 | M |  |
2 | > | HAS ACQ CONTEXT | CODE | EV (R-00254, SRT, "Specimen Type") | 1 | M |  | DCID 3520 “Blood Source Type”
3 | > | HAS ACQ CONTEXT | CODE | EV (G-C0E9, SRT, "Procedure site") | 1 | M |  | BCID 3630 “Cardiovascular Anatomic Locations”
4 | > |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (718-7, LN, "Hemoglobin") $Units = DT (g/dl, UCUM, "g/dl")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3524 “Blood Gas Pressures” $Units = DCID 3500 “Pressure Units”
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3525 “Blood Gas Content” $Units = DT (ml/dl, UCUM, "ml/dl")
8 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3528 “Blood pH” $Units = EV ([pH], UCUM, "pH")
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3526 “Blood Gas Saturation” $Units = EV (%, UCUM, "%")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3527 “Blood Base Excess” $Units = DT (meq/dl, UCUM, "meq/dl")
11 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122183, DCM, "Blood temperature") $Units = EV (Cel, UCUM, "C")

### TID 3520 Hemodynamic Clinical Context

The Clinical Context template allows the recording of information about the patient's clinical state that may affect interpretation of the hemodynamic measurements. This Template includes several coded conditions (e.g., complications, drugs, physiological challenges, etc.), each of which may have a Concept Modifier of "Relative Time". This Modifier indicates the temporal relationship of the hemodynamic measurements to the conditions recorded in this template. E.g., the Content Item structure:

[CONTAINER] "Findings"

>HAS ACQ CONTEXT "Cath Procedure Phase" "Post-intervention phase"

>HAS ACQ CONTEXT [CONTAINER] "Clinical Context"

>>CONTAINS [CODE] "Complication" "Arrhythmia"

>>>HAS CONCEPT MOD [CODE] "Relative Time" "After"

>CONTAINS [CONTAINER] "Arterial Measurements"…

conveys the semantics that this group of measurements for the post-intervention phase of a cath procedure was made after an occurrence of arrhythmia in the patient.

In the absence of a Relative Time modifier, the acquisition context is during the identified event or state.

Type:

  Extensible

Order:

  Significant

Table TID 3520. Hemodynamic Clinical Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122127, DCM, "Clinical Context") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (109054, DCM, "Patient State") | 1-n | U |  | BCID 3602 “Hemodynamic Patient State”
3 | > | CONTAINS | TEXT | EV (109054, DCM, "Patient State") | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (DD-60002, SRT, "Complication of Procedure") $ConditionValue = DCID 3413 “Adverse Outcomes”
5 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (122086, DCM, "Contrast administered") $ConditionValue = BCID 12 “Radiographic Contrast Agent”
6 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (122083, DCM, "Drug administered") $ConditionValue = BCID 10 “Interventional Drug”
7 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (109059, DCM, "Physiological challenges") $ConditionValue = BCID 3271 “Hemodynamic Physiological Challenges”
8 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (122138, DCM, "Circulatory Support") $ConditionValue = DCID 3553 “Circulatory Support”
9 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (P2-2200A, SRT, "Ventilatory assistance") $ConditionValue = DCID 3554 “Ventilation”
10 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (P2-35000, SRT, "Cardiac Pacing") $ConditionValue = BCID 3555 “Pacing”
11 | > | CONTAINS | INCLUDE | DTID 3521 “Relative Time” | 1-n | U |  | $ConditionName = EV (P0-0000, SRT, "Procedure") $ConditionValue = BCID 3405 “Procedure Action Values”

### TID 3521 Relative Time

The Relative Time template is invoked by 3520 Hemodynamic Acquisition Context template. It specifies an applicable clinical context condition and the time of the current measurements relative to that condition.

Table TID 3521. Parameters

Parameter Name | Parameter Usage
--- | ---
$ConditionName | Coded term for Concept Name of condition
$ConditionValue | Context Group for condition values

Type:

  Non-Extensible

Order:

  Significant

Table TID 3521. Relative Time

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | $ConditionName | 1 | M |  | $ConditionValue
2 | > | HAS CONCEPT MOD | CODE | EV (G-D709, SRT, "Relative time") | 1 | U |  | BCID 3600 “Relative Times”

### TID 3530 Hemodynamic Acquisition Context

The Hemodynamic Acquisition Context template is invoked by the various hemodynamic measurement templates.

Table TID 3530. Parameters

Parameter Name | Parameter Usage
--- | ---
$LocationName | Coded term for Concept Name of measurement location
$LocationValue | Context Group for appropriate anatomic locations

Type:

  Extensible

Order:

  Significant

Table TID 3530. Hemodynamic Acquisition Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS CONCEPT MOD | CODE | $LocationName | 1 | M |  | $LocationValue
2 | > | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | BCID 3019 “Cardiovascular Anatomic Location Modifiers”
3 |  | HAS ACQ CONTEXT | CODE | EV (G-C036, SRT, "Measurement Method ") | 1 | U |  | BCID 3241 “Hemodynamic Measurement Techniques”
4 |  | HAS ACQ CONTEXT | WAVEFORM | DT (121112, DCM, "Source of Measurement") | 1 | UC | XOR Row 5 |
5 |  | HAS ACQ CONTEXT | TCOORD | DT (121112, DCM, "Source of Measurement") | 1 | UC | XOR Row 4 |
6 | > | SELECTED FROM | WAVEFORM |  | 1 | M |  |

### TID 3550 Pressure Waveform Measurements

The Pressure Waveform Measurements template is invoked by the various hemodynamic measurement templates for recording general measurements made in conjunction with the specific required measurements for that anatomic location.

Type:

  Extensible

Order:

  Significant

Table TID 3550. Pressure Waveform Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3611 “Pressure Measurements” $Units = DCID 3500 “Pressure Units”
2 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3612 “Blood Velocity Measurements” $Units = EV (mm/s, UCUM, "mm/s")
3 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3613 “Hemodynamic Time Measurements” $Units = DT (ms, UCUM, "ms")
4 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32100, SRT, "Cardiac Output") $Units = EV (l/min, UCUM, "l/min")
5 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $Units = DT (ml, UCUM, "ml")
6 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (10230-1, LN, "LV Ejection Fraction") $Units = EV (%, UCUM, "%")
7 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8867-4, LN, "Heart rate") $Units = DT ({H.B.}/min, UCUM, "BPM")
8 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-043E7, SRT, "Respiration rate") $Units = DT (/min, UCUM, "breaths/min")
9 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (109025, DCM, "Max dp/dt") $Units = DT (mm[Hg]/s, UCUM, "mmHg/s")
10 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (109026, DCM, "Max neg dp/dt") $Units = DT (mm[Hg]/s, UCUM, "mmHg/s")
11 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122190, DCM, "Max dp/dt/P") $Units = DT (/s, UCUM, "/s")
12 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122192, DCM, "Indicator appearance time") $Units = DT (s, UCUM, "s")
13 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122193, DCM, "Maximum pressure acceleration") $Units = DT (mm[Hg]/s2, UCUM, "mmHg/s/s")
14 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3617 “Valve Flows” $Units = DT (ml/min, UCUM, "ml/min")

### TID 3560 Derived Hemodynamic Measurements

The Derived Hemodynamic Measurements Template consists of a CONTAINER containing measurements derived from one or more other measurements. These measurements are associated with a particular procedure phase, but not necessarily with a particular anatomic location.

Type:

  Extensible

Order:

  Significant

Table TID 3560. Derived Hemodynamic Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122126, DCM, "Derived Hemodynamic Measurements") | 1 | U |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3614 “Valve Areas, Non-mitral” $Units = EV (cm2, UCUM, "cm2") $Equation = DT (122262, DCM, "Area = Flow / 44.5 * sqrt(Gradient[mmHg]) ")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-02320, SRT, "Mitral Valve Area") $Units = EV (cm2, UCUM, "cm2") $Equation = DT (122263, DCM, "MVA = Flow / 38.0 * sqrt(Gradient[mmHg]) ")
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3615 “Valve Areas” $ModType = EV (121425, DCM, "Index") $ModValue = EV (8277-6, LN, "Body Surface Area") $Units = DT (cm2/m2, UCUM, "cm2/m2")
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3616 “Hemodynamic Period Measurements” $Units = DT (s/min, UCUM, "s/min")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3617 “Valve Flows” $Units = DT (ml/s, UCUM, "ml/s")
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32110, SRT, "Cardiac Index") $Units = DT (l/min/m2, UCUM, "l/min/m2")
8 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3529 “Arterial / Venous Content” $Units = DT (ml/dl, UCUM, "ml/dl")
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3618 “Hemodynamic Flows” $Units = DT (l/min, UCUM, "l/min")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8736-1, LN, "FICK Cardiac Output") $Units = DT (l/min, UCUM, "l/min")
11 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8750-2, LN, "FICK Cardiac Index") $Units = DT (l/min/m2, UCUM, "l/min/m2")
12 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122229, DCM, "Arteriovenous difference") $Units = DT (ml/dl, UCUM, "ml/dl")
13 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = BCID 3620 “Hemodynamic Ratios” $Units = DT ({ratio}, UCUM, "ratio")
14 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122237, DCM, "Corrected Sinus Node Recovery Time") $Units = DT (ms, UCUM, "ms")
15 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8861-7, LN, "Left Ventricular Stroke Work") $Units = DT (gf.m, UCUM, "gf.m")
16 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8862-5, LN, "Right Ventricular Stroke Work") $Units = DT (gf.m, UCUM, "gf.m")
17 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8863-3, LN, "Left Ventricular Stroke Work Index") $Units = DT (gf.m/m2, UCUM, "gf.m/m2")
18 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (8864-1, LN, "Right Ventricular Stroke Work Index") $Units = DT (gf.m/m2, UCUM, "gf.m/m2")
19 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122238, DCM, "Max volume normalized to 50mmHg pulse pressure") $Units = DT (ml, UCUM, "ml")
20 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122239, DCM, "Oxygen Consumption") $Units = DT (ml/min, UCUM, "ml/min") $Equation = BCID 3664 “Oxygen Consumption Equations and Tables”
21 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (19217-9, LN, "Oxygen partial pressure at 50% saturation (P50) ") $Units = DCID 3500 “Pressure Units” $Equation = BCID 3666 “P50 Equations”
22 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (11556-8, LN, "Blood Oxygen partial pressure") $Units = DCID 3500 “Pressure Units”
23 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3618 “Hemodynamic Flows” $ModType = EV (121425, DCM, "Index") $ModValue = EV (8277-6, LN, "Body Surface Area") $Units = DT (l/min/m2, UCUM, "l/min/m2")
24 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3619 “Hemodynamic Resistance Measurements” $Units = BCID 3502 “Hemodynamic Resistance Units”
25 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3619 “Hemodynamic Resistance Measurements” $ModType = EV (121425, DCM, "Index") $ModValue = EV (8277-6, LN, "Body Surface Area") $Units = BCID 3503 “Indexed Hemodynamic Resistance Units”
26 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122227, DCM, "Left to Right Flow") $Units = DT (l/min, UCUM, "l/min")
27 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122228, DCM, "Right to Left Flow") $Units = DT (l/min, UCUM, "l/min")
28 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $Units = DT (ml, UCUM, "ml")
29 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32120, SRT, "Stroke Volume") $ModType = EV (121425, DCM, "Index") $ModValue = EV (8277-6, LN, "Body Surface Area") $Units = DT (ml/m2, UCUM, "ml/m2")
30 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-042BA, SRT, "Total blood volume") $Units = DT (l, UCUM, "l")
31 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3667 “Framingham Scores” $Units = DT (%, UCUM, "%") $Equation = DCID 3668 “Framingham Tables”

### TID 3570 Summary, Hemodynamics

This Template allows the recording of any significant numeric measurements or findings.

Type:

  Extensible

Order:

  Significant

Table TID 3570. Summary, Hemodynamics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121111, DCM, "Summary") | 1 | U |  |
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  |
4 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3640 “Hypertension”
5 | >> |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
6 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
7 | > | CONTAINS | CODE | DCID 3641 “Hemodynamic Assessments” | 1-n | U |  | DCID 3642 “Degree Findings”
8 | >> | HAS PROPERTIES | CODE | EV (G-C0B2, SRT, "Condition") | 1 | U |  | EV (R-102B9, SRT, "Large v-wave")
9 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (R-102BA, SRT, "Diastolic pressure equalization")

### TID 3601 Procedure Context

The Procedure Context template describes acquisition context for measurements made or events recorded in a procedure.

Type:

  Extensible

Order:

  Significant

Table TID 3601. Procedure Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS ACQ CONTEXT | TEXT | EV (121065, DCM, "Procedure Description") | 1 | U |  | Defaults to Study Description (0008,1030) of the General Study Module
2 |  | HAS ACQ CONTEXT | CODE | EV (G-C0E8, SRT, "Has Intent") | 1 | U |  | BCID 3629 “Procedure Intent”
3 |  | HAS ACQ CONTEXT | CODE | EV (G-C09C, SRT, "Procedure Priority") | 1 | U |  | BCID 3414 “Procedure Urgency”
4 |  | HAS OBS CONTEXT | CODE | EV (121023, DCM, "Procedure Code") | 1-n | U |  | Defaults to Procedure Code Sequence (0008,1032) of General Study Module

### TID 3602 Cardiovascular Patient Characteristics

This template describes the characteristics of the patient that are specific to the current clinical presentation (visit). Patient Characteristic concepts in this template, which may replicate attributes in the Patient Study Module, are included here as possible targets of by-reference relationships from other content items in the SR tree.

### Note

Several of the concepts in this template duplicate concepts in TID 1007 “Subject Context, Patient”. The difference in use is that this template has those concepts as primary observations of the patient, while in TID 1007 “Subject Context, Patient” the concepts are used to set (or reset) the context for other observations.

Type:

  Extensible

Order:

  Significant

Table TID 3602. Cardiovascular Patient Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121118, DCM, "Patient Characteristics") | 1 | M |  |
2 | > | CONTAINS | NUM | EV (121033, DCM, "Subject Age") | 1 | M |  | UNITS = DCID 7456 “Units of Measure for Age”
3 | > | CONTAINS | CODE | EV (121032, DCM, "Subject Sex") | 1 | M |  | DCID 7455 “Sex”
4 | > | CONTAINS | NUM | EV (8302-2, LN, "Patient Height") | 1 | M |  | UNITS = EV (cm, UCUM, "cm")
5 | > | CONTAINS | NUM | EV (29463-7, LN, "Patient Weight") | 1 | M |  | UNITS = EV (kg, UCUM, "kg")
6 | > | CONTAINS | NUM | EV (122221, DCM, "Thorax diameter, sagittal") | 1 | U |  | UNITS = EV (cm, UCUM, "cm")
7 | > | CONTAINS | NUM | EV (8277-6, LN, "Body Surface Area") | 1 | MC | IF BSA used for indexed measurements in SOP Instance | UNITS = EV (m2, UCUM, "m2")
8 | >> | INFERRED FROM | CODE | EV (8278-4, LN, "Body Surface Area Formula") | 1 | U |  | BCID 3663 “Body Surface Area Equations”
9 | > | CONTAINS | NUM | EV (F-01860, SRT, "Body Mass Index") | 1 | U |  | UNITS = EV (kg/m2, UCUM, "kg/m2")
10 | >> | INFERRED FROM | CODE | EV (121420, DCM, "Equation") | 1 | U |  | DT (122265, DCM, "BMI = Wt/Ht^2")
11 | > | CONTAINS | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  | UNITS = EV ({H.B.}/min, UCUM, "BPM")
12 | > | CONTAINS | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
13 | > | CONTAINS | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
14 | > | CONTAINS | CODE | DT (8884-9, LN, "Cardiac Rhythm") | 1 | U |  | BCID 3415 “Cardiac Rhythms”
15 | > | CONTAINS | NUM | EV (F-03D8C, SRT, "Chest Circumference") | 1 | U |  | UNITS = EV (cm, UCUM, "cm")
16 | > | CONTAINS | TEXT | EV (F-009E4, SRT, "Breast size") | 1 | U |  | Bra size as text string
17 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1 | U |  | DCID 3202 “Chest Pain”
18 | > | CONTAINS | CODE | EV (F-04FCC, SRT, "Functional capacity") | 1 | U |  | DCID 3719 “Canadian Clinical Classification”
19 | > | CONTAINS | CODE | EV (F-04FCC, SRT, "Functional capacity") | 1 | U |  | DCID 3736 “NYHA Classification”
20 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  |
21 | > | CONTAINS | TEXT | EV (121110, DCM, "Patient Presentation") | 1 | U |  |

Content Item Descriptions

Rows 11-13 | Cardiac vital signs, for use when the SR SOP Instance does not record vital signs at multiple procedure phases or stages.
Row 16 | Breast size for interpretation of attenuation in nuclear medicine imaging

### TID 3603 Procedure Environmental Characteristics

Type:

  Extensible

Order:

  Significant

Table TID 3603. Procedure Environmental Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122222, DCM, "Procedure Environmental Characteristics") | 1 | M |  |
2 | > | CONTAINS | NUM | EV (122223, DCM, "Room oxygen concentration") | 1 | U |  | UNITS = EV (%, UCUM, "%")
3 | > | CONTAINS | NUM | EV (122224, DCM, "Room temperature") | 1 | U |  | UNITS = EV (Cel, UCUM, "C")
4 | > | CONTAINS | NUM | EV (122225, DCM, "Room Barometric pressure") | 1 | U |  | UNITS = DT (mbar, UCUM, "millibar")

## ECG Report Templates

### TID 3700 ECG Report

The ECG Report template is the root structure for the representation of analysis of an ECG waveform, potentially in comparison to a prior ECG waveform analysis. The analyzed waveform may or may not be stored as a DICOM SOP Instance.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 3700. ECG Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (28010-7, LN, "ECG Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | U |  | DCID 3670 “ECG Procedure Types”
3 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
5 | > | CONTAINS | CONTAINER | EV (121109, DCM, "Indications for Procedure") | 1 | U |  |
6 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3671 “Reason for ECG Exam”
7 | >> | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1 | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3802 “Cardiovascular Patient History” | 1 | U |  |
9 | > | CONTAINS | INCLUDE | DTID 3704 “Patient Characteristics for ECG” | 1 | U |  |
10 | > | CONTAINS | INCLUDE | DTID 3702 “Prior ECG Exam” | 1 | U |  |
11 | > | CONTAINS | INCLUDE | DTID 3708 “ECG Waveform Information” | 1 | M |  |
12 | > | CONTAINS | CONTAINER | EV (122144, DCM, "Quantitative Analysis") | 1 | M |  |
13 | >> | CONTAINS | INCLUDE | DTID 3713 “ECG Global Measurements” | 1 | U |  |
14 | >> | CONTAINS | INCLUDE | DTID 3714 “ECG Lead Measurements” | 1-n | U |  | One instantiation per reported lead
15 | > | CONTAINS | INCLUDE | DTID 3717 “ECG Qualitative Analysis” | 1 | U |  |
16 | > | CONTAINS | INCLUDE | DTID 3719 “Summary, ECG” | 1 | U |  |

### TID 3701 Clinical Context, ECG (Retired)

This Template has been retired (see PS3.16-2009).

### TID 3702 Prior ECG Exam

Type:

  Extensible

Order:

  Significant

Table TID 3702. Prior ECG Exam

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121066, DCM, "Prior Procedure Descriptions") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (122140, DCM, "Comparison with Prior Exam Done") | 1 | M |  | DCID 230 “Yes-No”
3 | > | CONTAINS | DATETIME | EV (122146, DCM, "Procedure DateTime") | 1 | U |  |
4 | > | CONTAINS | UIDREF | EV (121018, DCM, "Procedure Study Instance UID") | 1 | U |  |
5 | > | CONTAINS | COMPOSITE | EV (122075, DCM, "Prior report for current patient") | 1 | U |  |
6 | > | CONTAINS | WAVEFORM | EV (121112, DCM, "Source of Measurement") | 1 | U |  |

### TID 3704 Patient Characteristics for ECG

Type:

  Extensible

Order:

  Significant

Table TID 3704. Patient Characteristics for ECG

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121118, DCM, "Patient Characteristics") | 1 | M |  |
2 | > | CONTAINS | NUM | EV (121033, DCM, "Subject Age") | 1 | M |  | UNITS = DCID 7456 “Units of Measure for Age”
3 | > | CONTAINS | CODE | EV (121032, DCM, "Subject Sex") | 1 | M |  | DCID 7455 “Sex”
4 | > | CONTAINS | NUM | EV (8302-2, LN, "Patient Height") | 1 | U |  | UNITS = EV (cm, UCUM, "cm")
5 | > | CONTAINS | NUM | EV (29463-7, LN, "Patient Weight") | 1 | U |  | UNITS = EV (kg, UCUM, "kg")
6 | > | CONTAINS | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
7 | > | CONTAINS | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | U |  | UNITS = DCID 3500 “Pressure Units”
8 | > | CONTAINS | CODE | EV (109054, DCM, "Patient State") | 1 | U |  | DCID 3262 “ECG Patient State Values”
9 | > | CONTAINS | CODE | EV (R-00728, SRT, "Pacemaker in situ") | 1 | U |  | DCID 3672 “Pacemakers”
10 | > | CONTAINS | CODE | EV (R-0077C, SRT, "ICD in situ") | 1 | U |  | DCID 3692 “ICDs”

### TID 3708 ECG Waveform Information

The ECG Waveform Information template provides reference to, and important parameters of, the analyzed waveform.

Type:

  Extensible

Order:

  Significant

Table TID 3708. ECG Waveform Information

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121064, DCM, "Current Procedure Descriptions") | 1 | M |  |
2 | > | CONTAINS | WAVEFORM | EV (121112, DCM, "Source of Measurement") | 1 | U |  |
3 | > | CONTAINS | CODE | EV (10:11345, MDC, "Lead System") | 1 | U |  | DCID 3263 “Electrode Placement Values”
4 | > | CONTAINS | TEXT | EV (122142, DCM, "Acquisition Device Type") | 1 | U |  |
5 | > | CONTAINS | TEXT | EV (121122, DCM, "Equipment Identification") | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 1003 “Person Observer Identifying Attributes” | 1 | U |  | Person performing the ECG acquisition
7 | > | CONTAINS | TEXT | EV (121121, DCM, "Room Identification") | 1 | U |  |
8 | > | CONTAINS | DATETIME | EV (122146, DCM, "Procedure DateTime") | 1 | M |  |
9 | > | CONTAINS | NUM | DCID 3690 “ECG Control Variables Numeric” | 1-n | U |  |
10 | > | CONTAINS | TEXT | DCID 3691 “ECG Control Variables Text” | 1-n | U |  |
11 | > | CONTAINS | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | U |  |

### TID 3713 ECG Global Measurements

The ECG Global Measurements Template provides a structure for measurements calculated across the ECG waveform as a whole (multiple leads).

As an Extensible Template, applications may include any ECG global measurements, such as angles of the electrical vector of various ECG waves. The recommended vocabulary for such concepts is ISO/IEEE 11073-10102.

Type:

  Extensible

Order:

  Significant

Table TID 3713. ECG Global Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122158, DCM, "ECG Global Measurements") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 3715 “ECG Measurement Source” | 1 | U |  |
3 | > | CONTAINS | NUM | DT (2:16020, MDC, "Atrial Heart Rate") | 1 | U |  | UNITS = EV ({H.B.}/min, UCUM, "BPM")
4 | > | CONTAINS | NUM | DT (2:16016, MDC, "Ventricular Heart Rate") | 1 | M |  | UNITS = EV ({H.B.}/min, UCUM, "BPM")
5 | > | CONTAINS | NUM | DT (2:16160, MDC, "QT interval global") | 1 | M |  | UNITS = EV (ms, UCUM, "ms")
6 | > | CONTAINS | NUM | DT (2:15876, MDC, "QTc interval global") | 1 | U |  | UNITS = EV (ms, UCUM, "ms")
7 | >> | HAS PROPERTIES | CODE | DT (111001, DCM, "Algorithm Name") | 1 | U |  | DCID 3678 “Qt Correction Algorithms”
8 | > | CONTAINS | NUM | DT (2:15872, MDC, "PR interval global") | 1 | M |  | UNITS = EV (ms, UCUM, "ms")
9 | > | CONTAINS | NUM | DT (2:16156, MDC, "QRS duration global") | 1 | M |  | UNITS = EV (ms, UCUM, "ms")
10 | > | CONTAINS | NUM | DT (2:16168, MDC, "RR interval global") | 1 | M |  | UNITS = EV (ms, UCUM, "ms")
11 | > | CONTAINS | NUM | DCID 3689 “ECG Global Waveform Durations” | 1-n | U |  | UNITS = EV (ms, UCUM, "ms")
12 | > | CONTAINS | NUM | DCID 3229 “ECG Axis Measurements” | 1-n | U |  | UNITS = EV (deg, UCUM, "deg")
13 | > | CONTAINS | NUM | DT (2:16032, MDC, "Count of all beats") | 1 | U |  | UNITS = EV ({beats}, UCUM, "beats")
14 | > | CONTAINS | NUM | DT (122707, DCM, "Number of Ectopic Beats") | 1 | U |  | UNITS = EV ({beats}, UCUM, "beats")
15 | >> | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | U |  | BCID 3234 “Ectopic Beat Morphology”

### TID 3714 ECG Lead Measurements

The ECG Lead Measurements Template provides a structure for measurements calculated on individual ECG leads.

As an Extensible Template, applications may include any ECG per lead measurements, such as integrals over time of various ECG wave voltages. The recommended vocabulary for such concepts is ISO/IEEE 11073-10102.

Type:

  Extensible

Order:

  Significant

Table TID 3714. ECG Lead Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122159, DCM, "ECG Lead Measurements") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (122148, DCM, "Lead ID") | 1 | M |  | BCID 3001 “ECG Leads”
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 3715 “ECG Measurement Source” | 1 | U |  |
4 | > | CONTAINS | NUM | DCID 3687 “Electrophysiology Waveform Durations” | 1-n | U |  | UNITS = EV (ms, UCUM, "ms")
5 | > | CONTAINS | NUM | DCID 3688 “Electrophysiology Waveform Voltages” | 1-n | U |  | UNITS = EV (mV, UCUM, "mV")
6 | > | CONTAINS | CODE | EV (F-38035, SRT, "ST Segment Finding") | 1 | U |  | DCID 3233 “ST Segment Morphology”
7 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3680 “ECG Lead Noise Descriptions”

### TID 3715 ECG Measurement Source

The ECG Measurement Source Template provides a structure for identifying the particular cardiac cycle, or beat, in an analyzed ECG waveform used for the measurement group for which this template provides Observation Context. The cardiac cycle is identified by beat number, and optionally by specific temporal coordinates within a DICOM ECG Waveform SOP Instance.

Type:

  Extensible

Order:

  Significant

Table TID 3715. ECG Measurement Source

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (122149, DCM, "Beat Number") | 1 | U |  | Up to three numeric characters
2 |  |  | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | DCID 3676 “Lead Measurement Technique”
3 |  |  | TCOORD | EV (121112, DCM, "Source of Measurement") | 1 | U |  |
4 | > | SELECTED FROM | WAVEFORM |  | 1 | U |  |

Content Item Descriptions

Row 1 | Beat Number is specified as a numeric text string, and shall be treated as the ordinal of the beat (cardiac cycle) within the waveform acquisition for this lead that was analyzed for the measurements in this container (i.e., "1" for the first beat, "2" for the second, etc.). If absent, the measurements may have been made by a technique across multiple cycles as specified in Row 2 Measurement Method.
Rows 3 and 4 | Source of measurement identify the specific channel and time period within a DICOM ECG Waveform SOP Instance that was analyzed for the measurements in this container.

### TID 3717 ECG Qualitative Analysis

The ECG Qualitative Analysis template allows a free text qualitative interpretation of the analyzed ECG, as well as a structure for a coded analysis.

Type:

  Extensible

Order:

  Significant

Table TID 3717. Qualitative Analysis, ECG

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (122145, DCM, "Qualitative Analysis") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (F-00033, SRT, "ECG Finding") | 1 | MC | At least one of rows 2 and 3 shall be present |
3 | > | CONTAINS | CODE | EV (F-00033, SRT, "ECG Finding") | 1-n | MC | At least one of rows 2 and 3 shall be present | BCID 3230 “ECG Findings”
4 | >> | HAS CONCEPT MOD | TEXT | EV (121051, DCM, "Equivalent Meaning of Value") | 1 | U |  | Interpretive statement
5 | >> | INFERRED FROM | CODE | EV (F-00033, SRT, "ECG Finding") | 1-n | U |  | No BCID; may use implementation-specific codes
6 | >>> | HAS CONCEPT MOD | TEXT | EV (121051, DCM, "Equivalent Meaning of Value") | 1 | U |  | Interpretive statement

Content Item Descriptions

Row 3-4 | ECG Finding provides one or more coded interpretive statements using standard or implementation-specific codes. Each coded finding will include a Code Meaning (0008,0104) using the LO Value Representation (64 characters); longer human-readable text strings may be conveyed in the Row 4 Equivalent Meaning of Value Content Item..
Row 5-6 | Each primary ECG Finding of Row 3 may have multiple supporting coded findings in Row 5, with longer human-readable text strings if necessary in Row 6.

### TID 3718 ECG Interpretive Statement (Retired)

This Template is retired. See PS3.16-2009.

### TID 3719 Summary, ECG

Type:

  Extensible

Order:

  Significant

Table TID 3719. Summary, ECG

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121111, DCM, "Summary") | 1 | U |  |
3 | > | CONTAINS | CODE | DT (18810-2, LN, "ECG overall finding") | 1 | U |  | DCID 3677 “Summary Codes ECG”

## Cath Lab Clinical Report Templates

The templates that comprise the Cardiac Catheterization Report are interconnected as shown in Figure A-7.

Figure A-7. Cardiac Catheterization Report Template Hierarchy

### Note

Figure A-7 shows the relationship of templates; it does not show the structural hierarchy of Content Items in the IOD.

### TID 3800 Cardiac Catheterization Report Root

The Cardiac Cath Report provides the overall clinical results of the catheterization procedure and interventions. In many cases, more detailed information is optionally available in other reports (Hemodynamic Measurements, Procedure Log, etc.). That information is collected and summarized here (and referenced when available).

When a Discharge Summary section (row 12) is included, this report template covers the full set of information required for submission to the ACC NCDR™ (version 2.0) registry.

### Note

1. The information required for such a submission must sometimes be reformatted from a single concept in these templates to two data elements for the registry, or vice versa.

2. This Template is expected to be used with the Basic Text SR or Enhanced SR IOD.3. This Cardiac Cath Report template is a baseline template within these SOP Classes that may be replaced; it is therefore in no sense binding for exchange of this type of report. It is solely an example of a possible encoding of the Cardiac Cath Report.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 3800. Cardiac Catheterization Report Root

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (18745-0, LN, "Cardiac Catheterization Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DCID 3739 “Cath Procedure Type”
3 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
4 | > |  | INCLUDE | DTID 3601 “Procedure Context” | 1 | M |  |
5 | > | CONTAINS | INCLUDE | DTID 3802 “Cardiovascular Patient History” | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3803 “Patient Presentation, Cath” | 1 | M |  |
7 | > | CONTAINS | INCLUDE | DTID 3806 “Cath Procedure” | 1 | M |  |
8 | > | CONTAINS | INCLUDE | DTID 3810 “Cardiac Catheterization Findings” | 1 | M |  |
9 | > | CONTAINS | INCLUDE | DTID 3807 “Percutaneous Coronary Intervention Procedure” | 1 | U |  |
10 | > | CONTAINS | INCLUDE | DTID 3809 “Other Interventional Procedures” | 1-n | U |  |
11 | > | CONTAINS | INCLUDE | DTID 3820 “Adverse Outcomes, Cath” | 1 | M |  |
12 | > | CONTAINS | INCLUDE | DTID 3824 “Summary, Cath” | 1 | M |  |
13 | > | CONTAINS | INCLUDE | DTID 3828 “Discharge Summary, Cath” | 1 | U |  |

### TID 3802 Cardiovascular Patient History

This template contains information about a cardiovascular patient's past medical history that is relevant for the interpretation of the SR document of which it is part. This information is considered to have some degree of "persistence" across different episodes of care.

Type:

  Extensible

Order:

  Significant

Table TID 3802. Cardiovascular Patient History

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121060, DCM, "History") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121060, DCM, "History") | 1 | U |  |
3 | > | CONTAINS | CONTAINER | DT (11450-4, LN, "Problem List") | 1 | U |  |
4 | >> | CONTAINS | TEXT | DCID 3769 “Concern Types” | 1-n | U |  |
5 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1-n | U |  | $Problem = DCID 3756 “Cardiac Patient Risk Factors”
6 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1 | U |  | $Problem = EV (G-023F, SRT, "History of Diabetes mellitus") $Therapy = DCID 3722 “Diabetic Therapy”
7 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1 | U |  | $Problem = EV (G-0269, SRT, "History of Hypertension") $Therapy = DCID 3760 “Hypertension Therapy”
8 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1 | U |  | $Problem = EV (R-102B8, SRT, "History of Hypercholesterolemia") $Therapy = DCID 3761 “Antilipemic Agents”
9 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1 | U |  | $Problem = EV (D3-30000, SRT, "Arrhythmia") $Therapy = DCID 3762 “Antiarrhythmic Agents”
10 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1 | U |  | $Problem = EV (G-03AA, SRT, "History of Myocardial infarction") $ModType = DT (122170, DCM, "Type of Myocardial Infarction") $ModValue = DCID 3723 “MI Types” $Therapy = DCID 3764 “Myocardial Infarction Therapies”
11 | >> | CONTAINS | INCLUDE | DTID 3829 “Problem Properties” | 1 | U |  | $Problem = EV (G-0069, SRT, "History of Kidney disease") $Stage = DCID 3778 “Stages”
12 | > | CONTAINS | CONTAINER | DT (29762-2, LN, "Social History") | 1 | U |  |
13 | >> | CONTAINS | TEXT | EV (F-02455, SRT, "Social History") | 1 | U |  |
14 | >> | CONTAINS | TEXT | DCID 3774 “Social History” | 1-n | U |  |
15 | >> | CONTAINS | CODE | EV (F-93109, SRT, "Tobacco Smoking Behavior") | 1 | U |  | DCID 3724 “Smoking History”
16 | >> | CONTAINS | CODE | DT (F-931D4, SRT, "Drug misuse behavior") | 1 | U |  | DT (D9-30400, SRT, "Cocaine Abuse")
17 | > | CONTAINS | CONTAINER | DT (10167-5, LN, "Past Surgical History") | 1 | U |  |
18 | >> | CONTAINS | TEXT | EV (121060, DCM, "History") | 1-n | U |  |
19 | >> | CONTAINS | INCLUDE | DTID 3830 “Procedure Properties” | 1-n | U |  | $ProcType = DT (P0-009C3, SRT, "Surgical Procedure") $Procedure = DCID 3721 “Cardiovascular Surgeries”
20 | > | CONTAINS | CONTAINER | DT (30954-2, LN, "Relevant Diagnostic Tests and/or Laboratory Data") | 1 | U |  |
21 | >> | CONTAINS | TEXT | EV (121060, DCM, "History") | 1-n | U |  |
22 | >> | CONTAINS | INCLUDE | DTID 3830 “Procedure Properties” | 1-n | U |  | $ProcType = DT (P0-00002, SRT, "Diagnostic procedure") $Procedure = DCID 3757 “Cardiac Diagnostic Procedures”
23 | >> | CONTAINS | NUM | DT (2086-7, LN, "Cholesterol.in HDL") | 1 | U |  | UNITS = EV (mg/dl, UCUM, "mg/dl")
24 | >> | CONTAINS | NUM | DT (2089-1, LN, "Cholesterol.in LDL") | 1 | U |  | UNITS = EV (mg/dl, UCUM, "mg/dl")
25 | > | CONTAINS | CONTAINER | DT (10160-0, LN, "History of Medication Use") | 1 | U |  |
26 | >> | CONTAINS | TEXT | DT (111516, DCM, "Medication Type") | 1-n | U |  |
27 | >>> | HAS PROPERTIES | CODE | DT (33999-4, LN, "Status") | 1 | U |  | DCID 3773 “Use Status”
28 | >> | CONTAINS | CODE | DT (111516, DCM, "Medication Type") | 1-n | U |  |
29 | >>> | HAS PROPERTIES | NUM | DT (G-C0B7, SRT, "Dosage") | 1 | U |  |
30 | >>> | HAS PROPERTIES | CODE | DT (33999-4, LN, "Status") | 1 | U |  | DCID 3773 “Use Status”
31 | > | CONTAINS | CONTAINER | DT (10157-6, LN, "History of Family Member Diseases") | 1 | U |  |
32 | >> | CONTAINS | TEXT | EV (121060, DCM, "History") | 1-n | U |  |
33 | >> | CONTAINS | CODE | EV (G-03E5, SRT, "Family history of clinical finding") | 1-n | U |  | DCID 3758 “Cardiovascular Family History”
34 | >>> | HAS PROPERTIES | CODE | EV (G-C32E, SRT, "Subject relationship") | 1 | M |  | DCID 7451 “Family Member”
35 | > | CONTAINS | CONTAINER | DT (46264-8, LN, "History of medical device use") | 1 | U |  |
36 | >> | CONTAINS | TEXT | EV (121060, DCM, "History") | 1 | U |  |
37 | >> | CONTAINS | INCLUDE | DTID 3831 “Medical Device Use” | 1-n | U |  | $Device = DCID 3777 “Implanted Devices”

### TID 3803 Patient Presentation, Cath

This template describes the aspects of the patient that are specific to this clinical presentation (admission).

Type:

  Extensible

Order:

  Significant

Table TID 3803. Patient Presentation, Cath

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121110, DCM, "Patient Presentation") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (122128, DCM, "Patient Transferred From") | 1 | U |  |
3 | > | CONTAINS | DATETIME | EV (15, NCDR [2.0b], "Admission Date/Time") | 1 | U |  |
4 | > | CONTAINS | CODE | EV (17, NCDR [2.0b], "Admission Status") | 1 | U |  | DCID 3729 “Admission Status”
5 | > | CONTAINS | CODE | EV (18, NCDR [2.0b], "Insurance Payor Type") | 1 | U |  | DCID 3730 “Insurance Payor”
6 | > | CONTAINS | CODE | EV (46, NCDR [2.0b], "Congestive Heart Failure Prior to Procedure") | 1 | U |  | DCID 230 “Yes-No”
7 | > | CONTAINS | CODE | EV (47, NCDR [2.0b], "NYHA Classification") | 1 | UC | IFF Row 6 Value code meaning is <yes> | DCID 3736 “NYHA Classification”
8 | > | CONTAINS | CODE | EV (48, NCDR [2.0b], "Noninvasive Testing - Ischemia") | 1 | U |  | DCID 3737 “Non-invasive Test - Ischemia”
9 | > | CONTAINS | CODE | EV (49, NCDR [2.0b], "Pre-Cath Angina Type") | 1 | U |  | DCID 3738 “Pre-Cath Angina Type”
10 | > | CONTAINS | CODE | EV (50, NCDR [2.0b], "Pre-Cath Canadian Classification") | 1 | U |  | DCID 3719 “Canadian Clinical Classification”
11 | > | CONTAINS | CODE | EV (51, NCDR [2.0b], "Acute Coronary Syndrome Time Period") | 1 | UC | IFF Row 9 Value code meaning is <ACS> | DCID 3735 “Acute Coronary Syndrome Time Period”
12 | > | CONTAINS | CONTAINER | EV (121109, DCM, "Indications for Procedure") | 1 | U |  |
13 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3727 “Indications for Catheterization”
14 | > | CONTAINS | NUM | EV (10230-1, LN, "LV Ejection Fraction") | 1-n | U |  | UNITS = EV (%, UCUM, "%")
15 | >> | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement method") | 1 | U |  | DCID 3744 “EF Testing Method”
16 | >> |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
17 | > | CONTAINS | CODE | EV (121069, DCM, "Previous Finding") | 1-n | U |  | DCID 3700 “Cath Diagnosis”
18 | > | CONTAINS | TEXT | EV (121110, DCM, "Patient Presentation") | 1 | U |  |

### TID 3806 Cath Procedure

This template describes the patient-related information about this specific clinical encounter (catheterization).

Type:

  Extensible

Order:

  Significant

Table TID 3806. Cath Procedure

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121064, DCM, "Current Procedure Descriptions") | 1 | M |  |
2 | > | CONTAINS | DATETIME | EV (52, NCDR [2.0b], "Procedure DateTime") | 1 | M |  |
3 | > | CONTAINS | TEXT | EV (53, NCDR [2.0b], "Procedure Number in this admission") | 1 | U |  | Up to three numeric characters
4 | > | CONTAINS | TEXT | EV (121065, DCM, "Procedure Description") | 1 | U |  |
5 | > | CONTAINS | COMPOSITE | EV (121120, DCM, "Cath Lab Procedure Log") | 1-n | U |  |
6 | > | CONTAINS | NUM | EV (55, NCDR [2.0b], "Fluoroscopy Time") | 1 | U |  | UNITS = DT (min, UCUM, "min")
7 | > | CONTAINS | NUM | EV (122130, DCM, "Dose Area Product") | 1 | U |  | UNITS = DT (mGy.cm2, UCUM, "mGy.cm2")
8 | > | CONTAINS | PNAME | EV (76, NCDR [2.0b], "Catheterization Operator") | 1 | M |  |
9 | > | CONTAINS | PNAME | EV (121088, DCM, "Fellow") | 1-n | U |  |
10 | > | CONTAINS | PNAME | BCID 7453 “Performing Roles” | 1-n | U |  |
11 | > | CONTAINS | CODE | EV (122129, DCM, "PCI during this procedure") | 1 | U |  | DCID 230 “Yes-No”
12 | > | CONTAINS | CONTAINER | EV (F-04460, SRT, "Medication Given") | 1 | M |  |
13 | >> | CONTAINS | CODE | EV (57, NCDR [2.0b], "Thrombolytics") | 1 | U |  | DCID 3740 “Thrombolytic Administration”
14 | >> | CONTAINS | CODE | EV (58, NCDR [2.0b], "IIb/IIIa Blockade") | 1 | U |  | DCID 3741 “Medication Administration, Lab Visit”
15 | >> | CONTAINS | CODE | EV (59, NCDR [2.0b], "Heparin") | 1-n | U |  | DCID 3742 “Medication Administration, PCI”
16 | >> | CONTAINS | CODE | EV (60, NCDR [2.0b], "Aspirin") | 1 | U |  | DCID 3741 “Medication Administration, Lab Visit”
17 | >> | CONTAINS | CODE | EV (61, NCDR [2.0b], "Clopidogrel/Ticlopidine") | 1 | U |  | DCID 3743 “Clopidogrel/ticlopidine Administration”
18 | >> | CONTAINS | TEXT | EV (122083, DCM, "Drug administered") | 1-n | U |  |
19 | > | CONTAINS | CODE | EV (122138, DCM, "Circulatory Support") | 1-n | U |  | DCID 3553 “Circulatory Support”
20 | > | CONTAINS | CODE | EV (74, NCDR [2.0b], "Percutaneous Entry Site") | 1 | M |  | DCID 3746 “Percutaneous Entry Site”
21 | > | CONTAINS | CODE | EV (75, NCDR [2.0b], "Closure Device") | 1 | U |  | DCID 3747 “Percutaneous Closure”

Content Item Descriptions

Row 3 | Procedure Number (this admission) is specified as a numeric text string, and shall be treated as the ordinal of this catheterization procedure within the admission (i.e., "1" for the first catheterization, "2" for the second, etc.).

### TID 3807 Percutaneous Coronary Intervention Procedure

This template describes the various aspects of a coronary intervention.

Type:

  Extensible

Order:

  Significant

Table TID 3807. Percutaneous Coronary Intervention Procedure

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121064, DCM, "Current Procedure Descriptions") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (122061, DCM, "Percutaneous Coronary Intervention")
3 | > | CONTAINS | PNAME | EV (121114, DCM, "Performing Physician") | 1 | M |  |
4 | > | CONTAINS | PNAME | EV (121088, DCM, "Fellow") | 1-n | U |  |
5 | > | CONTAINS | PNAME | DCID 7452 “Organizational Roles” | 1-n | U |  |
6 | > | CONTAINS | CODE | EV (G-C09C, SRT, "Procedure Priority") | 1 | M |  | DCID 3414 “Procedure Urgency”
7 | > | CONTAINS | CONTAINER | EV (121109, DCM, "Indications for Procedure") | 1 | U |  |
8 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3726 “Indications for Coronary Intervention”
9 | >> | CONTAINS | CODE | EV (122172, DCM, "Acute MI Present") | 1 | U |  | DCID 230 “Yes-No”
10 | >>> | HAS PROPERTIES | CODE | DT (122170, DCM, "Type of Myocardial Infarction") | 1 | U |  | DCID 3723 “MI Types”
11 | >>> | HAS PROPERTIES | DATETIME | EV (122173, DCM, "ST Elevation Onset DateTime") | 1 | U |  |
12 | >> | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1-n | U |  |
13 | > | CONTAINS | NUM | EV (122175, DCM, "Number of lesion interventions attempted") | 1 | M |  | UNITS = EV (1, UCUM, "units")
14 | > | CONTAINS | NUM | EV (122176, DCM, "Number of lesion interventions successful") | 1 | M |  | UNITS = EV (1, UCUM, "units")
15 | > | CONTAINS | CODE | EV (122177, DCM, "Procedure Result") | 1 | M |  | DCID 3749 “PCI Procedure Result”
16 | > | CONTAINS | TEXT | EV (122177, DCM, "Procedure Result") | 1 | U |  |
17 | > | CONTAINS | INCLUDE | DTID 3808 “Lesion Intervention Information” | 1-n | M |  |

Content Item Descriptions

Rows 8 and 12 | Allow the recording of findings as either codes or as text; the same finding shall not be recorded as both.
Rows 15 and 16 | Allow the recording of procedure results as either codes or as text, but not as both.

### TID 3808 Lesion Intervention Information

Type:

  Extensible

Order:

  Significant

Table TID 3808. Lesion Intervention Information

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122178, DCM, "Lesion Intervention Information") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 3105 “Lesion Identification and Properties” | 1 | M |  |
3 | > | CONTAINS | CODE | EV (108, NDR [2.0b], "Previous Dilation") | 1 | U |  | DCID 3750 “Previously Dilated Lesion”
4 | > | CONTAINS | CODE | EV (103, NDR [2.0b], "Guidewire crossing lesion") | 1 | U |  | DCID 3752 “Guidewire Crossing”
5 | > | CONTAINS | CODE | EV (G-C50A, SRT, "Uses Equipment") | 1-n | M |  | DCID 3411 “Intracoronary Devices”
6 | >> | HAS CONCEPT MOD | CODE | EV (122111, DCM, "Primary Intervention Device") | 1 | MC | IF Device is Primary for this Lesion | DCID 230 “Yes-No”
7 | >> | HAS PROPERTIES | TEXT | EV (121145, DCM, "Description of Material") | 1 | U |  |
8 | >> | HAS PROPERTIES | NUM | DCID 3423 “Numeric Device Characteristics” | 1-n | U |  |
9 | >> | HAS PROPERTIES | NUM | DCID 3425 “Intervention Parameters” | 1-n | U |  |
10 | >> | HAS PROPERTIES | DATETIME | EV (122105, DCM, "DateTime of Intervention") | 1 | U |  |
11 | >> | HAS PROPERTIES | NUM | EV (122106, DCM, "Duration of Intervention") | 1 | U |  | UNITS = EV (s, UCUM, "s")
12 | > | CONTAINS | NUM | EV (R-101BB, SRT, "Lumen Diameter Stenosis") | 1 | M |  | UNITS = EV (%, UCUM, "%")
13 | >> | HAS CONCEPT MOD | CODE | EV (G-72BB, SRT, "Catheterization Procedure Phase") | 1 | M |  | EV (G-7298, SRT, "Post-intervention Phase")
14 | >> | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | U |  | DCID 3745 “Calculation Method”
15 | >> |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
16 | > | CONTAINS | CODE | EV (122110, DCM, "Post- Intervention TIMI Flow") | 1 | U |  | DCID 3713 “TIMI Flow Characteristics”
17 | > | CONTAINS | CODE | EV (115, NCDR [2.0b], "Dissection in segment observed") | 1 | U |  | DCID 230 “Yes-No”
18 | > | CONTAINS | CODE | EV (116, NCDR [2.0b], "Acute closure observed") | 1 | U |  | DCID 230 “Yes-No”
19 | > | CONTAINS | CODE | EV (117, NCDR [2.0b], "Acute closure re-opened") | 1 | UC | IFF Row 18 value is <yes> | DCID 230 “Yes-No”
20 | > | CONTAINS | CODE | EV (118, NCDR [2.0b], "Perforation occurred") | 1 | U |  | DCID 230 “Yes-No”
21 | > | CONTAINS | IMAGE | DT (121080, DCM, "Best illustration of finding") | 1 | U |  |
22 | > | CONTAINS | TEXT | DT (122177, DCM, "Procedure Result") | 1 | U |  |

### TID 3809 Other Interventional Procedures

Type:

  Extensible

Order:

  Significant

Table TID 3809. Other Interventional Procedures

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121064, DCM, "Current Procedure Descriptions") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | BCID 3406 “Non-coronary Transcatheter Interventions”
3 | > | CONTAINS | TEXT | EV (121065, DCM, "Procedure Description") | 1 | U |  |
4 | > | CONTAINS | CODE | DT (121065, DCM, "Procedure Description") | 1 | U |  | No BCID
5 | > | CONTAINS | TEXT | DT (122177, DCM, "Procedure Result") | 1 | U |  |

Content Item Descriptions

Rows 3 and 4 | Allow the recording of procedure description as either code or as text; the same description shall not be recorded as both.

### TID 3810 Cardiac Catheterization Findings

The Cardiac Catheterization Findings Template provides the structure for the diagnostic findings of the cath procedure, organized into sub-sections based on type of sub-procedure. It also provides for top-level summary findings and diagnoses.

Type:

  Extensible

Order:

  Significant

Table TID 3810. Cardiac Catheterization Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 3812 “Hemodynamic Findings” | 1 | U |  |
3 | > | CONTAINS | INCLUDE | DTID 3817 “Coronary Arteriography Findings” | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 3814 “Left Ventriculography Findings” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 3815 “Right Ventriculography Findings” | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3818 “Other Cardiographic Findings” | 1-n | U |  |
7 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3700 “Cath Diagnosis”
8 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
9 | > | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1-n | U |  |

Content Item Descriptions

Rows 7 and 9 | Allow the recording of findings as either codes or as text; the same finding shall not be recorded as both.

### TID 3812 Hemodynamic Findings

Type:

  Extensible

Order:

  Significant

Table TID 3812. Hemodynamic Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (PA-50030, SRT, "Hemodynamic measurements")
3 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3640 “Hypertension”
4 | >> |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
5 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
6 | > | CONTAINS | CODE | DCID 3641 “Hemodynamic Assessments” | 1-n | U |  | DCID 3642 “Degree Findings”
7 | >> | HAS PROPERTIES | CODE | EV (G-C0B2, SRT, "Condition") | 1 | U |  | EV (R-102B9, SRT, "Large v-wave")
8 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (R-102BA, SRT, "Diastolic pressure equalization")
9 | > | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | U |  | $Report = DT (122120, DCM, "Hemodynamics Report")

Content Item Descriptions

Row 4 | (Through TID 3819 “Common Findings”) may be used to encode any significant hemodynamic numeric measurements. For reference, see TID 3550 “Pressure Waveform Measurements” and TID 3560 “Derived Hemodynamic Measurements”.

### TID 3814 Left Ventriculography Findings

The information contained here about the left ventricle is relatively qualitative in nature. It is a high-level summary of the more detailed information that may be contained in an optional Quantitative Ventricular Analysis report. This template addresses findings about any ventricular septal defect (Row 7), the myocardial wall (Row 11), and about the aortic root (Row 16).

Type:

  Extensible

Order:

  Significant

Table TID 3814. Left Ventriculography Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (P5-30041, SRT, "Left Ventriculography")
3 | > | CONTAINS | CODE | EV (F-30117, SRT, "Left Ventricular Function - Finding") | 1 | M |  | DCID 242 “Normal-Abnormal”
4 | > |  | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (10230-1, LN, "LV Ejection Fraction") $Units = EV (%, UCUM, "%") $Method = DCID 3748 “Angiographic EF Testing Method” $Derivation = DCID 3745 “Calculation Method”
5 | > | CONTAINS | CODE | EV (F-0224E, SRT, "Left Ventricular Cavity Size") | 1 | U |  | DCID 3705 “Chamber Size”
6 | > | CONTAINS | CODE | EV (F-02225, SRT, "Left Ventricular Contractility") | 1 | U |  | DCID 3706 “Overall Contractility”
7 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1 | U |  | EV (D4-31159, SRT, "Ventricular Septal Defect")
8 | >> | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | U |  | DCID 3707 “VSD Description”
9 | > | CONTAINS | INCLUDE | DTID 3816 “Ventricular Assessment” | 1 | U |  |
10 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | U |  |
11 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (T-D0772, SRT, "Myocardial Wall")
12 | >> | CONTAINS | CODE | EV (18179-2, LN, "Wall Segment") | 1-n | M |  | BCID 3717 “Myocardial Wall Segments”
13 | >>> | HAS PROPERTIES | CODE | EV (F-32050, SRT, "Cardiac Wall Motion") | 1 | M |  | DCID 3703 “Wall Motion”
14 | >>> | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | U |  | DCID 3704 “Myocardium Wall Morphology Findings”
15 | >>> | HAS PROPERTIES | NUM | DT (G-C1E3, SRT, "Score") | 1 | U |  |
16 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | U |  |
17 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (F-04403, SRT, "Aortic Root")
18 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | M |  | DCID 3709 “Aortic Root Description”

### TID 3815 Right Ventriculography Findings

The information contained here about right ventricle is relatively qualitative in nature. It is a high-level summary of the more detailed information that may be contained in an optional Quantitative Ventricular Analysis report.

Type:

  Extensible

Order:

  Significant

Table TID 3815. Right Ventriculography Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (P5-3003F, SRT, "Right Ventriculography")
3 | > |  | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (10231-9, LN, "RV Ejection Fraction")) $Units = EV (%, UCUM, "%") $Method = DCID 3748 “Angiographic EF Testing Method” $Derivation = DCID 3745 “Calculation Method”
4 | > | CONTAINS | CODE | EV (F-022A1, SRT, "Right Ventricular Cavity Size") | 1 | U |  | DCID 3705 “Chamber Size”
5 | > | CONTAINS | CODE | EV (F-0227A, SRT, "Right Ventricular Contractility") | 1 | U |  | DCID 3706 “Overall Contractility”
6 | > | CONTAINS | INCLUDE | DTID 3816 “Ventricular Assessment” | 1 | U |  |

### TID 3816 Ventricular Assessment

Type:

  Extensible

Order:

  Significant

Table TID 3816. Ventricular Assessment

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | U |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 3701 “Cardiac Valves and Tracts”
3 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | M |  | DCID 3711 “Valvular Abnormalities”
4 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
5 | >> | HAS PROPERTIES | NUM | DT (G-C1E3, SRT, "Score") | 1 | U |  |
6 |  | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | U |  | $Report = DT (122292, DCM, "Quantitative Ventriculography Report")

### TID 3817 Coronary Arteriography Findings

The information contained here about with regards to coronary artery lesions is relatively qualitative in nature. It is a high-level summary of the more detailed information that may be contained in an optional Quantitative Coronary Arteriography report. This template addresses findings about the individual arteries (Row 4), and about individual lesions (Row 9).

Type:

  Extensible

Order:

  Significant

Table TID 3817. Coronary Arteriography Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (P5-30100, SRT, "Coronary Arteriography")
3 | > | CONTAINS | CODE | EV (F-04404, SRT, "Coronary artery feature ") | 1 | U |  | DCID 3710 “Coronary Dominance”
4 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | U |  |
5 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | BCID 3015 “Coronary Arteries”
6 | >>> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical modifier") | 1 | U |  | BCID 3019 “Cardiovascular Anatomic Location Modifiers”
7 | >> | CONTAINS | CODE | EV (122134, DCM, "Vessel Morphology") | 1-n | U |  | DCID 3712 “Vessel Descriptors”
8 | >> | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | U |  |
9 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | U |  |
10 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (R-002EF, SRT, "Coronary artery lesion (culprit)")
11 | >> | CONTAINS | INCLUDE | DTID 3105 “Lesion Identification and Properties” | 1 | M |  |
12 | >> | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | U |  | $Report = DT (122291, DCM, "Quantitative Arteriography Report")

### TID 3818 Other Cardiographic Findings

Type:

  Extensible

Order:

  Significant

Table TID 3818. Other Cardiographic Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | BCID 3428 “Imaging Procedures”
3 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | U |  | BCID 3630 “Cardiovascular Anatomic Locations”
4 | > | CONTAINS | INCLUDE | DTID 3819 “Common Findings” | 1-n | M |  |

### TID 3819 Common Findings

Table TID 3819. Parameters

Parameter Name | Parameter Usage
--- | ---
$Report | Title of composite object (evidence document) referenced

Type:

  Extensible

Order:

  Significant

Table TID 3819. Common Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (121071, DCM, "Finding") | 1 | U |  |
2 |  |  | IMAGE | DT (121080, DCM, "Best illustration of finding") | 1 | U |  |
3 |  |  | NUM | No BCID | 1 | U |  |
4 | > |  | INCLUDE | DTID 1000 “Quotation” | 1 | U |  |
5 |  |  | COMPOSITE | DT (122073, DCM, "Current procedure evidence") | 1 | U |  |
6 | > | HAS PROPERTIES | CODE | EV (121144, DCM, "Document Title") | 1 | U |  | $Report

Content Item Descriptions

Row 3 | May be used to encode any significant image- or waveform-based numeric measurements, with post-coordination of the Concept Name using child Content Items (with relationship HAS CONCEPT MOD), as permitted by Section 6.2.4. The source of the measurement may be noted using the Quotation Template in Row 4.

### TID 3820 Adverse Outcomes, Cath

Type:

  Extensible

Order:

  Significant

Table TID 3820. Adverse Outcomes, Cath

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121113, DCM, "Complications") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (DD-60002, SRT, "Complication of Procedure") | 1-n | U |  | DCID 3755 “Cath Complications”
3 | >> | HAS CONCEPT MOD | CODE | EV (G-D709, SRT, "Relative Time") | 1 | U |  | DCID 12102 “Temporal Periods Relating to Procedure or Therapy”
4 | > | CONTAINS | CODE | EV (DD-60002, SRT, "Complication of Procedure") | 1-n | U |  | DCID 3754 “Vascular Complications”
5 | >> | HAS CONCEPT MOD | CODE | EV (G-D709, SRT, "Relative Time") | 1 | U |  | DCID 12102 “Temporal Periods Relating to Procedure or Therapy”
6 | > | CONTAINS | TEXT | EV (DD-60002, SRT, "Complication of Procedure") | 1-n | U |  |
7 | > | CONTAINS | CODE | EV (122179, DCM, "Peri-procedural MI occurred") | 1 | U |  | DCID 230 “Yes-No”
8 | >> | INFERRED FROM | NUM | EV (122181, DCM, "CK-MB peak") | 1 | U |  | UNITS = EV ([iU], UCUM, "International unit")
9 | >>> | HAS PROPERTIES | NUM | EV (R-0038B, SRT, "Normal Range Upper Limit") | 1 | M |  | UNITS = EV ([iU], UCUM, "International unit")
10 | >> | INFERRED FROM | NUM | EV (122180, DCM, "CK-MB baseline") | 1 | M |  | UNITS = EV ([iU], UCUM, "International unit")
11 | > | CONTAINS | IMAGE | DT (121080, DCM, "Best illustration of finding") | 1-n | U |  |

Content Item Descriptions

Rows 2, 3 and 4 | Allow the recording of outcomes as either codes or as text; the same outcome shall not be recorded as both.

### TID 3824 Summary, Cath

Type:

  Extensible

Order:

  Non-Significant

Table TID 3824. Summary, Cath

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3728 “Cath Findings”
3 | >> | HAS PROPERTIES | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
4 | > | CONTAINS | TEXT | EV (121111, DCM, "Summary") | 1 | U |  |
5 | > | CONTAINS | TEXT | EV (121075, DCM, "Recommendation") | 1-n | U |  |

### TID 3828 Discharge Summary, Cath

Type:

  Extensible

Order:

  Significant

Table TID 3828. Discharge Summary, Cath

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121115, DCM, "Discharge Summary") | 1 | M |  |
2 | > | CONTAINS | DATETIME | EV (122163, DCM, "Discharge DateTime") | 1 | U |  |
3 | > | CONTAINS | CODE | EV (122164, DCM, "Coronary Artery Bypass During This Admission") | 1 | U |  | DCID 230 “Yes-No”
4 | >> | HAS PROPERTIES | CODE | EV (G-C09C, SRT, "Procedure Priority") | 1 | U |  | BCID 3414 “Procedure Urgency”
5 | >> | HAS PROPERTIES | DATETIME | EV (122146, DCM, "Procedure DateTime") | 1 | U |  |
6 | > | CONTAINS | CODE | EV (122166, DCM, "Death During This Admission") | 1 | U |  | DCID 230 “Yes-No”
7 | >> | HAS PROPERTIES | DATETIME | EV (122165, DCM, "Date of Death") | 1 | U |  |
8 | >> | HAS PROPERTIES | CODE | EV (25, NCDR [2.0b], "Primary Cause of Death") | 1 | U |  | DCID 3733 “Primary Cause of Death”
9 | >> | HAS PROPERTIES | CODE | EV (122167, DCM, "Death During Catheterization") | 1 | U |  | DCID 230 “Yes-No”
10 | > | CONTAINS | TEXT | EV (121111, DCM, "Summary") | 1 | U |  |

### TID 3829 Problem Properties

Table TID 3829. Parameters

Parameter Name | Parameter Usage
--- | ---
$Problem | Coded Value or Context Group for problem
$ModType | Modifier Name for Concept Name of problem
$ModValue | Modifier Value for Concept Name of problem
$Therapy | Coded Value or Context Group for therapy received for problem
$Stage | Coded Value or Context Group for problem or disease stage

Type:

  Extensible

Order:

  Significant

Table TID 3829. Problem Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121430, DCM, "Concern") | 1 | M |  |
2 | > | CONTAINS | DATETIME | DT (121431, DCM, "DateTime Concern Noted") | 1 | U |  |
3 | > | CONTAINS | DATETIME | DT (121432, DCM, "DateTime Concern Resolved") | 1 | U |  |
4 | > | CONTAINS | CODE | DCID 3769 “Concern Types” | 1 | M |  | $Problem
5 | >> | HAS CONCEPT MOD | CODE | $ModType | 1-n | U |  | $ModValue
6 | >> | HAS PROPERTIES | DATETIME | DT (111526, DCM, "DateTime Started") | 1 | U |  |
7 | >> | HAS PROPERTIES | DATETIME | DT (121433, DCM, "DateTime Problem Resolved") | 1 | U |  |
8 | >> | HAS PROPERTIES | CODE | DT (33999-4, LN, "Status") | 1 | U |  | DCID 3770 “Problem Status”
9 | >> | HAS PROPERTIES | CODE | DT (G-C197, SRT, "Severity") | 1 | U |  | DCID 3716 “Severity”
10 | >> | HAS PROPERTIES | CODE | DT (G-C16B, SRT, "Stage") | 1 | U |  | $Stage
11 | > | CONTAINS | CODE | DT (11323-3, LN, "Health status") | 1 | U |  | DCID 3772 “Health Status”
12 | > | CONTAINS | CODE | EV (P0-0000E, SRT, "Therapy") | 1-n | U |  | $Therapy
13 | >> | HAS PROPERTIES | CODE | DT (33999-4, LN, "Status") | 1 | U |  | DCID 3773 “Use Status”
14 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

### TID 3830 Procedure Properties

Table TID 3830. Parameters

Parameter Name | Parameter Usage
--- | ---
$ProcType | Coded Value for class of procedure
$Procedure | Coded Value or Context Group for procedure
$ModType | Modifier Name for Concept Name of procedure
$ModValue | Modifier Value for Concept Name of procedure

Type:

  Extensible

Order:

  Significant

Table TID 3830. Procedure Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | $ProcType | 1 | M |  | $Procedure
2 | > | HAS CONCEPT MOD | CODE | $ModType | 1-n | U |  | $ModValue
3 | > | HAS PROPERTIES | DATETIME | DT (111526, DCM, "DateTime Started") | 1 | U |  |
4 | > | HAS PROPERTIES | COMPOSITE | EV (R-42B89, SRT, "Clinical Report") | 1-n | U |  |
5 | >> | HAS PROPERTIES | CODE | EV (121144, DCM, "Document Title") | 1 | U |  |
6 | > | HAS PROPERTIES | TEXT | EV (R-42B89, SRT, "Clinical Report") | 1-n | U |  | Description of report with URL or other reference for report
7 | > | HAS PROPERTIES | TEXT | DT (121434, DCM, "Service Delivery Location") | 1 | U |  |
8 | > | HAS PROPERTIES | PNAME | DT (121435, DCM, "Service Performer") | 1 | UC | IF Service Performer is a person |
9 | > | HAS PROPERTIES | TEXT | DT (121435, DCM, "Service Performer") | 1 | UC | IF Service Performer is an organization |
10 | > | HAS PROPERTIES | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

### TID 3831 Medical Device Use

Table TID 3831. Parameters

Parameter Name | Parameter Usage
--- | ---
$Device | Coded Value for type of device

Type:

  Extensible

Order:

  Significant

Table TID 3831. Medical Device Use

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121436, DCM, "Medical Device Used") | 1 | M |  | $Device
2 | > | HAS PROPERTIES | DATETIME | DT (111526, DCM, "DateTime Started") | 1 | U |  |
3 | > | HAS PROPERTIES | DATETIME | DT (111527, DCM, "DateTime Ended") | 1 | U |  |
4 | > | HAS PROPERTIES | CODE | DT (33999-4, LN, "Status") | 1 | U |  | DCID 3773 “Use Status”
5 | > | HAS PROPERTIES | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

## CT/MR Cardiovascular Analysis Report Templates

### TID 3900 CT/MR Cardiovascular Analysis Report

Root Template of the Non-invasive Computed Tomography and Magnetic Resonance Cardiovascular Analysis Report.

This template contains the top level structure and includes sub-templates for the various analyses.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 3900. CT/MR Cardiovascular Analysis Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (122600, DCM, "Cardiovascular Analysis Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure Reported") | 1-n | M |  | DCID 3820 “Non-invasive Vascular Procedures”
3 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
5 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 3802 “Cardiovascular Patient History” | 1 | U |  |
7 | > | CONTAINS | INCLUDE | DTID 3901 “Procedure Summary” | 1 | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3902 “Vascular Analysis” | 1 | U |  | $AnalysisPerformed = EV (122605, DCM, "Vascular Morphological Analysis")
9 | > | CONTAINS | INCLUDE | DTID 3902 “Vascular Analysis” | 1 | U |  | $AnalysisPerformed = EV (122606, DCM, "Vascular Functional Analysis")
10 | > | CONTAINS | INCLUDE | DTID 3920 “Ventricular Analysis” | 1 | U |  |
11 | > | CONTAINS | INCLUDE | DTID 3927 “Report Summary” | 1-n | U |  |

### TID 3901 Procedure Summary

Contains summaries related to the performed procedures.

Type:

  Extensible

Order:

  Significant

Table TID 3901. Procedure Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121064, DCM, "Current Procedure Descriptions") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121065, DCM, "Procedure Description") | 1-n | M |  |
3 | > | CONTAINS | CODE | DT (RID11248, RADLEX, "Cardiac Gating") | 1 | U |  | DCID 3104 “Cardiac Synchronization Technique”

### TID 3902 Vascular Analysis

Contains either morphological or functional vascular measurement results of an analysis

Table TID 3902. Parameters

Parameter Name | Parameter Usage
--- | ---
$AnalysisPerformed | Analysis Performed

Type:

  Extensible

Order:

  Significant

Table TID 3902. Vascular Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | $AnalysisPerformed
3 | > | CONTAINS | INCLUDE | DTID 3905 “Calcium Scoring Results” | 1 | UC | IFF the value of row 2 equals EV (122605, DCM, "Vascular Morphological Analysis") |
4 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-40501, SRT, "Blood Vessel of Head") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12105 “Intracranial Cerebral Vessels” $AnalysisPerformed = $AnalysisPerformed
5 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-40501, SRT, "Blood Vessel of Head") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12105 “Intracranial Cerebral Vessels” $AnalysisPerformed = $AnalysisPerformed
6 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-40501, SRT, "Blood Vessel of Head") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 12106 “Intracranial Cerebral Vessels (unilateral)” $AnalysisPerformed = $AnalysisPerformed
7 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-45005, SRT, "Artery of Neck") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12104 “Extracranial Arteries” $AnalysisPerformed = $AnalysisPerformed
8 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-45005, SRT, "Artery of Neck") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12104 “Extracranial Arteries” $AnalysisPerformed = $AnalysisPerformed
9 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-47040, SRT, "Artery of Lower Extremity ") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12109 “Lower Extremity Arteries” $AnalysisPerformed = $AnalysisPerformed
10 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-47040, SRT, "Artery of Lower Extremity ") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12109 “Lower Extremity Arteries” $AnalysisPerformed = $AnalysisPerformed
11 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-49403, SRT, "Vein of Lower Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12110 “Lower Extremity Veins” $AnalysisPerformed = $AnalysisPerformed
12 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-49403, SRT, "Vein of Lower Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12110 “Lower Extremity Veins” $AnalysisPerformed = $AnalysisPerformed
13 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-47020, SRT, "Artery of Upper Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12107 “Upper Extremity Arteries” $AnalysisPerformed = $AnalysisPerformed
14 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-47020, SRT, "Artery of Upper Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12107 “Upper Extremity Arteries” $AnalysisPerformed = $AnalysisPerformed
15 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-49103, SRT, "Vein of Upper Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12108 “Upper Extremity Veins” $AnalysisPerformed = $AnalysisPerformed
16 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-49103, SRT, "Vein of Upper Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12108 “Upper Extremity Veins” $AnalysisPerformed = $AnalysisPerformed
17 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-71019, SRT, "Vascular Structure of Kidney") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12115 “Renal Vessels” $AnalysisPerformed = $AnalysisPerformed
18 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-71019, SRT, "Vascular Structure of Kidney") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12115 “Renal Vessels” $AnalysisPerformed = $AnalysisPerformed
19 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-46002, SRT, "Artery of Abdomen") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12111 “Abdominal Arteries (lateral)” $AnalysisPerformed = $AnalysisPerformed
20 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-46002, SRT, "Artery of Abdomen") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12111 “Abdominal Arteries (lateral)” $AnalysisPerformed = $AnalysisPerformed
21 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-46002, SRT, "Artery of Abdomen") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 12112 “Abdominal Arteries (unilateral)” $AnalysisPerformed = $AnalysisPerformed
22 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-487A0, SRT, "Vein of Abdomen") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12113 “Abdominal Veins (lateral)” $AnalysisPerformed = $AnalysisPerformed
23 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-487A0, SRT, "Vein of Abdomen") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12113 “Abdominal Veins (lateral)” $AnalysisPerformed = $AnalysisPerformed
24 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-487A0, SRT, "Vein of Abdomen") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 12114 “Abdominal Veins (unilateral)” $AnalysisPerformed = $AnalysisPerformed
25 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-44000, SRT, "Pulmonary Artery Structure") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 3829 “Pulmonary Arteries” $AnalysisPerformed = $AnalysisPerformed
26 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-43000, SRT, "Coronary Artery Structure") $Anatomy = DCID 3015 “Coronary Arteries” $AnalysisPerformed = $AnalysisPerformed
27 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-48400, SRT, "Cardiac Vein Structure") $Anatomy = DCID 3839 “Coronary Veins” $AnalysisPerformed = $AnalysisPerformed
28 | > | CONTAINS | INCLUDE | DTID 3906 “Vascular Section Measurements” | 1-n | U |  | $VascularSection = DT (T-48581, SRT, "Pulmonary Venous Structure") $Anatomy = DCID 3840 “Pulmonary Veins” $AnalysisPerformed = $AnalysisPerformed

### TID 3905 Calcium Scoring Results

Contains the calcium scoring results related to plaque findings, vessels or the whole body.

Type:

  Extensible

Order:

  Significant

Table TID 3905. Calcium Scoring Results

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122603, DCM, "Calcium Scoring Analysis")
3 | > | CONTAINS | NUM | EV (122657, DCM, "Agatston Score Threshold") | 1 | U |  | UNITS = DT ([hnsf'U], UCUM, "Hounsfield unit")
4 | > | CONTAINS | NUM | EV (122658, DCM, "Calcium Mass Threshold") | 1 | U |  | UNITS = DT (mg/cm3, UCUM, "mg/cm3")
5 | > | CONTAINS | NUM | EV (122659, DCM, "Calcium Scoring Calibration") | 1 | U |  | UNITS = DT (mg/[hnsf'U].cm3, UCUM, "mg/[hnsf'U].cm3")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (112058, DCM, "Calcium Score") $Method = EV (112055, DCM, "Agatston Scoring Method") $Units = DT (1, UCUM, "no units")
7 | > | CONTAINS | NUM | EV (122660, DCM, "Calcium Volume") | 1 | U |  | UNITS = EV (mm3, UCUM, "mm3")
8 | > | CONTAINS | NUM | EV (122661, DCM, "Calcium Mass") | 1 | U |  | UNITS = EV (mg, UCUM, "mg")
9 | > | CONTAINS | NUM | EV (F-02A3B, SRT, "Number of Lesions") | 1 | U |  | UNITS = DT ({lesions}, UCUM, "lesions")
10 | > | CONTAINS | INCLUDE | DTID 3909 “Best Illustration of Findings” | 1-n | U |  |

### TID 3906 Vascular Section Measurements

Sections of vascular measurements are section containers of an anatomical region consisting of measurement group containers containing the measurements.

Table TID 3906. Parameters

Parameter Name | Parameter Usage
--- | ---
$VascularSection | The concept name of the region or structure of which the anatomy is part
$SectionLaterality | The laterality (if any) of the anatomy in this section heading
$Anatomy | The concept name of the vascular anatomy
$AnalysisPerformed | The context of the measurements performed during the analysis

Type:

  Extensible

Order:

  Significant

Table TID 3906. Vascular Section Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | $VascularSection | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IFF $SectionLaterality has a value | $SectionLaterality
3 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | M |  |
4 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | $Anatomy
5 | >> | CONTAINS | CODE | EV (122686, DCM, "Parent Vessel Finding") | 1-n | U |  | DCID 3810 “Vascular Morphology”
6 | >>> |  | INCLUDE | DTID 1350 “Negation Modifier, Presence of Finding” | 1 | U |  |
7 | >> | CONTAINS | INCLUDE | DTID 3905 “Calcium Scoring Results” | 1 | UC | IF the value of $AnalysisPerformed equals (122605, DCM, "Vascular Morphological Analysis") |
8 | >> | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | U |  |
9 | >>> | HAS CONCEPT MOD | CODE | EV (125101, DCM, "Vessel Branch") | 1-n | UC | IF concept value of row 4 is not equal to (T-43000, SRT, "Coronary Artery Structure") | DCID 12117 “Vessel Branch Modifiers”
10 | >>> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical Modifier") | 1 | UC | IF concept value of row 4 is not equal to (T-43000, SRT, "Coronary Artery Structure") | DCID 12116 “Vessel Segment Modifiers”
11 | >>> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical Modifier") | 1 | UC | IF concept value of row 4 equals (T-43000, SRT, "Coronary Artery Structure") | DCID 3019 “Cardiovascular Anatomic Location Modifiers”
12 | >>> | CONTAINS | INCLUDE | DTID 3907 “Vessel Measurements” | 1 | U |  |
13 | >>> | CONTAINS | INCLUDE | DTID 3908 “Vascular Lesion” | 1-n | UC | IF the value of$AnalysisPerformed equals (122605, DCM, "Vascular Morphological Analysis") |
14 | >>> | CONTAINS | INCLUDE | DTID 3910 “Flow Quantification” | 1 | UC | IF the value of$AnalysisPerformed equals (122606, DCM, "Vascular Functional Analysis") |

Content Item Descriptions

Row 3 | This Findings container allows an application to group related vessels or branches
Row 5 | The characteristics associated with the parent vessel shall also be reported in the findings Container for the parent vessel. Negative findings (characteristics not present) need not be reported in the parent vessel Container.

### TID 3907 Vessel Measurements

Contains measurements made on vessel level.

Type:

  Extensible

Order:

  Significant

Table TID 3907. Vessel Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | NUM | EV (R-101BB, SRT, "Lumen Diameter Stenosis") | 1 | U |  | UNITS = DT (%, UCUM, "%")
2 |  | CONTAINS | NUM | EV (R-101BA, SRT, "Lumen Area Stenosis") | 1 | U |  | UNITS = DT (%, UCUM, "%")
3 |  | CONTAINS | NUM | EV (121206, DCM, "Distance") | 1-n | U |  | UNITS = DT (mm, UCUM, "mm")
4 | > | HAS CONCEPT MOD | CODE | EV (122340, DCM, "Fiducial Feature") | 2 | M |  |
5 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
6 | > |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1 | U |  |
7 |  | CONTAINS | NUM | EV (G-0364, SRT, "Vessel Lumen Diameter") | 1-n | U |  | UNITS = DT (mm, UCUM, "mm")
8 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | U |  | DCID 3488 “Min/Max/Mean”
9 | > | HAS PROPERTIES | NUM | EV (122337, DCM, "Relative Position") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
10 | >> | HAS CONCEPT MOD | CODE | EV (122340, DCM, "Fiducial Feature") | 1 | M |  | DCID 3837 “Fiducial Feature”

Content Item Descriptions

Rows 3-5 | The distance between two identified fiducial features
Rows 7-10 | Measurement of vessel diameter made at a position relative to a fiducial feature
Row 9 | A positive value indicates a distance in the direction of flow within the vessel

### TID 3908 Vascular Lesion

Specifies properties and the features of a vascular lesion detected during the analysis. In addition it is possible to reference or include growing of lesions over time by adding references to previous reports or by adding previous examination results.

Type:

  Extensible

Order:

  Significant

Table TID 3908. Vascular Lesion

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (F-00585, SRT, "Lesion Finding") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121151, DCM, "Lesion Identifier") | 1 | M |  |
3 | > | CONTAINS | INCLUDE | DTID 3909 “Best Illustration of Findings” | 1-n | U |  |
4 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1-n | U |  |
5 | > | CONTAINS | NUM | EV (122337, DCM, "Relative Position") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
6 | >> | HAS CONCEPT MOD | CODE | EV (122340, DCM, "Fiducial Feature") | 1 | M |  | DCID 3837 “Fiducial Feature”
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (G-0364, SRT, "Vessel Lumen Diameter") $Derivation = DCID 3838 “Diameter Derivation” $FindingSite = DCID 3486 “Vascular Measurement Sites” $Units = DT (mm, UCUM, "mm")
8 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Derivation = DCID 3838 “Diameter Derivation” $FindingSite = DCID 3486 “Vascular Measurement Sites” $Units = DT (mm2, UCUM, "mm2")
9 | > | CONTAINS | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | M |  | DCID 3810 “Vascular Morphology”
10 | >> |  | INCLUDE | DTID 3909 “Best Illustration of Findings” | 1-n | U |  |
11 | >> | HAS PROPERTIES | TEXT | EV (121106, DCM, "Comment") | 1-n | U |  |
12 | >> |  | INCLUDE | DTID 3911 “Plaque Properties” | 1 | MC | IFF value of row 9 equals (M-01470, SRT, "Plaque") |
13 | >> |  | INCLUDE | DTID 3912 “Stenosis Properties” | 1 | MC | IFF value of row 9 equals (M-34200, "Stenosis") |
14 | >> |  | INCLUDE | DTID 3913 “Aneurysm Properties” | 1 | MC | IFF value of row 9 equals (M-32200, SRT, "Aneurysm") |
15 | >> |  | INCLUDE | DTID 3914 “Arterial Dissection Properties” | 1 | MC | IFF value of row 9 equals (D3-81310, SRT, "Arterial Dissection") |
16 | >> | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | MC | IFF value of row 9 equals (M-520F8, SRT, "Vascular Sclerosis") | DCID 3817 “Vascular Sclerosis Types”
17 | >> |  | INCLUDE | DTID 3915 “Vascular Occlusion Properties” | 1 | MC | IFF value of row 9 equals EVM-34000, SRT, "Occlusion") |
18 | >> |  | INCLUDE | DTID 3916 “Stent Properties” | 1 | MC | IFF value of row 9 equals (A-25500, SRT, "Stent") |

Content Item Descriptions

Row 5 | A positive value indicates a distance in the direction of flow within the vessel For example: An aneurysm with relative position -4 mm from the renal arteries would begin superior to the renal arteries.
Row 7, 8 | These rows are associated with the position of the most significant effect of the lesion, i.e., maximum diameter of aneurysm or the minimum diameter of stenosis

### TID 3909 Best Illustration of Findings

Specification of images, waveforms, spatial and temporal coordinates used to illustrate findings.

Type:

  Extensible

Order:

  Significant

Table TID 3909. Best Illustration of Findings

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | IMAGE | EV (121080, DCM, "Best illustration of finding") | 1 | U |  |
2 |  | CONTAINS | WAVEFORM | EV (121080, DCM, "Best illustration of finding") | 1 | U |  |
3 |  | CONTAINS | SCOORD | EV (121080, DCM, "Best illustration of finding") | 1 | U |  |
4 | > | SELECTED FROM | IMAGE | no purpose of reference | 1 | M |  |
5 |  | CONTAINS | TCOORD | EV (121080, DCM, "Best illustration of finding") | 1 | U |  |
6 | > | SELECTED FROM | SCOORD | no purpose of reference | 1 | MC | XOR row 8, 9 |
7 | >> | SELECTED FROM | IMAGE | no purpose of reference | 1 | M |  | must be a multi-frame image
8 | > | SELECTED FROM | WAVEFORM | no purpose of reference | 1 | MC | XOR row 6, 9 |
9 | > | SELECTED FROM | IMAGE | no purpose of reference | 1 | MC | XOR row 6, 8 | must be a multi-frame image

### TID 3910 Flow Quantification

Contains the flow quantification measurement results

Type:

  Extensible

Order:

  Significant

Table TID 3910. Flow Quantification

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122604, DCM, "Flow Quantification")
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 3929 “Cardiovascular Analysis Observation Context” | 1 | U |  |
4 | > | HAS OBS CONTEXT | DATETIME | EV (G-D321, SRT, "Start Time") | 1 | M |  |
5 | > | HAS OBS CONTEXT | DATETIME | EV (G-D320, SRT, "Stop Time") | 1 | M |  |
6 | > | CONTAINS | INCLUDE | DTID 3990 “Two Dimensional Measurement Graph” | 1 | U |  | $MeasurementGraph = EV (122667, DCM, "Blood velocity vs. time of cardiac cycle") $X-Concept = EV (122666, DCM, "Time relative to R-wave peak") $Y-Concept = EV (F-0319E, SRT, "Arterial Velocity") $X-AxisUnits = DT (ms, UCUM, "ms") $Y-AxisUnits = DT (cm/s, UCUM, "cm/s")
7 | > | CONTAINS | NUM | EV (122642, DCM, "Velocity Encoding Minimum Value") | 1 | U |  | UNITS = DT (cm/s, UCUM, "cm/s")
8 | > | CONTAINS | NUM | EV (122643, DCM, "Velocity Encoding Maximum Value") | 1 | U |  | UNITS = DT (cm/s, UCUM, "cm/s")
9 | > | CONTAINS | CONTAINER | EV (125007, DCM, "Measurement Group") | 1-n | M |  |
10 | >> | HAS CONCEPT MOD | TEXT | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | XOR row 11 |
11 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | XOR row 10 |
12 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122207, DCM, "Blood velocity, peak") $Units = DT (cm/s, UCUM, "cm/s")
13 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122205, DCM, "Blood velocity, mean") $Units = DT (cm/s, UCUM, "cm/s")
14 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-39200, SRT, "Blood Flow") $Derivation = EV (R-00317, SRT, "Mean") $Units = DT (ml/s, UCUM, "ml/s")
15 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-2 | U |  | $Measurement = EV (F-39200, SRT, "Blood Flow") $ModType = EV (G-C048, SRT, "Direction of flow") $ModValue = DCID 12221 “Flow Direction” $Units = DT (ml/s, UCUM, "ml/s")
16 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122645, DCM, "Net Forward Volume") $Units = DT (ml, UCUM, "ml")
17 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122645, DCM, "Net Forward Volume") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8277-6, LN, "BSA") $Units = DT (ml/m2, UCUM, "ml/m2")
18 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Derivation = DCID 3488 “Min/Max/Mean” $Units = DT (mm2, UCUM, "mm2")

### TID 3911 Plaque Properties

Properties of a plaque finding

Type:

  Extensible

Order:

  Significant

Table TID 3911. Plaque Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | NUM | EV (122376, DCM, "Total Plaque Volume") | 1 | U |  | UNITS = DT (mm3, UCUM, "mm3")
2 |  | HAS PROPERTIES | CODE | EV (G-A428, SRT, "Margin") | 1 | U |  | DCID 3715 “Lesion Margin”
3 |  | HAS PROPERTIES | CODE | EV (M-01000, SRT, "Morphological Abnormal Structure") | 1-n | M |  | DCID 3802 “Plaque Structures”
4 |  | HAS PROPERTIES | INCLUDE | DTID 3905 “Calcium Scoring Results” | 1 | U |  |
5 |  | HAS PROPERTIES | CODE | EV (121071, DCM, "Finding") | 1 | U |  | DT (R-102DA, SRT, "Contrast Media Seen in Plaque")

### TID 3912 Stenosis Properties

Properties of a stenosis finding

Type:

  Extensible

Order:

  Significant

Table TID 3912. Stenosis Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | CODE | EV (G-C036, SRT, "Measurement method") | 1 | M |  | DCID 3804 “Stenosis Measurement Methods”
2 |  | HAS PROPERTIES | CODE | EV (G-D775, SRT, "Type of Stenosis") | 1 | U |  | DCID 3805 “Stenosis Types”
3 |  | HAS PROPERTIES | CODE | EV (G-C002, SRT, "Associated with") | 1 | U |  | DCID 3815 “Source of Vascular Finding”
4 |  | HAS PROPERTIES | CODE | EV (G-C2FE, SRT, "Shape") | 1 | U |  | DCID 3806 “Stenosis Shape”
5 |  | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (G-0364, SRT, "Vessel Lumen Diameter") $Derivation = DCID 3488 “Min/Max/Mean” $Units = DT (mm, UCUM, "mm")
6 |  | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Derivation = DCID 3488 “Min/Max/Mean” $Units = DT (mm2, UCUM, "mm2")
7 |  | HAS PROPERTIES | NUM | EV (R-101BC, SRT, "Stenotic Lesion Length") | 1 | U |  | UNITS = DT (mm, UCUM, "mm")
8 |  | HAS PROPERTIES | CODE | EV (R-101BC, SRT, "Stenotic Lesion Length") | 1 | U |  | DCID 3831 “Stenosis Length”
9 |  | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (R-101BA, SRT, "Lumen Area Stenosis") $Derivation = DCID 3488 “Min/Max/Mean” $Units = DT (%, UCUM, "%")
10 |  | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (R-101BB, SRT, "Lumen Diameter stenosis") $Derivation = DCID 3488 “Min/Max/Mean” $Units = DT (%, UCUM, "%")
11 |  | HAS PROPERTIES | CODE | EV (R-101BA, SRT, "Lumen Area Stenosis") | 1-n | U |  | DCID 3832 “Stenosis Grade”
12 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | M |  | DCID 3488 “Min/Max/Mean”
13 |  | HAS PROPERTIES | CODE | EV (R-101BB, SRT, "Lumen Diameter Stenosis") | 1-n | U |  | DCID 3832 “Stenosis Grade”
14 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | M |  | DCID 3488 “Min/Max/Mean”

### TID 3913 Aneurysm Properties

Properties of an aneurysm finding

Type:

  Extensible

Order:

  Significant

Table TID 3913. Aneurysm Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | M |  | DCID 3808 “Aneurysm Types”
2 |  | HAS PROPERTIES | CODE | EV (G-C002, SRT, "Associated with") | 1 | U |  | DCID 3815 “Source of Vascular Finding”
3 |  | HAS PROPERTIES | INCLUDE | DTID 3917 “Aneurysm Measurements” | 1 | U |  |
4 |  | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | UC | IFF value of row 1 equals (M-32206, SRT, "Compound Aneurysm") | DCID 3808 “Aneurysm Types”
5 | > | HAS PROPERTIES | INCLUDE | DTID 3917 “Aneurysm Measurements” | 1 | U |  |

### TID 3914 Arterial Dissection Properties

Properties of a arterial dissection finding

Type:

  Extensible

Order:

  Significant

Table TID 3914. Arterial Dissection Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | CODE | EV (122387, DCM, "Dissection Classification") | 1 | M |  | DCID 3492 “Vascular Dissection Classifications”
2 |  | HAS PROPERTIES | CODE | EV (G-C150, SRT, "Etiology") | 1 | U |  | DCID 3809 “Associated Conditions”
3 |  | HAS PROPERTIES | NUM | EV (G-A22A, SRT, "Length") | 1 | U |  | UNITS = DT (mm, UCUM, "mm")
4 |  | HAS PROPERTIES | CODE | EV (R-102DD, SRT, "Anatomic structure potentially involved in evolution of disease") | 1-n | U |  | DCID 3827 “Vessel Segments”

### TID 3915 Vascular Occlusion Properties

Properties of vascular occlusion finding

Type:

  Extensible

Order:

  Significant

Table TID 3915. Vascular Occlusion Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | CODE | EV (G-D775, SRT, "Type of Stenosis") | 1 | M |  | DCID 3805 “Stenosis Types”
2 |  | HAS PROPERTIES | CODE | EV (G-C002, SRT, "Associated with") | 1 | U |  | DCID 3815 “Source of Vascular Finding”
3 |  | HAS PROPERTIES | CODE | EV (G-C2FE, SRT, "Shape") | 1 | U |  | DCID 3806 “Stenosis Shape”
4 |  | HAS PROPERTIES | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (R-101BC, SRT, "Stenotic Lesion Length") $Method = DCID 3804 “Stenosis Measurement Methods” $Units = DT (mm, UCUM, "mm")

### TID 3916 Stent Properties

Properties of a stent finding

Type:

  Extensible

Order:

  Significant

Table TID 3916. Stent Properties

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | CODE | EV (122685, DCM, "Stent Composition") | 1-n | M |  | DCID 3814 “Stent Composition”
2 |  | HAS PROPERTIES | NUM | EV (R-101AD, SRT, "Vascular Stent Diameter") | 1 | U |  | UNITS = DT (mm, UCUM, "mm")
3 |  | HAS PROPERTIES | NUM | EV (R-101B0, SRT, "Vascular Stent Length") | 1 | U |  | UNITS = DT (mm, UCUM, "mm")
4 |  | HAS PROPERTIES | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3813 “Stent Findings”
5 | > |  | INCLUDE | DTID 3912 “Stenosis Properties” | 1 | MC | IFF value of row 4 equals (M-34200, SRT, "Stenosis") |

### TID 3917 Aneurysm Measurements

Measurements of aneurysms. TID 300 “Measurement” invoked from this template allows the measurement to reference an image used as the source of the measurement.

Type:

  Extensible

Order:

  Significant

Table TID 3917. Aneurysm Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-A22A, SRT, "Length") $ModType = EV (G-C093, SRT, "Extent") $ModValue = DT (G-A143, SRT, "Longitudinal") $Units = DT (mm, UCUM, "mm")
2 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-D705, SRT, "Volume") $Method = DCID 3807 “Volume Measurement Methods” $Units = DT (mm3, UCUM, "mm3")
3 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (R-102DB, SRT, "Vessel Lumen Cross-Sectional Area Increase") $Units = DT (%, UCUM, "%")
4 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (R-102DB, SRT, "Vessel Lumen Cross-Sectional Area Increase") $Units = DT (mm2, UCUM, "mm2")
5 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (R-102DC, SRT, "Vessel Lumen Cross-Sectional Diameter Increase") $Units = DT (%, UCUM, "%")
6 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (R-102DC, SRT, "Vessel Lumen Cross-Sectional Diameter Increase") $Units = DT (mm, UCUM, "mm")

### TID 3920 Ventricular Analysis

Contains the ventricular functional measurement results.

Type:

  Extensible

Order:

  Significant

Table TID 3920. Ventricular Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122601, DCM, "Ventricular Analysis")
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 3929 “Cardiovascular Analysis Observation Context” | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 3921 “Ventricular Measurements” | 1-n | U |  | $Ventricle = EV (T-32600, SRT, "Left Ventricle")
5 | > | CONTAINS | INCLUDE | DTID 3921 “Ventricular Measurements” | 1-n | U |  | $Ventricle = EV (T-32500, SRT, "Right Ventricle")
6 | > | CONTAINS | INCLUDE | DTID 3925 “Ventricular Thickening Analysis” | 1-n | U |  |
7 | > | CONTAINS | INCLUDE | DTID 3926 “Myocardial Perfusion Analysis” | 1-n | U |  |

### TID 3921 Ventricular Measurements

Ventricular measurement results related to the volume of a ventricle.

Table TID 3921. Parameters

Parameter Name | Parameter Usage
--- | ---
$Ventricle | Describes if either the left or the right ventricle was examined

Type:

  Extensible

Order:

  Significant

Table TID 3921. Ventricular Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | $Ventricle
3 | > | CONTAINS | INCLUDE | DTID 3922 “Absolute Values of Ventricular Measurements” | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 3923 “BSA-Normalized Ventricular Measurements” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 3924 “Heart Rate-Normalized Ventricular Measurements” | 1 | U |  |

### TID 3922 Absolute Values of Ventricular Measurements

Ventricular measurement results related to the absolute volume of a ventricle.

Type:

  Extensible

Order:

  Significant

Table TID 3922. Absolute Values of Ventricular Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (122608, DCM, "Absolute Values Of Ventricular Measurements") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3833 “Cardiac Ejection Fraction” $ModType = DT (122670, DCM, "Papillary Muscle Included/Excluded") $ModValue = DCID 3821 “Papillary Muscle Included/excluded” $Method = DCID 3807 “Volume Measurement Methods” $Units = DT (%, UCUM, "%")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3835 “Cardiac Volume Measurements” $ModType = DT (122670, DCM, "Papillary Muscle Included/Excluded") $ModValue = DCID 3821 “Papillary Muscle Included/excluded” $Units = DT (ml, UCUM, "ml")
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32100, SRT, "Cardiac Output") $ModType = DT (122670, DCM, "Papillary Muscle Included/Excluded") $ModValue = DCID 3821 “Papillary Muscle Included/excluded” $Units = DT (l/min, UCUM, "l/min")
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $ModType = DT (122670, DCM, "Papillary Muscle Included/Excluded") $ModValue = DCID 3821 “Papillary Muscle Included/excluded” $Units = DT (g, UCUM, "g")
6 | >> | HAS CONCEPT MOD | CODE | EV (R-4089A, SRT, "Cardiac Cycle Point") | 1 | U |  | DCID 12233 “Cardiac Phase”
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122616, DCM, "Peak Ejection Rate") $Units = DT (ml/s, UCUM, "ml/s")
8 | > | CONTAINS | NUM | EV (122617, DCM, "Peak Ejection Time") | 1 | U |  | UNITS = EV (s, UCUM, "s")
9 | >> | HAS CONCEPT MOD | CODE | EV (122611, DCM, "Reference Point") | 1 | M |  | EV (F-32011, SRT, "End-Diastolic")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122618, DCM, "Peak Filling Rate") $Units = DT (ml/s, UCUM, "ml/s")
11 | > | CONTAINS | NUM | EV (122619, DCM, "Peak Filling Time") | 1 | U |  | UNITS = DT (s, UCUM, "s")
12 | >> | HAS CONCEPT MOD | CODE | EV (122611, DCM, "Reference Point") | 1 | M |  | DT (109070, DCM, "End-Systolic")

### TID 3923 BSA-Normalized Ventricular Measurements

Ventricular measurement results normalized based on the Body Surface Area

Type:

  Extensible

Order:

  Significant

Table TID 3923. BSA-Normalized Ventricular Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (122609, DCM, "Normalized values of ventricular measurements") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121425, DCM, "Index") | 1 | M |  | DT (8277-6, LN, "Body Surface Area")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (8277-6, LN, "Body Surface Area") $Unit = DT (m2, UCUM, "m2")
4 | >> | INFERRED FROM | CODE | EV (8278-4, LN, "Body Surface Area Formula") | 1 | U |  | BCID 3663 “Body Surface Area Equations”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3835 “Cardiac Volume Measurements” $ModType = EV (121425, DCM, "Index") $ModValue = DT (8277-6, LN, "Body Surface Area") $Units = DT (ml/m2, UCUM, "ml/m2")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32110, SRT, "Cardiac Index") $Units = DT (ml/min/m2, UCUM, "(ml/min) /m2")
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-2 | U |  | $Measurement = EV (122447, DCM, "Wall Mass") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8277-6, LN, "Body Surface Area") $Units = DT (g/m2, UCUM, "g/m2")
8 | >> | HAS CONCEPT MOD | CODE | DT (122670, DCM, "Papillary Muscle Included/Excluded") | 1 | U |  | DCID 3821 “Papillary Muscle Included/excluded”
9 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122618, DCM, "Peak Filling Rate") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8277-6, LN, "Body Surface Area") $Units = DT (ml/s/m2, UCUM, "(ml/s) /m2")
10 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32070, SRT, "Peak Cardiac Ejection Fraction") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8277-6, LN, "Body Surface Area") $Units = DT (%/m2, UCUM, "%/m2")

### TID 3924 Heart Rate-Normalized Ventricular Measurements

Ventricular measurement results normalized based on the Heart Rate

Type:

  Extensible

Order:

  Significant

Table TID 3924. Heart Rate-Normalized Ventricular Measurements

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (122609, DCM, "Normalized values of ventricular measurements") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121425, DCM, "Index") | 1 | M |  | DT (8867-4, LN, "Heart Rate")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3835 “Cardiac Volume Measurements” $ModType = EV (121425, DCM, "Index") $ModValue = DT (8867-4, LN, "Heart Rate") $Units = DT (ml/{H.B.}/min, UCUM, "ml/BPM")
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32100, SRT, "Cardiac Output") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8867-4, LN, "Heart Rate") $Units = DT (ml/min/{H.B.}/min, UCUM, "(ml/min) /BPM")
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122618, DCM, "Peak Filling Rate") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8867-4, LN, "Heart Rate") $Units = DT (ml/s/{H.B.}/min, UCUM, "(ml/s) /BPM")
6 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-32070, SRT, "Peak Cardiac Ejection Fraction") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8867-4, LN, "Heart Rate") $Units = DT (%/{H.B.}/min, UCUM, "%/BPM")

### TID 3925 Ventricular Thickening Analysis

Data of a ventricular wall thickening analysis

Type:

  Extensible

Order:

  Significant

Table TID 3925. Thickening Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122607, DCM, "Thickening Analysis")
3 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | M |  |
4 | >> | CONTAINS | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | MC | XOR row 5 | DCID 3717 “Myocardial Wall Segments”
5 | >> | CONTAINS | TEXT | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | XOR row 4 |
6 | >> | CONTAINS | INCLUDE | DTID 3909 “Best Illustration of Findings” | 1-n | U |  |
7 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122445, DCM, "Wall Thickness") $ModType = EV (R-4089A, SRT, "Cardiac Cycle Point") $ModValue = DT (F-32011, SRT, "End-Diastolic") $Units = DT (mm, UCUM, "mm")
8 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = EV (122445, DCM, "Wall Thickness") $ModType = EV (R-4089A, SRT, "Cardiac Cycle Point") $ModValue = DT (109070, DCM, "End-Systolic") $Units = DT (mm, UCUM, "mm")
9 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122624, DCM, "Wall Thickness Ratio end-systolic to end-diastolic") $Units = DT (%, UCUM, "%")
10 | >> | CONTAINS | CODE | EV (F-32050, SRT, "Cardiac Wall Motion") | 1 | U |  | DCID 3703 “Wall Motion”
11 | >> | CONTAINS | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | U |  | DCID 3704 “Myocardium Wall Morphology Findings”

### TID 3926 Myocardial Perfusion Analysis

Myocardial perfusion analysis results.

Perfusion measurements may be performed either for one or more ventricular segments (row 4) or for substructures inside ventricular segments (row 14)

Type:

  Extensible

Order:

  Significant

Table TID 3926. Myocardial Perfusion Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122602, DCM, "Myocardial Perfusion Analysis")
3 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | M |  |
4 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1-n | MC | XOR row 6 | DCID 3717 “Myocardial Wall Segments”
5 | >>> | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical Modifier") | 1 | U |  | DCID 3843 “Myocardial Subsegment”
6 | >> | HAS CONCEPT MOD | TEXT | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | XOR row 4 |
7 | >> | HAS ACQ CONTEXT | CODE | EV (109054, DCM, "Patient State") | 1 | U |  | DCID 3101 “Cardiac Procedural State Values”
8 | >> | HAS ACQ CONTEXT | INCLUDE | DTID 3106 “Drugs/Contrast Administered” | 1-n | U |  |
9 | >> | CONTAINS | TEXT | EV (122627, DCM, "Curve Fit Method") | 1 | U |  |
10 | >> | CONTAINS | INCLUDE | DTID 3909 “Best Illustration of Findings” | 1-n | U |  |
11 | >> | CONTAINS | TEXT | EV (122628, DCM, "Baseline Result Correction") | 1 | U |  |
12 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3836 “Time-based Perfusion Measurements” $Units = EV (s, UCUM, "s")
13 | >> | CONTAINS | NUM | EV (122640, DCM, "Image Interval") | 1 | U |  | UNITS = EV (ms, UCUM, "ms")
14 | >> | CONTAINS | NUM | EV (122635, DCM, "MR Perfusion Peak") | 1 | U |  | UNITS = DT (1, UCUM, "No units")
15 | >> | CONTAINS | NUM | EV (122636, DCM, "MR Perfusion Slope") | 1 | U |  | UNITS = DT (1, UCUM, "No units")
16 | >> | CONTAINS | NUM | EV (122637, DCM, "MR Perfusion Time Integral") | 1 | U |  | UNITS = DT (1, UCUM, "No units")
17 | >> | CONTAINS | CONTAINER | EV (125007, DCM, "Measurement Group") | 1-n | U |  |
18 | >>> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 3836 “Time-based Perfusion Measurements” $Units = EV (s, UCUM, "s")
19 | >>> | CONTAINS | NUM | EV (122635, DCM, "MR Perfusion Peak") | 1 | U |  | UNITS = DT (1, UCUM, "No units")
20 | >>> | CONTAINS | NUM | EV (122636, DCM, "MR Perfusion Slope") | 1 | U |  | UNITS = DT (1, UCUM, "No units")
21 | >>> | CONTAINS | NUM | EV (122637, DCM, "MR Perfusion Time Integral") | 1 | U |  | UNITS = DT (1, UCUM, "No units")
22 | >> | CONTAINS | CODE | EV (122664, DCM, "Late Contrast Enhancement") | 1 | U |  | DCID 230 “Yes-No”
23 | >>> | HAS ACQ CONTEXT | NUM | EV (122665, DCM, "Time after start of injection of contrast bolus") | 1 | M |  | UNITS = DT (s, UCUM, "s")
24 | >>> | HAS ACQ CONTEXT | NUM | EV (122668, DCM, "Time interval since detection of contrast bolus") | 1 | U |  | UNITS = DT (s, UCUM, "s")

Content Item Descriptions

Row 12 | Image Interval is appropriate only for equally time-spaced images

### TID 3927 Report Summary

Contains summary elements based on the findings of the report

Type:

  Extensible

Order:

  Non-Significant

Table TID 3927. Report Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | BCID 7001 “Diagnostic Imaging Report Headings” | 1 | M |  |
2 | > | CONTAINS | CODE | BCID 7002 “Diagnostic Imaging Report Elements” | 1-n | U |  |
3 | >> |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1-n | U |  |
4 | >> |  | INCLUDE | DTID 321 “Waveform or Temporal Coordinates” | 1-n | U |  |
5 | > | CONTAINS | TEXT | BCID 7002 “Diagnostic Imaging Report Elements” | 1-n | U |  |
6 | >> |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1-n | U |  |
7 | >> |  | INCLUDE | DTID 321 “Waveform or Temporal Coordinates” | 1-n | U |  |

### TID 3929 Cardiovascular Analysis Observation Context

Defines the observation context for cardiovascular Functional Analysis

Type:

  Extensible

Order:

  Significant

Table TID 3929. Cardiovascular Analysis Observation Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS OBS CONTEXT | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  | UNITS = DT ({H.B.}/min, UCUM, "BPM")
2 |  | HAS OBS CONTEXT | CODE | EV (8884-9, LN, "Cardiac Rhythm") | 1 | U |  | DCID 3826 “Heart Rhythm”
3 |  | HAS OBS CONTEXT | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | U |  | UNITS = DT (mm[Hg], UCUM, "mmHg")
4 |  | HAS OBS CONTEXT | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | U |  | UNITS = DT (mm[Hg], UCUM, "mmHg")
5 |  | HAS OBS CONTEXT | CODE | EV (F-043E6, SRT, "Respiration Observable") | 1 | U |  | DCID 3823 “Respiratory Status”
6 |  | HAS ACQ CONTEXT | INCLUDE | DTID 3106 “Drugs/Contrast Administered” | 1-n | U |  |

### TID 3990 Two Dimensional Measurement Graph

Generic template representing arbitrary two-dimensional graphs.

Table TID 3990. Parameters

Parameter Name | Parameter Usage
--- | ---
$MeasurementGraph | Describes what the graph is about
$X-Concept | Concept of the X-Axis of the graph
$Y-Concept | Concept of the Y-Axis of the graph
$X-AxisUnit | Unit of the x-axis data elements
$Y-AxisUnit | Unit of the y-axis data elements

Type:

  Extensible

Order:

  Significant

Table TID 3990. Two Dimensional Measurement Graph

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | $MeasurementGraph | 1 | M |  |
2 | > | CONTAINS | CODE | EV (122698, DCM, "X-Concept") | 1 | M |  | $X-Concept
3 | > | CONTAINS | CODE | EV (122699, DCM, "Y-Concept") | 1 | M |  | $Y-Concept
4 | > | CONTAINS | CONTAINER | no concept name | 1-n | MC | IF Row 7, 8, or 9 not present |
5 | >> | CONTAINS | NUM | $X-Concept | 1 | M |  | UNITS = $X-AxisUnit
6 | >> | CONTAINS | NUM | $Y-Concept | 1 | M |  | UNITS = $Y-AxisUnit
7 | > | CONTAINS | IMAGE | $MeasurementGraph | 1 | U |  |
8 | > | CONTAINS | WAVEFORM | $MeasurementGraph | 1 | U |  |
9 | > | CONTAINS | COMPOSITE | $MeasurementGraph | 1 | U |  |

Content Item Descriptions

Rows 5-6 | The X-Concept values shall be monotonically increasing.
Row 7 | Secondary Capture Image containing a bitmap representation of the graph
Row 8 | Waveform containing a representation of the graph
Row 9 | Composite Object containing a rendered representation of the graph

## Mammography CAD SR IOD Templates

The templates that comprise the Mammography CAD SR IOD are interconnected as in Figure A-8:

Figure A-8. Mammography CAD SR IOD Template Structure

In Figure A-8, '···' indicates possible recursive application of subordinate templates.

### TID 4000 Mammography CAD Document Root

This template forms the top of a content tree that allows a mammography CAD device to describe the results of detection and analysis of Mammographic evidence. This template, together with its subordinate templates, describes both the results for presentation to radiologists and partial product results for consumption by mammography CAD devices in subsequent mammography CAD reports.

This template defines a Container that contains an Image Library, the mammography CAD results, and summaries of the detection and analysis algorithms performed. The Image Library contains the Image SOP Class and Instance UIDs, and selected attributes for each image referenced in either the algorithm summaries or mammography CAD results.

The Summary of Detections and Summary of Analyses sub-trees gather lists of algorithms attempted, grouped by success/failure status. Algorithms not attempted are not mentioned in these sub-trees. This information forms the basis for understanding why a mammography CAD report may produce no (or fewer than anticipated) results. Mammography CAD results are constructed bottom-up, starting from Single Image Findings (see TID 4006 “Mammography CAD Single Image Finding”), associated as Composite Features (see TID 4004 “Mammography CAD Composite Feature”), and from which Individual and Overall Impressions are formed.

See Figure E.1-1 “Top Levels of Mammography CAD SR Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  Yes

Table TID 4000. Mammography CAD Document Root

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111036, DCM, "Mammography CAD Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | CONTAINS | CONTAINER | EV (111028, DCM, "Image Library") | 1 | M |  |
4 | >> | CONTAINS | INCLUDE | DTID 4020 “CAD Image Library Entry” | 1-n | M |  | $ImageLaterality = DCID 6022 “Side” $ImageView = DCID 4014 “View for Mammography” $ImageViewMod = DCID 4015 “View Modifier for Mammography”
5 | > | CONTAINS | INCLUDE | DTID 4001 “Mammography CAD Overall Impression/Recommendation” | 1 | M |  |
6 | > | CONTAINS | CODE | EV (111064, DCM, "Summary of Detections") | 1 | M |  | DCID 6042 “Status of Results”
7 | >> | INFERRED FROM | INCLUDE | DTID 4015 “CAD Detections Performed” | 1 | MC | Shall be present unless the value of (111064, DCM, "Summary of Detections") is (111225, DCM, "Not Attempted") | $DetectionCode = DCID 6014 “Mammography Single Image Finding”
8 | > | CONTAINS | CODE | EV (111065, DCM, "Summary of Analyses") | 1 | M |  | DCID 6042 “Status of Results”
9 | >> | INFERRED FROM | INCLUDE | DTID 4016 “CAD Analyses Performed” | 1 | MC | Shall be present unless the value of (111065, DCM, "Summary of Analyses") is (111225, DCM, "Not Attempted") | $AnalysisCode = DCID 6043 “Types of Mammography CAD Analysis”

Content Item Descriptions

Image Library | The "Image Library" section of the Content Tree (TID 4000 “Mammography CAD Document Root”, row 3) shall include all Image SOP Instances from the Current Requested Procedure Evidence Sequence (0040,A375) attribute of the SR Document General module. If a portion of another instance of a Mammography CAD SR IOD is duplicated in the "Overall Impression/ Recommendation" section of the Content Tree, the "Image Library" shall also include all Image Library Entries referenced from the duplicated portions of the Mammography CAD SR.
Detections Performed | The "Detections Performed" and "Analyses Performed" sections of the Content Tree (TID 4000 “Mammography CAD Document Root”, rows 6 and 8) together shall reference all Image SOP Instances included in the Current Requested Procedure Evidence Sequence (0040,A375) attribute of the SR Document General module.
Analyses Performed

### TID 4001 Mammography CAD Overall Impression/Recommendation

This template forms the top of the mammography CAD results sub-tree. The contents of this template describe the overall impression the mammography CAD device had for the mammographic evidence presented and any recommendations that the mammography CAD device made. The details of the overall impression and recommendation are expressed in this instance of the Mammography CAD Impression/Recommendation Body (see TID 4002 “Mammography CAD Impression/Recommendation Body”). The data from which the details are inferred, are expressed in the Mammography CAD Individual Impression/Recommendations (see TID 4003 “Mammography CAD Individual Impression/Recommendation”), of which there may be several.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4001. Mammography CAD Overall Impression/Recommendation

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111017, DCM, "CAD Processing and Findings Summary") | 1 | M |  | DCID 6047 “CAD Processing and Findings Summary”
2 | > | HAS PROPERTIES | INCLUDE | DTID 4002 “Mammography CAD Impression/Recommendation Body” | 1 | U |  |
3 | > | INFERRED FROM | INCLUDE | DTID 4003 “Mammography CAD Individual Impression/Recommendation” | 1-n | MC | Shall be present if 1 or more (111059, DCM, "Single Image Finding") or (111015, DCM, "Composite Feature") content items are reported. |

Content Item Descriptions

CAD Processing and Findings Summary | This code value is used to express if and why the Overall Impression/Recommendation sub-tree is empty. The Summary of Detections and Summary of Analyses sub-trees of the Document Root node contain detail about which (if any) algorithms succeeded or failed. If the code value indicates that there were no findings, then the code value can be used to determine whether mammography CAD processing occurred successfully, without parsing the Summary of Detections and Summary of Analyses sub-trees.
Row 3 | There are no constraints regarding the 1-n multiplicity of the inclusion of TID 4003 “Mammography CAD Individual Impression/Recommendation” or its underlying structure, other than the TID 4001 “Mammography CAD Overall Impression/Recommendation” and TID 4003 “Mammography CAD Individual Impression/Recommendation” requirements. Individual Impression/Recommendation containers may be organized, for example per image, per finding or composite feature, or some combination thereof.

### TID 4002 Mammography CAD Impression/Recommendation Body

The details of an impression and recommendation are expressed in this template. It is applied to both Mammography CAD Overall Impression/Recommendation (TID 4001 “Mammography CAD Overall Impression/Recommendation”) and Mammography CAD Individual Impression/Recommendation (TID 4003 “Mammography CAD Individual Impression/Recommendation”).

Type:

  Non-Extensible

Order:

  Significant

Table TID 4002. Mammography CAD Impression/Recommendation Body

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111005, DCM, "Assessment Category") | 1-n | MC | At least one of rows 1, 3, 5, 6, 8, 9 shall be present. | DCID 6026 “Mammography Assessment”
2 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 6022 “Side”
3 |  |  | CODE | EV (111023, DCM, "Differential Diagnosis/ Impression") | 1-n | MC | At least one of rows 1, 3, 5, 6, 8, 9 shall be present. | DCID 6002 “Change Since Last Mammogram Or Prior Surgery”
4 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 6022 “Side”
5 |  |  | TEXT | EV (111033, DCM, "Impression Description") | 1 | MC | At least one of rows 1, 3, 5, 6, 8, 9 shall be present. |
6 |  |  | CODE | EV (111053, DCM, "Recommended Follow-up") | 1-n | MC | At least one of rows 1, 3, 5, 6, 8, 9 shall be present. | DCID 6028 “Mammography Recommended Follow-up”
7 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 6022 “Side”
8 |  |  | NUM | EV (111055, DCM, "Recommended Follow-up Interval") | 1 | MC | At least one of rows 1, 3, 5, 6, 8, 9 shall be present. May be present only if (111054, DCM, "Recommended Follow-up Date") is not present. | UNITS = DCID 6046 “Units of Follow-up Interval” Values = Integer ≥ 0, where 0 = immediate follow-up
9 |  |  | DATE | EV (111054, DCM, "Recommended Follow-up Date") | 1 | MC | At least one of rows 1, 3, 5, 6, 8, 9 shall be present. May be present only if (111055, DCM, "Recommended Follow-up Interval") is not present. | Shall be later than date of exam
10 |  |  | NUM | EV (111013, DCM, "Certainty of impression") | 1 | UC | May be present only if (111005, DCM, "Assessment Category"), (111023, DCM, "Differential Diagnosis/Impression") or (111033, DCM, "Impression Description") is present. | UNITS = EV (%, UCUM, "Percent") Values = 0 - 100
11 |  |  | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1-n | M |  |
12 |  |  | NUM | DCID 6142 “Calculated Value” | 1-n | U |  |
13 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 6022 “Side”
14 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | M |  | DCID 6140 “Calculation Methods”
15 | > | INFERRED FROM | TEXT | EV (112034, DCM, "Calculation Description") | 1 | U |  |

Content Item Descriptions

Certainty of Impression | The certainty that the device populating the Mammography CAD SR report places on this impression, where 0 equals no certainty and 100 equals certainty.
Impression Description | Free-form text describing the overall or an individual impression

### TID 4003 Mammography CAD Individual Impression/Recommendation

This template collects an individual impression the mammography CAD device had for a lesion, non-lesion object, or correlation of related objects. The details of the impression and recommendation are expressed in the Mammography CAD Impression/Recommendation Body (see TID 4002 “Mammography CAD Impression/Recommendation Body”). The data from which the details are inferred are expressed in the Composite Features (see TID 4004 “Mammography CAD Composite Feature”) and/or Single Image Findings (see TID 4006 “Mammography CAD Single Image Finding”) of which there may be several.

The sub-tree headed by this template is illustrated in Figure E.1-3 “Example of Individual Impression/Recommendation Levels of Mammography CAD SR Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Table TID 4003. Mammography CAD Individual Impression/Recommendation

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111034, DCM, "Individual Impression/ Recommendation") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
3 | > | CONTAINS | INCLUDE | DTID 4002 “Mammography CAD Impression/Recommendation Body” | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 4004 “Mammography CAD Composite Feature” | 1-n | MC | At least one of rows 4, 5 shall be present. |
5 | > | CONTAINS | INCLUDE | DTID 4006 “Mammography CAD Single Image Finding” | 1-n | MC | At least one of rows 4, 5 shall be present. |

Content Item Descriptions

Rendering Intent | This content item constrains the SCP receiving the Mammography CAD SR IOD in its use of the contents of this template and its target content items. Mammography CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent mammography CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .

### TID 4004 Mammography CAD Composite Feature

This template collects a composite feature for a lesion, non-lesion object, or correlation of related objects. The details of the composition are expressed in the Mammography CAD Composite Feature Body (see TID 4005 “Mammography CAD Composite Feature Body”). The data from which the details are inferred, are expressed in the Composite Features (see TID 4004 “Mammography CAD Composite Feature”) and/or Single Image Findings (see TID 4006 “Mammography CAD Single Image Finding”), of which there may be several.

A Composite Feature shall be INFERRED FROM any combination of two or more Composite Features or Single Image Findings or mixture thereof.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4004. Mammography CAD Composite Feature

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111015, DCM, "Composite Feature") | 1 | M |  | DCID 6016 “Mammography Composite Feature”
2 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
4 | > | HAS PROPERTIES | INCLUDE | DTID 4005 “Mammography CAD Composite Feature Body” | 1 | M |  |
5 | > | INFERRED FROM | INCLUDE | DTID 4004 “Mammography CAD Composite Feature” | 1-n | MC | At least two items shall be present: two of row 5, two of row 6, or one of each. |
6 | > | INFERRED FROM | INCLUDE | DTID 4006 “Mammography CAD Single Image Finding” | 1-n | MC | At least two items shall be present: two of row 5, two of row 6, or one of each. |
7 | > | HAS OBS CONTEXT | INCLUDE | DTID 4022 “CAD Observation Context” | 1 | MC | Shall be present only if this feature is incorporated from a different report than its parent. |

Content Item Descriptions

Rendering Intent | This content item constrains the SCP receiving the Mammography CAD SR IOD in its use of the contents of this template and its target content items. Mammography CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent mammography CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .

### TID 4005 Mammography CAD Composite Feature Body

The details of a composite feature are expressed in this template. It is applied to Mammography CAD Composite Feature (TID 4004 “Mammography CAD Composite Feature”).

Type:

  Non-Extensible

Order:

  Significant

Table TID 4005. Mammography CAD Composite Feature Body

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111016, DCM, "Composite type") | 1 | M |  | DCID 6035 “Composite Feature Relations” The value shall be (111155, DCM, "Target content items are related contra-laterally") if the parent content item has code value (F-01792, SRT, "Focal asymmetric breast tissue") or (F-01793, SRT, "Asymmetric breast tissue").
2 |  |  | CODE | EV (111057, DCM, "Scope of Feature") | 1 | M |  | DCID 6036 “Scope of Feature”
3 |  |  | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
4 |  |  | NUM | EV (111011, DCM, "Certainty of Feature") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
5 |  |  | NUM | EV (111047, DCM, "Probability of cancer") | 1 | UC | May be present only if value of parent is not (111102, DCM, "Non-lesion") | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
6 |  |  | CODE | EV (111042, DCM, "Pathology") | 1-n | U |  | BCID 6030 “Mammography Pathology Codes”
7 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1400) "Linear Measurement" shall be used.
8 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1401) "Area Measurement" shall be used.
9 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1402) "Volume Measurement" shall be used.
10 |  |  | INCLUDE | DTID 4021 “Mammography CAD Geometry” | 1-n | U |  |
11 |  |  | NUM | DCID 6037 “Mammography Quantitative Temporal Difference Type” | 1-n | UC | May be present only if the value of (111016, DCM, "Composite type") is (111153, DCM, "Target content items are related temporally") | UNITS = DCID 7460 “Units of Linear Measurement” UNITS = DCID 7461 “Units of Area Measurement” UNITS = DCID 7462 “Units of Volume Measurement” UNITS = DT (1, UCUM, "no units")
12 | > | R-INFERRED FROM | NUM |  | 2 | U |  | The referenced numeric values shall have the same Concept Name. Their UNITS shall be the same as row 11
13 |  |  | CODE | EV (111049, DCM, "Qualitative Difference") | 1-n | UC | May be present only if the value of (111016, DCM, "Composite type") is (111153, DCM, "Target content items are related temporally") | DCID 6038 “Mammography Qualitative Temporal Difference Type”
14 | > | HAS PROPERTIES | TEXT | EV (111021, DCM, "Description of Change") | 1 | U |  |
15 | > | R-INFERRED FROM | CODE |  | 2 | M |  | The referenced code values shall have the same Concept Name and be from the same context group.
16 |  |  | CODE | EV (111048, DCM, "Quadrant location") | 1 | U |  | DCID 6020 “Quadrant Location”
17 |  |  | CODE | EV (111014, DCM, "Clockface or region") | 1 | U |  | DCID 6018 “Clockface Location or Region”
18 |  |  | CODE | EV (111020, DCM, "Depth") | 1 | U |  | DCID 6024 “Depth”
19 |  |  | CODE | EV (111035, DCM, "Lesion Density") | 1 | UC | May be present only if value of parent is (F-01791, SRT, "Mammographic breast mass") or (F-01796, SRT, "Mammography breast density") | DCID 6008 “Density Modifier”
20 |  |  | CODE | EV (M-020F9, SRT, "Shape") | 1 | UC | May be present only if value of parent is (F-01791, SRT, "Mammographic breast mass") or (F-01796, SRT, "Mammography breast density") | DCID 6004 “Mammography Characteristics of Shape”
21 |  |  | CODE | EV (111037, DCM, "Margins") | 1-n | UC | May be present only if value of parent is (F-01791, SRT, "Mammographic breast mass") or (F-01796, SRT, "Mammography breast density") | DCID 6006 “Mammography Characteristics of Margin”
22 |  |  | CODE | EV (111009, DCM, "Calcification Type") | 1-n | UC | May be present only if value of parent is (F-01775, SRT, "Calcification Cluster") or (F-01776, SRT, "Individual Calcification") | DCID 6010 “Mammography Calcification Types”
23 |  |  | CODE | EV (111008, DCM, "Calcification Distribution") | 1 | UC | May be present only if value of parent is (F-01775, SRT, "Calcification Cluster") | DCID 6012 “Calcification Distribution Modifier”
24 |  |  | NUM | EV (111038, DCM, "Number of calcifications") | 1 | UC | May be present only if value of parent is (F-01775, SRT, "Calcification Cluster") | UNITS = EV (1, UCUM, "no units") Value = Integer 1 - n
25 |  |  | NUM | DCID 6142 “Calculated Value” | 1-n | U |  |
26 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | M |  | DCID 6140 “Calculation Methods”
27 | > | INFERRED FROM | TEXT | EV (112034, DCM, "Calculation Description") | 1 | U |  |

Content Item Descriptions

Certainty of Feature | The likelihood that the feature analyzed, and classified by the CODE specified in the Composite Feature parent template, is in fact that type of feature.
Volume Measurement | If dimensions for a volume are to be stated in terms of length, width, and depth, then one shall use 3 instances of TID (1400) Linear Measurement.
Row 11 | Values ≤ 0 are allowed. The two referenced numeric values are target content items of the first generation Composite Feature or Single Image Finding children of this composite feature. Given the equation, A - B, the value representing A shall be referenced first.
Qualitative Difference | The two referenced code values are target content items of the first generation Composite Feature or Single Image Finding children of this composite feature.

### TID 4006 Mammography CAD Single Image Finding

This template describes a single image finding for a lesion or other object. The details of the finding are expressed in this template and/or more specific templates. The details from which a single image Calcification Cluster is inferred may be expressed in a number of Single Image Findings (see TID 4006 “Mammography CAD Single Image Finding”) of type Individual Calcification.

A Single Image Finding of type Breast Composition may be INFERRED FROM by-reference to a Single Image Finding of type Breast Geometry.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4006. Mammography CAD Single Image Finding

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111059, DCM, "Single Image Finding") | 1 | M |  | DCID 6014 “Mammography Single Image Finding”
2 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
3 | >> | HAS PROPERTIES | NUM | EV (111071, DCM, "CAD Operating Point") | 1 | UC | IFF value of row 2 is (111151, DCM, "Presentation Optional") and row 9 of TID 4017 “CAD Detection Performed” is present | UNITS = DT ({1:n}, UCUM, "range: 1:n"), where n is the maximum specified in Row 9 of TID 4017 “CAD Detection Performed”. Value is restricted to being an integer
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
5 | > | HAS PROPERTIES | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
6 | > | HAS PROPERTIES | NUM | EV (111012, DCM, "Certainty of Finding") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
7 | > | HAS PROPERTIES | NUM | EV (111047, DCM, "Probability of cancer") | 1 | UC | May be present unless value of parent is (F-01710, SRT, "Breast composition"), (111100, DCM, "Breast geometry"), (T-04100, SRT, "Nipple"), (111099, DCM, "Selected region"), (111101, DCM, "Image quality") or (111102, DCM, "Non-lesion") | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
8 | > | HAS PROPERTIES | INCLUDE | DTID 4021 “Mammography CAD Geometry” | 1 | MC | Shall be present unless value of parent is (F-01710, SRT, "Breast composition"), (111100, DCM, "Breast geometry") or (111101, DCM, "Image quality") |
9 | > | HAS PROPERTIES | INCLUDE | DTID 4007 “Mammography CAD Breast Composition” | 1 | MC | Shall be present only if value of parent is (F-01710, SRT, "Breast composition") |
10 | > | R-INFERRED FROM | CODE |  | 1-n | UC | May be present only if value of parent is (F-01710, SRT, "Breast composition") | Shall reference a (111059, DCM, "Single Image Finding") of value: EV (111100, DCM, "Breast geometry")
11 | > | HAS PROPERTIES | INCLUDE | DTID 4008 “Mammography CAD Breast Geometry” | 1 | MC | Shall be present only if value of parent is (111100, DCM, "Breast geometry") |
12 | > | HAS PROPERTIES | INCLUDE | DTID 4009 “Mammography CAD Individual Calcification” | 1 | UC | May be present only if value of parent is (F-01776, SRT, "Individual Calcification") |
13 | > | HAS PROPERTIES | INCLUDE | DTID 4010 “Mammography CAD Calcification Cluster” | 1 | UC | May be present only if value of parent is (F-01775, SRT, "Calcification Cluster") |
14 | > | HAS PROPERTIES | INCLUDE | DTID 4011 “Mammography CAD Density” | 1 | UC | May be present only if value of parent is (F-01796, SRT, "Mammography breast density") |
15 | > | HAS PROPERTIES | CODE | EV (111297, DCM, "Nipple Characteristic") | 1 | UC | May be present only if value of parent is (T-04100, SRT, "Nipple") | DCID 6039 “Nipple Characteristic”
16 | > | HAS PROPERTIES | INCLUDE | DTID 4012 “Mammography CAD Non-Lesion” | 1 | MC | Shall be present only if value of parent is (111102, DCM, "Non-lesion") |
17 | > | HAS PROPERTIES | INCLUDE | DTID 4013 “Mammography CAD Selected Region” | 1 | MC | Shall be present only if value of parent is (111099, DCM, "Selected Region") |
18 | > | R-INFERRED FROM | IMAGE |  | 1 | MC | IF value of parent is (111101, DCM, "Image quality") and IFF row 19 is not present | Shall reference an IMAGE content item in the (111028, DCM, "Image Library")
19 | > | HAS PROPERTIES | SCOORD | EV (111030, DCM, "Image Region") | 1-n | MC | IF value of parent is (111101, DCM, "Image quality") and IFF row 18 is not present |
20 | >> | R-SELECTED FROM | IMAGE |  | 1 | M |  | All the (111030, DCM, "Image Region") content items in a single invocation of this template shall reference the same IMAGE content item in the (111028, DCM, "Image Library")
21 | > | HAS PROPERTIES | INCLUDE | DTID 4014 “CAD Image Quality” | 1-n | MC | Shall be present only if value of parent is (111101, DCM, "Image quality") | $QualityFinding = DCID 6041 “Mammography Image Quality Finding” $QualityStandard = DCID 6045 “Mammography Types of Quality Control Standard”
22 | > | HAS PROPERTIES | NUM | DCID 6142 “Calculated Value” | 1-n | U |  |
23 | >> | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | M |  | DCID 6140 “Calculation Methods”
24 | >> | INFERRED FROM | TEXT | EV (112034, DCM, "Calculation Description") | 1 | U |  |
25 | > | INFERRED FROM | INCLUDE | DTID 4006 “Mammography CAD Single Image Finding” | 1-n | UC | May be present only if value of parent is (F-01775, SRT, "Calcification Cluster") | EV (F-01776, SRT, "Individual Calcification")
26 | > | HAS OBS CONTEXT | INCLUDE | DTID 4022 “CAD Observation Context” | 1 | MC | Shall be present only if this finding is incorporated from a different report than its parent. |

Content Item Descriptions

Rendering Intent | This content item constrains the SCP receiving the Mammography CAD SR IOD in its use of the contents of this template and its target content items. Mammography CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent mammography CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .
CAD Operating Point | Additional information to use when Rendering Intent is "Presentation Optional". A CAD Operating Point of zero is not sent, and is encoded as a Rendering Intent of "Presentation Required". See Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 and Section E.4 “CAD Operating Point” in PS3.17 .
Single Image Finding | A Single Image Finding (whose parent is a Single Image Finding of type Calcification Cluster) allows one level of nesting for the definition of individual calcifications within the cluster. To use this template recursively, this Single Image Finding code value shall be "Individual Calcification".
Certainty of Finding | The likelihood that the finding detected, and classified by the CODE specified in the Single Image Finding parent template, is in fact that type of finding.

### TID 4007 Mammography CAD Breast Composition

Type:

  Non-Extensible

Order:

  Significant

Table TID 4007. Mammography CAD Breast Composition

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (F-01710, SRT, "Breast composition") | 1 | MC | At least one of row 1 or 2 shall be present | DCID 6000 “Overall Breast Composition”
2 |  |  | NUM | EV (111046, DCM, "Percent Fibroglandular Tissue") | 1 | MC | At least one of row 1 or 2 shall be present | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100

Content Item Descriptions

Percent Fibroglandular Tissue | Percent of breast area that is mammographically dense, excluding pectoralis muscle.

### TID 4008 Mammography CAD Breast Geometry

Type:

  Non-Extensible

Order:

  Significant

Table TID 4008. Mammography CAD Breast Geometry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | SCOORD | EV (111007, DCM, "Breast Outline Including Pectoral Muscle Tissue") | 1 | M |  | GRAPHIC TYPE = {POLYLINE}
2 | > | R-SELECTED FROM | IMAGE |  | 1 | M |  | Shall reference an IMAGE content item in the (111028, DCM, "Image Library")
3 |  |  | SCOORD | EV (111045, DCM, "Pectoral Muscle Outline") | 1 | U |  | GRAPHIC TYPE = {POLYLINE}
4 | > | R-SELECTED FROM | IMAGE |  | 1 | M |  | Shall reference the same node as row 2

Content Item Descriptions

Figure A-8a. Example of Breast Outline Including Pectoral Muscle Tissue | Figure A-8b. Example of Pectoral Muscle Outline

### TID 4009 Mammography CAD Individual Calcification

This template provides the detail specific to an individual calcification.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4009. Mammography CAD Individual Calcification

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111009, DCM, "Calcification Type") | 1-n | MC | At least one of rows 1, 2, 3 shall be present | DCID 6010 “Mammography Calcification Types”
2 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | MC | At least one of rows 1, 2, 3 shall be present | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1400) Linear Measurement shall be used.
3 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | MC | At least one of rows 1, 2, 3 shall be present | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1401) Area Measurement shall be used.
4 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1402) Volume Measurement shall be used.

### TID 4010 Mammography CAD Calcification Cluster

This template provides the detail specific to a calcification cluster.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4010. Mammography CAD Calcification Cluster

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111009, DCM, "Calcification Type") | 1-n | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | DCID 6010 “Mammography Calcification Types”
2 |  |  | CODE | EV (111008, DCM, "Calcification Distribution") | 1 | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | DCID 6012 “Calcification Distribution Modifier”
3 |  |  | NUM | EV (111038, DCM, "Number of calcifications") | 1 | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | UNITS = EV (1, UCUM, "no units") Value = Integer > = 1
4 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1400) Linear Measurement shall be used.
5 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1401) Area Measurement shall be used.
6 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1402) Volume Measurement shall be used.

### TID 4011 Mammography CAD Density

This template provides the detail specific to a density.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4011. Mammography CAD Density

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111035, DCM, "Lesion Density") | 1 | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | DCID 6008 “Density Modifier”
2 |  |  | CODE | EV (M-020F9, SRT, "Shape") | 1 | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | DCID 6004 “Mammography Characteristics of Shape”
3 |  |  | CODE | EV (111037, DCM, "Margins") | 1-n | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | DCID 6006 “Mammography Characteristics of Margin”
4 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1400) Linear Measurement shall be used.
5 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | MC | At least one of rows 1, 2, 3, 4, 5 shall be present | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1401) Area Measurement shall be used.
6 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1402) Volume Measurement shall be used.

### TID 4012 Mammography CAD Non-Lesion

This template provides the detail specific to a finding other than a lesion (see CID 6040 “Non-lesion Object Type”).

Type:

  Non-Extensible

Order:

  Significant

Table TID 4012. Mammography CAD Non-Lesion

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111039, DCM, "Object type") | 1 | M |  | DCID 6040 “Non-lesion Object Type”
2 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1400) Linear Measurement shall be used.
3 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1401) Area Measurement shall be used.
4 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1402) Volume Measurement shall be used.

### TID 4013 Mammography CAD Selected Region

This template provides the detail specific to a selected region. A selected region is any mammography CAD derived arbitrary region of the image, whether within the breast outline or not. This can be use to delineate regions such as the intramammary fold.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4013. Mammography CAD Selected Region

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (111058, DCM, "Selected Region Description") | 1 | M |  |
2 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1400) Linear Measurement shall be used.
3 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1401) Area Measurement shall be used.
4 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  | If the measured path is encoded, the SCOORD and its by-reference relationship to the IMAGE in TID (1402) Volume Measurement shall be used.

### TID 4014 CAD Image Quality

This template provides the detail specific to image quality. It allows the encoding of descriptors of image quality (e.g., CID 6041 “Mammography Image Quality Finding”) for a given image or region of an image. For instance, images with partial motion blur can be identified with the region noted.

Table TID 4014. Parameters

Parameter Name | Parameter Usage
--- | ---
$QualityFinding | Coded term or Context Group for Quality Finding
$QualityStandard | Coded term or Context Group for Quality Control Standard

Type:

  Non-Extensible

Order:

  Significant

Table TID 4014. CAD Image Quality

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111052, DCM, "Quality Finding") | 1 | M |  | $QualityFinding
2 | > | HAS PROPERTIES | CODE | EV (111050, DCM, "Quality Assessment") | 1 | U |  | DCID 6044 “Types of Image Quality Assessment”
3 | > | HAS PROPERTIES | CODE | EV (111051, DCM, "Quality Control Standard") | 1 | UC | Shall be present if row 2 is present. | $QualityStandard
4 | > | HAS PROPERTIES | NUM | EV (111029, DCM, "Image Quality Rating") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100

Content Item Descriptions

Image Quality Rating | A numeric value in the range 0 to 100, inclusive, where 0 is worst quality and 100 is best quality.

### TID 4015 CAD Detections Performed

This template gathers two lists of detection algorithms attempted, grouped by success/failure status. Algorithms not attempted are not mentioned in this sub-tree of the Document Root (e.g., TID 4000 “Mammography CAD Document Root”). This information forms the basis for understanding why a CAD report may produce no (or fewer than anticipated) detection results.

The sub-tree formed by this template is illustrated in Figure E.1-2 “Summary of Detections and Analyses Levels of Mammography CAD SR Content Tree” in PS3.17 .

Table TID 4015. Parameters

Parameter Name | Parameter Usage
--- | ---
$DetectionCode | Coded term or Context Group for Detection Performed

Type:

  Non-Extensible

Order:

  Significant

Table TID 4015. CAD Detections Performed

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111063, DCM, "Successful Detections") | 1 | MC | Shall be present only if value of parent is (111222, DCM, "Succeeded") or (111223, DCM, "Partially Succeeded") |
2 | > | CONTAINS | INCLUDE | DTID 4017 “CAD Detection Performed” | 1-n | M |  | $DetectionCode = $DetectionCode
3 |  |  | CONTAINER | EV (111025, DCM, "Failed Detections") | 1 | MC | Shall be present only if value of parent is (111224, DCM, "Failed") or (111223, DCM, "Partially Succeeded") |
4 | > | CONTAINS | INCLUDE | DTID 4017 “CAD Detection Performed” | 1-n | M |  | $DetectionCode = $DetectionCode

### TID 4016 CAD Analyses Performed

This template gathers two lists of analysis algorithms attempted, grouped by success/failure status. Algorithms not attempted are not mentioned in this sub-tree of the Document Root (e.g., TID 4000 “Mammography CAD Document Root”). This information forms the basis for understanding why a CAD report may produce no (or fewer than anticipated) analysis results.

The sub-tree formed by this template is illustrated in Figure E.1-2 “Summary of Detections and Analyses Levels of Mammography CAD SR Content Tree” in PS3.17 .

Table TID 4016. Parameters

Parameter Name | Parameter Usage
--- | ---
$AnalysisCode | Coded term or Context Group for Analysis Performed

Type:

  Non-Extensible

Order:

  Significant

Table TID 4016. CAD Analyses Performed

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111062, DCM, "Successful Analyses") | 1 | MC | Shall be present only if value of parent is (111222, DCM, "Succeeded") or (111223, DCM, "Partially Succeeded") |
2 | > | CONTAINS | INCLUDE | DTID 4018 “CAD Analysis Performed” | 1-n | M |  | $AnalysisCode = $AnalysisCode
3 |  |  | CONTAINER | EV (111024, DCM, "Failed Analyses") | 1 | MC | Shall be present only if value of parent is (111224, DCM, "Failed") or (111223, DCM, "Partially Succeeded") |
4 | > | CONTAINS | INCLUDE | DTID 4018 “CAD Analysis Performed” | 1-n | M |  | $AnalysisCode = $AnalysisCode

### TID 4017 CAD Detection Performed

This template fully identifies a detection algorithm and the images and/or image regions on which it operated (see TID 4015 “CAD Detections Performed”).

Table TID 4017. Parameters

Parameter Name | Parameter Usage
--- | ---
$DetectionCode | Coded term or Context Group for Detection Performed

Type:

  Non-Extensible

Order:

  Significant

Table TID 4017. CAD Detection Performed

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111022, DCM, "Detection Performed") | 1 | M |  | $DetectionCode
2 | > | HAS PROPERTIES | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
3 | > | HAS PROPERTIES | IMAGE |  | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present |
4 | > | R-HAS PROPERTIES | IMAGE |  | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present | Shall reference IMAGE content item(s) in the (111028, DCM, "Image Library")
5 | > | HAS PROPERTIES | UIDREF | EV (112002, DCM, "Series Instance UID") | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present |
6 | > | HAS PROPERTIES | SCOORD | EV (111030, DCM, "Image Region") | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present |
7 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR row 8 |
8 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR row 7 | Shall reference an IMAGE content item in the (111028, DCM, "Image Library")
9 | > |  | INCLUDE | DTID 4023 “CAD Operating Points” | 1 | U |  |

Content Item Descriptions

CAD Algorithm Identification | If more than one detection algorithm has the same "Detection Performed" code value (e.g., CID 6014 “Mammography Single Image Finding”) then the "CAD Algorithm Identification" shall unambiguously distinguish between algorithms.
Rows 5, 6 | Mammography CAD SR: When this template is invoked for the Mammography CAD SR, the Image Library is mandatory, thus only row 4 and/or row 6 shall be present. Chest CAD SR: When this template is invoked for the Chest CAD SR, the Image Library is optional, thus any combination of rows 3, 4, 5 and 6 may be present. Colon CAD SR: When this template is invoked for the Colon CAD SR, the Image Library does not exist, thus rows 3, 5, and/or 6 may be present and row 4 shall not be present.
Rows 5, 6 | Mammography CAD SR: When this template is invoked for the Mammography CAD SR, the Image Library is mandatory, thus only row 8 shall be present. Chest CAD SR: When this template is invoked for the Chest CAD SR, the Image Library is optional, thus row 7 or 8 may be present. Colon CAD SR: When this template is invoked for the Colon CAD SR, the Image Library does not exist, thus only row 7 may be present.

### TID 4018 CAD Analysis Performed

This template fully identifies an analysis algorithm and the images and/or image regions on which it operated (see TID 4016 “CAD Analyses Performed”).

Table TID 4018. Parameters

Parameter Name | Parameter Usage
--- | ---
$AnalysisCode | Coded term or Context Group for Analysis Performed

Type:

  Non-Extensible

Order:

  Significant

Table TID 4018. CAD Analysis Performed

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | $AnalysisCode
2 | > | HAS PROPERTIES | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
3 | > | HAS PROPERTIES | IMAGE |  | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present |
4 | > | R-HAS PROPERTIES | IMAGE |  | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present | Shall reference IMAGE content item(s) in the (111028, DCM, "Image Library")
5 | > | HAS PROPERTIES | UIDREF | EV (112002, DCM, "Series Instance UID") | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present |
6 | > | HAS PROPERTIES | SCOORD | EV (111030, DCM, "Image Region") | 1-n | MC | At least one of row 3, 4, 5 or 6 shall be present |
7 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 8 |
8 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR Row 7 | Shall reference an IMAGE content item in the (111028, DCM, "Image Library")
9 | > |  | INCLUDE | DTID 4023 “CAD Operating Points” | 1 | U |  |

Content Item Descriptions

CAD Algorithm Identification | If more than one analysis algorithm has the same "Analysis Performed" code value (e.g., CID 6043 “Types of Mammography CAD Analysis”) then the "CAD Algorithm Identification" shall unambiguously distinguish between algorithms.
Rows 5, 6 | Mammography CAD SR: When this template is invoked for the Mammography CAD SR, the Image Library is mandatory, and a total of at least two instances of row 4 or row 6 shall be present. Chest CAD SR: When this template is invoked for the Chest CAD SR, the Image Library is optional, thus any combination of rows 3, 4, 5 and 6 may be present. Colon CAD SR: When this template is invoked for the Colon CAD SR, the Image Library does not exist, thus rows 3, 5 and/or 6 may be present and row 4 shall not be present.
Rows 5, 6 | Mammography CAD SR: When this template is invoked for the Mammography CAD SR, the Image Library is mandatory, thus only row 8 shall be present. Chest CAD SR: When this template is invoked for the Chest CAD SR, the Image Library is optional, thus row 7 or 8 may be present. Colon CAD SR: When this template is invoked for the Colon CAD SR, the Image Library does not exist, thus only row 7 may be present.

### TID 4019 CAD Algorithm Identification

This template details the algorithm unambiguously. Re-state the software identification from the General Equipment Module of the SR IOD if all algorithms are unambiguously defined by that module.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4019. CAD Algorithm Identification

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (111001, DCM, "Algorithm Name") | 1 | M |  |
2 |  |  | TEXT | EV (111003, DCM, "Algorithm Version") | 1 | M |  |
3 |  |  | TEXT | EV (111002, DCM, "Algorithm Parameters") | 1-n | U |  |

### TID 4020 CAD Image Library Entry

Each instance of the Image Library Entry template contains the Image SOP Class and Instance UIDs, and selected attributes for an image that facilitate spatial analysis without having to retrieve the entire set of referenced images. If values for the attributes are not present in the Image SOP Instance, then as many of the attributes as possible should be derived.

Table TID 4020. Parameters

Parameter Name | Parameter Usage
--- | ---
$ImageLaterality | Coded term or Context Group for Image Laterality
$ImageView | Coded term or Context Group for Image View
$ImageViewMod | Coded term or Context Group for Image View Modifier

Type:

  Extensible

Order:

  Significant

Table TID 4020. CAD Image Library Entry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | IMAGE |  | 1 | M |  |
2 | > | HAS ACQ CONTEXT | CODE | EV (111027, DCM, "Image Laterality") | 1 | MC | Shall be present if (0020,0062) is in the Image Instance | $ImageLaterality
3 | > | HAS ACQ CONTEXT | CODE | EV (111031, DCM, "Image View") | 1 | MC | Shall be present if (0054,0220) is in the Image Instance | $ImageView
4 | >> | HAS CONCEPT MOD | CODE | EV (111032, DCM, "Image View Modifier") | 1-n | MC | Shall be present if (0054,0222) is in the Image Instance | $ImageViewMod
5 | > | HAS ACQ CONTEXT | TEXT | EV (111044, DCM, "Patient Orientation Row") | 1 | MC | Shall be present if (0020,0020) is in the Image Instance |
6 | > | HAS ACQ CONTEXT | TEXT | EV (111043, DCM, "Patient Orientation Column") | 1 | MC | Shall be present if (0020,0020) is in the Image Instance |
7 | > | HAS ACQ CONTEXT | DATE | EV (111060, DCM, "Study Date") | 1 | MC | Shall be present if (0008,0020) is in the Image Instance |
8 | > | HAS ACQ CONTEXT | TIME | EV (111061, DCM, "Study Time") | 1 | MC | Shall be present if (0008,0030) is in the Image Instance |
9 | > | HAS ACQ CONTEXT | DATE | EV (111018, DCM, "Content Date") | 1 | MC | Shall be present if (0008,0023) is in the Image Instance |
10 | > | HAS ACQ CONTEXT | TIME | EV (111019, DCM, "Content Time") | 1 | MC | Shall be present if (0008,0033) is in the Image Instance |
11 | > | HAS ACQ CONTEXT | NUM | EV (111026, DCM, "Horizontal Pixel Spacing") | 1 | MC | Shall be present if (0018,1164) or (0028,0030) is in the Image Instance | UNITS = EV (um, UCUM, "micrometer") UNITS = EV (mm, UCUM, "millimeter")
12 | > | HAS ACQ CONTEXT | NUM | EV (111066, DCM, "Vertical Pixel Spacing") | 1 | MC | Shall be present if (0018,1164) or (0028,0030) is in the Image Instance | UNITS = EV (um, UCUM, "micrometer") UNITS = EV (mm, UCUM, "millimeter")
13 | > | HAS ACQ CONTEXT | NUM | EV (112011, DCM, "Positioner Primary Angle") | 1 | UC | May be present if (0018,1510) is in the Image Instance |
14 | > | HAS ACQ CONTEXT | NUM | EV (112012, DCM, "Positioner Secondary Angle") | 1 | UC | May be present if (0018,1511) is in the Image Instance |
15 | > | HAS ACQ CONTEXT | NUM | EV (112226, DCM, "Spacing between slices") | 1 | UC | May be computed from the Image Position (Patient) (0020,0032) projected onto the normal to the Image Orientation (Patient) (0020,0037) if present; may or may not be the same as the Spacing Between Slices (0018,0088) if present. | UNITS = EV (mm, UCUM, "millimeter")
16 | > | HAS ACQ CONTEXT | NUM | EV (112225, DCM, "Slice Thickness") | 1 | UC | May be present if Slice Thickness (0018,0050) is in the Image Instance. | UNITS = EV (mm, UCUM, "millimeter")
17 | > | HAS ACQ CONTEXT | UIDREF | EV (112227, DCM, "Frame of Reference UID") | 1 | UC | May be present if Frame of Reference UID (0020,0052) is in the Image Instance. |
18 | > | HAS ACQ CONTEXT | NUM | EV (110901, DCM, "Image Position (Patient) X") | 1 | UC | May be present if Image Position (Patient) (0020,0032) is in the Image Instance, and is the first value of Image Position (Patient) (0020,0032) for the referenced image or frame. | UNITS = EV (mm, UCUM, "millimeter")
19 | > | HAS ACQ CONTEXT | NUM | EV (110902, DCM, "Image Position (Patient) Y") | 1 | MC | Shall be present if Row 18 is present, and is the second value of Image Position (Patient) (0020,0032) in the Image Instance for the referenced image or frame. | UNITS = EV (mm, UCUM, "millimeter")
20 | > | HAS ACQ CONTEXT | NUM | EV (110903, DCM, "Image Position (Patient) Z") | 1 | MC | Shall be present if Row 18 is present, and is the second value of Image Position (Patient) (0020,0032) in the Image Instance for the referenced image or frame. | UNITS = EV (mm, UCUM, "millimeter")
21 | > | HAS ACQ CONTEXT | NUM | EV (110904, DCM, "Image Orientation (Patient) Row X") | 1 | UC | May be present if Image Position (Patient) (0020,0037) is in the Image Instance, and is the first value of Image Orientation (Patient) (0020,0037) for the referenced image or frame. | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
22 | > | HAS ACQ CONTEXT | NUM | EV (110905, DCM, "Image Orientation (Patient) Row Y") | 1 | MC | Shall be present if Row 21 is present, and is the second value of Image Orientation (Patient) (0020,0037) in the Image Instance for the referenced image or frame. | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
23 | > | HAS ACQ CONTEXT | NUM | EV (110906, DCM, "Image Orientation (Patient) Row Z") | 1 | MC | Shall be present if Row 21 is present, and is the third value of Image Orientation (Patient) (0020,0037) in the Image Instance for the referenced image or frame. | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
24 | > | HAS ACQ CONTEXT | NUM | EV (110907, DCM, "Image Orientation (Patient) Column X") | 1 | MC | Shall be present if Row 21 is present, and is the fourth value of Image Orientation (Patient) (0020,0037) in the Image Instance for the referenced image or frame. | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
25 | > | HAS ACQ CONTEXT | NUM | EV (110908, DCM, "Image Orientation (Patient) Column Y") | 1 | MC | Shall be present if Row 21 is present, and is the fifth value of Image Orientation (Patient) (0020,0037) in the Image Instance for the referenced image or frame. | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
26 | > | HAS ACQ CONTEXT | NUM | EV (110909, DCM, "Image Orientation (Patient) Column Z") | 1 | MC | Shall be present if Row 21 is present, and is the sixth value of Image Orientation (Patient) (0020,0037) in the Image Instance for the referenced image or frame. | UNITS = EV ({-1:1}, UCUM, "{-1:1}")
27 | > | HAS ACQ CONTEXT | NUM | EV (110910, DCM, "Pixel Data Rows") | 1 | UC | May be present if Rows (0028,0010) is in the Image Instance. | UNITS = EV ({pixels}, UCUM, "pixels")
28 | > | HAS ACQ CONTEXT | NUM | EV (110911, DCM, "Pixel Data Columns") | 1 | MC | Shall be present if Row 27 is present, and is the value of Columns (0028,0011) in the Image Instance. | UNITS = EV ({pixels}, UCUM, "pixels")

Content Item Descriptions

Patient Orientation Row | First (row) and second (column) components of Patient Orientation (0020,0020) in the Image IOD. See Section C.7.6.1.1.1 in PS3.3 .
Patient Orientation Column
Horizontal Imager Pixel Spacing | The row (first) component of Imager Pixel Spacing (0018,1164) in the Image IOD. See Section C.8.11.4 in PS3.3 .
Vertical Imager Pixel Spacing | The column (second) component of Imager Pixel Spacing (0018,1164) in the Image IOD. See Section C.8.11.4 in PS3.3 .

### TID 4021 Mammography CAD Geometry

All geometry template invocations require specification of the location of the center of the object. Outline is optional.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4021. Mammography CAD Geometry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | SCOORD | EV (111010, DCM, "Center") | 1 | M |  | GRAPHIC TYPE = {POINT}
2 | > | R-SELECTED FROM | IMAGE |  | 1 | M |  | Shall reference an IMAGE content item in the (111028, DCM, "Image Library")
3 |  |  | SCOORD | EV (111041, DCM, "Outline") | 1 | U |  |
4 | > | R-SELECTED FROM | IMAGE |  | 1 | M |  | Shall reference the same content item as row 2
5 |  |  | SCOORD | DCID 6166 “CAD Geometry Secondary Graphical Representation” | 1-n | U |  |
6 | > | R-SELECTED FROM | IMAGE |  | 1 | M |  | Shall reference the same content item as row 2

### TID 4022 CAD Observation Context

This template is invoked when a content item, which may be the "root" of a sub-tree, is paraphrased from a prior SR document.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4022. CAD Observation Context

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | COMPOSITE | EV (111040, DCM, "Original Source") | 1 | MC | Shall be present if the original source is a DICOM object. |
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 |  |  | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |

### TID 4023 CAD Operating Points

This template describes CAD operating points. The description is deliberately left flexible and optional to allow implementation at differing levels of complexity.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4023. CAD Operating Points

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | HAS PROPERTIES | NUM | EV (111072, DCM, "Maximum CAD Operating Point") | 1 | M |  | UNITS = DT ([arb'U], UCUM, "arbitrary unit") Value is restricted to being an integer
2 |  | HAS PROPERTIES | NUM | EV (111092, DCM, "Recommended CAD Operating Point") | 1 | U |  | UNITS = DT ({0:n}, UCUM, "range: 0:n"), where n is the value specified in row 1 Value is restricted to being an integer
3 |  | HAS PROPERTIES | CONTAINER | EV (111093, DCM, "CAD Operating Point Table") | 1 | U |  |
4 | > | CONTAINS | CODE | EV (122698, DCM, "X-Concept") | 1 | M |  | DCID 6048 “CAD Operating Point Axis Label”
5 | > | CONTAINS | CODE | EV (122699, DCM, "Y-Concept") | 1 | M |  | DCID 6048 “CAD Operating Point Axis Label”
6 | > | CONTAINS | NUM | EV (111071, DCM, "CAD Operating Point") | 1-n | M | Number of instances of this row shall equal value of row 1, plus 1. | UNITS = DT ({0:n}, UCUM, "range: 0:n"), where n is the value of Row 1. Value is restricted to being an integer that is unique within the invocation of this template.
7 | >> | HAS PROPERTIES | TEXT | EV (111081, DCM, "CAD Operating Point Description") | 1 | U |  |
8 | >> | HAS PROPERTIES | NUM | The value of Row 4 | 1 | U |  |
9 | >> | HAS PROPERTIES | NUM | The value of Row 5 | 1 | U |  |

Content Item Descriptions

Maximum CAD Operating Point | The maximum possible value of CAD Operating Point for this type of Detection Performed. No CAD Operating Point value recorded in the CAD Processing and Findings Summary sub-tree of the report for this type of Detection Performed shall exceed this value. The report may or may not contain Rendering Intent = "Presentation Optional" detections that are assigned the maximum value.
Recommended CAD Operating Point | A number indicating which of the CAD operating points is recommended by the creator of a CAD SR instance as the first operating point to be used when rendering the CAD SR instance contents. Subsequent changes to the displayed operating point are implementation dependent.

## Chest CAD SR IOD Templates

The templates that comprise the Chest CAD SR IOD are interconnected as in Figure A-9.

Figure A-9. Chest CAD SR IOD Template Structure

In Figure A-9, '···' indicates possible recursive application of subordinate templates.

### TID 4100 Chest CAD Document Root

This template forms the top of a content tree that allows a chest CAD device to describe the results of detection and analysis of chest evidence. This template, together with its subordinate templates, describes both the results for presentation to radiologists and partial product results for consumption by chest CAD devices in subsequent chest CAD reports.

This template defines a Container that contains an Image Library, the CAD results, and summaries of the detection and analysis algorithms performed. The Image Library contains the Image SOP Class and Instance UIDs, and selected attributes for each image referenced in either the algorithm summaries or chest CAD results.

The atomic CAD results of Single Image Findings and Composite Features are described in the Chest CAD Findings Summary sub-tree.

The Summary of Detections and Summary of Analyses sub-trees gather lists of algorithms attempted, grouped by success/failure status. Algorithms not attempted are not mentioned in these sub-trees. This information forms the basis for understanding why a chest CAD report may produce no (or fewer than anticipated) results. Chest CAD results are constructed bottom-up, starting from Single Image Findings (see TID 4104 “Chest CAD Single Image Finding”), associated as Composite Features (see TID 4102 “Chest CAD Composite Feature”).

See Figure F.1-1 “Top Levels of Chest CAD SR Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  Yes

Table TID 4100. Chest CAD Document Root

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (112000, DCM, "Chest CAD Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | CONTAINS | CONTAINER | EV (111028, DCM, "Image Library") | 1 | U |  |
4 | >> | CONTAINS | INCLUDE | DTID 4020 “CAD Image Library Entry” | 1-n | M |  | $ImageLaterality = DCID 244 “Laterality” $ImageView = DCID 4010 “DX View” $ImageViewMod = DCID 4011 “DX View Modifier”
5 | > | CONTAINS | INCLUDE | DTID 4101 “Chest CAD Findings Summary” | 1 | M |  |
6 | > | CONTAINS | CODE | EV (111064, DCM, "Summary of Detections") | 1 | M |  | DCID 6042 “Status of Results”
7 | >> | INFERRED FROM | INCLUDE | DTID 4015 “CAD Detections Performed” | 1 | MC | Shall be present unless the value of row 6 is (111225, DCM, "Not Attempted") | $DetectionCode = DCID 6101 “Chest Finding Or Feature”, DCID 6102 “Chest Finding Or Feature Modifier”
8 | > | CONTAINS | CODE | EV (111065, DCM, "Summary of Analyses") | 1 | M |  | DCID 6042 “Status of Results”
9 | >> | INFERRED FROM | INCLUDE | DTID 4016 “CAD Analyses Performed” | 1 | MC | Shall be present unless the value of row 8 is (111225, DCM, "Not Attempted") | $AnalysisCode = DCID 6137 “Types of CAD Analysis”

Content Item Descriptions

Image Library | The "Image Library" section of the Content Tree (TID 4100 “Chest CAD Document Root”, row 3) may include all Image SOP Instances from the Current Requested Procedure Evidence Sequence (0040,A375) attribute of the SR Document General module. If a portion of another instance of a Chest CAD SR IOD is duplicated in the "Chest CAD Findings Summary" section of the Content Tree, the "Image Library" may also include all Image Library Entries referenced from the duplicated portions of the Chest CAD SR. The Image Library is intended to be used in cases where the acquisition context content items differ from image to image, such as different views and/or laterality in projection X-Ray.
Detections Performed | The "Detections Performed" and "Analyses Performed" sections of the Content Tree (TID 4100 “Chest CAD Document Root”, rows 7 and 9) together shall reference all Image SOP Instances included in the Current Requested Procedure Evidence Sequence (0040,A375) attribute of the SR Document General module.
Analyses Performed

### TID 4101 Chest CAD Findings Summary

The contents of this template describe the findings and aggregate features that the chest CAD device detected for the chest evidence presented. This template forms the chest CAD results sub-tree of the Chest CAD Document Root (TID 4100 “Chest CAD Document Root”). The data from which the details are inferred are expressed in the Composite Features (see TID 4102 “Chest CAD Composite Feature”) and/or Single Image Findings (see TID 4104 “Chest CAD Single Image Finding”), of which there may be several.

The sub-tree headed by this template is illustrated in Figure F.1-2 “Example of CAD Processing and Findings Summary Sub-Tree of Chest CAD SR Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Table TID 4101. Chest CAD Findings Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111017, DCM, "CAD Processing and Findings Summary") | 1 | M |  | DCID 6047 “CAD Processing and Findings Summary”
2 | > | INFERRED FROM | INCLUDE | DTID 4102 “Chest CAD Composite Feature” | 1-n | U |  |
3 | > | INFERRED FROM | INCLUDE | DTID 4104 “Chest CAD Single Image Finding” | 1-n | U |  |
4 | > | HAS PROPERTIES | INCLUDE | DTID 4106 “Response Evaluation” | 1-n | U |  |

Content Item Descriptions

CAD Processing and Findings Summary | This code value is used to express if and why the Chest CAD Findings Summary sub-tree is empty. The Summary of Detections and Summary of Analyses sub-trees of the Document Root node contain detail about which (if any) algorithms succeeded or failed. If the code value indicates that there were no findings, then the code value can be used to determine whether chest CAD processing occurred successfully, without parsing the Summary of Detections and Summary of Analyses sub-trees.

### TID 4102 Chest CAD Composite Feature

This template collects a composite feature for a lesion, anatomy, non-lesion object, or correlation of related objects (see TID 4101 “Chest CAD Findings Summary”). The details of the composition are expressed in the Chest CAD Composite Feature Body (see TID 4103 “Chest CAD Composite Feature Body”). The data from which the details are inferred, are expressed in the Composite Features (see TID 4102 “Chest CAD Composite Feature”) and/or Single Image Findings (see TID 4104 “Chest CAD Single Image Finding”), of which there may be several.

A Composite Feature shall be INFERRED FROM any combination of two or more Composite Features or Single Image Findings or mixture thereof.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4102. Chest CAD Composite Feature

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111015, DCM, "Composite Feature") | 1 | M |  | DCID 6101 “Chest Finding Or Feature”
2 | > | HAS CONCEPT MOD | CODE | EV (112023, DCM, "Composite Feature Modifier") | 1 | U |  | DCID 6102 “Chest Finding Or Feature Modifier”
3 | > | HAS CONCEPT MOD | TEXT | EV (112050, DCM, "Anatomic Identifier") | 1 | U |  |
4 | > | HAS CONCEPT MOD | CODE | EV (112003, DCM, "Associated Chest Component") | 1 | MC | Shall be present IFF value of row 1 is (112005, DCM, "Radiographic anatomy") | DCID 6100 “Chest Component Categories”
5 | > | HAS CONCEPT MOD | CODE | EV (112037, DCM, "Non-lesion Modifier") | 1 | UC | May be present IFF value of row 1 is (111102, DCM, "Non-lesion") | DCID 6139 “Non-lesion Modifiers”
6 | > | HAS CONCEPT MOD | CODE | EV (112038, DCM, "Osseous Modifier") | 1 | UC | May be present IFF value of row 2 is from DCID 6114 “Osseous Anatomy Finding Or Feature” | DCID 6115 “Osseous Anatomy Modifiers”
7 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
8 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
9 | > | HAS OBS CONTEXT | CODE | EV (112016, DCM, "Baseline Category") | 1 | U |  | DCID 6145 “Baseline Category”
10 | > | HAS OBS CONTEXT | INCLUDE | DTID 4022 “CAD Observation Context” | 1 | MC | Shall be present IFF this feature is duplicated from a different report than its parent. |
11 | > | HAS OBS CONTEXT | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
12 | > | HAS PROPERTIES | INCLUDE | DTID 4103 “Chest CAD Composite Feature Body” | 1 | M |  |
13 | > | INFERRED FROM | INCLUDE | DTID 4102 “Chest CAD Composite Feature” | 1-n | MC | At least two items shall be present: two of row 13, two of row 14, or one of each. |
14 | > | INFERRED FROM | INCLUDE | DTID 4104 “Chest CAD Single Image Finding” | 1-n | MC | At least two items shall be present: two of row 13, two of row 14, or one of each. |

Content Item Descriptions

Anatomic Identifier | An identifier of an anatomic feature when a multiplicity of features of that type may be present, such as "Rib 1", "Rib 2" or thoracic vertebrae "T1" or "T2".
Rendering Intent | This content item constrains the SCP receiving the Chest CAD SR IOD in its use of the contents of this template and its target content items. Chest CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent chest CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .

### TID 4103 Chest CAD Composite Feature Body

The details of a composite feature are expressed in this template. It is applied to Chest CAD Composite Feature (TID 4102 “Chest CAD Composite Feature”).

Type:

  Non-Extensible

Order:

  Significant

Table TID 4103. Chest CAD Composite Feature Body

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111016, DCM, "Composite type") | 1 | M |  | DCID 6035 “Composite Feature Relations”
2 |  |  | CODE | EV (111057, DCM, "Scope of Feature") | 1 | M |  | DCID 6036 “Scope of Feature”
3 |  |  | NUM | EV (111011, DCM, "Certainty of feature") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
4 |  |  | INCLUDE | DTID 4107 “Chest CAD Geometry” | 1 | U |  |
5 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  |
6 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  |
7 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  |
8 |  |  | INCLUDE | DTID 4105 “Chest CAD Descriptors” | 1 | U |  |
9 |  |  | NUM | DCID 6133 “Chest Quantitative Temporal Difference Type” | 1-n | UC | May be present IFF the value of row 1 is (111153, DCM, "Target content items are related temporally") |
10 | > | R-INFERRED FROM | NUM |  | 2 | U |  | The referenced numeric values shall have the same Concept Name. Their UNITS shall be the same as row 9
11 |  |  | CODE | EV (111049, DCM, "Qualitative Difference") | 1-n | UC | May be present only if the value of row 1 is (111153, DCM, "Target content items are related temporally") | DCID 6134 “Chest Qualitative Temporal Difference Type”
12 | > | HAS PROPERTIES | TEXT | EV (111021, DCM, "Description of Change") | 1 | U |  |
13 | > | R-INFERRED FROM | CODE |  | 2 | M |  | The referenced content items shall have the same Concept Name and their code values shall be from the same context group.

Content Item Descriptions

Certainty of Feature | The certainty of the CAD device that the feature analyzed and classified by the CODE, as specified in the Composite Feature parent template, is in fact that type of feature.
Volume Measurement | If dimensions for a volume are to be stated in terms of length, width, and depth, then one shall use 3 instances of TID (1400) Linear Measurement.
Row 9 | Values ≤ 0 are allowed. The two referenced numeric values are target content items of the first generation Composite Feature or Single Image Finding children of this composite feature. Given the equation, A - B, the value representing A shall be referenced first.
Qualitative Difference | The two referenced code values are target content items of the first generation Composite Feature or Single Image Finding children of this composite feature.

### TID 4104 Chest CAD Single Image Finding

This template describes a single image finding for a lesion or other object. The details of the finding are expressed in this template and/or more specific templates.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4104. Chest CAD Single Image Finding

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111059, DCM, "Single Image Finding") | 1 | M |  | DCID 6101 “Chest Finding Or Feature”
2 | > | HAS CONCEPT MOD | CODE | EV (112024, DCM, "Single Image Finding Modifier") | 1 | U |  | DCID 6102 “Chest Finding Or Feature Modifier”
3 | > | HAS CONCEPT MOD | TEXT | EV (112050, DCM, "Anatomic Identifier") | 1 | U |  |
4 | > | HAS CONCEPT MOD | CODE | EV (112003, DCM, "Associated Chest Component") | 1 | MC | Shall be present IFF value of row 1 is (112005, DCM, "Radiographic anatomy") | DCID 6100 “Chest Component Categories”
5 | > | HAS CONCEPT MOD | CODE | EV (112037, DCM, "Non-lesion Modifier") | 1 | UC | May be present IFF value of row 1 is (111102, DCM, "Non-lesion") | DCID 6139 “Non-lesion Modifiers”
6 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
7 | >> | HAS PROPERTIES | NUM | EV (111071, DCM, "CAD Operating Point") | 1 | UC | IFF value of row 6 is (111151, DCM, "Presentation Optional") and row 1 of TID 4023 “CAD Operating Points” is present for the finding identified in row 1 | UNITS = DT ({1:n}, UCUM, "range: 1:n"), where n is the maximum specified in Row 1 of TID 4023 “CAD Operating Points” for the finding identified in row 1. Value is restricted to being an integer
8 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
9 | > | HAS OBS CONTEXT | CODE | EV (112016, DCM, "Baseline Category") | 1 | U |  | DCID 6145 “Baseline Category”
10 | > | HAS OBS CONTEXT | INCLUDE | DTID 4022 “CAD Observation Context” | 1 | MC | Shall be present IFF this finding is duplicated from a different report than its parent. |
11 | > | HAS OBS CONTEXT | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
12 | > | HAS PROPERTIES | NUM | EV (111012, DCM, "Certainty of Finding") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
13 | > | HAS PROPERTIES | TEXT | EV (111058, DCM, "Selected Region Description") | 1 | MC | Shall be present IFF value of row 1 is (111099, DCM, "Selected region") |
14 | > | HAS PROPERTIES | INCLUDE | DTID 4107 “Chest CAD Geometry” | 1 | MC | Shall be present unless value of row 1 is (111101, DCM, "Image quality") |
15 | > | HAS PROPERTIES | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  |
16 | > | HAS PROPERTIES | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  |
17 | > | HAS PROPERTIES | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  |
18 | > | HAS PROPERTIES | INCLUDE | DTID 4105 “Chest CAD Descriptors” | 1 | U |  |
19 | > | INFERRED FROM | IMAGE |  | 1 | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") and rows 20 and 21 are not present |
20 | > | R-INFERRED FROM | IMAGE |  | 1 | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") and rows 19 and 21 are not present | Shall reference an IMAGE content item in the (111028, DCM, "Image Library")
21 | > | INFERRED FROM | SCOORD | EV (111030, DCM, "Image Region") | 1-n | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") and rows 19 and 20 are not present |
22 | >> | SELECTED FROM | IMAGE |  | 1 | MC | XOR row 23 | All the row 21 content items in a single invocation of this template shall reference the same IMAGE
23 | >> | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR row 22 | All the row 21 content items in a single invocation of this template shall reference the same IMAGE content item in the (111028, DCM, "Image Library")
24 | > | HAS PROPERTIES | INCLUDE | DTID 4014 “CAD Image Quality” | 1 | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") | $QualityFinding = DCID 6135 “Image Quality Finding” $QualityStandard = DCID 6136 “Chest Types of Quality Control Standard”

Content Item Descriptions

Anatomic Identifier | An identifier of an anatomic feature when a multiplicity of features of that type may be present, such as "Rib 1", "Rib 2" or thoracic vertebrae "T1" or "T2".
Rendering Intent | This content item constrains the SCP receiving the Chest CAD SR IOD in its use of the contents of this template and its target content items. Chest CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent chest CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .
CAD Operating Point | Additional information to use when Rendering Intent is "Presentation Optional". A CAD Operating Point of zero is not sent, and is encoded as a Rendering Intent of "Presentation Required". See Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 and Section E.4 “CAD Operating Point” in PS3.17 .
Certainty of Finding | The certainty of the CAD device that the finding detected and classified by the Single Image Finding CODE specified is in fact that type of finding.

### TID 4105 Chest CAD Descriptors

This template provides qualitative detail for a Single Image Finding or Composite Feature. It is applied to Chest CAD Composite Feature (TID 4102 “Chest CAD Composite Feature”) and Chest CAD Single Image Finding (TID 4104 “Chest CAD Single Image Finding”).

Type:

  Non-Extensible

Order:

  Significant

Table TID 4105. Chest CAD Descriptors

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (112025, DCM, "Size Descriptor") | 1 | U |  | DCID 6118 “Size Descriptor”
2 |  |  | CODE | EV (112026, DCM, "Width Descriptor") | 1 | U |  | DCID 6107 “Width Descriptor”
3 |  |  | CODE | EV (112015, DCM, "Border shape") | 1 | U |  | DCID 6119 “Chest Border Shape”
4 |  |  | CODE | EV (112007, DCM, "Border definition") | 1 | U |  | DCID 6120 “Chest Border Definition”
5 |  |  | CODE | EV (112014, DCM, "Orientation Descriptor") | 1 | U |  | DCID 6121 “Chest Orientation Descriptor”
6 |  |  | CODE | EV (112009, DCM, "Type of Content") | 1-n | U |  | DCID 6122 “Chest Content Descriptor”
7 |  |  | CODE | EV (112027, DCM, "Opacity Descriptor") | 1 | U |  | DCID 6123 “Chest Opacity Descriptor”
8 |  |  | CODE | EV (112013, DCM, "Location in Chest") | 1 | U |  | DCID 6124 “Location in Chest”
9 |  |  | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
10 |  |  | CODE | EV (112006, DCM, "Distribution Descriptor") | 1-n | U |  | DCID 6128 “Chest Distribution Descriptor”
11 |  |  | CODE | EV (112028, DCM, "Abnormal Distribution of Anatomic Structure") | 1 | U |  | DCID 6108 “Chest Anatomic Structure Abnormal Distribution”
12 |  |  | CODE | EV (112008, DCM, "Site involvement") | 1-n | U |  | DCID 6129 “Chest Site Involvement”
13 |  |  | CODE | EV (G-C197, SRT, "Severity") | 1 | U |  | DCID 6130 “Severity Descriptor”
14 |  |  | CODE | EV (112010, DCM, "Texture Descriptor") | 1 | U |  | DCID 6131 “Chest Texture Descriptor”
15 |  |  | CODE | EV (112030, DCM, "Calcification Descriptor") | 1 | U |  | DCID 6132 “Chest Calcification Descriptor”
16 |  |  | NUM | DCID 6142 “Calculated Value” | 1-n | U |  |
17 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | M |  | DCID 6140 “Calculation Methods”
18 | > | INFERRED FROM | NUM | EV (112032, DCM, "Threshold Attenuation Coefficient") | 1 | U |  | UNITS = EV ([hnsf'U], UCUM, "Hounsfield unit")
19 | > | INFERRED FROM | TEXT | EV (112034, DCM, "Calculation Description") | 1 | U |  |
20 |  |  | NUM | DCID 6141 “Attenuation Coefficient Measurements” | 1-n | U |  | UNITS = EV ([hnsf'U], UCUM, "Hounsfield unit")

### TID 4106 Response Evaluation

This template provides a means to report response evaluation to cancer treatment, based on a method such as RECIST or WHO.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4106. Response Evaluation

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (112020, DCM, "Response Evaluation") | 1 | M |  |
2 | > | HAS OBS CONTEXT | CODE | EV (112021, DCM, "Response Evaluation Method") | 1 | M |  | DT (112022, DCM, "RECIST") or DT (112029, DCM, "WHO")
3 | > | CONTAINS | CODE | EV (112048, DCM, "Current Response") | 1 | U |  | DCID 6143 “Lesion Response”
4 | > | CONTAINS | CODE | EV (112049, DCM, "Best Overall Response") | 1 | U |  | DCID 6143 “Lesion Response”
5 | > | CONTAINS | NUM | EV (112051, DCM, "Measurement of Response") | 1 | U |  | UNITS not specified

### TID 4107 Chest CAD Geometry

All geometry template invocations require specification of either the location of the center of the object, the outline, or both. Geometry is a property of single image findings (see TID 4104 “Chest CAD Single Image Finding”) and composite features (see TID 4103 “Chest CAD Composite Feature Body”).

Type:

  Non-Extensible

Order:

  Significant

Table TID 4107. Chest CAD Geometry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | SCOORD | EV (111010, DCM, "Center") | 1 | MC | At least one of rows 1, 4 shall be present. | GRAPHIC TYPE = {POINT}
2 | > | SELECTED FROM | IMAGE |  | 1 | MC | XOR row 3 |
3 | > | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR row 2 | Shall reference an IMAGE content item in the (111028, DCM, "Image Library")
4 |  |  | SCOORD | EV (111041, DCM, "Outline") | 1 | MC | At least one of rows 1, 4 shall be present. |
5 | > | SELECTED FROM | IMAGE |  | 1 | MC | XOR row 6 | Shall reference the same content item as row 2
6 | > | R-SELECTED FROM | IMAGE |  | 1 | MC | XOR row 5 | Shall reference the same content item as row 3

### TID 4108 Tracking Identifier

This template provides a means to identify an object for longitudinal tracking, potentially across multiple Structured Reports, over time.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4108. Tracking Identifier

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | TEXT | EV (112039, DCM, "Tracking Identifier") | 1 | MC | At least one of row 1 or 2 shall be present. | A string of characters with case being non-significant. Leading and trailing spaces and control characters are forbidden.
2 |  |  | UIDREF | EV (112040, DCM, "Tracking Unique Identifier") | 1 | MC | At least one of row 1 or 2 shall be present. |

Content Item Descriptions

Tracking Identifier | A human readable identifier for longitudinal tracking, e.g., "Watchlist Nodule 1".
Tracking Unique Identifier | This is distinct from the Observation UID (0040,A171) that may be present in the data set for each content item, which identifies only a specific observation, not an object tracked over time, and each tracked object may have many observations.

## Colon CAD SR IOD Templates

The templates that comprise the Colon CAD SR IOD are interconnected as in Figure A-9b. In Figure A-9b, '···' indicates possible recursive application of subordinate templates.

Figure A-9b. Colon CAD SR IOD Template Structure

### TID 4120 Colon CAD Document Root

This template forms the top of a content tree that allows a colon CAD device to describe the results of detection and analysis of colon evidence. This template, together with its subordinate templates, describes both the results for presentation to radiologists and partial product results for consumption by colon CAD devices in subsequent colon CAD reports.

This template defines a Container that contains the CAD results and summaries of the detection and analysis algorithms performed.

The atomic CAD results of Single Image Findings and Composite Features are described in the Colon CAD Findings Summary sub-tree.

The Summary of Detections and Summary of Analyses sub-trees gather lists of algorithms attempted, grouped by success/failure status. Algorithms not attempted are not mentioned in these sub-trees. This information forms the basis for understanding why a colon CAD report may produce no (or fewer than anticipated) results. Colon CAD results are constructed bottom-up, starting from Single Image Findings (see TID 4127 “Colon CAD Single Image Finding”), associated as Composite Features (see TID 4125 “Colon CAD Composite Feature”).

See Figure SS.1-1 “Top Levels of Colon CAD SR Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  Yes

Table TID 4120. Colon CAD Document Root

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (112220, DCM, "Colon CAD Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | CONTAINS | INCLUDE | DTID 4122 “CAD Common Image Properties Entry” | 1-n | M |  |
4 | > | CONTAINS | INCLUDE | DTID 4121 “Colon CAD Findings Summary” | 1 | M |  |
5 | > | CONTAINS | CODE | EV (111064, DCM, "Summary of Detections") | 1 | M |  | DCID 6042 “Status of Results”
6 | >> | INFERRED FROM | INCLUDE | DTID 4015 “CAD Detections Performed” | 1 | MC | Shall be present unless the value of row 5 is (111225, DCM, "Not Attempted") | $DetectionCode = DCID 6201 “Colon Finding Or Feature”
7 | > | CONTAINS | CODE | EV (111065, DCM, "Summary of Analyses") | 1 | M |  | DCID 6042 “Status of Results”
8 | >> | INFERRED FROM | INCLUDE | DTID 4016 “CAD Analyses Performed” | 1 | MC | Shall be present unless the value of row 7 is (111225, DCM, "Not Attempted") | $AnalysisCode = DCID 6137 “Types of CAD Analysis”

Content Item Descriptions

Detections Performed | The "Detections Performed" and "Analyses Performed" sections of the Content Tree (TID 4120 “Colon CAD Document Root”, rows 6 and 8) together shall reference all Image SOP Instances included in the Current Requested Procedure Evidence Sequence (0040,A375) attribute of the SR Document General module.
Analyses Performed

### TID 4121 Colon CAD Findings Summary

The contents of this template describe the findings and aggregate features that the colon CAD device detected for the colon evidence presented. This template forms the colon CAD results sub-tree of the Colon CAD Document Root (TID 4120 “Colon CAD Document Root”). The data from which the details are inferred are expressed in the Composite Features (see TID 4125 “Colon CAD Composite Feature”) and/or Single Image Findings (see TID 4127 “Colon CAD Single Image Finding”), of which there may be several.

The sub-tree headed by this template is illustrated in Figure F.1-2 “Example of CAD Processing and Findings Summary Sub-Tree of Chest CAD SR Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Table TID 4121. Colon CAD Findings Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111017, DCM, "CAD Processing and Findings Summary") | 1 | M |  | DCID 6047 “CAD Processing and Findings Summary”
2 | > | HAS PROPERTIES | CODE | EV (112222, DCM, "Colon Overall Assessment") | 1 | U |  | DCID 6200 “Colon Overall Assessment”
3 | > | INFERRED FROM | INCLUDE | DTID 4125 “Colon CAD Composite Feature” | 1-n | U |  |
4 | > | INFERRED FROM | INCLUDE | DTID 4127 “Colon CAD Single Image Finding” | 1-n | U |  |

Content Item Descriptions

CAD Processing and Findings Summary | This code value is used to express if and why the Colon CAD Findings Summary sub-tree is empty. The Summary of Detections and Summary of Analyses sub-trees of the Document Root node contain detail about which (if any) algorithms succeeded or failed. If the code value indicates that there were no findings, then the code value can be used to determine whether colon CAD processing occurred successfully, without parsing the Summary of Detections and Summary of Analyses sub-trees.

### TID 4122 CAD Common Image Properties Entry

Each instance of the CAD Common Image Properties Entry template contains selected attributes for a set of parallel contiguous equally spaced slices (with identical properties) from which CAD findings are derived.

Type:

  Extensible

Order:

  Significant

Table TID 4122. CAD Common Image Properties Entry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (112224, DCM, "Image Set Properties") | 1 | M |  |
2 | > | CONTAINS | UIDREF | EV (112227, DCM, "Frame of Reference UID") | 1 | M |  |
3 | > | CONTAINS | UIDREF | EV (110180, DCM, "Study Instance UID") | 1 | M |  |
4 | > | CONTAINS | DATE | EV (111060, DCM, "Study Date") | 1 | M |  | Shall be taken from Study Date (0008,0020) in the Image Instances.
5 | > | CONTAINS | TIME | EV (111061, DCM, "Study Time") | 1 | M |  | Shall be taken from Study Time (0008,0030) in the Image Instances.
6 | > | CONTAINS | CODE | EV (121139, DCM, "Modality") | 1 | M |  | Shall be taken from Modality (0008,0060) in the Image Instances.
7 | > | CONTAINS | NUM | EV (111026, DCM, "Horizontal Pixel Spacing") | 1 | M |  | Shall be taken from value 1 of Pixel Spacing (0028,0030) in the Image Instances. UNITS = EV (mm/{pixel}, UCUM, "millimeters per pixel")
8 | > | CONTAINS | NUM | EV (111066, DCM, "Vertical Pixel Spacing") | 1 | M |  | Shall be taken from value 2 of Pixel Spacing (0028,0030) in the Image Instances. UNITS = EV (mm/{pixel}, UCUM, "millimeters per pixel")
9 | > | CONTAINS | NUM | EV (112225, DCM, "Slice Thickness") | 1 | M |  | Shall be taken from Slice Thickness (0018,0050) in the Image Instances. UNITS = EV (mm, UCUM, "millimeter")
10 | > | CONTAINS | NUM | EV (112226, DCM, "Spacing between slices") | 1 | M |  | Shall be computed from the Image Position (Patient) (0020,0032) projected onto the normal to the Image Orientation (Patient) (0020,0037); may or may not be the same as the Spacing Between Slices (0018,0088) if present. UNITS = EV (mm, UCUM, "millimeter")
11 | > | CONTAINS | CODE | EV (112228, DCM, "Recumbent Patient Position with respect to gravity") | 1 | MC | Required if Patient Position (0018,5100) is present in the image instances and has a value. | Shall be derived from Patient Position (0018,5100) in the Image Instances. DCID 6206 “Recumbent Patient Orientation for Colon”

### TID 4125 Colon CAD Composite Feature

This template collects a composite feature for a lesion, non-lesion object, or correlation of related objects (see TID 4121 “Colon CAD Findings Summary”). The details of the composition are expressed in the Colon CAD Composite Feature Body (see TID 4126 “Colon CAD Composite Feature Body”). The data from which the details are inferred, are expressed in the Composite Features (see TID 4125 “Colon CAD Composite Feature”) and/or Single Image Findings (see TID 4127 “Colon CAD Single Image Finding”), of which there may be several.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4125. Colon CAD Composite Feature

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111015, DCM, "Composite Feature") | 1 | M |  | DCID 6201 “Colon Finding Or Feature”
2 | > | HAS CONCEPT MOD | CODE | EV (112023, DCM, "Composite Feature Modifier") | 1 | U |  | DCID 6202 “Colon Finding Or Feature Modifier”
3 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
4 | >> | HAS PROPERTIES | NUM | EV (111071, DCM, "CAD Operating Point") | 1 | UC | IFF value of row 3 is (111151, DCM, "Presentation Optional") and row 1 of TID 4023 “CAD Operating Points” is present for the feature identified in row 1. | UNITS = DT ({1:n}, UCUM, "range: 1:n"), where n is the maximum specified in Row 1 of TID 4023 “CAD Operating Points” for the feature identified in row 1. Value is restricted to being an integer.
5 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
6 | > | HAS OBS CONTEXT | INCLUDE | DTID 4022 “CAD Observation Context” | 1 | MC | Shall be present IFF this feature is duplicated from a different report than its parent. |
7 | > | HAS OBS CONTEXT | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
8 | > | HAS PROPERTIES | INCLUDE | DTID 4126 “Colon CAD Composite Feature Body” | 1 | M |  |
9 | > | INFERRED FROM | INCLUDE | DTID 4125 “Colon CAD Composite Feature” | 1-n | U |  |
10 | > | INFERRED FROM | INCLUDE | DTID 4127 “Colon CAD Single Image Finding” | 1-n | U |  |

Content Item Descriptions

Rendering Intent | This content item constrains the SCP receiving the Colon CAD SR IOD in its use of the contents of this template and its target content items. Colon CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent colon CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .

### TID 4126 Colon CAD Composite Feature Body

The details of a composite feature are expressed in this template. It is applied to Colon CAD Composite Feature (TID 4125 “Colon CAD Composite Feature”).

Type:

  Non-Extensible

Order:

  Significant

Table TID 4126. Colon CAD Composite Feature Body

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111016, DCM, "Composite type") | 1 | M |  | DCID 6035 “Composite Feature Relations”
2 |  |  | CODE | EV (111057, DCM, "Scope of Feature") | 1 | M |  | DCID 6036 “Scope of Feature”
3 |  |  | NUM | EV (111011, DCM, "Certainty of feature") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
4 |  |  | INCLUDE | DTID 4129 “Colon CAD Geometry” | 1 | U |  |
5 |  |  | INCLUDE | DTID 4128 “Colon CAD Descriptors” | 1 | U |  |
6 |  |  | NUM | DCID 6207 “Colon Quantitative Temporal Difference Type” | 1-n | UC | May be present IFF the value of row 1 is (111153, DCM, "Target content items are related temporally") |
7 | > | R-INFERRED FROM | NUM |  | 2 | U |  | The referenced numeric values shall have the same Concept Name. Their UNITS shall be the same as row 6
8 |  |  | CODE | EV (111049, DCM, "Qualitative Difference") | 1-n | UC | May be present only if the value of row 1 is (111153, DCM, "Target content items are related temporally") | DCID 6134 “Chest Qualitative Temporal Difference Type”
9 | > | HAS PROPERTIES | TEXT | EV (111021, DCM, "Description of Change") | 1 | U |  |
10 | > | R-INFERRED FROM | CODE |  | 2 | M |  | The referenced content items shall have the same Concept Name and their code values shall be from the same context group.

Content Item Descriptions

Certainty of Feature | The CAD device's certainty that the feature analyzed and classified by the CODE, as specified in the Composite Feature parent template is, in fact, that type of feature.
Row 6 | Values ≤ 0 are allowed. The two referenced numeric values are target content items of the first generation Composite Feature or Single Image Finding children of this composite feature. Given the equation, A - B, the value representing A shall be referenced first.
Qualitative Difference | The two referenced code values are target content items of the first generation Composite Feature or Single Image Finding children of this composite feature.

### TID 4127 Colon CAD Single Image Finding

This template describes a single image finding for a lesion or other object. The details of the finding are expressed in this template and/or more specific templates.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4127. Colon CAD Single Image Finding

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111059, DCM, "Single Image Finding") | 1 | M |  | DCID 6201 “Colon Finding Or Feature”
2 | > | HAS CONCEPT MOD | CODE | EV (112024, DCM, "Single Image Finding Modifier") | 1 | U |  | DCID 6202 “Colon Finding Or Feature Modifier”
3 | > | HAS CONCEPT MOD | CODE | EV (111056, DCM, "Rendering Intent") | 1 | M |  | DCID 6034 “Intended Use of CAD Output”
4 | >> | HAS PROPERTIES | NUM | EV (111071, DCM, "CAD Operating Point") | 1 | UC | IFF value of row 3 is (111151, DCM, "Presentation Optional") and row 1 of TID 4023 “CAD Operating Points” is present for the finding identified in row 1 | UNITS = DT ({1:n}, UCUM, "range: 1:n"), where n is the maximum specified in Row 1 of TID 4023 “CAD Operating Points” for the finding identified in row 1. Value is restricted to being an integer.
5 | > | HAS OBS CONTEXT | INCLUDE | DTID 4108 “Tracking Identifier” | 1 | U |  |
6 | > | HAS OBS CONTEXT | INCLUDE | DTID 4022 “CAD Observation Context” | 1 | MC | Shall be present IFF this finding is duplicated from a different report than its parent. |
7 | > | HAS OBS CONTEXT | INCLUDE | DTID 4019 “CAD Algorithm Identification” | 1 | M |  |
8 | > | HAS PROPERTIES | NUM | EV (111012, DCM, "Certainty of Finding") | 1 | U |  | UNITS = EV (%, UCUM, "Percent") Value = 0 - 100
9 | > | HAS PROPERTIES | TEXT | EV (111058, DCM, "Selected Region Description") | 1 | MC | Shall be present IFF value of row 1 is (111099, DCM, "Selected region") |
10 | > | HAS PROPERTIES | INCLUDE | DTID 4129 “Colon CAD Geometry” | 1 | MC | Shall be present unless value of row 1 is (111101, DCM, "Image quality") |
11 | > | HAS PROPERTIES | INCLUDE | DTID 4128 “Colon CAD Descriptors” | 1 | U |  |
12 | > | INFERRED FROM | IMAGE |  | 1 | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") and row 13 is not present |
13 | > | INFERRED FROM | SCOORD | EV (111030, DCM, "Image Region") | 1-n | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") and row 12 is not present |
14 | >> | SELECTED FROM | IMAGE |  | 1 | M |  | All the row 13 content items in a single invocation of this template shall reference the same IMAGE
15 | > | HAS PROPERTIES | INCLUDE | DTID 4014 “CAD Image Quality” | 1 | MC | Shall be present IFF value of row 1 is (111101, DCM, "Image quality") | $QualityFinding = DCID 6135 “Image Quality Finding” $QualityStandard = DCID 6208 “Colon Types of Quality Control Standard”

Content Item Descriptions

Rendering Intent | This content item constrains the SCP receiving the Colon CAD SR IOD in its use of the contents of this template and its target content items. Colon CAD devices may opt to use data marked "Not for Presentation" or "Presentation Optional" as input to subsequent colon CAD processing steps. Refer to Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 .
CAD Operating Point | Additional information to use when Rendering Intent is "Presentational Optional". A CAD Operating Point of zero is not sent, and is encoded as a Rendering Intent of "Presentation Required". See Section O.2 “Structured Reporting Storage SOP Class SCU and SCP Behavior” in PS3.4 and Section E.4 “CAD Operating Point” in PS3.17 .
Certainty of Finding | The certainty of the CAD device that the finding detected and classified by the Single Image Finding CODE specified is in fact that type of finding.

### TID 4128 Colon CAD Descriptors

This template provides qualitative detail for a Single Image Finding or Composite Feature. It is applied to Colon CAD Composite Feature (TID 4125 “Colon CAD Composite Feature”) and Colon CAD Single Image Finding (TID 4127 “Colon CAD Single Image Finding”).

Type:

  Non-Extensible

Order:

  Significant

Table TID 4128. Colon CAD Descriptors

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (G-C504, SRT, "Associated Morphology") | 1-n | U |  | DCID 6209 “Colon Morphology Descriptor”
2 |  |  | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | U |  | DCID 6210 “Location in Intestinal Tract”
3 |  |  | CODE | EV (111014, DCM, "Clockface or region") | 1 | U |  | DCID 6205 “Clockface Location for Colon”
4 |  | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 6212 “Calculated Value for Colon Findings” $Derivation = DCID 6140 “Calculation Methods” $DerivationParameter = EV (112032, DCM, "Threshold Attenuation Coefficient") $DerivationParameterUnits = EV ([hnsf'U], UCUM, "Hounsfield unit")
5 |  |  | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  |
6 |  |  | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  |
7 |  |  | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  |
8 |  |  | INCLUDE | DTID 1406 “Three Dimensional Linear Measurement” | 1-n | U |  |
9 |  |  | NUM | DCID 6141 “Attenuation Coefficient Measurements” | 1-n | U |  | UNITS = EV ([hnsf'U], UCUM, "Hounsfield unit")
10 | > | HAS PROPERTIES | CODE | EV (112009, DCM, "Type of Content") | 1 | U |  | DCID 6211 “Colon CAD Material Description”

Content Item Descriptions

Row 3 | 12 o'clock position is the anterior direction of the patient regardless of the positioning with respect to gravity; clockwise is from the point of view of an observer located closer to the anus than the finding being observed.

### TID 4129 Colon CAD Geometry

All geometry template invocations require specification of either the location of the center of the object, the outline, or both. Geometry is a property of single image findings (see TID 4127 “Colon CAD Single Image Finding”) and composite features (see TID 4125 “Colon CAD Composite Feature”).

Type:

  Non-Extensible

Order:

  Significant

Table TID 4129. Colon CAD Geometry

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | SCOORD | EV (111010, DCM, "Center") | 1 | MC | At least one of rows 1, 3, 4, 6 or 10 shall be present. | GRAPHIC TYPE = {POINT}
2 | > | SELECTED FROM | IMAGE |  | 1 | M |  |
3 |  |  | SCOORD3D | EV (111010, DCM, "Center") | 1 | MC | At least one of rows 1, 3, 4, 6 or 10 shall be present. | GRAPHIC TYPE = {POINT}
4 |  |  | SCOORD | EV (111041, DCM, "Outline") | 1 | MC | At least one of rows 1, 3, 4, 6 or 10 shall be present. |
5 | > | SELECTED FROM | IMAGE |  | 1 | M |  |
6 |  |  | SCOORD3D | EV (111041, DCM, "Outline") | 1 | MC | At least one of rows 1, 3, 4, 6 or 10 shall be present. |
7 |  |  | SCOORD | DCID 6166 “CAD Geometry Secondary Graphical Representation” | 1-n | U |  |
8 | > | SELECTED FROM | IMAGE |  | 1 | M |  |
9 |  |  | SCOORD3D | DCID 6166 “CAD Geometry Secondary Graphical Representation” | 1-n | U |  |
10 |  |  | IMAGE | EV (112229, DCM, "Identifying Segment") | 1 | MC | At least one of rows 1, 3, 4, 6 or 10 shall be present. | Referenced image shall be a Segmentation and the content item shall include Referenced Segment Number (0062,000B)

## Breast Imaging Report Templates

The templates that comprise the Breast Imaging Report are interconnected as in Figure A-10.

Figure A-10. Breast Imaging Report Template Structure

### TID 4200 Breast Imaging Report

This template forms the content tree that allows a Breast Imaging Report device to describe the results of a radiologist's diagnostic interpretation of Breast Imaging (e.g., X-Ray mammography or breast ultrasound) evidence. This template, together with its subordinate templates, describes the results for presentation to clinicians, or for consumption by Breast Imaging Report devices for subsequent Breast Imaging Reports.

This template shall be instantiated at the Root node only.

See Figure Q.1-1 “Top Level of Breast Imaging Report Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Root:

  Yes

Table TID 4200. Breast Imaging Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111400, DCM, "Breast Imaging Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | CONTAINS | INCLUDE | DTID 4202 “Breast Imaging Report Narrative” | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 4208 “Breast Imaging Report Supplementary Data” Breast Imaging Report Supplementary Data | 1 | U |  |

### TID 4201 Breast Imaging Procedure Reported

A procedure that is reported in a Breast Imaging Report is expressed in this template. The results of more than one procedure may be included in a single report instance (see TID 4208 “Breast Imaging Report Supplementary Data”).

See Figure Q.1-2 “Breast Imaging Procedure Reported Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Table TID 4201. Breast Imaging Procedure Reported

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DCID 6050 “Breast Procedure Reported”
2 | > | HAS CONCEPT MOD | CODE | EV (111464, DCM, "Procedure Modifier") | 1-n | U |  | DCID 6058 “Procedure Modifiers for Breast”
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | DCID 6022 “Side”
4 | > | HAS PROPERTIES | CODE | EV (111401, DCM, "Reason for procedure") | 1 | U |  | DCID 6051 “Breast Procedure Reason”
5 | >> | HAS CONCEPT MOD | CODE | EV (G-D709, SRT, "Relative time") | 1 | U |  | DCID 12102 “Temporal Periods Relating to Procedure or Therapy”
6 | >> | HAS CONCEPT MOD | CODE | EV (111402, DCM, "Clinical Finding") | 1-n | UC | IFF row 4 value is "Clinical Finding" | DCID 6055 “Breast Clinical Finding Or Indicated Problem” Breast Clinical Finding or Indicated Problem
7 | >>> | HAS PROPERTIES | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 6022 “Side”
8 | > | HAS PROPERTIES | DATE | EV (111060, DCM, "Study Date") | 1 | U |  |

Content Item Descriptions

Row 5 "Relative time" | This content item indicates whether the value of "Reason for procedure" (row 4) is modified with "pre-" or "follow-up".

### TID 4202 Breast Imaging Report Narrative

This template contains the narrative text sub-tree of the content tree of a Breast Imaging Report. The narrative summary may be subdivided into sections with section headings.

See Figure Q.1-3 “Breast Imaging Report Narrative Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Table TID 4202. Breast Imaging Report Narrative

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111412, DCM, "Narrative Summary") | 1 | M |  |
2 | > | CONTAINS | CONTAINER | BCID 6052 “Breast Imaging Report Section Title” | 1-n | M |  |
3 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
4 | >> | CONTAINS | TEXT | BCID 6053 “Breast Imaging Report Elements” | 1 | M |  |
5 | >>> | INFERRED FROM | INCLUDE | DTID 350 “References to Supporting Evidence” | 1 | U |  |

### TID 4203 Breast Imaging Assessment

This template provides the content of a Breast Imaging Assessment, for an overall assessment section for the entire report (see TID 4208 “Breast Imaging Report Supplementary Data”) or an assessment of a particular finding (see TID 4206 “Breast Imaging Report Finding Section”). This template defines a code-based assessment of the interpretation results.

See Figure Q.1-5 “Breast Imaging Assessment Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Table TID 4203. Breast Imaging Assessment

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (111005, DCM, "Assessment Category") | 1 | M |  | DCID 6026 “Mammography Assessment”
2 |  |  | CODE | EV (111053, DCM, "Recommended Follow-up") | 1-n | U |  | BCID 6028 “Mammography Recommended Follow-up”
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 6022 “Side”
4 | > | HAS PROPERTIES | NUM | EV (111055, DCM, "Recommended Follow-up Interval") | 1 | U |  | UNITS = DCID 6046 “Units of Follow-up Interval” Values = Integer ≥ 0, where 0 = immediate follow-up
5 | > | HAS PROPERTIES | DATE | EV (111054, DCM, "Recommended Follow-up Date") | 1 | U |  |
6 | > | HAS PROPERTIES | INCLUDE | DTID 4207 “Breast Imaging Pathology Results” | 1-n | U |  |

### TID 4204 Breast Imaging Report Intervention Section

This template defines a supplementary data section for an Intervention of the breast, for the Breast Imaging Report. It is included from TID 4208 “Breast Imaging Report Supplementary Data”.

Type:

  Extensible

Order:

  Significant

Table TID 4204. Breast Imaging Report Intervention Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111463, DCM, "Supplementary Data for Intervention") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 4201 “Breast Imaging Procedure Reported” | 1 | M |  |
3 | > | CONTAINS | CODE | EV (A-00110, SRT, "Instrument") | 1 | U |  |
4 | >> | HAS PROPERTIES | TEXT | EV (111465, DCM, "Needle Gauge") | 1 | UC | XOR row 5 |
5 | >> | HAS PROPERTIES | CODE | EV (111465, DCM, "Needle Gauge") | 1 | UC | XOR row 4 |
6 | >> | HAS PROPERTIES | NUM | EV (111467, DCM, "Needle Length") | 1 | U |  | UNITS = EV (cm, UCUM, "centimeter")
7 | > | CONTAINS | NUM | EV (111436, DCM, "Number of passes") | 1 | U |  | UNITS = EV ({passes}, UCUM, "passes")
8 | > | CONTAINS | NUM | EV (111437, DCM, "Number of specimens") | 1 | U |  | UNITS = EV ({specimens}, UCUM, "specimens")
9 | > | CONTAINS | CODE | EV (111431, DCM, "Instrument Approach") | 1-n | U |  | DCID 6065 “Instrument Approach”
10 | > | CONTAINS | CODE | EV (111438, DCM, "Needle in target") | 1 | U |  | DCID 230 “Yes-No”
11 | > | CONTAINS | NUM | EV (111439, DCM, "Number of needles around target") | 1 | U |  | UNITS = EV ({needles}, UCUM, "needles")
12 | > | CONTAINS | CODE | EV (F-04460, SRT, "Medication given") | 1-n | U |  |
13 | > | CONTAINS | CODE | EV (111440, DCM, "Incision made") | 1 | U |  | DCID 230 “Yes-No”
14 | > | CONTAINS | CODE | EV (111123, DCM, "Marker placement") | 1 | U |  | DCID 230 “Yes-No”
15 | > | CONTAINS | CODE | EV (111442, DCM, "Confirmation of target") | 1 | U |  | DCID 6066 “Target Confirmation”
16 | > | CONTAINS | CODE | EV (DD-60002, SRT, "Complication of procedure") | 1-n | U |  | DCID 6062 “Interventional Procedure Complications”
17 | >> | HAS PROPERTIES | CODE | EV (111466, DCM, "Severity of Complication") | 1 | U |  | DCID 251 “Severity of Complication”
18 | > | CONTAINS | CONTAINER | EV (121027, DCM, "Specimen") | 1-n | U |  |
19 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1009 “Subject Context, Specimen” | 1 | U |  |
20 | >> | CONTAINS | CODE | EV (F-00E6D, SRT, "Color of fluid") | 1 | U |  | DCID 6067 “Fluid Color”
21 | >> | CONTAINS | CODE | EV (111456, DCM, "Action on fluid") | 1 | U |  | DT (111457, DCM, "Sent for analysis") DT (111458, DCM, "Discarded")
22 | >> | CONTAINS | CODE | EV (111455, DCM, "Occult blood test result") | 1 | U |  | DCID 250 “Positive-Negative”
23 | >> | CONTAINS | INCLUDE | DTID 4207 “Breast Imaging Pathology Results” | 1-n | U |  |
24 | > | CONTAINS | INCLUDE | DTID 350 “References to Supporting Evidence” | 1 | U |  |

### TID 4205 Breast Composition Section

This template defines a Breast Composition section for the supplementary data sub-tree of the Breast Imaging Report. It is included from TID 4208 “Breast Imaging Report Supplementary Data”.

Type:

  Non-Extensible

Order:

  Significant

Table TID 4205. Breast Composition Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (F-01710, SRT, "Breast composition") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 4201 “Breast Imaging Procedure Reported” | 1-n | U |  |
3 | > | CONTAINS | CODE | EV (F-01710, SRT, "Breast composition") | 1-n | MC | At least one of row 3, 5 shall be present | DCID 6000 “Overall Breast Composition”
4 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | DCID 6022 “Side”
5 | > | CONTAINS | NUM | EV (111046, DCM, "Percent Fibroglandular Tissue") | 1-n | MC | At least one of row 3, 5 shall be present | UNITS = EV (%, UCUM, "Percent")
6 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | DCID 6022 “Side”
7 | > | CONTAINS | CODE | EV (111350, DCM, "Breast background echo texture") | 1-2 | U |  | DCID 6151 “Background Echotexture”
8 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | DCID 6022 “Side”

### TID 4206 Breast Imaging Report Finding Section

This template defines a supplementary data section for the Findings of the Breast Imaging Report. It is included from TID 4208 “Breast Imaging Report Supplementary Data”.

Type:

  Extensible

Order:

  Significant

Table TID 4206. Breast Imaging Report Finding Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
3 | > | CONTAINS | INCLUDE | DTID 4201 “Breast Imaging Procedure Reported” | 1 | M |  |
4 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | M |  | DCID 6054 “Breast Imaging Findings”
5 | >> | HAS CONCEPT MOD | CODE | EV (111405, DCM, "Implant type") | 1-n | UC | May be present if value of row 4 is (A-04010, SRT, "Implant") | DCID 6059 “Breast Implant Types”
6 | >> | HAS PROPERTIES | INCLUDE | DTID 4203 “Breast Imaging Assessment” | 1 | U |  |
7 | >> | HAS PROPERTIES | CODE | EV (111014, DCM, "Clockface or region") | 1 | U |  | DCID 6018 “Clockface Location or Region”
8 | >> | HAS PROPERTIES | CODE | EV (111048, DCM, "Quadrant location") | 1 | U |  | DCID 6020 “Quadrant Location”
9 | >> | HAS PROPERTIES | INCLUDE | DTID 1400 “Linear Measurement” | 1-n | U |  |
10 | >> | HAS PROPERTIES | INCLUDE | DTID 1401 “Area Measurement” | 1-n | U |  |
11 | >> | HAS PROPERTIES | INCLUDE | DTID 1402 “Volume Measurement” | 1-n | U |  |
12 | >> | HAS PROPERTIES | CODE | EV (111020, DCM, "Depth") | 1 | U |  | DCID 6024 “Depth”
13 | >> | HAS PROPERTIES | CODE | EV (111035, DCM, "Lesion Density") | 1 | U |  | DCID 6008 “Density Modifier”
14 | >> | HAS PROPERTIES | CODE | EV (M-020F9, SRT, "Shape") | 1-n | U |  | DCID 6004 “Mammography Characteristics of Shape”
15 | >> | HAS PROPERTIES | CODE | EV (111037, DCM, "Margins") | 1-n | U |  | DCID 6006 “Mammography Characteristics of Margin”
16 | >> | HAS PROPERTIES | CODE | EV (111009, DCM, "Calcification Type") | 1-n | U |  | DCID 6010 “Mammography Calcification Types”
17 | >> | HAS PROPERTIES | CODE | EV (111008, DCM, "Calcification Distribution") | 1 | U |  | DCID 6012 “Calcification Distribution Modifier”
18 | >> | HAS PROPERTIES | NUM | EV (111038, DCM, "Number of calcifications") | 1 | U |  | UNITS = EV ({calcifications}, UCUM, "calcifications") Value = Integer 1 - n
19 | >> | HAS PROPERTIES | CODE | EV (111407, DCM, "Implant finding") | 1-n | U |  | DCID 6072 “Breast Implant Findings”
20 | >> | HAS PROPERTIES | CODE | EV (G-C189, SRT, "Associated Finding") | 1-n | U |  | DCID 6056 “Associated Findings for Breast”
21 | >> | HAS PROPERTIES | NUM | EV (111406, DCM, "Number of similar findings") | 1 | U |  | UNITS = EV ({findings}, UCUM, "findings") Value = Integer 2 - n
22 | >> | HAS PROPERTIES | CODE | EV (F-01720, SRT, "Change since last mammogram") | 1-n | U |  | DCID 6002 “Change Since Last Mammogram Or Prior Surgery”
23 | >> | HAS PROPERTIES | CODE | EV (111354, DCM, "Orientation") | 1 | U |  | DCID 6152 “Orientation”
24 | >> | HAS PROPERTIES | CODE | EV (111357, DCM, "Lesion boundary") | 1 | U |  | DCID 6153 “Lesion Boundary”
25 | >> | HAS PROPERTIES | CODE | EV (111360, DCM, "Echo pattern") | 1 | U |  | DCID 6154 “Echo Pattern”
26 | >> | HAS PROPERTIES | CODE | EV (111366, DCM, "Posterior acoustic features") | 1 | U |  | DCID 6155 “Posterior Acoustic Features”
27 | >> | HAS PROPERTIES | CODE | EV (111371, DCM, "Identifiable effect on surrounding tissues") | 1 | U |  | DCID 6015 “Single Image Finding From BI-RADS®”
28 | >> | HAS PROPERTIES | CODE | EV (111372, DCM, "Vascularity") | 1 | U |  | DCID 6157 “Vascularity”
29 | >> | HAS PROPERTIES | CODE | EV (111380, DCM, "Correlation to Other Findings") | 1-n | U |  | DCID 6158 “Correlation to Other Findings”
30 | >> | INFERRED FROM | INCLUDE | DTID 350 “References to Supporting Evidence” | 1 | U |  |

### TID 4207 Breast Imaging Pathology Results

This template defines the pathology results for a procedure. It may be applied to a Breast Imaging Assessment (see TID 4203 “Breast Imaging Assessment”), or a Breast Imaging Intervention (see TID 4204 “Breast Imaging Report Intervention Section”).

Type:

  Extensible

Order:

  Significant

Table TID 4207. Breast Imaging Pathology Results

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111468, DCM, "Pathology Results") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 4201 “Breast Imaging Procedure Reported” | 1 | U |  |
3 | > | CONTAINS | DATETIME | EV (111469, DCM, "Sampling DateTime") | 1 | M |  |
4 | > | CONTAINS | CODE | EV (122177, DCM, "Procedure Result") | 1 | M |  | DCID 6063 “Interventional Procedure Results”
5 | > | CONTAINS | CODE | EV (111042, DCM, "Pathology") | 1-n | U |  | BCID 6030 “Mammography Pathology Codes”
6 | >> | HAS PROPERTIES | CODE | EV (111388, DCM, "Malignancy Type") | 1 | U |  | DCID 6159 “Malignancy Type”
7 | >> | HAS PROPERTIES | NUM | DCID 6165 “Breast Linear Measurements” | 1-n | U |  | UNITS = EV (mm, UCUM, "millimeter")
8 | >> | HAS PROPERTIES | CODE | EV (F-02900, SRT, "Histological grade finding") | 1 | U |  | BCID 6069 “Nottingham Combined Histologic Grade” BCID 6070 “Bloom-Richardson Histologic Grade”
9 | >>> | HAS CONCEPT MOD | CODE | EV (R-00258, SRT, "Histologic grade") | 1 | U |  | BCID 6071 “Histologic Grading Method”
10 | >> | HAS PROPERTIES | CODE | EV (R-00274, SRT, "Tumor margin status") | 1 | U |  | DT (111470, DCM, "Uninvolved"), DT (111471, DCM, "Involved")
11 | >> | HAS PROPERTIES | CODE | EV (111472, DCM, "Nipple involved") | 1 | U |  | DCID 230 “Yes-No”
12 | >> | HAS PROPERTIES | NUM | EV (111473, DCM, "Number of nodes removed") | 1 | U |  | UNITS = EV ({nodes}, UCUM, "nodes")
13 | >> | HAS PROPERTIES | NUM | EV (111474, DCM, "Number of nodes positive") | 1 | MC | Shall be present IFF value of row 12 is > 0 | UNITS = EV ({nodes}, UCUM, "nodes")
14 | >> | HAS PROPERTIES | CODE | EV (R-00465, SRT, "pT category finding") | 1 | U |  | DCID 6160 “Breast Primary Tumor Assessment From AJCC”
15 | >> | HAS PROPERTIES | CODE | EV (R-00463, SRT, "Node stage finding") | 1 | U |  | DCID 6161 “Clinical Regional Lymph Node Assessment for Breast”
16 | >> | HAS PROPERTIES | CODE | EV (R-00461, SRT, "Metastasis stage finding") | 1 | U |  | DCID 6162 “Assessment of Metastasis for Breast”
17 | >> | HAS PROPERTIES | CODE | EV (R-00443, SRT, "Tumor stage finding") | 1 | U |  | BCID 6068 “Tumor Stages From AJCC”
18 | >> | HAS PROPERTIES | CODE | EV (111475, DCM, "Estrogen receptor") | 1 | U |  | DCID 250 “Positive-Negative”
19 | >> | HAS PROPERTIES | CODE | EV (111476, DCM, "Progesterone receptor") | 1 | U |  | DCID 250 “Positive-Negative”
20 | >> | HAS PROPERTIES | NUM | EV (111477, DCM, "S Phase") | 1 | U |  | UNITS = EV (%, UCUM, "percent")

### TID 4208 Breast Imaging Report Supplementary Data

This template forms a supplementary data sub-tree of the content tree of a Breast Imaging Report. Each subsection provides a specific type of supporting evidence to the narrative text sub-tree, for example, as coded and numeric data.

See Figure Q.1-4 “Breast Imaging Report Supplementary Data Content Tree” in PS3.17 .

Type:

  Non-Extensible

Order:

  Significant

Table TID 4208. Breast Imaging Report Supplementary Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111414, DCM, "Supplementary Data") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 4201 “Breast Imaging Procedure Reported” | 1-n | M |  |
3 | > | CONTAINS | CODE | EV (111403, DCM, "Baseline screening mammogram") | 1 | U |  | DCID 230 “Yes-No”
4 | > | CONTAINS | CODE | EV (111404, DCM, "First mammogram ever") | 1 | U |  | DCID 230 “Yes-No”
5 | > | CONTAINS | INCLUDE | DTID 4205 “Breast Composition Section” | 1 | U |  |
6 | > | CONTAINS | INCLUDE | DTID 4206 “Breast Imaging Report Finding Section” | 1-n | U |  |
7 | > | CONTAINS | INCLUDE | DTID 4204 “Breast Imaging Report Intervention Section” | 1-n | U |  |
8 | > | CONTAINS | CONTAINER | EV (111413, DCM, "Overall Assessment") | 1 | U |  |
9 | >> | CONTAINS | INCLUDE | DTID 4203 “Breast Imaging Assessment” | 1 | M |  |

## OB-GYN Report Templates

### TID 5000 OB-GYN Ultrasound Procedure Report

This is the template for the root of the content tree for the OB-GYN ultrasound procedure report.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 5000. OB-GYN Ultrasound Procedure Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125000, DCM, "OB-GYN Ultrasound Procedure Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 5001 “OB-GYN Patient Characteristics” | 1 | U |  |
5 | > | CONTAINS | CONTAINER | DT (111028, DCM, "Image Library") | 1 | U |  |
6 | >> | CONTAINS | IMAGE | No purpose of reference | 1-n | M |  |
7 | > | CONTAINS | INCLUDE | DTID 5002 “OB-GYN Procedure Summary Section” | 1 | U |  |
8 | > | CONTAINS | INCLUDE | DTID 5004 “Fetal Biometry Ratio Section” | 1-n | U |  |
9 | > | CONTAINS | INCLUDE | DTID 5005 “Fetal Biometry Section” | 1-n | U |  |
10 | > | CONTAINS | INCLUDE | DTID 5006 “Fetal Long Bones Section” | 1-n | U |  |
11 | > | CONTAINS | INCLUDE | DTID 5007 “Fetal Cranium Section” | 1-n | U |  |
12 | > | CONTAINS | INCLUDE | DTID 5009 “Fetal Biophysical Profile Section” | 1-n | U |  |
13 | > | CONTAINS | INCLUDE | DTID 5011 “Early Gestation Section” | 1-n | U |  |
14 | > | CONTAINS | INCLUDE | DTID 5010 “Amniotic Sac Section” | 1 | U |  |
15 | > | CONTAINS | INCLUDE | DTID 5015 “Pelvis and Uterus Section” | 1 | U |  |
16 | > | CONTAINS | INCLUDE | DTID 5012 “Ovaries Section” | 1 | U |  |
17 | > | CONTAINS | INCLUDE | DTID 5013 “Follicles Section” | 1 | U |  | $Laterality = EV (G-A101, SRT, "Left") $Number = EV (11879-4, LN, "Number of follicles in left ovary")
18 | > | CONTAINS | INCLUDE | DTID 5013 “Follicles Section” | 1 | U |  | $Laterality = EV (G-A100, SRT, "Right") $Number = EV (11880-2, LN, "Number of follicles in right ovary")
19 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | U |  |
20 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | EV (T-F6800, SRT, "Embryonic Vascular Structure")
21 | >> | CONTAINS | INCLUDE | DTID 5025 “OB-GYN Fetal Vascular Ultrasound Measurement Group” | 1 | M |  | $AnatomyGroup = DCID 12141 “Fetal Vasculature Anatomical Location”
22 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1-n | U |  |
23 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | EV (T-D6007, SRT, "Pelvic Vascular Structure")
24 | >> | CONTAINS | INCLUDE | DTID 5026 “OB-GYN Pelvic Vascular Ultrasound Measurement Group” | 1 | M |  | $AnatomyGroup = DCID 12140 “Pelvic Vasculature Anatomical Location”

### TID 5001 OB-GYN Patient Characteristics

Patient Characteristic concepts in this template, which may replicate attributes in the Patient Study Module, are included here as possible targets of by-reference relationships from other content items in the SR tree.

### Note

Several of the concepts in this template duplicate concepts in TID 1007 “Subject Context, Patient”. The difference in use is that this template has those concepts as primary observations of the patient, while in TID 1007 “Subject Context, Patient” the concepts are used to set (or reset) the context for other observations.

Type:

  Extensible

Order:

  Significant

Table TID 5001. OB-GYN Patient Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121118, DCM, "Patient Characteristics") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
3 | > | CONTAINS | NUM | EV (8302-2, LN, "Patient Height") | 1 | U |  |
4 | > | CONTAINS | NUM | EV (29463-7, LN, "Patient Weight") | 1 | U |  |
5 | > | CONTAINS | NUM | EV (11996-6, LN, "Gravida") | 1 | U |  |
6 | > | CONTAINS | NUM | EV (11977-6, LN, "Para") | 1 | U |  |
7 | > | CONTAINS | NUM | EV (11612-9, LN, "Aborta") | 1 | U |  |
8 | > | CONTAINS | NUM | EV (33065-4, LN, "Ectopic Pregnancies") | 1 | U |  |

### TID 5002 OB-GYN Procedure Summary Section

Observations of the procedure of immediate clinical interest.

Type:

  Extensible

Order:

  Significant

Table TID 5002. OB-GYN Procedure Summary Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | DATE | DCID 12003 “OB-GYN Dates” | 1-n | U |  |
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = BCID 12018 “OB-GYN Summary”
4 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1-n | U |  |
5 | >> |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1-n | U |  |
6 | > | CONTAINS | INCLUDE | BTID 5003 “OB-GYN Fetus Summary” | 1-n | UC | No more than 1 inclusion per fetus |

### TID 5003 OB-GYN Fetus Summary

The Fetus Summary template is a container for summary data of a fetus.

Type:

  Extensible

Order:

  Significant

Table TID 5003. OB-GYN Procedure Fetus Summary

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125008, DCM, "Fetus Summary") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1-n | U |  |
4 | >> |  | INCLUDE | DTID 320 “Image or Spatial Coordinates” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 12019 “OB-GYN Fetus Summary” $Equation = DCID 12012 “OB Equations and Tables”

### TID 5004 Fetal Biometry Ratio Section

The Fetal Biometry Section Ratio template is a container for common biometric ratios.

Type:

  Extensible

Order:

  Significant

Table TID 5004. Fetal Biometry Ratio Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125001, DCM, "Fetal Biometry Ratios") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | NUM | DCID 12004 “Fetal Biometry Ratios” | 1-n | M |  |
4 | >> | R-INFERRED FROM | NUM |  | 2 | U |  |
5 | >> | HAS PROPERTIES | INCLUDE | DTID 312 “Normal Range Properties” | 1 | U |  |

Content Item Descriptions

Row 3 | Numeric ratio related to fetal growth
Row 4 | Reference to the numerator and denominator of the ratio.

### TID 5005 Fetal Biometry Section

The Fetal Biometry Section template is a container for common biometric groups.

Type:

  Extensible

Order:

  Significant

Table TID 5005. Fetal Biometry Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125002, DCM, "Fetal Biometry") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | INCLUDE | DTID 5008 “Fetal Biometry Group” | 1-n | M |  | $BiometryType = MemberOf {DCID 12005 “Fetal Biometry Measurements” $TargetSite = DCID 12020 “Fetal Biometry Anatomic Sites”

Content Item Descriptions

Row 3 | The group of measurements. Only one group per biometry type.

### TID 5006 Fetal Long Bones Section

The Long Bones template is a container for biometric data of long bones.

Type:

  Extensible

Order:

  Significant

Table TID 5006. Fetal Long Bones Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125003, DCM, "Fetal Long Bones") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | INCLUDE | DTID 5008 “Fetal Biometry Group” | 1-n | M |  | $BiometryType = MemberOf {DCID 12006 “Fetal Long Bones Biometry Measurements” $TargetSite = DCID 12021 “Fetal Long Bone Anatomic Sites”

Content Item Descriptions

Row 3 | The group of measurements. Only one group per biometry type.

### TID 5007 Fetal Cranium Section

The Fetal Cranium template is a container for groups of biometric data of the fetal cranium.

Type:

  Extensible

Order:

  Significant

Table TID 5007. Fetal Cranium Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125004, DCM, "Fetal Cranium") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | INCLUDE | DTID 5008 “Fetal Biometry Group” | 1-n | M |  | $BiometryType = MemberOf {DCID 12007 “Fetal Cranium” $TargetSite = DCID 12022 “Fetal Cranium Bone Anatomic Sites”

Content Item Descriptions

Row 3 | The group of measurements. Only one group per biometry type.

### TID 5008 Fetal Biometry Group

The Biometry Group template is container for a biometric value and its associated growth metrics.

Table TID 5008. Parameters

Parameter Name | Parameter Usage
--- | ---
$BiometryType | The concept name of the biometry measurement
$TargetSite | Value for Anatomic Location of the biometry measurement

Type:

  Extensible

Order:

  Significant

Table TID 5008. Fetal Biometry Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125005, DCM, "Biometry Group") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | MC | At least one of row 2 and 3 shall be present | $Measurement = $BiometryType $TargetSite = $TargetSite $Derivation = DCID 3627 “Measurement Type”
3 | > | CONTAINS | NUM | EV (18185-9, LN, "Gestational Age") | 1 | MC | At least one of row 2 and 3 shall be present | UNITS = EV (d, UCUM, "days")
4 | >> | INFERRED FROM | CODE | DCID 228 “Equation or Table” | 1 | U |  | DCID 12013 “Gestational Age Equations and Tables”
5 | >> | R-INFERRED FROM | NUM |  | 1-n | U |  |
6 | >> | HAS PROPERTIES | NUM | DCID 226 “Population Statistical Descriptors” | 1-n | U |  |
7 | > | CONTAINS | NUM | DCID 12017 “Growth Distribution Rank” | 1 | U |  |
8 | >> | INFERRED FROM | CODE | DCID 228 “Equation or Table” | 1 | U |  | DCID 12015 “Fetal Growth Equations and Tables”

Content Item Descriptions

Row 1 | Container to segregate biometry data by measurement type
Row 2 | The discrete measurements of the biometry type including derived measurements such as mean. One of the measurements may be flagged as selected for derived measurements. The anatomic location may be precoordinated in the measurement type, but may also be explicitly conveyed in the $TargetSite parameter, which then also allows laterality to be encoded within TID 300 “Measurement”.
Row 3 | The estimated gestational age derived from an equation or table based on the explicitly referenced R-INFERRED FROM content item, selected measurement or mean, in that order of preference.
Row 4 | The reference that defines the equation or table of GA derivation
Row 6 | The uncertainty/confidence limits of the gestational age
Row 7 | Expresses the rank of the selected or mean measurement of row 2 relative to the distribution specified in row 8.
Row 8 | This row specifies the CODE reference used to compute the percentile or Z-score.

### TID 5009 Fetal Biophysical Profile Section

This template encodes scoring observations for fetal well-being evaluation as described by Manning, Antepartum Fetal Evaluation: Development of a Fetal Biophysical Profile Score, Am. J Obsetet Gynecol, 1980;136:787.

Type:

  Extensible

Order:

  Significant

Table TID 5009. Fetal Biophysical Profile Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125006, DCM, "Biophysical Profile") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | NUM | EV (11631-9, LN, "Gross Body Movement") | 1 | MC | At least one of row 3-7 shall be present | UNITS = DT ({0:2}, UCUM, "range 0:2")
4 | > | CONTAINS | NUM | EV (11632-7, LN, "Fetal Breathing") | 1 | MC | At least one of row 3-7 shall be present | UNITS = DT ({0:2}, UCUM, "range 0:2")
5 | > | CONTAINS | NUM | EV (11635-0, LN, "Fetal Tone") | 1 | MC | At least one of row 3-7 shall be present | UNITS = DT ({0:2}, UCUM, "range 0:2")
6 | > | CONTAINS | NUM | EV (11635-5, LN, "Fetal Heart Reactivity") | 1 | MC | At least one of row 3-7 shall be present | UNITS = DT ({0:2}, UCUM, "range 0:2")
7 | > | CONTAINS | NUM | EV (11630-1, LN, "Amniotic Fluid Volume") | 1 | MC | At least one of row 3-7 shall be present | UNITS = DT ({0:2}, UCUM, "range 0:2")
8 | > | CONTAINS | NUM | DT (11634-3, LN, "Biophysical Profile Sum Score") | 1 | U |  |

Content Item Descriptions

Row 3-7 | The numeric profile score of range 0-2
Row 8 | The sum of rows 3-7. The range is from 0 to the maximum possible score according the items scored in rows 3-7.

### TID 5010 Amniotic Sac Section

This template specifies a container for amniotic sac quadrant diameters and a derived index.

Type:

  Extensible

Order:

  Significant

Table TID 5010. Amniotic Sac Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (T-F1300, SRT, "Amniotic Sac")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = DT (11627-7, LN, "Amniotic Fluid Index")
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 4 | U |  | $Measurement = DCID 12008 “OB-GYN Amniotic Sac”

Content Item Descriptions

Row 3 | The sum of the 4 quadrant diameters
Row 4 | The four amniotic sac quadrant diameters

### TID 5011 Early Gestation Section

The Early Gestation Section template is a container for common, first trimester biometric groups.

Type:

  Extensible

Order:

  Significant

Table TID 5011. Early Gestation Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125009, DCM, "Early Gestation") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | INCLUDE | DTID 5008 “Fetal Biometry Group” | 1-n | M |  | $BiometryType = Member of {DCID 12009 “Early Gestation Biometry Measurements”}

### TID 5012 Ovaries Section

This template contains metrics of ovary size.

Type:

  Extensible

Order:

  Significant

Table TID 5012. Ovaries Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (T-87000, SRT, "Ovary")
3 | > | CONTAINS | INCLUDE | DTID 5016 “LWH Volume Group” | 1 | U |  | $GroupName = EV (T-87000, SRT, "Ovary") $Width = EV (11829-9, LN, "Left Ovary Width") $Length = EV (11840-6, LN, "Left Ovary Length") $Height = EV (11857-0, LN, "Left Ovary Height") $Volume = EV (12164-0, LN, "Left Ovary Volume")
4 | > | CONTAINS | INCLUDE | DTID 5016 “LWH Volume Group” | 1 | U |  | $GroupName = EV (T-87000, SRT, "Ovary") $Width = EV (11830-7, LN, "Right Ovary Width") $Length = EV (11841-4, LN, "Right Ovary Length") $Height = EV (11858-8, LN, "Right Ovary Height") $Volume = EV (12165-7, LN, "Right Ovary Volume")

### TID 5013 Follicles Section

This template contains follicle metrics for left or right ovarian follicles.

Table TID 5013. Parameters

Parameter Name | Parameter Usage
--- | ---
$Laterality | Ovary laterality
$Number | The number of follicles

Type:

  Extensible

Order:

  Significant

Table TID 5013. Follicles Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (T-87600, SRT, "Ovarian Follicle")
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | $Laterality
4 | > | CONTAINS | NUM | $Number | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 5014 “Follicle Measurement Group” | 1-n | U |  |

### TID 5014 Follicle Measurement Group

This template contains metrics for one ovarian follicle.

Type:

  Extensible

Order:

  Significant

Table TID 5014. Follicle Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125007, DCM, "Measurement Group") | 1 | M |  |
2 | > | HAS OBS CONTEXT | TEXT | EV (125010, DCM, "Identifier") | 1 | U |  | Unique among all groups of same laterality
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (G-D705, SRT, "Volume")
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (11793-7, LN, "Follicle Diameter") $Derivation = DCID 3627 “Measurement Type”

### TID 5015 Pelvis and Uterus Section

This template contains general measurements in the pelvis and uterus.

Type:

  Extensible

Order:

  Significant

Table TID 5015. Pelvis and Uterus Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125011, DCM, "Pelvis and Uterus") | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 5016 “LWH Volume Group” | 1 | U |  | $GroupName = EV (T-83000, SRT, "Uterus") $Width = EV (11865-3, LN, "Uterus Width") $Length = EV (11842-2, LN, "Uterus Length") $Height = EV (11859-6, LN, "Uterus Height") $Volume = EV (33192-6, LN, "Uterus Volume")
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 12011 “Ultrasound Pelvis and Uterus” $TargetSite = DCID 12023 “Pelvis and Uterus Anatomic Sites” $Derivation = DCID 3627 “Measurement Type”

### TID 5016 LWH Volume Group

This template is a container for a group of measurements that assess the size of an anatomical structure using a volume derived from perpendicular diameters.

Table TID 5016. Parameters

Parameter Name | Parameter Usage
--- | ---
$GroupName | The name of the volume group that is an anatomical structure
$Volume | Concept name of volume measurement
$Length | Concept name of length measurement
$Width | Concept name of width measurement
$Height | Concept name of height measurement

Type:

  Extensible

Order:

  Significant

Table TID 5016. LWH Volume Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | $GroupName | 1 | M |  |
2 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | MC | At least one of row 2, 3, 4, 5 shall be present | $Measurement = $Volume $TargetSite = $GroupName
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | MC | At least one of row 2, 3, 4, 5 shall be present | $Measurement = $Length $TargetSite = $GroupName $Derivation = DCID 3627 “Measurement Type”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | MC | At least one of row 2, 3, 4, 5 shall be present | $Measurement = $Width $TargetSite = $GroupName $Derivation = DCID 3627 “Measurement Type”
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | MC | At least one of row 2, 3, 4, 5 shall be present | $Measurement = $Height $TargetSite = $GroupName $Derivation = DCID 3627 “Measurement Type”

### TID 5025 OB-GYN Fetal Vascular Ultrasound Measurement Group

This template is an anatomy specific container of OB-GYN fetal vascular measurements.

Table TID 5025. Parameters

Parameter Name | Parameter Usage
--- | ---
$AnatomyGroup | The concept name of the vascular anatomy

Type:

  Extensible

Order:

  Significant

Table TID 5025. OB-GYN Fetal Vascular Ultrasound Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | $AnatomyGroup | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this template is invoked more than once to describe more than one fetus |
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IFF anatomy has laterality | DCID 244 “Laterality”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $MeasType = DCID 12119 “Vascular Ultrasound Property” $Derivation = DCID 3627 “Measurement Type”

Content Item Descriptions

Anatomy Group | Specifies the anatomical context of the observations in the group.

### TID 5026 OB-GYN Pelvic Vascular Ultrasound Measurement Group

This template is an anatomy specific container of OB-GYN pelvic vascular measurements.

Table TID 5026. Parameters

Parameter Name | Parameter Usage
--- | ---
$AnatomyGroup | The concept name of the vascular anatomy

Type:

  Extensible

Order:

  Significant

Table TID 5026. OB-GYN Pelvic Vascular Ultrasound Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | $AnatomyGroup | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IFF anatomy has laterality | DCID 244 “Laterality”
3 | > | HAS CONCEPT MOD | TEXT | EV (112050, DCM, "Anatomic Identifier") | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $MeasType = DCID 12119 “Vascular Ultrasound Property” $Derivation = DCID 3627 “Measurement Type”
5 | >> | HAS CONCEPT MOD | CODE | EV (125105, DCM, "Measurement Orientation") | 1 | U |  | DCID 12118 “Measurement Orientation”
6 | >> | HAS PROPERTIES | NUM | EV (125106, DCM, "Doppler Angle") | 1 | U |  | UNIT = EV (deg, UCUM, "deg")
7 | >> | HAS PROPERTIES | NUM | EV (125107, DCM, "Sample Volume Depth") | 1 | U |  | UNIT = EV (cm, UCUM, "cm")

Content Item Descriptions

Row 1 | Specifies the anatomical context of the observations in the group.
Row 3 | Differentiates between multiple structures such as the two umbilical arteries.

## Vascular Ultrasound Report Templates

### TID 5100 Vascular Ultrasound Report

This is the template for the root the content tree for the vascular ultrasound procedure report.

Type:

  Extensible

Order:

  Significant

Table TID 5100. Vascular Ultrasound Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125100, DCM, "Vascular Ultrasound Procedure Report") | 1 | M |  |
2 | > | HAS OBS CONTEXT | CODE | EV (R-40FB8, SRT, "Temporal periods Relating to Procedure") | 1 | U |  | DCID 12102 “Temporal Periods Relating to Procedure or Therapy”
3 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
4 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
5 | > | CONTAINS | INCLUDE | DTID 5101 “Vascular Patient Characteristics” | 1 | U |  |
6 | > | CONTAINS | CONTAINER | EV (111028, DCM, "Image Library") | 1 | U |  |
7 | >> | CONTAINS | IMAGE | No purpose of reference | 1-n | M |  |
8 | > | CONTAINS | INCLUDE | DTID 5102 “Vascular Procedure Summary Section” | 1 | U |  |
9 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-40501, SRT, "Blood Vessel of Head") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12105 “Intracranial Cerebral Vessels”
10 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-40501, SRT, "Blood Vessel of Head") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12105 “Intracranial Cerebral Vessels”
11 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-40501, SRT, "Blood Vessel of Head") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 12106 “Intracranial Cerebral Vessels (unilateral)”
12 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-45005, SRT, "Artery of neck") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12104 “Extracranial Arteries” $AnatomyRatio = DCID 12123 “Carotid Ratios”
13 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-45005, SRT, "Artery of neck") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12104 “Extracranial Arteries” $AnatomyRatio = DCID 12123 “Carotid Ratios”
14 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-47040, SRT, "Artery of Lower Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12109 “Lower Extremity Arteries”
15 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-47040, SRT, "Artery of Lower Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12109 “Lower Extremity Arteries”
16 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-49403, SRT, "Vein of Lower Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12110 “Lower Extremity Veins”
17 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-49403, SRT, "Vein of Lower Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12110 “Lower Extremity Veins”
18 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-47020, SRT, "Artery Of Upper Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12107 “Upper Extremity Arteries”
19 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-47020, SRT, "Artery Of Upper Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12107 “Upper Extremity Arteries”
20 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-49103, SRT, "Vein Of Upper Extremity") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12108 “Upper Extremity Veins”
21 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-49103, SRT, "Vein Of Upper Extremity") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12108 “Upper Extremity Veins”
22 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-71019, SRT, "Vascular Structure Of Kidney") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12115 “Renal Vessels” $AnatomyRatio = DCID 12124 “Renal Ratios”
23 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-71019, SRT, "Vascular Structure Of Kidney") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12115 “Renal Vessels” $AnatomyRatio = DCID 12124 “Renal Ratios”
24 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-46002, SRT, "Artery of Abdomen") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12111 “Abdominal Arteries (lateral)”
25 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-46002, SRT, "Artery of Abdomen") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12111 “Abdominal Arteries (lateral)”
26 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-46002, SRT, "Artery of Abdomen") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 12112 “Abdominal Arteries (unilateral)”
27 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-487A0, SRT, "Vein of Abdomen") $SectionLaterality = EV (G-A101, SRT, "Left") $Anatomy = DCID 12113 “Abdominal Veins (lateral)”
28 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-487A0, SRT, "Vein of Abdomen") $SectionLaterality = EV (G-A100, SRT, "Right") $Anatomy = DCID 12113 “Abdominal Veins (lateral)”
29 | > | CONTAINS | INCLUDE | DTID 5103 “Vascular Ultrasound Section” | 1 | U |  | $SectionScope = DT (T-487A0, SRT, "Vein of Abdomen") $SectionLaterality = EV (G-A103, SRT, "Unilateral") $Anatomy = DCID 12114 “Abdominal Veins (unilateral)”
30 | > | CONTAINS | INCLUDE | DTID 5105 “Ultrasound Graft Section” | 1 | U |  |

### TID 5101 Vascular Patient Characteristics

Patient Characteristic concepts in this template, which may replicate attributes in the Patient Study Module, are included here as possible targets of by-reference relationships from other content items in the SR tree.

### Note

Several of the concepts in this template duplicate concepts in TID 1007 “Subject Context, Patient”. The difference in use is that this template has those concepts as primary observations of the patient, while in TID 1007 “Subject Context, Patient” the concepts are used to set (or reset) the context for other observations.

Type:

  Extensible

Order:

  Significant

Table TID 5101. Vascular Patient Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121118, DCM, "Patient Characteristics") | 1 | M |  |
2 | > | CONTAINS | NUM | EV (121033, DCM, "Subject Age") | 1 | U |  | UNITS = DCID 7456 “Units of Measure for Age”
3 | > | CONTAINS | CODE | EV (121032, DCM, "Subject Sex") | 1 | U |  | DCID 7455 “Sex”
4 | > | CONTAINS | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  |
5 | > | CONTAINS | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | U |  |
6 | > | CONTAINS | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | U |  |

### TID 5102 Vascular Procedure Summary Section

Comments and observations of the procedure of immediate clinical interest.

Type:

  Extensible

Order:

  Significant

Table TID 5102. Vascular Procedure Summary Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | TEXT | DCID 12101 “Vascular Summary” | 1-n | M |  |

### TID 5103 Vascular Ultrasound Section

Sections of a vascular ultrasound report are section containers of an anatomic region consisting of measurement group containers that contain the measurements.

Table TID 5103. Parameters

Parameter Name | Parameter Usage
--- | ---
$SectionScope | The concept name of the section heading modifier
$SectionLaterality | The laterality (if any) of the anatomy in this section heading
$Anatomy | The concept name of the vascular anatomy
$AnatomyRatio | The concept name of anatomy-coordinated ratio concepts

Type:

  Extensible

Order:

  Significant

Table TID 5103. Vascular Ultrasound Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | $SectionScope
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | $SectionLaterality
4 | > | CONTAINS | INCLUDE | DTID 5104 “Vascular Ultrasound Measurement Group” | 1-n | M |  | $AnatomyGroup = $Anatomy
5 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = $AnatomyRatio

### TID 5104 Vascular Ultrasound Measurement Group

This template is an anatomy specific container of measurements.

Table TID 5104. Parameters

Parameter Name | Parameter Usage
--- | ---
$AnatomyGroup | The concept name of the vascular anatomy

Type:

  Extensible

Order:

  Significant

Table TID 5104. Vascular Ultrasound Measurement Group

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | $AnatomyGroup | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-A1F8, SRT, "Topographical Modifier") | 1 | U |  | DCID 12116 “Vessel Segment Modifiers”
3 | > | HAS CONCEPT MOD | CODE | EV (125101, DCM, "Vessel Branch") | 1-n | U |  | DCID 12117 “Vessel Branch Modifiers”
4 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = DCID 12119 “Vascular Ultrasound Property” $Derivation = DCID 3627 “Measurement Type”
5 | >> | HAS CONCEPT MOD | CODE | EV (R-4089A, SRT, "Cardiac Cycle Point") | 1 | U |  | DCID 12233 “Cardiac Phase”
6 | >> | HAS CONCEPT MOD | CODE | EV (R-41FFC, SRT, "Temporal period related to eating") | 1 | U |  | DT (G-A491, SRT, "Post-prandial")
7 | >> | HAS CONCEPT MOD | CODE | EV (125105, DCM, "Measurement Orientation") | 1 | U |  | DCID 12118 “Measurement Orientation”
8 | >> | HAS PROPERTIES | NUM | EV (125106, DCM, "Doppler Angle") | 1 | U |  | UNIT = EV (deg, UCUM, "degrees")
9 | >> | HAS PROPERTIES | NUM | EV (125107, DCM, "Sample Volume Depth") | 1 | U |  | UNIT = EV (cm, UCUM, "cm")

Content Item Descriptions

Row 1 | Specifies the anatomic context of the observations in the group.
Row 2 | Details the anatomical location, e.g., proximal, middle, or distal
Row 3 | The particular vessel branch, such as the inferior, medial or lateral
Row 5 | Cardiac phase (systolic, diastolic), especially for aorta measurements
Row 6 | Eating phase, especially for mesenteric and celiac measurements

### TID 5105 Ultrasound Graft Section

This template is a container of measurements on a vascular graft.

Type:

  Extensible

Order:

  Significant

Table TID 5105. Ultrasound Graft Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (T-D000F, SRT, "Vascular Graft")
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | DCID 244 “Laterality”
4 | > | HAS CONCEPT MOD | CODE | DT (G-D871, SRT, "Proximal anastomosis") | 1 | M |  | BCID 12103 “Vascular Ultrasound Anatomic Location”
5 | > | HAS CONCEPT MOD | CODE | DT (G-D872, SRT, "Distal Anastomosis") | 1 | M |  | BCID 12103 “Vascular Ultrasound Anatomic Location”
6 | > | HAS CONCEPT MOD | CODE | DT (125102, DCM, "Graft Type") | 1 | U |  | No BCID
7 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | M |  | $Measurement = DCID 12119 “Vascular Ultrasound Property”

Content Item Descriptions

Proximal anastomosis | The proximal location of the graft
Distal anastomosis | The distal location of the graft
Graft type | The type of graft, e.g., "in situ", "prosthetic", "autogenous"

## Echocardiography Procedure Report Templates

### TID 5200 Echocardiography Procedure Report

This template forms the top of a content tree that allows an ultrasound device to describe the results of an adult echocardiography imaging procedure. It is instantiated at the root node. It can also be included in other templates that need to incorporate echocardiography findings into another report as quoted evidence.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 5200. Echocardiography Procedure Report

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125200, DCM, "Adult Echocardiography Procedure Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
4 | > | CONTAINS | CONTAINER | DT (55111-9, LN, "Current Procedure Descriptions") | 1 | U |  |
5 | >> | CONTAINS | CODE | DT (125203, DCM, "Acquisition Protocol") | 1-n | M |  | BCID 12001 “Ultrasound Protocol Types”
6 | > | CONTAINS | INCLUDE | DTID 5201 “Echocardiography Patient Characteristics” | 1 | U |  |
7 | > | CONTAINS | CONTAINER | EV (111028, DCM, "Image Library") | 1 | U |  |
8 | >> | CONTAINS | IMAGE | No purpose of reference | 1-n | M |  |
9 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-32600, SRT, "Left Ventricle") $MeasType = DCID 12200 “Echocardiography Left Ventricle”
10 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-32500, SRT, "Right Ventricle") $MeasType = DCID 12204 “Echocardiography Right Ventricle”
11 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-32300, SRT, "Left Atrium") $MeasType = DCID 12205 “Echocardiography Left Atrium”
12 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-32200, SRT, "Right Atrium") $MeasType = DCID 12206 “Echocardiography Right Atrium”
13 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-35400, SRT, "Aortic Valve") $MeasType = DCID 12211 “Echocardiography Aortic Valve”
14 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-35300, SRT, "Mitral Valve") $MeasType = DCID 12207 “Echocardiography Mitral Valve”
15 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-35200, SRT, "Pulmonic Valve") $MeasType = DCID 12209 “Echocardiography Pulmonic Valve”
16 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-35100, SRT, "Tricuspid Valve") $MeasType = DCID 12208 “Echocardiography Tricuspid Valve”
17 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-42000, SRT, "Aorta") $MeasType = DCID 12212 “Echocardiography Aorta”
18 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-44000, SRT, "Pulmonary artery") $MeasType = DCID 12210 “Echocardiography Pulmonary Artery”
19 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-48600, SRT, "Vena Cava") $MeasType = DCID 12215 “Echocardiography Vena Cavae”
20 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-48581, SRT, "Pulmonary Venous Structure") $MeasType = DCID 12214 “Echocardiography Pulmonary Veins”
21 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (T-39050, SRT, "Pericardial cavity") $MeasType = DCID 12250 “Cardiac Ultrasound Common Linear Measurements”
22 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (P5-30031, SRT, "Cardiac Shunt Study") $MeasType = DCID 12217 “Echocardiography Cardiac Shunt”
23 | > | CONTAINS | INCLUDE | DTID 5202 “Echo Section” | 1 | U |  | $SectionSubject = EV (D4-30000, SRT, "Congenital Anomaly of Cardiovascular System") $MeasType = DCID 12218 “Echocardiography Congenital”
24 | > | CONTAINS | INCLUDE | DTID 5204 “Wall Motion Analysis” | 1-n | U |  | $Procedure = DT (P5-B3121, SRT, "Echocardiography for Determining Ventricular Contraction")

Content Item Descriptions

Row 24 | The wall motion findings of stress stage. There may be multiple Template instances to report wall motion findings of multiple stages.

### TID 5201 Echocardiography Patient Characteristics

Patient Characteristic concepts in this template, which may replicate attributes in the Patient Study Module, are included here as possible targets of by-reference relationships from other content items in the SR tree.

### Note

Several of the concepts in this template duplicate concepts in TID 1007 “Subject Context, Patient”. The difference in use is that this template has those concepts as primary observations of the patient, while in TID 1007 “Subject Context, Patient” the concepts are used to set (or reset) the context for other observations.

Type:

  Extensible

Order:

  Significant

Table TID 5201. Echocardiography Patient Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121118, DCM, "Patient Characteristics") | 1 | M |  |
2 | > | CONTAINS | NUM | EV (121033, DCM, "Subject Age") | 1 | U |  | UNITS = DCID 7456 “Units of Measure for Age”
3 | > | CONTAINS | CODE | EV (121032, DCM, "Subject Sex") | 1 | U |  | DCID 7455 “Sex”
4 | > | CONTAINS | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  |
5 | > | CONTAINS | NUM | EV (F-008EC, SRT, "Systolic Blood Pressure") | 1 | U |  |
6 | > | CONTAINS | NUM | EV (F-008ED, SRT, "Diastolic Blood Pressure") | 1 | U |  |
7 | > | CONTAINS | NUM | EV (8277-6, LN, "Body Surface Area") | 1 | M |  |
8 | >> | INFERRED FROM | CODE | EV (8278-4, LN, "Body Surface Area Formula") | 1 | U |  | BCID 3663 “Body Surface Area Equations”

### TID 5202 Echo Section

This is a generic section heading Template for any of the anatomical headings. Measurements within a section heading appear as groups (by image mode, acquisition protocol, and/or protocol stage).

Table TID 5202. Parameters

Parameter Name | Parameter Usage
--- | ---
$SectionSubject | The subject modifier of the section heading container
$MeasType | The concept name of the measurement

Type:

  Extensible

Order:

  Significant

Table TID 5202. Echo Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | $SectionSubject
3 | > | CONTAINS | CONTAINER | DT (125007, DCM, "Measurement Group") | 1-n | M |  |
4 | >> | HAS CONCEPT MOD | CODE | EV (G-0373, SRT, "Image Mode") | 1 | U |  | BCID 12224 “Ultrasound Image Modes”
5 | >> | HAS CONCEPT MOD | CODE | DT (125203, DCM, "Acquisition Protocol") | 1 | U |  |
6 | >> | HAS CONCEPT MOD | TEXT | DT (125203, DCM, "Acquisition Protocol") | 1 | U |  |
7 | >> | HAS ACQ CONTEXT | CODE | EV (18139-6, LN, "Stage") | 1 | U |  | BCID 12002 “Ultrasound Protocol Stage Types”
8 | >> | CONTAINS | INCLUDE | DTID 5203 “Echo Measurement” | 1-n | M |  | $Measurement = $MeasType $Method = CID 12227 “Echocardiography Measurement Method”

Content Item Descriptions

Rows 4, 5 | Type of measurement group. May be grouped by image mode, or acquisition protocol, or some other user or manufacturer designated classification
Row 7 | For measurements acquired in a staged protocol, all measurements in a measurement group are acquired at the identified stage.

### TID 5203 Echo Measurement

Table TID 5203. Parameters

Parameter Name | Parameter Usage
--- | ---
$Measurement | Coded term or Context Group for Concept Name of measurement
$Method | Value for Measurement Method

Type:

  Extensible

Order:

  Significant

Table TID 5203. Echo Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = $Measurement $Method = $Method $TargetSite = BCID 12236 “Echo Anatomic Sites” $TargetSiteMod = BCID 12237 “Echocardiography Anatomic Site Modifiers”
2 | > | HAS CONCEPT MOD | CODE | EV (G-C048, SRT, "Flow Direction") | 1 | U |  | BCID 12221 “Flow Direction”
3 | > | HAS CONCEPT MOD | CODE | EV (R-40899, SRT, "Respiratory Cycle Point") | 1 | U |  | DCID 12234 “Respiration State”
4 | > | HAS CONCEPT MOD | CODE | EV (R-4089A, SRT, "Cardiac Cycle Point") | 1 | U |  | DCID 12233 “Cardiac Phase”
5 | > | HAS ACQ CONTEXT | CODE | EV (G-0373, SRT, "Image Mode") | 1 | U |  | DCID 12224 “Ultrasound Image Modes”
6 | > | HAS ACQ CONTEXT | CODE | EV (111031, DCM, "Image View") | 1 | U |  | BCID 12226 “Echocardiography Image View”
7 | > | HAS ACQ CONTEXT | CODE | EV (18139-6, LN, "Stage") | 1 | U |  | BCID 12002 “Ultrasound Protocol Stage Types”

Content Item Descriptions

Row 1 | TID 300 specifies an "Equivalent Meaning of Concept Name" that allows the creating application to specify the preferred composed concept name representing the measurement and the associated post-coordination Concept Modifiers (e.g., the ASE terminology described in Section N.3 “Illustrative Mapping to ASE Concepts” in PS3.17 ).

### TID 5204 Wall Motion Analysis

The Wall Motion Analysis Template is used to document wall motion scoring for any imaging modality.

Table TID 5204. Parameters

Parameter Name | Parameter Usage
--- | ---
$Procedure | The imaging procedure used for wall motion analysis.

Type:

  Extensible

Order:

  Significant

Table TID 5204. Wall Motion Analysis

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | $Procedure
3 | > | HAS ACQ CONTEXT | CODE | EV (18139-6, LN, "Stage") | 1 | U |  | BCID 3207 “Stress Test Procedure Phases”
4 | > | CONTAINS | IMAGE | EV (125201, DCM, "Illustration of Finding") | 1 | U |  |
5 | > | CONTAINS | TEXT | EV (18118-0, LN, "LV Wall Motion Segmental Findings") | 1 | U |  |
6 | > | CONTAINS | NUM | DT (125202, DCM, "LV Wall Motion Score Index") | 1 | U |  |
7 | >> | HAS CONCEPT MOD | CODE | EV (G-E048, SRT, "Assessment Scale") | 1 | M |  | BCID 12238 “Wall Motion Scoring Schemes”
8 | > | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | U |  |
9 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DT (T-D0772, SRT, "Myocardial Wall")
10 | >> | CONTAINS | CODE | EV (18179-2, LN, "Wall Segment") | 1-n | M |  | BCID 3717 “Myocardial Wall Segments”
11 | >>> | HAS PROPERTIES | CODE | EV (F-32050, SRT, "Cardiac Wall Motion") | 1 | MC | IF row 12 is absent | DCID 3703 “Wall Motion”
12 | >>> | HAS PROPERTIES | CODE | EV (G-C504, SRT, "Associated Morphology") | 1 | MC | IF row 11 is absent | DCID 3704 “Myocardium Wall Morphology Findings”
13 | >>> | HAS PROPERTIES | NUM | DT (G-C1E3, SRT, "Score") | 1 | U |  |
14 | >>> | HAS PROPERTIES | NUM | EV (122624, DCM, "Wall Thickness Ratio end-systolic to end-diastolic") | 1 | U |  | UNITS = DT (%, UCUM, "%")

Content Item Descriptions

Row 3 | The stage of the protocol at which these findings were scored. This row may be absent if this is a generic, non-staged scoring.
Row 4 | Image that graphically depicts the segments and their scores.
Row 5 | Text narration accompanying this stage.
Row 6 | The composite score computed from the average of the scored segments
Row 7 | The type of scoring scheme used to score this exam.
Row 8 | A container of all of the individual segment findings for this stage. The container shall be present if the observer makes an assessment, including the assessment of Not Visualized. It shall not be present if no evaluation was made.
Rows 11, 12 | Scar/thinning (in Row 12) may accompany akinesis and dyskinesis (in Row 11).
Row 13 | A numeric designation for the score. Score ranges vary, typically 0-4 or 0-5. Numeric scores may depend on wall motion findings as well as morphology findings. See Table 5204-1 for conventional numeric assignment schemes. The UCUM annotation code enables specifying the numeric range, ({L:N}, UCUM, "scale L:N"), where L and N are the lower and upper ends of the range.

A description of the scoring schemes described in Table 5204-1 is available in Recommendations for Quantitation of the Left Ventricle by Two-Dimensional Echocardiography, Journal of the American Society of Echocardiography, Vol 2, No 5 358-367, Oct 1989.

Table 5204-1. Numeric Score Assignment for Segmental Findings

Conventional Numeric Assignment | Wall Motion Finding or Morphology Finding
--- | ---
4 Point | 5 Point | 5 Point with Graded Hypokinesis
-1 | (F-32050, SRT, "Cardiac Wall Motion") = (R-0030D SRT, "Hyperkinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (R-0030D, SRT, "Hyperkinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (R-0030D, SRT, "Hyperkinesis")
0 | (F-32050, SRT, "Cardiac Wall Motion") = (122288, DCM, "Not Visualized") | (F-32050, SRT, "Cardiac Wall Motion") = (122288, DCM, "Not Visualized") | (F-32050, SRT, "Cardiac Wall Motion") = (122288, DCM, "Not Visualized")
1 | (F-32050, SRT, "Cardiac Wall Motion") = (R-00344, SRT, "Normal Wall Motion") | (F-32050, SRT, "Cardiac Wall Motion") = (R-00344, SRT, "Normal Wall Motion") | (F-32050, SRT, "Cardiac Wall Motion") = (R-00344, SRT, "Normal Wall Motion")
1.5 |  |  | (F-32050, SRT, "Cardiac Wall Motion") = (R-00327, SRT, "Mild Hypokinesis")
2 | (F-32050, SRT, "Cardiac Wall Motion") = (F-32056, SRT, "Hypokinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (F-32056, SRT, "Hypokinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (R-0032F, SRT, "Moderate Hypokinesis")
2.5 |  |  | (F-32050, SRT, "Cardiac Wall Motion") = (R-00370, SRT, "Severe Hypokinesis")
3 | (F-32050, SRT, "Cardiac Wall Motion") = (F-30004, SRT, "Akinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (F-30004, SRT, "Akinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (F-30004, SRT, "Akinesis")
4 | (F-32050, SRT, "Cardiac Wall Motion") = (F-32052, SRT, "Dyskinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (F-32052, SRT, "Dyskinesis") | (F-32050, SRT, "Cardiac Wall Motion") = (F-32052, SRT, "Dyskinesis")
5 |  | (G-C504, SRT, "Associated Morphology") = (D3-10510, SRT, "Ventricular Aneurysm") | (G-C504, SRT, "Associated Morphology") = (D3-10510, SRT, "Ventricular Aneurysm")

### TID 5220 Pediatric, Fetal and Congenital Cardiac Ultrasound Reports

This template forms the top of a content tree that allows an ultrasound application to describe the results of a Cardiac Ultrasound imaging procedure. It is instantiated at the root node.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 5220. Pediatric, Fetal and Congenital Cardiac Ultrasound Reports

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DCID 12245 “Cardiac Ultrasound Report Titles” | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
4 | > | CONTAINS | CONTAINER | EV (121109, DCM, "Indications for Procedure") | 1 | U |  |
5 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 12246 “Cardiac Ultrasound Indication for Study”
6 | >> | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1 | U |  |
7 | > | CONTAINS | INCLUDE | DTID 3802 “Cardiovascular Patient History” | 1 | U |  |
8 | > | CONTAINS | INCLUDE | DTID 3602 “Cardiovascular Patient Characteristics” | 1 | U |  |
9 | > | CONTAINS | INCLUDE | DTID 5225 “Cardiac Ultrasound Fetal Characteristics” | 1-n | U |  | No more than one inclusion per fetus
10 | > | CONTAINS | INCLUDE | DTID 5226 “Cardiac Ultrasound Summary Section” | 1 | U |  |
11 | > | CONTAINS | INCLUDE | DTID 5227 “Cardiac Ultrasound Fetal Summary Section” | 1-n | U |  | No more than one inclusion per fetus
12 | > | CONTAINS | CONTAINER | EV (111028, DCM, "Image Library") | 1 | U |  |
13 | >> | CONTAINS | IMAGE | No purpose of reference | 1-n | M |  |
14 | > | CONTAINS | INCLUDE | DTID 5221 “Cardiac Ultrasound Pediatric Echo Measurement Section” | 1 | U |  |
15 | > | CONTAINS | INCLUDE | DTID 5228 “Cardiac Ultrasound Fetal Measurement Section” | 1-n | UC | For Fetal Report only. | No more than one inclusion per fetus

Content Item Descriptions

Row 3 | For Fetal Report, this row establishes the subject context of the mother.
Row 7 | For Fetal Report, this row will be the patient history of the mother.
Row 8 | For Fetal Report, this row will be the Patient Characteristics for the mother.
Row 10 | For Fetal Report, this row will be the Summary Section for the mother.

### TID 5221 Cardiac Ultrasound Pediatric Echo Measurement Section

Type:

  Extensible

Order:

  Significant

Table TID 5221. Cardiac Ultrasound Pediatric Echo Measurement Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12282 “Cardiac Ultrasound Venous Return Systemic Finding Sites” $MeasType = DCID 12264 “Cardiac Ultrasound Venous Return Systemic Measurements”
2 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12283 “Cardiac Ultrasound Venous Return Pulmonary Finding Sites” $MeasType = DCID 12263 “Cardiac Ultrasound Venous Return Pulmonary Measurements”
3 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12284 “Cardiac Ultrasound Atria and Atrial Septum Finding Sites” $MeasType = DCID 12265 “Cardiac Ultrasound Atria and Atrial Septum Measurements”
4 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12285 “Cardiac Ultrasound Atrioventricular Valves Finding Sites” $MeasType = DCID 12268 “Cardiac Ultrasound Atrioventricular Valves Measurements”
5 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12286 “Cardiac Ultrasound Interventricular Septum Finding Sites” $MeasType = DCID 12269 “Cardiac Ultrasound Interventricular Septum Measurements”
6 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12287 “Cardiac Ultrasound Ventricles Finding Sites” $MeasType = DCID 12259 “Cardiac Ultrasound Ventricles Measurements”
8 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12288 “Cardiac Ultrasound Outflow Tracts Finding Sites” $MeasType = DCID 12271 “Cardiac Ultrasound Outflow Tracts Measurements”
9 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12289 “Cardiac Ultrasound Semilunar Valves, Annulate and Sinuses Finding Sites” $MeasType = DCID 12272 “Cardiac Ultrasound Semilunar Valves, Annulate and Sinuses Measurements”
10 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12290 “Cardiac Ultrasound Pulmonary Arteries Finding Sites” $MeasType = DCID 12260 “Cardiac Ultrasound Pulmonary Artery”
11 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12291 “Cardiac Ultrasound Aorta Finding Sites” $MeasType = DCID 12274 “Cardiac Ultrasound Aorta Measurements”
12 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12292 “Cardiac Ultrasound Coronary Arteries Finding Sites” $MeasType = DCID 12275 “Cardiac Ultrasound Coronary Arteries Measurements”
13 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12293 “Cardiac Ultrasound Aortopulmonary Connections Finding Sites” $MeasType = DCID 12276 “Cardiac Ultrasound Aorto Pulmonary Connections Measurements”
14 |  |  | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = DCID 12294 “Cardiac Ultrasound Pericardium and Pleura Finding Sites” $MeasType = DCID 12277 “Cardiac Ultrasound Pericardium and Pleura Measurements”

### TID 5222 Pediatric, Fetal and Congenital Cardiac Ultrasound Section

This is a generic section heading Template for any of the anatomical headings. Measurements within a section heading appear as groups (by image mode or acquisition protocol).

Table TID 5222. Parameters

Parameter Name | Parameter Usage
--- | ---
$SectionSubject | The subject modifier of the section heading container
$MeasType | The concept name of the measurement

Type:

  Extensible

Order:

  Significant

Table TID 5222. Pediatric, Fetal and Congenital Cardiac Ultrasound Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | $SectionSubject
3 | >> | HAS CONCEPT MOD | CODE | EV (P1-32006, SRT, "Heart valve replacement - prosthesis") | 1 | U |  | DCID 230 “Yes-No”
4 | > | CONTAINS | CONTAINER | DT (125007, DCM, "Measurement Group") | 1-n | M |  |
5 | >> | HAS CONCEPT MOD | CODE | EV (G-0373, SRT, "Image Mode") | 1 | U |  | BCID 12224 “Ultrasound Image Modes”
6 | >> | HAS CONCEPT MOD | TEXT | DT (125203, DCM, "Acquisition Protocol") | 1 | U |  |
7 | >> | CONTAINS | INCLUDE | DTID 5223 “Pediatric, Fetal and Congenital Cardiac Ultrasound Measurement” | 1-n | M |  | $Measurement = $MeasType $Method = CID 12227 “Echocardiography Measurement Method”

### TID 5223 Pediatric, Fetal and Congenital Cardiac Ultrasound Measurement

This Template provides for the post-coordination of a measurement with a variety of concept modifiers and acquisition context observations. When invoked from TID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section”, the measurement concept is implicitly post-coordinated with the concept modifiers of the Measurement Group (TID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” Rows 5 and 6), and with the Finding Site of the report section (TID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” Row 2). The finding site may be further specified within this Template by the Target Site and Target Site Modifiers (CID 12280 “Cardiac Ultrasound Target Sites” and CID 12281 “Cardiac Ultrasound Target Site Modifiers”).

The implicit finding site inherited from TID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” can be made explicit by using the same finding site concept in the Target Site (the measurement concept modifier), rather than a term from CID 12280 “Cardiac Ultrasound Target Sites”. This explicit post-coordination allows the use of one of the modifiers of CID 12281 “Cardiac Ultrasound Target Site Modifiers” to that finding site, as the Target Site Modifier requires an explicit Target Site in the measurement structure (TID 300 “Measurement” Rows 5 and 7). In fact, any child concept of the finding site in the SNOMED hierarchy may be used as the measurement Target Site.

The finding or target site may be identified by a concept from the SNOMED "clinical finding" or "morphological anomaly" hierarchies (e.g., D4-31220 "Atrial Septal Defect", or M-36700 "Effusion"), rather than the "anatomical structure" hierarchy. In this case, the meaning is inferred as "the anatomic location of the clinical finding or morphological anomaly, within the constraints of other implicit or explicit post-coordinated finding site concepts."

### Note

Thus when TID 5221 “Cardiac Ultrasound Pediatric Echo Measurement Section” Row 14 invokes TID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” with the section finding site concept (T-39000, SRT, "Pericardium"), and TID 5223 “Pediatric, Fetal and Congenital Cardiac Ultrasound Measurement” Row 1 applies the target site (M-36700, SRT, "Effusion"), the effective finding site is "pericardial effusion".

Table TID 5223. Parameters

Parameter Name | Parameter Usage
--- | ---
$Measurement | Coded term or Context Group for Concept Name of measurement
$Method | Value for Measurement Method

Type:

  Extensible

Order:

  Significant

Table TID 5223. Pediatric, Fetal and Congenital Cardiac Ultrasound Measurement

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | INCLUDE | DTID 300 “Measurement” | 1 | M |  | $Measurement = $Measurement $Method = $Method $TargetSite = BCID 12280 “Cardiac Ultrasound Target Sites” $TargetSiteMod = BCID 12281 “Cardiac Ultrasound Target Site Modifiers” $Derivation = DCID 3838 “Diameter Derivation”
2 | > | HAS CONCEPT MOD | CODE | EV (121425, DCM, "Index") | 1 | U |  | DCID 3455 “Index Methods”
3 | > | HAS CONCEPT MOD | CODE | EV (G-C048, SRT, "Flow Direction") | 1 | U |  | BCID 12221 “Flow Direction”
4 | > | HAS CONCEPT MOD | CODE | EV (R-40899, SRT, "Respiratory Cycle Point") | 1 | U |  | DCID 12234 “Respiration State”
5 | > | HAS CONCEPT MOD | CODE | EV (R-4089A, SRT, "Cardiac Cycle Point") | 1 | U |  | DCID 12233 “Cardiac Phase”
6 | > | HAS ACQ CONTEXT | CODE | EV (G-0373, SRT, "Image Mode") | 1 | U |  | DCID 12224 “Ultrasound Image Modes”
7 | > | HAS ACQ CONTEXT | CODE | EV (111031, DCM, "Image View") | 1 | U |  | BCID 12226 “Echocardiography Image View”

Content Item Descriptions

Row 1 | For an index type of measurement, the concept name of this row 1 will still be the original measurement concept name; it is row 2 that gives the indication that row 1 is actually an index type of measurement. When this happens, the measurement value of row 1 should be a value after being indexed and the measurement unit of row 1 should be an index type of unit. For example, to insert a "Stroke Volume Index" measurement to this SR object, the concept name of row 1 will be "Stroke Volume", its numerical value will be the calculation result of "Stroke Volume /BSA" and its units are "ml/cm2".
Row 2 | When this row is available, the row 1 is an index calculation of the object.

### TID 5225 Cardiac Ultrasound Fetal Characteristics

Contains a list of Fetus Specific characteristics.

Type:

  Extensible

Order:

  Significant

Table TID 5225. Cardiac Ultrasound Fetal Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125015, DCM, "Fetus Characteristics") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this template is invoked more than once to describe more than one fetus. |
3 | > | CONTAINS | NUM | EV (18185-9, LN, "Gestational Age") | 1 | U |  | UNITS = DCID 7456 “Units of Measure for Age”
4 | > | CONTAINS | DATE | EV (11778-8, LN, "EDD") | 1 | U |  |
5 | > | CONTAINS | NUM | EV (8867-4, LN, "Heart Rate") | 1 | U |  |

### TID 5226 Cardiac Ultrasound Summary Section

Comments and observations of the procedure of immediate clinical interest.

Type:

  Extensible

Order:

  Significant

Table TID 5226. Cardiac Ultrasound Summary Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121111, DCM, "Summary") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | MC | IF row 3 does not exist | BCID 12248 “Cardiac Ultrasound Summary Codes”
3 | > | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1-n | MC | IF row 2 does not exist |
4 | > | CONTAINS | CODE | EV (P0-009C3, SRT, "Surgical Procedure") | 1-n | U |  | BCID 12247 “Pediatric, Fetal and Congenital Cardiac Surgical Interventions”

### TID 5227 Cardiac Ultrasound Fetal Summary Section

Comments and observations of the procedure of immediate clinical interest.

Type:

  Extensible

Order:

  Significant

Table TID 5227. Cardiac Ultrasound Fetal Summary Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (125008, DCM, "Fetus Summary") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this template is invoked more than once to describe more than one fetus |
3 | > | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | MC | IF row 4 does not exist | BCID 12249 “Cardiac Ultrasound Fetal Summary Codes”
4 | > | CONTAINS | TEXT | EV (121071, DCM, "Finding") | 1-n | MC | IF row 3 does not exist |
5 | > | CONTAINS | CODE | EV (P0-009C3, SRT, "Surgical Procedure") | 1-n | U |  | BCID 12247 “Pediatric, Fetal and Congenital Cardiac Surgical Interventions”

### TID 5228 Cardiac Ultrasound Fetal Measurement Section

Type:

  Extensible

Order:

  Significant

Table TID 5228. Cardiac Ultrasound Fetal Measurement Section

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (125016, DCM, "Fetal Measurements") | 1 | M |  |
2 | > | HAS OBS CONTEXT | INCLUDE | DTID 1008 “Subject Context, Fetus” | 1 | MC | IF this template is invoked more than once to describe more than one fetus. |
3 | > | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = DCID 12279 “Cardiac Ultrasound Fetal General Measurements”
4 | > | CONTAINS | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = EV (T-F6845, SRT, "Ductus arteriosus") $MeasType = DCID 12218 “Echocardiography Congenital”
5 | > | CONTAINS | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = EV (T-F6806, SRT, "Ductus venosus") $MeasType = DCID 12218 “Echocardiography Congenital”
6 | > | CONTAINS | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = EV (T-F1810, SRT, "Umbilical artery") $MeasType = DCID 12218 “Echocardiography Congenital”
7 | > | CONTAINS | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = EV (T-D03B4, SRT, "Umbilical vein") $MeasType = DCID 12218 “Echocardiography Congenital”
8 | > | CONTAINS | INCLUDE | DTID 5222 “Pediatric, Fetal and Congenital Cardiac Ultrasound Section” | 1-n | U |  | $SectionSubject = EV (T-45600, SRT, "Middle cerebral artery") $MeasType = DCID 12218 “Echocardiography Congenital”

## Implantation Plan SR Document Templates

The templates that comprise the Implantation Plan SR Document IOD are interconnected as in Figure A-11.

Figure A-11. Implantation Plan SR Document IOD Template Structure

### TID 7000 Implantation Plan

This template contains all the necessary information to position an Implant Assembly and its Components in a patient. Therefore, all the Components that comprise an Implant Assembly are listed. If the Implant Assembly consists of more than one Component, the relation between the Components will be described as well. It is also possible to describe the registration between the Components and the patient and between the Components themselves.

To reference the Components within this document the Implantation Plan Component ID is used.

The Component Connection links two Implantation Plan Components in a commutative way. This means that for each link between A and B only one Component Connection has to be defined and not two for A-B and B-A.

The terminology used is defined by illustration using the example in Figure A-12.

Figure A-12. Implant Assembly and Components Terminology

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 7000. Implantation Plan

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (112345, DCM, "Implantation Plan") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | U |  |
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 1001 “Observation Context” | 1 | M |  |
4 | > | CONTAINS | INCLUDE | DTID 351 “Previous Reports” | 1 | MC | IFF previous Implantation Plan Documents exist | Shall only reference other Implantation Plan Documents
5 | > | CONTAINS | INCLUDE | DTID 7001 “Related Implantation Reports” | 1 | MC | IFF related Implantation Plan Documents exist that are not referenced by row 4 | Shall only reference other Implantation Plan Documents
6 | > | CONTAINS | CONTAINER | EV (112360, DCM, "Implant Component List") | 1 | M |  |
7 | >> | CONTAINS | COMPOSITE | EV (112366, DCM, "Implant Assembly Template") | 1 | U |  | References an Implant Assembly Template SOP Instance
8 | >> | CONTAINS | CONTAINER | EV (112346, DCM, "Selected Implant Component") | 1-n | M |  |
9 | >>> | CONTAINS | TEXT | EV (112347, DCM, "Component ID") | 1 | M |  |
10 | >>> | CONTAINS | CODE | EV (112370, DCM, "Component Type") | 1 | MC | IFF Row 8 contains more than one item. | DCID 7306 “Human Hip Implant Planning Landmarks”
11 | >>> | CONTAINS | COMPOSITE | No purpose of reference | 1 | M |  | References an Implant Template Storage SOP Instance
12 | >>> | CONTAINS | UIDREF | EV (112227, DCM, "Frame Of Reference UID") | 1 | M |  |
13 | >>> | CONTAINS | COMPOSITE | EV (112371, DCM, "Manufacturer Implant Template") | 1 | M |  | References an Implant Template Storage SOP Instance
14 | > | CONTAINS | CONTAINER | EV (112355, DCM, "Assembly") | 1-n | U |  |
15 | >> | CONTAINS | CONTAINER | EV (112350, DCM, "Component Connection") | 1-n | M |  |
16 | >>> | CONTAINS | CONTAINER | EV (112374, DCM, "Connected Implantation Plan Component") | 2 | M |  |
17 | >>>> | CONTAINS | TEXT | EV (112347, DCM, "Component ID") | 1 | M |  | Defined in the Implant Component List CONTAINER
18 | >>>> | CONTAINS | TEXT | EV (112351, DCM, "Mating Feature Set ID") | 1 | M |  | Only one Component Connection per Mating Feature Set is allowed
19 | >>>> | CONTAINS | TEXT | EV (112352, DCM, "Mating Feature ID") | 1 | M |  |
20 | >>>> | CONTAINS | CONTAINER | EV (112362, DCM, "Degrees of Freedom Specification") | 1-n | U |  |
21 | >>>>> | CONTAINS | TEXT | EV (112363, DCM, "Degree of Freedom ID") | 1 | M |  |
22 | >>>>> | CONTAINS | NUM | EV (112376, DCM, "Degree of Freedom Exact Translational Value") | 1 | MC | IFF Row 23, 24, 25, 26 and 27 are absent | UNITS = EV (mm, UCUM, "mm")
23 | >>>>> | CONTAINS | NUM | EV (112377, DCM, "Degree of Freedom Minimum Translational Value") | 1 | MC | IFF row 22, 25, 26, and 27 are absent | UNITS = EV (mm, UCUM, "mm")
24 | >>>>> | CONTAINS | NUM | EV (112378, DCM, "Degree of Freedom Maximum Translational Value") | 1 | MC | IFF row 22, 25, 26, and 27 are absent | UNITS = EV (mm, UCUM, "mm")
25 | >>>>> | CONTAINS | NUM | EV (112379, DCM, "Degree of Freedom Exact Rotational Value") | 1 | MC | IFF row 22, 23, 24, 26 and 27 are absent | UNITS = EV (deg, UCUM, "degree")
26 | >>>>> | CONTAINS | NUM | EV (112380, DCM, "Degree of Freedom Minimum Rotational Value") | 1 | MC | IFF row 22, 23, 24 and 25 are absent | UNITS = EV (deg, UCUM, "degree")
27 | >>>>> | CONTAINS | NUM | EV (112381, DCM, "Degree of Freedom Maximum Rotational Value") | 1 | MC | IFF row 22, 23, 24 and 25 are absent | UNITS = EV (deg, UCUM, "degree")
28 | > | CONTAINS | CONTAINER | EV (112358, DCM, "Information used for planning") | 1 | U |  |
29 | >> | CONTAINS | CODE | EV (112375, DCM, "Planning Method") | 1 | U |  | BCID 7320 “Planning Methods”
30 | >> | CONTAINS | IMAGE | EV (112354, DCM, "Patient Image") | 1-n | U |  |
31 | >>> | HAS PROPERTIES | NUM | EV (111026, DCM, "Horizontal Pixel Spacing") | 1 | M |  | UNITS = EV (mm/{pixel}, UCUM, "mm/pixel")
32 | >>> | HAS PROPERTIES | NUM | EV (111026, DCM, "Vertical Pixel Spacing") | 1 | M |  | UNITS = EV (mm/{pixel}, UCUM, "mm/pixel")
33 | >> | CONTAINS | COMPOSITE | EV (112361, DCM, "Patient Data Used During Planning") | 1-n | U |  | References SOP Instances except Images
34 | >>> | HAS PROPERTIES | UIDREF | EV (112356, DCM, "User Selected Fiducial") | 1-n | MC | IFF row 33 references a Fiducial SOP Instance |
35 | >>>> | HAS CONCEPT MOD | TEXT | EV (112369, DCM, "Fiducial Intent") | 1 | U |  |
36 | > | CONTAINS | CONTAINER | EV (112367, DCM, "Planning Information for Intraoperative Usage") | 1 | U |  |
37 | >> | CONTAINS | TEXT | EV (121173, DCM, "Physician Note") | 1-n | U |  |
38 | >> | CONTAINS | COMPOSITE | EV (112359, DCM, "Supporting Information") | 1 | U |  | SOP Class UID shall be Encapsulated PDF Storage
39 | >> | CONTAINS | COMPOSITE | EV (112372, DCM, "Derived Planning Images") | 1-n | U |  |
40 | >> | CONTAINS | COMPOSITE | EV (112353, DCM, "Spatial Registration") | 1-n | U |  | References Spatial Registration SOP Instances and Deformable Spatial Registration SOP Instances
41 | >>> | HAS PROPERTIES | UIDREF | EV (112227, DCM, "Frame of Reference UID") | 1-n | U |  |
42 | >> | CONTAINS | COMPOSITE | EV (112373, DCM, "Derived Planning Data") | 1-n | U |  | References SOP Instances except Images and Spatial Registrations
43 | >>> | HAS PROPERTIES | UIDREF | EV (112357, DCM, "Derived Fiducial") | 1-n | MC | IFF row 42 references a Fiducial SOP Instance |
44 | >>>> | HAS CONCEPT MOD | TEXT | EV (112369, DCM, "Fiducial Intent") | 1 | U |  |
45 | >> | CONTAINS | COMPOSITE | EV (112364, DCM, "Related Patient Data Not Used During Planning") | 1-n | U |  |

Content Item Descriptions

Row 7 | If an Implant Assembly Template was used for the planning, it should be referenced here.
Row 9 | ID given to this Implant Component. Used to reference this specific Component within the Implantation Plan.
Row 10 | See description of Component Type Code Sequence (0076,0034) Attribute in Section C.29.2.1 “Implant Assembly Template Module” in PS3.3
Row 11 | Reference to the Template that describes that component. May be the same Implant Template as referenced in row 13. The target of the reference may not be needed or available during implantation. e.g., if the plan is opened in another hospital where those implant templates are not used.
Row 12 | This Frame of Reference is the Frame of Reference of the Implant Component (Frame of Reference UID (0020,0052) Attribute in the “Generic Implant Template Description Module” in PS3.3 ). This may help to find the right registration information (row 43).
Row 13 | References the Original Template that was the basis for the Derived Template. May be the same Implant Template as referenced in row 11. The target of the reference may not be needed or available during implantation, e.g., if the plan is opened in another hospital where those implant templates are not used.
Row 14 | If there is no Component Connection between sets of Implant Components, one Assembly must be used for each set.
Row 17 | The ID of a planned Component that is defined in this document and that is part of this Relation.
Row 18 | See description of Mating Feature Set ID (0068,63C0) Attribute in the “Generic Implant Template Mating Features Module” in PS3.3
Row 19 | See description of Mating Feature ID (0068,63F0) Attribute in the “Generic Implant Template Mating Features Module” in PS3.3
Row 21 | See description of Degree of Freedom ID (0068,6410) Attribute in the “Generic Implant Template Mating Features Module” in PS3.3
Row 22 - 27 | Defines the range or exact value that was selected or calculated by the planning application.
Row 31 | Defines the calibrated Horizontal Pixel Spacing that was used by the planning application, which may be different from the spacing encoded in the referenced Image SOP Instance.
Row 32 | Defines the calibrated Vertical Pixel Spacing that was used by the planning application, which may be different from the spacing encoded in the referenced Image SOP Instance.
Row 33 | Any patient data other than Image IEs used for the planning, e.g., Surface Segmentations.
Row 34 | Fiducials selected by the user for registration of implant components referenced in the parent content item.
Row 35 | User comment about the Fiducial. This may be the reason it was selected, the intended use, the anatomical or non-anatomical structure that the Fiducial represents, or any other intent.
Row 38 | All kinds of information in PDF form that are created by a planning application may be referenced here, e.g., drawings.
Row 39 | All kinds of images that are created by a planning application should be referenced here, e.g., images that show patient images overlaid with contour information of the Implant Component, or images that show how several implant components may be composed, or merged patient images.
Row 40 | References registration objects that contain registration data that is relevant for this Implantation Plan, e.g., registration of Implant Components.
Rows 5, 6 | Identifies one or more items within the sequence of referenced Frames of Reference (Registration Sequence (0070,0308) in the Spatial Registration Module or Deformable Registration Sequence (0064,0002) in the “Deformable Spatial Registration Module” in PS3.3 ) that are relevant for this Implantation Plan. See Figure A-13.
Row 42 | Any patient data created during the planning process that is not referenced in row 39 and 40, e.g., Surface Segmentation Instances created by the planning application.
Row 43 | These Fiducials are derived from the Fiducials identified in Row 34.
Row 45 | References to any relevant patient data containing IOD instances that were not used in planning or derived from it but belong to the patient model. Might be reports, images, surface segmentations, or other.

Figure A-13. References to Registration Objects

### TID 7001 Related Implantation Reports

This general template provides a means to reference related Implantation Plan SR Document instances that are not previous Reports. Other Implantations that are planned to be done during the same intervention should be referenced here.

Type:

  Extensible

Order:

  Significant

Table TID 7001. Related Implantation Reports

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (112365, DCM, "Related Implantation Reports") | 1 | M |  |
2 | > | CONTAINS | COMPOSITE |  | 1-n | M |  |

## Relevant Patient Information Templates

### TID 9000 Relevant Patient Information for Breast Imaging

This template collects a patient's relevant information as it relates to breast imaging. This template, together with its subordinate templates, describes the history of a patient's reproductive system, hormone medications, past procedures, risk factors, and indicated problems as they relate to breast health.

Type:

  Extensible

Order:

  Significant

Table TID 9000. Relevant Patient Information for Breast Imaging

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111511, DCM, "Relevant Patient Information for Breast Imaging") | 1 | M |  |
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | CONTAINS | INCLUDE | DTID 3114 “Patient Assessment” | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 9001 “Gynecological History” | 1 | U |  |
5 | > | CONTAINS | INCLUDE | DTID 9002 “Medication, Substance, Environmental Exposure” | 1 | U |  | $ContainerConcept = EV (10160-0, LN, "History Of Medication Use") $CodeConcept = EV (111516, DCM, "Medication Type") $CodeValue = DCID 6080 “Gynecological Hormones”
6 | > | CONTAINS | INCLUDE | DTID 9003 “Previous Procedure” | 1 | U |  | $ProcedureList = DCID 6083 “Procedures for Breast” $ProcedureModifier = DCID 6058 “Procedure Modifiers for Breast” $NumConceptName = DCID 6095 “Numeric Properties of Procedures” $LateralityValue = DCID 6022 “Side” $ProcedureResult = DCID 6063 “Interventional Procedure Results” $ComplicationValue = DCID 6062 “Interventional Procedure Complications”
7 | > | CONTAINS | INCLUDE | DTID 9004 “Indicated Problem” | 1 | U |  | $ProblemList = DCID 6055 “Breast Clinical Finding Or Indicated Problem” $LateralityValue = DCID 6022 “Side” $LocationValue = DCID 6018 “Clockface Location or Region”, DCID 6020 “Quadrant Location”
8 | > | CONTAINS | INCLUDE | DTID 9005 “Risk Factor” | 1 | U |  | $RiskList = DCID 6081 “Breast Cancer Risk Factors” $FamilyList = DCID 7451 “Family Member”

### TID 9001 Gynecological History

This general template collects the details of a patient's reproductive system history, such as number of births, and gynecological surgery history.

Type:

  Extensible

Order:

  Significant

Table TID 9001. Gynecological History

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (R-20767, SRT, "Gynecological History") | 1 | M |  |
2 | > | HAS OBS CONTEXT | CODE | EV (111534, DCM, "Role of person reporting") | 1 | U |  | DCID 7450 “Person Roles”
3 | > | CONTAINS | DATE | EV (11955-2, LN, "Date of last menstrual period") | 1 | U |  |
4 | > | CONTAINS | NUM | EV (111518, DCM, "Age when first menstrual period occurred") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
5 | > | CONTAINS | NUM | EV (111519, DCM, "Age at First Full Term Pregnancy") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
6 | > | CONTAINS | NUM | EV (11977-6, LN, "Para") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
7 | > | CONTAINS | NUM | EV (11639-2, LN, "Term") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
8 | > | CONTAINS | NUM | EV (11637-6, LN, "Preterm") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
9 | > | CONTAINS | NUM | EV (11636-8, LN, "Live Births") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
10 | > | CONTAINS | NUM | EV (111593, DCM, "LBW or IUGR") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
11 | > | CONTAINS | NUM | EV (11996-6, LN, "Gravida") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
12 | > | CONTAINS | NUM | EV (11612-9, LN, "Aborta") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
13 | > | CONTAINS | NUM | EV (33065-4, LN, "Ectopic Pregnancies") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
14 | > | CONTAINS | NUM | EV (111520, DCM, "Age at Menopause") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
15 | > | CONTAINS | NUM | EV (111521, DCM, "Age when hysterectomy performed") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
16 | >> | HAS CONCEPT MOD | CODE | EV (R-404ED, SRT, "Extent") | 1 | U |  | EV (R-404F1, SRT, "Complete") EV (R-404FE, SRT, "Partial")
17 | > | CONTAINS | NUM | EV (111522, DCM, "Age when left ovary removed") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
18 | > | CONTAINS | NUM | EV (111523, DCM, "Age when right ovary removed") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
19 | > | CONTAINS | CODE | EV (111543, DCM, "Breast feeding history") | 1 | U |  | DCID 230 “Yes-No”
20 | >> | HAS PROPERTIES | NUM | EV (111544, DCM, "Average breast feeding period") | 1 | U |  | UNITS = EV (wk, UCUM, "Week")
21 | > | CONTAINS | CODE | EV (111532, DCM, "Pregnancy Status") | 1 | U |  | DCID 6096 “Pregnancy Status”
22 | > | CONTAINS | CODE | EV (111391, DCM, "Menstrual Cycle Phase") | 1 | U |  | DCID 6163 “Menstrual Cycle Phase”

### TID 9002 Medication, Substance, Environmental Exposure

This general template provides detailed information on a patient's medication or substance use, or exposure to environmental factors, including type and duration of use or exposure.

Table TID 9002. Parameters

Parameter Name | Parameter Usage
--- | ---
$ContainerConcept | Coded term for the concept name of the CONTAINER, identifying it as medication, substance, or environmental exposure history.
$CodeConcept | Coded term for the concept name of the CODE, identifying it as medication, substance, or environmental exposure.
$CodeValue | Coded term or Context Group for value of the medication, substance, or environmental exposure.

Type:

  Extensible

Order:

  Significant

Table TID 9002. Medication, Substance, Environmental Exposure

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | $ContainerConcept | 1 | M |  |
2 | > | CONTAINS | CODE | $CodeConcept | 1-n | M |  | $CodeValue
3 | >> | HAS CONCEPT MOD | CODE | EV (G-C032, SRT, "Classification") | 1 | U |  |
4 | >> | HAS OBS CONTEXT | CODE | EV (111534, DCM, "Role of person reporting") | 1 | U |  | DCID 7450 “Person Roles”
5 | >> | HAS PROPERTIES | NUM | EV (111524, DCM, "Age Started") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
6 | >> | HAS PROPERTIES | NUM | EV (111525, DCM, "Age Ended") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
7 | >> | HAS PROPERTIES | DATETIME | EV (111526, DCM, "DateTime Started") | 1 | U |  |
8 | >> | HAS PROPERTIES | DATETIME | EV (111527, DCM, "DateTime Ended") | 1 | U |  |
9 | >> | HAS PROPERTIES | NUM | EV (G-7290, SRT, "Duration") | 1 | U |  | UNITS = DCID 6046 “Units of Follow-up Interval”
10 | >> | HAS PROPERTIES | CODE | EV (111528, DCM, "Ongoing") | 1 | U |  | DCID 230 “Yes-No”
11 | >> | HAS PROPERTIES | TEXT | EV (111529, DCM, "Brand Name") | 1 | U |  |
12 | >> | HAS PROPERTIES | NUM | DCID 6092 “Quantitative Concepts for Usage, Exposure” | 1 | U |  | The unit of measure shall be quantity per unit of time
13 | >> | HAS PROPERTIES | CODE | DCID 6093 “Qualitative Concepts for Usage, Exposure Amount” | 1 | U |  | DCID 6090 “Relative Usage, Exposure Amount”
14 | >> | HAS PROPERTIES | CODE | DCID 6094 “Qualitative Concepts for Usage, Exposure Frequency” | 1 | U |  | DCID 6091 “Relative Frequency of Event Values”

Content Item Descriptions

Row 3 "Classification" | No context group is provided for the value set, but it is recommended that values from a standard external coding scheme, such as SRT or NDC, be used.
Rows 13 & 14 | If both of these content items are instantiated, the concept names selected for each should match. For example, use "Relative dose amount" as the concept name for row 13 with "Relative dose frequency" as the concept name for row 14.

### TID 9003 Previous Procedure

This general template provides detailed information on a patient's previous procedure, surgery, or treatment.

Table TID 9003. Parameters

Parameter Name | Parameter Usage
--- | ---
$ProcedureList | Coded term or Context Group for value of Previous Procedure
$ProcedureModifier | Coded term or Context Group for value of Previous Procedure Modifier
$NumConceptName | Coded term or Context Group for the concept name of a numeric property of the Previous Procedure
$LateralityValue | Coded term or Context Group for value of Laterality
$ProcedureResult | Coded term or Context Group for value of Result of Procedure
$ComplicationValue | Coded term or Context Group for value of Complication

Type:

  Extensible

Order:

  Significant

Table TID 9003. Previous Procedure

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111513, DCM, "Relevant Previous Procedures") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (111531, DCM, "Previous Procedure") | 1-n | M |  | $ProcedureList
3 | >> | HAS CONCEPT MOD | CODE | EV (111464, DCM, "Procedure Modifier") | 1-n | U |  | $ProcedureModifier
4 | >> | HAS OBS CONTEXT | CODE | EV (111534, DCM, "Role of person reporting") | 1 | U |  | DCID 7450 “Person Roles”
5 | >> | HAS PROPERTIES | NUM | $NumConceptName | 1-n | U |  |
6 | >> | HAS PROPERTIES | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | $LateralityValue
7 | >> | HAS PROPERTIES | DATETIME | EV (122146, DCM, "Procedure DateTime") | 1 | UC | XOR row 8 |
8 | >> | HAS PROPERTIES | CODE | EV (111395, DCM, "Estimated Timeframe") | 1 | UC | XOR row 7 | BCID 6164 “Time Intervals”
9 | >> | HAS PROPERTIES | NUM | EV (R-42009, SRT, "Number of occurrences") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
10 | >> | HAS PROPERTIES | CODE | EV (DD-60002, SRT, "Complication of procedure") | 1-n | U |  | $ComplicationValue
11 | >>> | HAS PROPERTIES | CODE | EV (111466, DCM, "Severity of Complication") | 1 | U |  | DCID 251 “Severity of Complication”
12 | >> | HAS PROPERTIES | CODE | EV (122177, DCM, "Procedure Result") | 1 | U |  | $ProcedureResult
13 | >> | HAS PROPERTIES | INCLUDE | DTID 4207 “Breast Imaging Pathology Results” | 1-n | U |  |

### TID 9004 Indicated Problem

This general template provides information about indicated problems presented by a patient. For example, indicated breast problems relating to the purpose for a mammographic examination.

Table TID 9004. Parameters

Parameter Name | Parameter Usage
--- | ---
$ProblemList | Coded term or Context Group for value of Indicated Problem
$LateralityValue | Coded term or Context Group for value of Laterality
$LocationValue | Coded term or Context Group for value of Location

Type:

  Extensible

Order:

  Significant

Table TID 9004. Indicated Problem

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (11450-4, LN, "Problem List") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (111533, DCM, "Indicated Problem") | 1-n | M |  | $ProblemList
3 | >> | HAS OBS CONTEXT | CODE | EV (111534, DCM, "Role of person reporting") | 1 | U |  | DCID 7450 “Person Roles”
4 | >> | HAS OBS CONTEXT | DATETIME | EV (111535, DCM, "DateTime problem observed") | 1 | U |  |
5 | >> | HAS PROPERTIES | CODE | EV (G-C171, SRT, "Laterality") | 1 | U |  | $LateralityValue
6 | >> | HAS PROPERTIES | CODE | EV (G-C0E3, SRT, "Finding site") | 1 | U |  | $LocationValue
7 | >> | HAS PROPERTIES | NUM | EV (G-7290, SRT, "Duration") | 1 | U |  |
8 | >> | HAS PROPERTIES | CODE | EV (R-407E7, SRT, "Frequency") | 1 | U |  | DCID 6091 “Relative Frequency of Event Values”
9 | >> | HAS PROPERTIES | DATETIME | EV (111536, DCM, "DateTime of last evaluation") | 1 | U |  |
10 | >> | HAS PROPERTIES | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

### TID 9005 Risk Factor

This general template provides detailed information on the risk factors for a patient, related to medical history for themselves and family members.

Table TID 9005. Parameters

Parameter Name | Parameter Usage
--- | ---
$RiskList | Coded term or Context Group for value of Risk Factor
$FamilyList | Coded term or Context Group for value of Family Member with Risk Factor

Type:

  Extensible

Order:

  Significant

Table TID 9005. Risk Factor

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111515, DCM, "Relevant Risk Factors") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (F-01500, SRT, "Risk factor") | 1-n | M |  | $RiskList
3 | >> | HAS CONCEPT MOD | CODE | EV (111530, DCM, "Risk Factor modifier") | 1 | U |  | EV (G-0002, SRT, "Family history of")
4 | >> | HAS PROPERTIES | NUM | EV (18185-9, LN, "Gestational Age") | 1 | UC | IFF value of row 2 is (G-0305, SRT, "History of - premature delivery") |
5 | >> | HAS OBS CONTEXT | CODE | EV (111534, DCM, "Role of person reporting") | 1 | U |  | DCID 7450 “Person Roles”
6 | >> | HAS PROPERTIES | NUM | EV (111538, DCM, "Age at Occurrence") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
7 | >> | HAS PROPERTIES | NUM | EV (G-7290, SRT, "Duration") | 1 | U |  | UNITS = DCID 6046 “Units of Follow-up Interval”
8 | >> | HAS PROPERTIES | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
9 | >> | INFERRED FROM | CODE | EV (111537, DCM, "Family Member with Risk Factor") | 1-n | U |  | $FamilyList
10 | >>> | HAS PROPERTIES | NUM | EV (111538, DCM, "Age at Occurrence") | 1 | U |  | UNITS = EV (a, UCUM, "Year")
11 | >>> | HAS CONCEPT MOD | CODE | EV (111539, DCM, "Menopausal phase") | 1 | U |  | DCID 6086 “Menopausal Phase”
12 | >>> | HAS CONCEPT MOD | CODE | EV (111540, DCM, "Side of Family") | 1 | U |  | DCID 6097 “Side of Family”

### TID 9006 Obstetric History

This general template collects the details of a patient's obstetric history for a current pregnancy. Information regarding previous pregnancies is conveyed using the Gynecological History Template.

Type:

  Extensible

Order:

  Significant

Table TID 9006. Obstetric History

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (R-20658, SRT, "Obstetric History") | 1 | M |  |
2 | > | CONTAINS | DATE | DCID 12003 “OB-GYN Dates” | 1-n | U |  |
3 | > | CONTAINS | NUM | EV (18185-9, LN, "Gestational Age") | 1 | U |  | UNITS = EV (d, UCUM, "day")
4 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1-n | U |  |

Content Item Descriptions

Row 3 "Gestational Age" | Observation DateTime (0040,A032) for Content Item shall be present, in order to convey the date and time at which this Gestational Age was established.

### TID 9007 General Relevant Patient Information

This template collects a patient's relevant information for general purpose use. This template, together with its subordinate templates, describes the history of a patient's reproductive system, medications, substance use, environmental exposure, past procedures, risk factors, and indicated problems.

Type:

  Extensible

Order:

  Significant

Table TID 9007. General Relevant Patient Information

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (111517, DCM, "Relevant Patient Information") | 1 | M |  |
2 | > | HAS CONCEPT MOD | INCLUDE | DTID 1204 “Language of Content Item and Descendants” | 1 | M |  |
3 | > | CONTAINS | INCLUDE | DTID 3114 “Patient Assessment” | 1 | U |  |
4 | > | CONTAINS | INCLUDE | DTID 9002 “Medication, Substance, Environmental Exposure” | 1 | U |  | $ContainerConcept = EV (10160-0, LN, "History Of Medication Use") $CodeConcept = EV (111516, DCM, "Medication Type")
5 | > | CONTAINS | INCLUDE | DTID 9002 “Medication, Substance, Environmental Exposure” | 1 | U |  | $ContainerConcept = EV (111545, DCM, "Substance Use History") $CodeConcept = EV (111546, DCM, "Used Substance Type") $CodeValue = BCID 6089 “Substances”
6 | > | CONTAINS | INCLUDE | DTID 9002 “Medication, Substance, Environmental Exposure” | 1 | U |  | $ContainerConcept = EV (111547, DCM, "Environmental Exposure History") $CodeConcept = EV (111548, DCM, "Environmental Factor")
7 | > | CONTAINS | INCLUDE | DTID 9003 “Previous Procedure” | 1 | U |  | $LateralityValue = BCID 244 “Laterality”
8 | > | CONTAINS | INCLUDE | DTID 9004 “Indicated Problem” | 1 | U |  | $LateralityValue = BCID 244 “Laterality”
9 | > | CONTAINS | INCLUDE | DTID 9005 “Risk Factor” | 1 | U |  | $RiskList = BCID 6087 “General Risk Factors” $FamilyList = DCID 7451 “Family Member”
10 | > | CONTAINS | INCLUDE | DTID 9001 “Gynecological History” | 1 | U |  |
11 | > | CONTAINS | INCLUDE | DTID 9006 “Obstetric History” | 1 | U |  |
12 | > | CONTAINS | INCLUDE | DTID 3802 “Cardiovascular Patient History” | 1 | U |  |
13 | > | CONTAINS | INCLUDE | DTID 351 “Previous Reports” | 1 | U |  |

## X-Ray Radiation Dose SR IOD Templates

The templates that comprise the X-Ray Radiation Dose SR are interconnected as in Figure A-14.

Figure A-14. X-Ray Radiation Dose SR IOD Template Structure

### TID 10001 Projection X-Ray Radiation Dose

This template defines a container (the root) with subsidiary content items, each of which represents a single projection X-Ray irradiation event entry or plane-specific dose accumulations. There is a defined recording observer (the system or person responsible for recording the log, generally the system). A Biplane irradiation event will be recorded as two individual events, one for each plane. Accumulated values will be kept separate for each plane.

Type:

  Extensible

Order:

  Non-Significant

Root:

  Yes

Table TID 10001. Projection X-Ray Radiation Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113701, DCM, "X-Ray Radiation Dose Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | DT (113704, DCM, "Projection X-Ray") DT (P5-40010, SRT, "Mammography")
3 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E8, SRT, "Has Intent") | 1 | M |  | DCID 3629 “Procedure Intent”
4 | > | CONTAINS | CODE | EV (122142, DCM, "Acquisition Device Type") | 1 | U |  | DCID 10032 “Projection X-Ray Acquisition Device Types”
5 | > |  | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
6 | > | HAS OBS CONTEXT | CODE | EV (113705, DCM, "Scope of Accumulation") | 1 | M |  | DCID 10000 “Scope of Accumulation”
7 | >> | HAS PROPERTIES | UIDREF | DCID 10001 “UID Types” | 1 | M |  |
8 | > | CONTAINS | CODE | EV (113945, DCM, "X-Ray Detector Data Available") | 1 | U |  | DCID 230 “Yes-No”
9 | > | CONTAINS | CODE | EV (113943, DCM, "X-Ray Source Data Available") | 1 | U |  | DCID 230 “Yes-No”
10 | > | CONTAINS | CODE | EV (113944, DCM, "X-Ray Mechanical Data Available") | 1 | U |  | DCID 230 “Yes-No”
11 | > | CONTAINS | INCLUDE | DTID 10002 “Accumulated X-Ray Dose” | 1 | MC | IFF Single Plane system | $Plane = EV (113622, DCM, "Single Plane")
12 | > | CONTAINS | INCLUDE | DTID 10002 “Accumulated X-Ray Dose” | 1 | MC | IFF Biplane system | $Plane = EV (113620, DCM, "Plane A")
13 | > | CONTAINS | INCLUDE | DTID 10002 “Accumulated X-Ray Dose” | 1 | MC | IFF Biplane system | $Plane = EV (113621, DCM, "Plane B")
14 | > | CONTAINS | INCLUDE | DTID 10003 “Irradiation Event X-Ray Data” | 1-n | MC | IF any of the values of TID (10001) Row 18 are not (113858, DCM, "MPPS Content"), (113866, DCM, "Copied From Image Attributes") or (113867, DCM, "Computed From Image Attributes") |
15 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
16 | > | CONTAINS | IMAGE | EV (121342, DCM, "Dose Image") | 1-n | U |  |
17 | > | CONTAINS | INCLUDE | DTID 1020 “Person Participant” | 1 | U |  | $PersonProcedureRole = EV (113850, DCM, "Irradiation Authorizing")
18 | > | CONTAINS | CODE | EV (113854, DCM, "Source of Dose Information") | 1-n | M |  | DCID 10020 “Source of Projection X-Ray Dose Information”

Content Item Descriptions

Row 2 | "Projection X-Ray" refers to procedures performed on either integrated equipment (where information is passed between the X-Ray source (generator and tube), detector, and mechanical systems), or non-integrated equipment (where data might not be available for one or more components such as cassette-based systems). The data availability can be described in Rows 8, 9 and 10. The specific type of equipment can be described in Row 4. The coded term for "Mammography" is intended to encompass all types of projection X-Ray imaging of the breast. ### Note Mammography exams are distinguished by a different value in this attribute, Angiography exams are distinguished by the Irradiation Event Type attribute = Fluoroscopy, CR/DR exams are distinguished by one or more of the Data Availability Flags = No, and CT exams are distinguished by the use of a different template.
Row 5 | The observer context may include both a Person Observer identification, as well as the identity of the equipment providing the values for the irradiation event (Device Observer identification), if not inherited.
Row 8 | A value of "No" indicates that details associated with the X-Ray Detector are not available to the device generating this report. For example, an X-Ray Source system might lack any communication with the associated cassette-based X-Ray detector or any method of entering such information. A value of "Yes" or the absence of this row means that the details are available.
Row 9 | A value of "No" indicates that details associated with the X-Ray Source are not available to the device generating this report. For example, a cassette-based X-Ray detector might lack any communication with the associated X-Ray Source or any method of entering such information. A value of "Yes" or the absence of this row means that the details are available.
Row 10 | A value of "No" indicates that details associated with the Gantry and/or Table are not available to the device generating this report. For example, a cassette-based X-Ray detector might lack any communication with the associated gantry/table or any method of entering such information. A value of "Yes" or the absence of this row means that the details are available.
Row 11 | Cassette-based X-Ray systems should consider themselves to be Single Plane systems.
Row 14 | Details of the underlying irradiation events. If Row 18 has a value of "MPPS Content" then a TID 10003 “Irradiation Event X-Ray Data” item may be generated for each item in the MPPS Exposure Dose Sequence (0040,030E), but since this is an optional element in MPPS, if it is absent, empty or incomplete, there may be no irradiation event level information available. Alternatively, the information may be copied or computed from the images.
Row 16 | The Dose Image references a graphic representation of the radiation dose distribution. This may be a Secondary Capture scan of a dosimetry film.
Row 17 | The physician responsible for determining that the irradiating procedure was appropriate for the indications. The value may come from Requesting Physician (0032,1032), Requesting Physician Identification Sequence (0032,1031) or somewhere else based on hospital policies.
Row 18 | The primary source of information from which this dose object was constructed. The Source of Dose Information (Row 18) is independent of the Scope Of Accumulation (Row 6); e.g., it would be typical to have a scope of (113016, DCM, "Performed Procedure Step"), but a source of (113856, DCM, "Automated Data Collection") rather than (113858, DCM, "MPPS Content").

### TID 10002 Accumulated X-Ray Dose

This general template provides detailed information on projection X-Ray dose value accumulations over several irradiation events from the same equipment (typically a study or a performed procedure step).

Table TID 10002. Parameters

Parameter Name | Parameter Usage
--- | ---
$Plane | Coded term identifying to which acquisition plane the encoded information belongs.

Type:

  Extensible

Order:

  Non-Significant

Table TID 10002. Accumulated X-Ray Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113702, DCM, "Accumulated X-Ray Dose Data") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (113764, DCM, "Acquisition Plane") | 1 | M |  | $Plane
3 | > | CONTAINS | CONTAINER | EV (122505, DCM, "Calibration") | 1-n | MC | IFF Calibration Data is available |
4 | >> | HAS CONCEPT MOD | CODE | EV (113794, DCM, "Dose Measurement Device") | 1 | M |  | DCID 10010 “Dose Measurement Devices”
5 | >> | CONTAINS | DATETIME | EV (113723, DCM, "Calibration Date") | 1 | M |  |
6 | >> | CONTAINS | NUM | EV (122322, DCM, "Calibration Factor") | 1 | M |  | UNITS = EV (1, UCUM, "no units")
7 | >> | CONTAINS | NUM | EV (113763, DCM, "Calibration Uncertainty") | 1 | M |  | UNITS = EV (%, UCUM, "Percent")
8 | >> | CONTAINS | TEXT | EV (113724, DCM, "Calibration Responsible Party") | 1 | M |  |
9 | >> | CONTAINS | TEXT | EV (113720, DCM, "Calibration Protocol") | 1 | U |  |
10 | > | CONTAINS | INCLUDE | DTID 10004 “Accumulated Fluoroscopy and Acquisition Projection X-Ray Dose” | 1 | MC | IFF TID (10001) Row 4 = (113957, DCM, "Fluoroscopy-Guided Projection Radiography System") or TID (10001) Row 2 = (113704, DCM, "Projection X-Ray") and TID (10001) Row 4 is absent) |
11 | > | CONTAINS | INCLUDE | DTID 10005 “Accumulated Mammography X-Ray Dose” | 1 | MC | IFF TID (10001) Row 2 = (P5-40010, SRT, "Mammography") |
12 | > | CONTAINS | INCLUDE | DTID 10007 “Accumulated Total Projection Radiography Dose” | 1 | MC | IFF TID (10001) Row 4 = (113958, DCM, "Integrated Projection Radiography System") or TID (10001) Row 4 = (113957, DCM, "Fluoroscopy-Guided Projection Radiography System") or TID (10001) Row 2 = (113704, DCM, "Projection X-Ray") and TID (10001) Row 4 is absent) |
13 | > | CONTAINS | INCLUDE | DTID 10006 “Accumulated Cassette-based Projection Radiography Dose” | 1 | MC | IFF TID (10001) Row 4 = (113959, DCM, "Cassette-based Projection Radiography System") |
14 | > | CONTAINS | INCLUDE | DTID 1021 “Device Participant” | 1 | MC | Required if the irradiating device is not the recording device and the dose was accumulated on a single device. | $DeviceProcedureRole = EV (113859, DCM, "Irradiating Device")

Content Item Descriptions

Row 5 | Date that the calibration of the equipment's dose indicators was performed
Row 6 | Typically a value provided by the medical physicist. The recorded dose or dose area product values in this report can be multiplied by this factor to obtain estimated real-world values. ### Note It is important that this value must not be applied to the measured values before storing them in the report.
Row 7 | Value range from 0 to 100 percent. Uncertainty of the 'actual' value expressed as+/- of the mean.
Row 8 | Identifies Individual or organization responsible for calibration
Row 9 | Describes calibration protocol according to equipment standards or local guidelines.
Row 14 | The device that produced the irradiation accumulated in this template. I.e., the X-Ray source. This is not required to be present if the information is the same as that already recorded in TID 1004 “Device Observer Identifying Attributes” encoded via the inclusion of TID 1002 “Observer Context” in TID 10001 “Projection X-Ray Radiation Dose” Row 5, which in turn may be absent if identical to the content in the Enhanced General Equipment Module, or if more than one device produced the accumulated irradiation.

### TID 10003 Irradiation Event X-Ray Data

This template conveys the dose and equipment parameters of a single irradiation event.

The template and requirements are structured to consider equipment with various levels of integration between the components (X-Ray Source, Plate or Detector, and Gantry/Table) of the equipment.

An irradiation event is the loading of X-Ray equipment caused by a single continuous actuation of the equipment's irradiation switch, from the start of the loading time of the first pulse until the loading time trailing edge of the final pulse. The irradiation event is the "smallest" information entity to be recorded in the realm of Radiation Dose reporting. Individual Irradiation Events are described by a set of accompanying physical parameters that are sufficient to understand the "quality" of irradiation that is being applied. This set of parameters may be different for the various types of equipment that are able to create irradiation events. Any automatic on-off switching of the irradiation source during the event shall not be treated as separate events, rather the event includes the time between start and stop of irradiation as triggered by the user. E.g., a pulsed fluoro X-Ray acquisition shall be treated as a single irradiation event.

As described in Section 6.2.4, measurement concepts may be post-coordinated, even though not explicitly specified in the Template. In particular, post-coordination using modifier concept (121401, DCM, "Derivation"), with modifier values drawn from CID 10009 “Measured/calculated” would be appropriate to encode indications of measured or of calculated values.

Type:

  Extensible

Order:

  Non-Significant

Table TID 10003. Irradiation Event X-Ray Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113706, DCM, "Irradiation Event X-Ray Data") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (113764, DCM, "Acquisition Plane") | 1 | M |  | DCID 10003 “Equipment Plane Identification”
3 | > | CONTAINS | UIDREF | EV (113769, DCM, "Irradiation Event UID") | 1 | M |  |
4 | > | CONTAINS | TEXT | EV (113605, DCM, "Irradiation Event Label") | 1 | U |  |
5 | >> | HAS CONCEPT MOD | CODE | EV (113606, DCM, "Label Type") | 1 | MC | IF the value of Row 4 is the value of an Attribute in the images. | DCID 10022 “Label Types”
6 | > | CONTAINS | DATETIME | DT (111526, DCM, "DateTime Started") | 1 | M |  |
7 | > | CONTAINS | CODE | EV (113721, DCM, "Irradiation Event Type") | 1 | M |  | DCID 10002 “Irradiation Event Types”
8 | > | CONTAINS | TEXT | EV (125203, DCM, "Acquisition Protocol") | 1 | U |  |
9 | > | CONTAINS | CODE | EV (T-D0005, SRT, "Anatomical structure") | 1 | U |  | DCID 4009 “DX Anatomy Imaged”
10 | >> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | UC | If anatomy is bi-lateral | DCID 244 “Laterality”
11 | > | CONTAINS | CODE | EV (111031, DCM, "Image View") | 1 | U |  | DCID 4010 “DX View” DCID 4014 “View for Mammography”
12 | >> | HAS CONCEPT MOD | CODE | EV (111032, DCM, "Image View Modifier") | 1-n | U |  | DCID 4011 “DX View Modifier” DCID 4015 “View Modifier for Mammography”
13 | >> | CONTAINS | CODE | EV (113946, DCM, "Projection Eponymous Name") | 1 | U |  | DCID 4012 “Projection Eponymous Name”
14 | > | CONTAINS | CODE | EV (113745, DCM, "Patient Table Relationship") | 1 | U |  | DCID 21 “Patient Equipment Relationship”
15 | > | CONTAINS | CODE | EV (113743, DCM, "Patient Orientation") | 1 | U |  | DCID 19 “Patient Orientation”
16 | >> | HAS CONCEPT MOD | CODE | EV (113744, DCM, "Patient Orientation Modifier") | 1 | M |  | DCID 20 “Patient Orientation Modifier”
17 | > | CONTAINS | CODE | EV (123014, DCM, "Target Region") | 1 | M |  | DCID 4031 “Common Anatomic Regions”
18 | > | CONTAINS | NUM | EV (122130, DCM, "Dose Area Product") | 1 | MC | IFF TID (10001) Row 2 = (113704, DCM, "Projection X-Ray") | UNITS = EV (Gy.m2, UCUM, "Gy.m2")
19 | > | CONTAINS | NUM | EV (111634, DCM, "Half Value Layer") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
20 | > | CONTAINS | NUM | EV (111638, DCM, "Patient Equivalent Thickness") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
21 | > | CONTAINS | NUM | EV (111636, DCM, "Entrance Exposure at RP") | 1 | MC | IF TID (10001) Row 2 = (P5-40010, SRT, "Mammography") and (TID (10001) Row 9 is absent or value is (R-0038D, SRT, "Yes")) and (TID (10001) Row 10 is absent or value is (R-0038D, SRT, "Yes")) | UNITS = EV (mGy, UCUM, "mGy")
22 | > | CONTAINS | TEXT | EV (113780, DCM, "Reference Point Definition") | 1 | MC | IF Row 21 is present and Row 23 is not present |
23 | > | CONTAINS | CODE | EV (113780, DCM, "Reference Point Definition") | 1 | MC | IF Row 21 is present and Row 22 is not present | DCID 10025 “Radiation Dose Reference Points”
24 | > | CONTAINS | INCLUDE | DTID 4007 “Mammography CAD Breast Composition” | 1 | U |  |
25 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
26 | > | CONTAINS | INCLUDE | DTID 1020 “Person Participant” | 1-n | U |  | $PersonProcedureRole = EV (113851, DCM, "Irradiation Administering")
27 | > | CONTAINS | INCLUDE | DTID 10003A “Irradiation Event X-Ray Detector Data” | 1 | MC | IFF TID (10001) Row 8 is absent or has a value of (R-0038D, SRT, "Yes") |
28 | > | CONTAINS | INCLUDE | DTID 10003B “Irradiation Event X-Ray Source Data” | 1 | MC | IFF TID (10001) Row 9 is absent or has a value of (R-0038D, SRT, "Yes") |
29 | > | CONTAINS | INCLUDE | DTID 10003C “Irradiation Event X-Ray Mechanical Data” | 1 | MC | IFF TID (10001) Row 10 is absent or has a value of (R-0038D, SRT, "Yes") |

Content Item Descriptions

Row 3 | If the image generating entity does not assign a DICOM UID to the irradiation event (e.g., for non-digital imaging equipment), the application generating this report shall assign a UID. In the case of non-integrated cassette-based equipment, a standalone Detector will generate UIDs for the Events it observes. If the X-Ray Source component of the equipment also reports information, it too will generate UIDs for the Events it creates. A downstream system (e.g., a workstation or the Dose Information Reporter itself) may combine the two reports into a composite report, and match up the events based on details such as the time information, and use the UIDs of the X-Ray Source.
Row 6 | Provide DateTime the application of X-Ray started. This shall correspond to the start of the first irradiation in the Irradiation Event, which defines the starting point for the calculation of Row 36 "Irradiation Duration".
Row 17 | The target region is the anatomy exposed.
Row 21 | A text definition of the Reference Point (RP) used for RP-related dose values.
Row 22 | A coded definition of the Reference Point (RP) used for RP-related dose values
Row 25 | People responsible for the administration of the radiation reported in the irradiation event. May include values that would appear in Performing Physicians' Name (0008,1050), Performing Physician Identification Sequence (0008,1052), Operators' Name (0008,1070) and/or Operator Identification Sequence (0008,1072).

### TID 10003A Irradiation Event X-Ray Detector Data

This template contains data that is expected to be available to the X-Ray detector or plate reader component of the equipment.

Type:

  Extensible

Order:

  Non-Significant

Table TID 10003A. Irradiation Event X-Ray Detector Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | EV (113845, DCM, "Exposure Index") | 1 | MC | IF the value is displayable to the X-Ray system operator. | UNITS = EV (1, UCUM, "no units")
2 |  |  | NUM | EV (113846, DCM, "Target Exposure Index") | 1 | MC | IF the value is displayable to the X-Ray system operator. | UNITS = EV (1, UCUM, "no units")
3 |  |  | NUM | EV (113847, DCM, "Deviation Index") | 1 | MC | IF the value is displayable to the X-Ray system operator. | UNITS = EV (1, UCUM, "no units")
4 |  |  | INCLUDE | DTID 1021 “Device Participant” | 1 | U |  | $DeviceProcedureRole = EV (113942, DCM, "X-Ray Reading Device")
5 |  |  | IMAGE | EV (113795, DCM, "Acquired Image") | 1-n | MC | IFF Image Object is created for this irradiation event |

Content Item Descriptions

Row 4 | The device that read the detector of this Irradiation Event. E.g., the CR Plate Reader.
Row 5 | Reference to Image instances created during this event, if any. The UID reference(s) provided here shall be the values at the time the images were initially created. (Note that image UIDs may be changed as the images are managed over a long term.)

### TID 10003B Irradiation Event X-Ray Source Data

This template contains data that is expected to be available to the X-Ray source component of the equipment.

Type:

  Extensible

Order:

  Non-Significant

Table TID 10003B. Irradiation Event X-Ray Source Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | EV (113738, DCM, "Dose (RP) ") | 1 | MC | IF TID (10001) Row 2 = (113704, DCM, "Projection X-Ray") AND any of the values of TID (10001) Row 18 are not (113858, DCM, "MPPS Content") | UNITS = EV (Gy, UCUM, "Gy")
2 |  |  | TEXT | EV (113780, DCM, "Reference Point Definition") | 1 | MC | IF Row 1 is present and Row 3 is not present |
3 |  |  | CODE | EV (113780, DCM, "Reference Point Definition") | 1 | MC | IF Row 1 is present and Row 2 is not present | DCID 10025 “Radiation Dose Reference Points”
4 |  |  | NUM | EV (111631, DCM, "Average Glandular Dose") | 1 | MC | IFF TID (10001) Row 2 = (P5-40010, SRT, "Mammography") | UNITS = EV (mGy, UCUM, "mGy")
5 |  |  | CODE | EV (113732, DCM, "Fluoro Mode") | 1 | UC | IFF TID (10003) Row 7 value = (P5-06000, SRT, "Fluoroscopy") | DCID 10004 “Fluoro Modes”
6 |  |  | NUM | EV (113791, DCM, "Pulse Rate") | 1 | MC | IFF Row 5 value = (113631, DCM, "Pulsed") | UNITS = EV ({pulse}/s, UCUM, "pulse/s")
7 |  |  | NUM | EV (113768, DCM, "Number of Pulses") | 1 | MC | IFF Row 5 is not present or Row 5 is present and equals (113631, DCM, "Pulsed") | UNITS = EV (1, UCUM, "no units")
8 | > | HAS CONCEPT MOD | CODE | EV (121401, DCM, "Derivation") | 1 | MC | IFF count of pulses in Row 7 is estimated | EV (R-10260, SRT, "Estimated")
9 |  |  | NUM | EV (113793, DCM, "Pulse Width") | 1-n | U |  | UNITS = EV (ms, UCUM, "ms")
10 |  |  | NUM | EV (113742, DCM, "Irradiation Duration") | 1 | U |  | UNITS = EV (s, UCUM, "s")
11 |  |  | NUM | EV (113733, DCM, "KVP") | 1-n | M |  | UNITS = EV (kV, UCUM, "kV")
12 |  |  | NUM | EV (113734, DCM, "X-Ray Tube Current") | 1-n | MC | IF Row 15 is not present | UNITS = EV (mA, UCUM, "mA")
13 |  |  | NUM | EV (113767, DCM, "Average X-Ray Tube Current") | 1 | U |  | UNITS = EV (mA, UCUM, "mA")
14 |  |  | NUM | EV (113824, DCM, "Exposure Time") | 1 | MC | IF Row 15 is not present | UNITS = EV (ms, UCUM, "ms")
15 |  |  | NUM | EV (113736, DCM, "Exposure") | 1-n | MC | IF Row 12 and 14 are not present | UNITS = EV (uA.s, UCUM, "uA.s")
16 |  |  | NUM | EV (113766, DCM, "Focal Spot Size") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
17 |  |  | CODE | EV (111632, DCM, "Anode Target Material") | 1 | U |  | DCID 10016 “Anode Target Material”
18 |  |  | CONTAINER | EV (113771, DCM, "X-Ray Filters") | 1-n | U |  |
19 | > | CONTAINS | CODE | EV (113772, DCM, "X-Ray Filter Type") | 1 | U |  | DCID 10007 “X-Ray Filter Types”
20 | > | CONTAINS | CODE | EV (113757, DCM, "X-Ray Filter Material") | 1 | U |  | DCID 10006 “X-Ray Filter Materials”
21 | > | CONTAINS | NUM | EV (113758, DCM, "X-Ray Filter Thickness Minimum") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
22 | > | CONTAINS | NUM | EV (113773, DCM, "X-Ray Filter Thickness Maximum") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
23 |  |  | NUM | EV (113790, DCM, "Collimated Field Area") | 1 | U |  | UNITS = EV (m2, UCUM, "m2")
24 |  |  | NUM | EV (113788, DCM, "Collimated Field Height") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
25 |  |  | NUM | EV (113789, DCM, "Collimated Field Width") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
26 |  |  | CODE | EV (111635, DCM, "X-Ray Grid") | 1-n | U |  | DCID 10017 “X-Ray Grid”
27 |  |  | INCLUDE | DTID 1021 “Device Participant” | 1 | MC | Required if the irradiating device is not the recording device. | $DeviceProcedureRole = EV (113859, DCM, "Irradiating Device")

Content Item Descriptions

Row 1 | Dose applied by this irradiation event, relative to defined reference point.
Row 7 | If a precise count of pulses is not available, an estimated number shall be provided, and the Row 8 Concept Modifier shall indicate "Estimated"
Row 9 | Pulse width as measured/recorded by the system, either as a single total value, or as multiple values. If multiple values are provided, their number shall match the value in Row 7 "Number of Pulses".
Row 11 | KVP value as measured/recorded by system, either as a single mean value, or as multiple values. If multiple values are provided, their number shall match the value in Row 7 "Number of Pulses".
Row 12 | Tube current as measured/recorded by system, either as a single mean value, or as multiple values. If multiple values are provided, their number shall match the value in Row 7 "Number of Pulses".
Row 14 | Exposure time as measured/recorded by the system.
Row 15 | Exposure as measured/recorded by system, either as a single total value, or as multiple values. If multiple values are provided, their number shall match the value in Row 7 "Number of Pulses". The Exposure will be affected by the shape of the pulse and other factors, and may not be a simple multiplication of tube current and exposure time.
Row 18 | If one or more Filter(s) were applied during this irradiation event
Row 23 | Collimated area at the receptor plane.
Row 27 | The device that produced the irradiation in this Irradiation Event. I.e., the X-Ray source. This is not required to be present if the information is the same as that already recorded in TID 1004 “Device Observer Identifying Attributes” encoded via the inclusion of TID 1002 “Observer Context” in TID 10001 “Projection X-Ray Radiation Dose” Row 5, which in turn may be absent if identical to the content in the Enhanced General Equipment Module.

### TID 10003C Irradiation Event X-Ray Mechanical Data

This template contains data that is expected to be available to the gantry or mechanical component of the equipment.

Type:

  Extensible

Order:

  Non-Significant

Table TID 10003C. Irradiation Event X-Ray Mechanical Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (113956, DCM, "CR/DR Mechanical Configuration") | 1 | U |  | DCID 10031 “CR/DR Mechanical Configuration”
2 |  |  | NUM | EV (112011, DCM, "Positioner Primary Angle") | 1 | UC | XOR Row 6 | UNITS = EV (deg, UCUM, "deg")
3 |  |  | NUM | EV (112012, DCM, "Positioner Secondary Angle") | 1 | UC | XOR Row 6 | UNITS = EV (deg, UCUM, "deg")
4 |  |  | NUM | EV (113739, DCM, "Positioner Primary End Angle") | 1 | UC | IFF TID (10003) Row 7 value = (113613, DCM, "Rotational Acquisition") | UNITS = EV (deg, UCUM, "deg")
5 |  |  | NUM | EV (113740, DCM, "Positioner Secondary End Angle") | 1 | UC | IFF TID (10003) Row 7 value = (113613, DCM, "Rotational Acquisition") | UNITS = EV (deg, UCUM, "deg")
6 |  |  | NUM | EV (113770, DCM, "Column Angulation") | 1 | UC | XOR Rows 2, 3 | UNITS = EV (deg, UCUM, "deg")
7 |  |  | NUM | EV (113754, DCM, "Table Head Tilt Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
8 |  |  | NUM | EV (113755, DCM, "Table Horizontal Rotation Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
9 |  |  | NUM | EV (113756, DCM, "Table Cradle Tilt Angle") | 1 | U |  | UNITS = EV (deg, UCUM, "deg")
10 |  |  | NUM | EV (111633, DCM, "Compression Thickness") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
11 |  |  | NUM | DCID 10008 “Dose Related Distance Measurements” | 1-n | U |  | UNITS = EV (mm, UCUM, "mm")

Content Item Descriptions

Row 2 | Angle in patient's "equatorial" plane (LAO to RAO). For dynamically changing angle during the event, the start value shall be provided. Equivalent to (0018,1510) in an image instance.
Row 3 | Angle in patient's "sagittal" plane (CRAN to CAUD). For dynamically changing angle during the event, the start value shall be provided. Equivalent to (0018,1511) in an image instance.
Row 4 | In case of motion during irradiation event, Positioner Primary ending angle
Row 5 | In case of motion during irradiation event., Positioner Secondary ending angle
Row 6 | Column device Angle in equipment based coordinates

### TID 10004 Accumulated Fluoroscopy and Acquisition Projection X-Ray Dose

This general template provides detailed information on projection X-Ray dose value accumulations over several irradiation events from the same equipment (typically a study or a performed procedure step).

Type:

  Extensible

Order:

  Non-Significant

Table TID 10004. Accumulated Fluoroscopy and Acquisition Projection X-Ray Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | EV (113726, DCM, "Fluoro Dose Area Product Total") | 1 | MC | IFF TID (10003) Row 7 value = (P5-06000, SRT, "Fluoroscopy") for at least one irradiation event | UNITS = EV (Gy.m2, UCUM, "Gy.m2")
2 |  |  | NUM | EV (113728, DCM, "Fluoro Dose (RP) Total") | 1 | MC | IFF TID (10003) Row 7 value = (P5-06000, SRT, "Fluoroscopy") for at least one irradiation event AND any of the values of TID (10001) Row 18 are not (113858, DCM, "MPPS Content"). | UNITS = EV (Gy, UCUM, "Gy")
3 |  |  | NUM | EV (113730, DCM, "Total Fluoro Time") | 1 | MC | IFF TID (10003) Row 7 value = (P5-06000, SRT, "Fluoroscopy") for at least one irradiation event. | UNITS = EV (s, UCUM, "s")
4 |  |  | NUM | EV (113727, DCM, "Acquisition Dose Area Product Total") | 1 | M |  | UNITS = EV (Gy.m2, UCUM, "Gy.m2")
5 |  |  | NUM | EV (113729, DCM, "Acquisition Dose (RP) Total") | 1 | MC | IF any of the values of TID (10001) Row 18 are not (113858, DCM, "MPPS Content"). May be present otherwise. | UNITS = EV (Gy, UCUM, "Gy")
6 |  |  | NUM | EV (113855, DCM, "Total Acquisition Time") | 1 | M |  | UNITS = EV (s, UCUM, "s")

Content Item Descriptions

Rows 1-3 | Fluoroscopic component only
Row 3 | Total clock time of Fluoroscopy accumulated over the defined scope of accumulation (i.e., the sum of the Irradiation Duration values for accumulated fluoroscopy irradiation events)
Rows 4-6 | Acquisition component only
Row 6 | Total clock time of acquisitions accumulated over the defined scope of accumulation (i.e., the sum of the Irradiation Duration values for accumulated acquisition irradiation events)

### TID 10005 Accumulated Mammography X-Ray Dose

This modality specific template provides detailed information on breast imaging projection X-Ray dose value accumulations over several irradiation events from the same equipment (typically a study or a performed procedure step).

Type:

  Extensible

Order:

  Significant

Table TID 10005. Accumulated Mammography X-Ray Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | EV (111637, DCM, "Accumulated Average Glandular Dose") | 1-2 | M |  | UNITS = EV (mGy, UCUM, "mGy")
2 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | M |  | DCID 6022 “Side”

### TID 10006 Accumulated Cassette-based Projection Radiography Dose

This template provides information on Projection Radiography dose values accumulated on Cassette-based systems over one or more irradiation events (typically a study or a performed procedure step) from the same equipment.

Type:

  Extensible

Order:

  Non-Significant

Table TID 10006. Accumulated Cassette-Based Projection Radiography Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CODE | EV (113947, DCM, "Detector Type") | 1 | MC | IF TID (10001) Row 8 is absent or value is (R-0038D, SRT, "Yes") | DCID 10030 “Detector Types”
2 |  |  | NUM | EV (113731, DCM, "Total Number of Radiographic Frames") | 1 | MC | IF TID (10001) Row 8 is absent or value is (R-0038D, SRT, "Yes") | UNITS = EV (1, UCUM, "no units")

Content Item Descriptions

Row 2 | The number of radiographic frames recorded by the X-Ray detector or the number of exposures recorded by the X-Ray source, whichever is known to be greater.

### TID 10007 Accumulated Total Projection Radiography Dose

This template provides information on total Projection Radiography dose values accumulated on Integrated or combined fluoroscopy/acquisition systems over one or more irradiation events (typically a study or a performed procedure step) from the same equipment.

Type:

  Extensible

Order:

  Non-Significant

Table TID 10007. Accumulated Total Projection Radiography Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | EV (113722, DCM, "Dose Area Product Total") | 1 | M |  | UNITS = EV (Gy.m2, UCUM, "Gy.m2")
2 |  |  | NUM | EV (113725, DCM, "Dose (RP) Total") | 1 | MC | IF TID (10001) Row 4 = (113958, DCM, "Integrated Projection Radiography System") or any of the values of TID (10001) Row 18 are not (113858, DCM, "MPPS Content"). | UNITS = EV (Gy, UCUM, "Gy")
3 |  |  | NUM | EV (113737, DCM, "Distance Source to Reference Point") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
4 |  |  | NUM | EV (113731, DCM, "Total Number of Radiographic Frames") | 1 | U |  | UNITS = EV (1, UCUM, "no units")
5 |  |  | CODE | EV (113780, DCM, "Reference Point Definition") | 1 | MC | IF any of (113725, DCM, "Dose (RP) Total"), (113728, DCM, "Fluoro Dose (RP) Total") or (113729, DCM, "Acquisition Dose (RP) Total") are present, and Row 6 is not present. | DCID 10025 “Radiation Dose Reference Points”
6 |  |  | TEXT | EV (113780, DCM, "Reference Point Definition") | 1 | MC | IF any of (113725, DCM, "Dose (RP) Total"), (113728, DCM, "Fluoro Dose (RP) Total") or (113729, DCM, "Acquisition Dose (RP) Total") are present, and Row 5 is not present. |

Content Item Descriptions

Row 1 | Accumulated Dose Area Product
Row 2 | Accumulated dose relative to reference point.
Row 3 | A single value for Radiography systems calculating reference point dose based on fixed distance.
Row 5 | A coded definition of the Reference Point (RP) used for RP-related dose values.
Row 6 | A text definition of the Reference Point (RP) used for RP-related dose values.

## CT Radiation Dose SR IOD Templates

The templates that comprise the CT Radiation Dose SR are interconnected as in Figure A-15.

Figure A-15. CT Radiation Dose SR IOD Template Structure

### TID 10011 CT Radiation Dose

This template defines a container (the root) with subsidiary content items, each of which corresponds to a single CT X-Ray irradiation event entry. There is a defined recording observer (the system or person responsible for recording the log, generally the system). Accumulated values shall be kept for a whole Study or at least a part of a Study, if the Study is divided in the workflow of the examination, or a performed procedure step. Multiple CT Radiation Dose objects may be created for one Study.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 10011. CT Radiation Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113701, DCM, "X-Ray Radiation Dose Report") | 1 | M |  | Root node
2 | > | HAS CONCEPT MOD | CODE | EV (121058, DCM, "Procedure reported") | 1 | M |  | EV (P5-08000, SRT, "Computed Tomography X-Ray")
3 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E8, SRT, "Has Intent") | 1 | M |  | DCID 3629 “Procedure Intent”
4 | > |  | INCLUDE | DTID 1002 “Observer Context” | 1-n | M |  |
5 | > | HAS OBS CONTEXT | DATETIME | EV (113809, DCM, "Start of X-Ray Irradiation") | 1 | M |  |
6 | > | HAS OBS CONTEXT | DATETIME | EV (113810, DCM, "End of X-Ray Irradiation") | 1 | M |  |
7 | > | HAS OBS CONTEXT | CODE | EV (113705, DCM, "Scope of Accumulation") | 1 | M |  | DCID 10000 “Scope of Accumulation”
8 | >> | HAS PROPERTIES | UIDREF | DCID 10001 “UID Types” | 1 | M |  |
9 | > | CONTAINS | INCLUDE | DTID 10012 “CT Accumulated Dose Data” | 1 | M |  |
10 | > | CONTAINS | INCLUDE | DTID 10013 “CT Irradiation Event Data” | 1-n | M |  |
11 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
12 | > | CONTAINS | CODE | EV (113854, DCM, "Source of Dose Information") | 1-n | M |  | DCID 10021 “Source of CT Dose Information”
13 | > | CONTAINS | INCLUDE | DTID 1020 “Person Participant” | 1 | U |  | $PersonProcedureRole = EV (113850, DCM, "Irradiation Authorizing")

Content Item Descriptions

Row 4 | The observer context may include both a Person Observer identification, as well as the identity of the equipment providing the values for the irradiation event (Device Observer identification), if not inherited.
Row 5 | Start, Date Time of the first CT Irradiation Event of the accumulation
Row 6 | End, Date Time of the last CT Irradiation Event of the accumulation
Row 12 | The primary source of information from which this dose object was constructed.
Row 13 | The physician responsible for determining that the irradiating procedure was appropriate for the indications. The value may come from Requesting Physician (0032,1032), Requesting Physician Identification Sequence (0032,1031) or somewhere else based on hospital policies.

### TID 10012 CT Accumulated Dose Data

This general template provides detailed information on CT X-Ray dose value accumulations over several irradiation events from the same equipment and over the scope of accumulation specified for the report (typically a Study or a Performed Procedure Step).

Type:

  Extensible

Order:

  Significant

Table TID 10012. CT Accumulated Dose Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113811, DCM, "CT Accumulated Dose Data") | 1 | M |  |
2 | > | CONTAINS | NUM | EV (113812, DCM, "Total Number of Irradiation Events") | 1 | M |  | UNITS = EV ({events}, UCUM, "events")
3 | > | CONTAINS | NUM | EV (113813, DCM, "CT Dose Length Product Total") | 1 | M |  | UNITS = EV (mGy.cm, UCUM, "mGy.cm")
4 | > | CONTAINS | NUM | EV (113814, DCM, "CT Effective Dose Total") | 1 | U |  | UNITS = EV (mSv, UCUM, "mSv")
5 | >> | HAS PROPERTIES | TEXT | EV (121406, DCM, "Reference Authority") | 1 | MC | XOR row 6 |
6 | >> | HAS PROPERTIES | CODE | EV (121406, DCM, "Reference Authority") | 1 | MC | XOR row 5 | DCID 10015 “CT Dose Reference Authorities”
7 | >> | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | M |  | DCID 10011 “Effective Dose Evaluation Method”
8 | >> | HAS PROPERTIES | TEXT | EV (113815, DCM, "Patient Model") | 1 | MC | IF the value of row 7 equals (113800, DCM, "DLP to E conversion via MC computation") or equals (113801, DCM, "CTDIfreeair to E conversion via MC computation") |
9 | >> | HAS PROPERTIES | CONTAINER | EV (113816, DCM, "Condition Effective Dose measured") | 1 | MC | IF the value of row 7 equals (113802, DCM, "DLP to E conversion via measurement") or equals (113803, DCM, "CTDIfreeair to E conversion via measurement") |
10 | >>> | CONTAINS | TEXT | EV (113817, DCM, "Effective Dose Phantom Type") | 1 | M |  |
11 | >>> | CONTAINS | TEXT | EV (113818, DCM, "Dosimeter Type") | 1 | M |  |
12 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
13 | > | CONTAINS | INCLUDE | DTID 1021 “Device Participant” | 1 | MC | Required if the irradiating device is not the recording device and the dose was accumulated on a single device. | $DeviceProcedureRole = EV (113859, DCM, "Irradiating Device")

Content Item Descriptions

Rows 5, 6 | Total Number of CT irradiation events. A CT irradiation event is one continuous irradiation procedure and is defined through consistent acquisition parameters. In the case of dose modulation the calculations are based on the effective parameters (e.g., the effective mA recorded in the Mean X-Ray Tube Current), and these acquisition parameters are consistent.
Rows 5, 6 | The Dose Length Product (DLP) is calculated for every irradiation event. The Dose Length Product Total is the sum of the DLP values. The calculation is based on the CTDIvol result of each irradiation event.
Rows 5, 6 | Effective dose (E, in units of mSv) evaluated as a total over the scope is defined in Row 6 of template TID 10011 “CT Radiation Dose”. Effective dose is defined by the reference in Rows 5 or 6 of this template. It may be calculated from a product of DLP and an 'Effective Dose Conversion Factor' (E/DLP). Or it may be calculated from a product of the Mean CTDIfreeair and the ratio E/CTDIfreeair. The ratios E/DLP or E/CTDIfreeair may be evaluated either from computer simulations applying Monte Carlo (MC) sampling techniques or from dosimetric measurements in an anthropomorphic phantom, e.g., the Alderson-Rando phantom.. The specific method used is identified in Rows 7 through 11.
Row 5 - 6 | Reference of the base publication defining the Effective Dose, either as a coded value, or a textual bibliographic reference. ICRP Publications shall be referenced using their assigned coded values.
Row 7 | Description of the method used for Effective Dose evaluations.
Row 8 | Description of the reference-patient mathematical or computational model used when Effective Dose is derived via Monte Carlo simulations of radiation transport in such models. Examples of publications that specify particular reference patient models are NUREG/CR-1159, ORNL/NUREG/TM-367 (1980); NRPB-R186 (1985); GSF-Bericht S-885 (1986); Fill et al., Health Physics Vol. 86 (3): 253-272 (2004).
Row 9 | Description of the condition Effective Dose measured
Row 10 | Type of Effective Dose phantom used, e.g., Alderson-Rando
Row 11 | Type of dosimeter used, e.g., TLD (Thermo Luminescence Dosimeter)
Row 13 | The device that produced the irradiation accumulated in this template. I.e., the CT Scanner. This is not required to be present if the information is the same as that already recorded in TID 1004 “Device Observer Identifying Attributes” encoded via the inclusion of TID 1002 “Observer Context” in TID 10011 “CT Radiation Dose” Row 4, which in turn may be absent if identical to the content in the Enhanced General Equipment Module, or if more than one device produced the accumulated irradiation.

### TID 10013 CT Irradiation Event Data

This template conveys the dose and equipment parameters of a single irradiation event.

A CT irradiation event is the loading of X-Ray equipment caused by a single continuous actuation of the equipment's irradiation switch, from the start of the loading time of the first pulse until the loading time trailing edge of the final pulse. Any on-off switching of the radiation source during the event shall not be treated as separate events; rather the event includes the time between start and stop of radiation as triggered by the user, e.g., a single sequence of scanning comprised of multiple slices acquired with successive tube rotations and table increments shall be treated as a single irradiation event. Depending on the examination workflow and the anatomical target region the CT irradiation event data may split into multiple instances of this template for better dose estimation. The irradiation event is the "smallest" information entity to be recorded in the realm of Radiation Dose reporting. Individual Irradiation Events are described by a set of accompanying physical parameters that are sufficient to understand the "quality" of irradiation that is being applied. This set of parameters may be different for the various types of equipment that are able to create irradiation events.

Type:

  Extensible

Order:

  Significant

Table TID 10013. CT Irradiation Event Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113819, DCM, "CT Acquisition") | 1 | M |  |
2 | > | CONTAINS | TEXT | EV (125203, DCM, "Acquisition Protocol") | 1 | U |  |
3 | > | CONTAINS | CODE | EV (123014, DCM, "Target Region") | 1 | M |  | DCID 4030 “CT, MR and PET Anatomy Imaged”
4 | > | CONTAINS | CODE | EV (113820, DCM, "CT Acquisition Type") | 1 | M |  | DCID 10013 “CT Acquisition Type”
4b | >> | CONTAINS | CODE | EV (113961, DCM, "Reconstruction Algorithm") | 1-n | U |  | DCID 10033 “CT Reconstruction Algorithm”
5 | > | CONTAINS | CODE | EV (G-C32C, SRT, "Procedure Context") | 1 | U |  | DCID 10014 “Contrast Imaging Technique”
6 | > | CONTAINS | UIDREF | EV (113769, DCM, "Irradiation Event UID") | 1 | M |  |
6b | > | CONTAINS | TEXT | EV (113605, DCM, "Irradiation Event Label") | 1 | U |  |
6c | >> | HAS CONCEPT MOD | CODE | EV (113606, DCM, "Label Type") | 1 | MC | IF the value of Row 6b is the value of an Attribute in the images. | DCID 10022 “Label Types”
7 | > | CONTAINS | CONTAINER | EV (113822, DCM, "CT Acquisition Parameters") | 1 | M |  |
8 | >> | CONTAINS | NUM | EV (113824, DCM, "Exposure Time") | 1 | M |  | UNITS = EV (s, UCUM, "s")
9 | >> | CONTAINS | INCLUDE | DTID 10014 “Scanning Length” | 1 | M |  |
10 | >> | CONTAINS | NUM | EV (113826, DCM, "Nominal Single Collimation Width") | 1 | M |  | UNITS = EV (mm, UCUM, "mm")
11 | >> | CONTAINS | NUM | EV (113827, DCM, "Nominal Total Collimation Width") | 1 | M |  | UNITS = EV (mm, UCUM, "mm")
12 | >> | CONTAINS | NUM | EV (113828, DCM, "Pitch Factor") | 1 | MC | IF row 4 equals (P5-08001, SRT, "Spiral Acquisition") or equals (113804, DCM, "Sequenced Acquisition") | UNITS = EV ({ratio}, UCUM, "ratio")
13 | >> | CONTAINS | NUM | EV (113823, DCM, "Number of X-Ray Sources") | 1 | M |  | UNITS = EV ({X-Ray sources}, UCUM, "X-Ray sources")
14 | >> | CONTAINS | CONTAINER | EV (113831, DCM, "CT X-Ray Source Parameters") | 1-n | M |  |
15 | >>> | CONTAINS | TEXT | EV (113832, DCM, "Identification of the X-Ray Source") | 1 | M |  |
16 | >>> | CONTAINS | NUM | EV (113733, DCM, "KVP") | 1 | M |  | UNITS = EV (kV, UCUM, "kV")
17 | >>> | CONTAINS | NUM | EV (113833, DCM, "Maximum X-Ray Tube Current") | 1 | M |  | UNITS = EV (mA, UCUM, "mA")
18 | >>> | CONTAINS | NUM | EV (113734, DCM, "X-Ray Tube Current") | 1 | M |  | UNITS = EV (mA, UCUM, "mA")
19 | >>> | CONTAINS | NUM | EV (113834, DCM, "Exposure Time per Rotation") | 1 | MC | IF row 4 does not equal (113805, DCM, "Constant Angle Acquisition") | UNITS = EV (s, UCUM, "s")
20 | >>> | CONTAINS | NUM | EV (113821, DCM, "X-Ray Filter Aluminum Equivalent") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
21 | > | CONTAINS | CONTAINER | EV (113829, DCM, "CT Dose") | 1 | MC | IF row 4 does not equal (113805, DCM, "Constant Angle Acquisition") |
22 | >> | CONTAINS | NUM | EV (113830, DCM, "Mean CTDIvol") | 1 | M |  | UNITS = EV (mGy, UCUM, "mGy")
23 | >> | CONTAINS | CODE | EV (113835, DCM, "CTDIw Phantom Type") | 1 | M |  | DCID 4052 “Phantom Devices”
24 | >> | CONTAINS | NUM | EV (113836, DCM, "CTDIfreeair Calculation Factor") | 1 | U |  | UNITS = EV (mGy/mA.s, UCUM, "mGy/mA.s")
25 | >> | CONTAINS | NUM | EV (113837, DCM, "Mean CTDIfreeair") | 1 | U |  | UNITS = EV (mGy, UCUM, "mGy")
26 | >> | CONTAINS | NUM | EV (113838, DCM, "DLP") | 1 | M |  | UNITS = EV (mGy.cm, UCUM, "mGy.cm")
27 | >> | CONTAINS | NUM | EV (113839, DCM, "Effective Dose") | 1 | U |  | UNITS = EV (mSv, UCUM, "mSv")
28 | >>> | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | MC | IF row 27 is present | DCID 10011 “Effective Dose Evaluation Method”
29 | >>>> | HAS PROPERTIES | NUM | EV (113840, DCM, "Effective Dose Conversion Factor") | 1 | MC | IF row 28 is present and equals (113800, DCM, "DLP to E conversion via MC computation") or equals (113802, DCM, "DLP to E conversion via measurement") | UNITS = EV (mSv/mGy.cm, UCUM, "mSv/mGy.cm")
30 | >> | CONTAINS | NUM | EV (113930, DCM, "Size Specific Dose Estimation") | 1-n | U |  | UNITS = EV (mGy, UCUM, "mGy")
31 | >>> | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | M |  | DCID 10023 “Size Specific Dose Estimation Method for CT”
32 | >>>> | INFERRED FROM | NUM | EV (113931, DCM, "Measured Lateral Dimension") | 1 | MC | IF row 31 equals (113934, DCM, "AAPM 204 Lateral Dimension") or (113936, DCM, "AAPM 204 Sum of Lateral and AP Dimension") | UNITS = EV (mm, UCUM, "mm")
33 | >>>> | INFERRED FROM | NUM | EV (113932, DCM, "Measured AP Dimension") | 1 | MC | IF row 31 equals (113935, DCM, "AAPM 204 AP Dimension") or (113936, DCM, "AAPM 204 Sum of Lateral and AP Dimension") | UNITS = EV (mm, UCUM, "mm")
34 | >>>> | INFERRED FROM | NUM | EV (113933, DCM, "Derived Effective Diameter") | 1 | MC | IF row 31 equals (113934, DCM, "AAPM 204 Lateral Dimension") or (113935, DCM, "AAPM 204 AP Dimension") or (113936, DCM, "AAPM 204 Sum of Lateral and AP Dimension") or (113937, DCM, "AAPM 204 Effective Diameter Estimated From Patient Age") | UNITS = EV (mm, UCUM, "mm")
35 | >> | CONTAINS | INCLUDE | DTID 10015 “CT Dose Check Details” | 1 | M |  |
36 | > | CONTAINS | TEXT | EV (113842, DCM, "X-Ray Modulation Type") | 1 | U |  |
37 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |
38 | > | CONTAINS | INCLUDE | DTID 1020 “Person Participant” | 1-n | U |  | $PersonProcedureRole = EV (113851, DCM, "Irradiation Administering")
39 | > | CONTAINS | INCLUDE | DTID 1021 “Device Participant” | 1 | MC | Required if the irradiating device is not the recording device. | $DeviceProcedureRole = EV (113859, DCM, "Irradiating Device")

Content Item Descriptions

Row 2 | User-defined type of clinical acquisition protocol for creating images or image-derived measurements. May be taken from Protocol Name (0018,1030) or from Performed Procedure Step Description (0040,0254).
Row 3 | The target region is the anatomy exposed.
Row 4 | Description of the method used during acquisition of this CT irradiation event, may be derived from Acquisition Type (0018,9302).
Row 4b | Though not a characteristic of the acquisition per se, the type of reconstruction intended has a bearing on the technique used. If multiple types of reconstruction are performed, multiple values can be listed. These values should correspond to the values of Reconstruction Algorithm (0018,9315) in the reconstructed images.
Row 5 | The acquisition was performed with or without contrast medium application.
Row 8 | Total time the patient has received X-Ray exposure during the irradiation event.
Row 10 | The value of the nominal width (referenced to the location of the isocenter along the z axis) of a single collimated slice in mm.
Row 11 | The value of the nominal width (referenced to the location of the isocenter along the z axis) of the nominal total collimation in mm over the area of active X-Ray detection (z-coverage).
Row 12 | Pitch Factor: For Spiral Acquisition, the Pitch Factor is the ratio of the Table Feed per Rotation to the Nominal Total Collimation Width. For Sequenced Acquisition, the Pitch Factor is the ratio of the Table Feed per single sequenced scan to the Nominal Total Collimation Width.
Row 14 | CT X-Ray source parameters related to the acquisition. For each X-Ray source an item must be present. For multi-energy acquisitions, multiple items may be present for each X-Ray source, each item describing one energy level.
Row 15 | Identification of the X-Ray source. Identifies the particular X-Ray source (in a multi-source CT system) for which the set of X-Ray source parameter values is reported.
Row 16 | KVP value as measured/recorded by system.
Row 18 | Mean tube current as measured/recorded by system.
Row 19 | Exposure time as measured/recorded by the system per rotation.
Row 20 | Thickness of an equivalent filter constructed from aluminum, in case of multi-source CT systems AND if Row 4 is not present
Row 21 | CT Dose for one acquisition
Rows 5, 6 | "Mean CTDIvol" refers to the average value of the CTDIvol applied within this acquisition. CTDIvol is the volume CTDIw, where CTDIw is the weighted computed tomography dose index 100 as defined in IEC 60601-2-44. For Sequenced and Spiral scanning, CTDIvol = CTDIw / Pitch Factor. For Stationary and Free scanning, CTDIvol = CTDIw ✕ Cumulative Exposure Time/ Exposure Time Per Rotation. According to IEC 60601-2-44 Ed 3 for Constant Angle Acquisition may be calculated as CTDIvol = (CTDIw / Current Time Product (mAs)) ✕ X-Ray Tube Current (mA) x (Nominal Total Collimation Width (mm) / Table Speed (mm/s)). ### Note The ratio CTDIw / Current Time Product is evaluated independently of the Constant Angle Acquisition but with the same settings of tube voltage and Total Collimation Width as those of the Constant Angle Acquisition. See also CTDIvol (0018,9345) and Spiral Pitch Factor (0018,9311) in the “Enhanced Computed Tomography Image IOD” in PS3.3 .
Row 23 | The type of phantom used for CTDI measurement according to IEC 60601-2-44(e.g., Head 16 cm diameter PMMA, Body 32 cm diameter PMMA).
Rows 5, 6 | The CTDIfree air Calculation Factor is the CTDIfree air per mAs, expressed in units of mGy/mAs. The CTDIfree air Calculation Factor may be used in one method calculating Dose. For example, for this acquisition, Effective Dose = Mean X-Ray Tube Current ✕ Cumulative Exposure Time ✕ CTDIfree air Calculation Factor ✕ (Effective Dose/ CTDIfree air).
Rows 5, 6 | MeanCTDIfree air is the mean CTDI for this acquisition, evaluated free-in-air according to IEC 60601-2-44. MeanCTDIfree air = Mean X-Ray Tube Current ✕ Cumulative Exposure Time ✕ CTDIfree air Calculation Factor. The CTDIfree air may be used in one method of calculating Effective Dose.
Rows 5, 6 | For Spiral scanning, DLP = CTDIvol ✕ Scanning Length. For Sequenced scanning, DLP = CTDIvol ✕ Nominal Total Collimation Width ✕ Cumulative Exposure Time / Exposure Time per Rotation. For Stationary and Free scanning, DLP = CTDIvol ✕ Nominal Total Collimation Width (according to IEC 60601-2-44).
Row 27 | Effective Dose in mSv of the single continuous time-frame of the irradiation computed as described in TID 10012 “CT Accumulated Dose Data”.
Row 29 | The Effective Dose Conversion Factor is the ratio of the Effective Dose to the DLP, expressed in units of mSv/mGy.cm, and it is used as a factor in one method of estimating Effective Dose. Monte Carlo Simulations (or dosimetric measurements in an anthropomorphic phantom, e.g., the Alderson-Rando phantom) may be used as a basis for the evaluation of Effective Dose Conversion Factors.
Row 30 | More than one Size Specific Dose Estimation may be included, for example if different computation methods are used.
Row 31 | The methods of AAPM Report 204 are listed in CID 10023 “Size Specific Dose Estimation Method for CT”; other methods may be used. The phantom size (16cm or 32cm) used for the calculation is available from the phantom type defined in Row 23.
Row 32 | The condition specifies inclusion of the Measured Lateral Dimension if it was used in the calculation.
Row 33 | The condition specifies inclusion of the Measured AP Dimension if it was used in the calculation.
Row 34 | The Derived Effective Diameter is conditionally included, whether it was derived from measurements or estimated from age, but may not be used for other (non-AAPM Report 204) methods.
Row 35 | Record of details associated with using the NEMA Dose Check Standard (NEMA XR-25-2010).
Row 36 | The type of exposure modulation. May use the value of Exposure Modulation Type (0018,9323) from CT Exposure Macro or from CT Image Module.
Row 38 | People responsible for the administration of the radiation reported in the irradiation event. May include values that would appear in Performing Physicians' Name (0008,1050), Performing Physician Identification Sequence (0008,1052), Operators' Name (0008,1070) and/or Operator Identification Sequence (0008,1072).
Row 39 | The device that produced the irradiation in this Irradiation Event. I.e., the CT scanner. This is not required to be present if the information is the same as that already recorded in the TID 1004 “Device Observer Identifying Attributes” encoded via the inclusion of TID 1002 “Observer Context” in TID 10011 “CT Radiation Dose” Row 4, which in turn may be absent if identical to the content in the Enhanced General Equipment Module.

### TID 10014 Scanning Length

Type:

  Extensible

Order:

  Significant

Table TID 10014. Scanning Length

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | NUM | EV (113825, DCM, "Scanning Length") | 1 | M |  | UNITS = EV (mm, UCUM, "mm")
2 |  |  | NUM | EV (113893, DCM, "Length of Reconstructable Volume") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
3 |  |  | NUM | EV (113899, DCM, "Exposed Range") | 1 | UC | IFF TID 10013 “CT Irradiation Event Data” row 4 CT Acquisition Type equals (P5-08001, SRT, "Spiral Acquisition") | UNITS = EV (mm, UCUM, "mm")
4 |  |  | NUM | EV (113895, DCM, "Top Z Location of Reconstructable Volume") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
5 |  |  | NUM | EV (113896, DCM, "Bottom Z Location of Reconstructable Volume") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
6 |  |  | NUM | EV (113897, DCM, "Top Z Location of Scanning Length") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
7 |  |  | NUM | EV (113898, DCM, "Bottom Z Location of Scanning Length") | 1 | U |  | UNITS = EV (mm, UCUM, "mm")
8 |  |  | UIDREF | EV (112227, DCM, "Frame of Reference UID") | 1 | MC | IF any of Rows 4 through 7 are present. | If present, shall be the same UID as in the images reconstructed from this irradiation event.

Content Item Descriptions

Row 1 | For Spiral scanning, the scanning length is normally the table travel in mm during the tube loading (see Figure A-16). For Sequenced scanning, the scanning length is the table travel between consecutive scans times the number of scans. For Stationary and Free scanning, the scanning length is the nominal width of the total collimation.
Row 2 | The length of the reconstructable volume is the maximum z-range between the outermost edges of the top and bottom slices that can be reconstructed from the acquisition. For Spiral scanning, the length of reconstructable volume is the z-range between the outermost beginning of the first reconstructable slice and the outermost end of the last reconstructable slice (see Figure A-16). For Sequenced scanning, the length of reconstructable volume is the z-range between the outermost beginning of the first slice and the outermost end of the last slice (i.e., including any skip). For Stationary and Free scanning, the length of reconstructable volume is the nominal width of the total collimation.
Row 3 | For Spiral scanning, the exposed range is as defined in IEC 60601-2-44 (Ed. 3) 302.115(b) (see Figure A-16). Exposed range is not defined for other modes of scanning.
Rows 4-5 | The Top and Bottom Z Locations of the Reconstructable Volume are independent of the slice width of any actual reconstructed slices. They are measured from the edges of the volume, and hence are not equal to the Z locations encoded in the images of any actual reconstructed slices, which are recorded as the center of the slice.
Rows 4-7 | These locations are patient (not table or gantry) relative, to allow them to be defined in the Patient Coordinate System and hence related to the Image Position (Patient) in the reconstructed images. They are also defined in terms of the top (towards the patient's head), and bottom (towards the patient's feet) of the corresponding ranges, in order to make them independent of whether the scan starts at the top or the bottom or shuttles back and forth in between (see Figure A-16).

Figure A-16. Spiral Acquisition Parameters

### TID 10015 CT Dose Check Details

This template records details related to the use of the NEMA Dose Check Standard (NEMA XR-25-2010).

Type:

  Extensible

Order:

  Significant

Table TID 10015. CT Dose Check Details

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113900, DCM, "Dose Check Alert Details") | 1 | MC | IF the scanning device has implemented dose alerts |
2 | > | CONTAINS | CODE | EV (113901, DCM, "DLP Alert Value Configured") | 1 | M |  | DCID 230 “Yes-No”
3 | > | CONTAINS | CODE | EV (113902, DCM, "CTDIvol Alert Value Configured") | 1 | M |  | DCID 230 “Yes-No”
4 | > | CONTAINS | NUM | EV (113903, DCM, "DLP Alert Value") | 1 | MC | IFF value of Row 2 is (R-0038D, SRT, "Yes") | UNITS = EV (mGy.cm, UCUM, "mGy.cm")
5 | > | CONTAINS | NUM | EV (113904, DCM, "CTDIvol Alert Value") | 1 | MC | IFF value of Row 3 is (R-0038D, SRT, "Yes") | UNITS = EV (mGy, UCUM, "mGy")
6 | > | CONTAINS | NUM | EV (113905, DCM, "Accumulated DLP Forward Estimate") | 1 | MC | IF Accumulated DLP Forward Estimate (Row 6) exceeds DLP Alert Value (Row 4) | UNITS = EV (mGy.cm, UCUM, "mGy.cm")
7 | > | CONTAINS | NUM | EV (113906, DCM, "Accumulated CTDIvol Forward Estimate") | 1 | MC | IF Accumulated CTDIvol Forward Estimate (Row 7) exceeds CTDIvol Alert Value (Row 5) | UNITS = EV (mGy, UCUM, "mGy")
8 | > | CONTAINS | TEXT | EV (113907, DCM, "Reason for Proceeding") | 1 | UC | IFF Accumulated DLP Forward Estimate (Row 6) exceeds DLP Alert Value (Row 4) or Accumulated CTDIvol Forward Estimate (Row 7) exceeds CTDIvol Alert Value (Row 5) |
9 | > | CONTAINS | INCLUDE | DTID 1020 “Person Participant” | 1 | MC | IF Accumulated DLP Forward Estimate (Row 6) exceeds DLP Alert Value (Row 4) or Accumulated CTDIvol Forward Estimate (Row 7) exceeds CTDIvol Alert Value (Row 5) | $PersonProcedureRole = EV (113850, DCM, "Irradiation Authorizing")
10 |  |  | CONTAINER | EV (113908, DCM, "Dose Check Notification Details") | 1 | MC | IF the scanning device has implemented dose notifications |
11 | > | CONTAINS | CODE | EV (113909, DCM, "DLP Notification Value Configured") | 1 | M |  | DCID 230 “Yes-No”
12 | > | CONTAINS | CODE | EV (113910, DCM, "CTDIvol Notification Value Configured") | 1 | M |  | DCID 230 “Yes-No”
13 | > | CONTAINS | NUM | EV (113911, DCM, "DLP Notification Value") | 1 | MC | IFF value of Row 11 is (R-0038D, SRT, "Yes") | UNITS = EV (mGy.cm, UCUM, "mGy.cm")
14 | > | CONTAINS | NUM | EV (113912, DCM, "CTDIvol Notification Value") | 1 | MC | IFF value of Row 12 is (R-0038D, SRT, "Yes") | UNITS = EV (mGy, UCUM, "mGy")
15 | > | CONTAINS | NUM | EV (113913, DCM, "DLP Forward Estimate") | 1 | MC | IF DLP Forward Estimate (Row 15) exceeds DLP Notification Value (Row 13) | UNITS = EV (mGy.cm, UCUM, "mGy.cm")
16 | > | CONTAINS | NUM | EV (113914, DCM, "CTDIvol Forward Estimate") | 1 | MC | IF CTDIvol Forward Estimate (Row 16) exceeds CTDIvol Notification Value (Row 14) | UNITS = EV (mGy, UCUM, "mGy")
17 | > | CONTAINS | TEXT | EV (113907, DCM, "Reason for Proceeding") | 1 | UC | IFF DLP Forward Estimate (Row 15) exceeds DLP Notification Value (Row 13) or CTDIvol Forward Estimate (Row 16) exceeds CTDIvol Notification Value (Row 14) |
18 | > | CONTAINS | INCLUDE | DTID 1020 “Person Participant” | 1 | UC | IFF DLP Forward Estimate (Row 15) exceeds DLP Notification Value (Row 13) or CTDIvol Forward Estimate (Row 16) exceeds CTDIvol Notification Value (Row 14) | $PersonProcedureRole = EV (113850, DCM, "Irradiation Authorizing")

Content Item Descriptions

Row 1 | Container for Dose Check Alert details.
Row 2 | Indicates whether a DLP Alert Value was configured (e.g., by the institution) for the exam to which this irradiation event belongs.
Row 3 | Indicates whether a CTDIvol Alert Value was configured (e.g., by the institution) for the exam to which this irradiation event belongs.
Row 4 | The configured value applicable to the current exam that would trigger an alert if the accumulated DLP were projected to exceed it.
Row 5 | The configured value applicable to the current exam that would trigger an alert if the Accumulated CTDIvol at any given location were projected to exceed it.
Row 6 | The value estimated prior to performing this irradiation event of the projected DLP accumulated during this exam, including this irradiation event. The estimate may include assumptions such as those described in NEMA XR 25-2010.
Row 7 | The value estimated prior to performing this irradiation event of the projected CTDIvol accumulated during this exam, including this irradiation event. The value is for the location with the highest estimated accumulation. The actual location is not recorded. The estimate may include assumptions such as those described in NEMA XR 25-2010.
Row 8 | The reason provided by the operator for proceeding with an irradiation event projected to exceed an alert value.
Row 9 | Person responsible for authorizing irradiation projected to exceed an alert value.
Row 10 | Container for Dose Check Notification details.
Row 11 | Indicates whether a DLP Notification Value was configured (e.g., by the institution) for the Protocol Element Group to which this irradiation event corresponds.
Row 12 | Indicates whether a CTDIvol Notification Value was configured (e.g., by the institution) for the Protocol Element Group to which this irradiation event corresponds.
Row 13 | The configured value applicable to the current irradiation event that would trigger a notification if the DLP were projected to exceed it.
Row 14 | The configured value applicable to the current irradiation event that would trigger a notification if the CTDIvol were projected to exceed it.
Row 15 | The value estimated prior to performing this irradiation event of the DLP for this irradiation event. The estimate may include assumptions such as those described in NEMA XR 25-2010.
Row 16 | The value estimated prior to performing this irradiation event of the CTDIvol for this irradiation event. The value is for the location with the highest estimated value. The actual location is not recorded. The estimate may include assumptions such as those described in NEMA XR 25-2010.
Row 17 | The reason provided by the operator for proceeding with an irradiation event projected to exceed a notification value.
Row 18 | Person responsible for authorizing irradiation projected to exceed a notification value.

## Radiopharmaceutical Radiation Dose SR IOD Templates

The templates that comprise the Radiopharmaceutical Radiation Dose SR are interconnected as in Figure A-17.

Figure A-17. Radiopharmaceutical Radiation Dose SR IOD Template Structure

### TID 10021 Radiopharmaceutical Radiation Dose

This template defines a container (the root) with subsidiary content items, each of which corresponds to a single Radiopharmaceutical Administration Dose event entry. There is a defined recording observer (the system and/or person responsible for recording the assay of the radiopharmaceutical, and the person administered the radiopharmaceutical). Multiple Radiopharmaceutical Radiation Dose objects may be created for one study. Radiopharmaceutical Start DateTime in TID 10022 “Radiopharmaceutical Administration Event Data” will convey the order of administrations.

Type:

  Extensible

Order:

  Significant

Root:

  Yes

Table TID 10021. Radiopharmaceutical Radiation Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113500, DCM, "Radiopharmaceutical Radiation Dose Report") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C2D0, SRT, "Associated Procedure") | 1 | M |  | DCID 3108 “NM/PET Procedures”
3 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E8, SRT, "Has Intent") | 1 | M |  | DCID 3629 “Procedure Intent”
4 | > | CONTAINS | INCLUDE | DTID 10022 “Radiopharmaceutical Administration Event Data” | 1 | M |  |
5 | > | CONTAINS | INCLUDE | DTID 10024 “Radiopharmaceutical Administration Patient Characteristics” | 1 | U |  |
6 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

Content Item Descriptions

Row 2 | The associated procedure is the procedure performed, or if no procedure was performed the procedure that was ordered.

### TID 10022 Radiopharmaceutical Administration Event Data

The Radiopharmaceutical Administration Event conveys the dose and assay and time information of a single radiopharmaceutical event. A Radiopharmaceutical Administration event is one radioactive pharmaceutical administered to a patient.

Type:

  Extensible

Order:

  Significant

Table TID 10022. Radiopharmaceutical Administration Event Data

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113502, DCM, "Radiopharmaceutical Administration") | 1 | M |  |
2 | > | CONTAINS | CODE | EV (F-61FDB, SRT, "Radiopharmaceutical agent") | 1 | M |  | DCID 25 “Radiopharmaceuticals” DCID 4021 “PET Radiopharmaceutical”
3 | >> | HAS PROPERTIES | CODE | EV (C-10072, SRT, "Radionuclide") | 1 | M |  | DCID 18 “Isotopes in Radiopharmaceuticals” DCID 4020 “PET Radionuclide”
4 | >> | HAS PROPERTIES | NUM | EV (R-42806, SRT, "Radionuclide Half Life") | 1 | M |  | UNITS = EV (s, UCUM, "seconds")
5 | > | CONTAINS | NUM | EV (123007, DCM, "Radiopharmaceutical Specific Activity") | 1 | U |  | UNITS = EV (Bq/mmol, UCUM, "Bq/mmol")
6 | > | CONTAINS | UIDREF | EV (113503, DCM, "Radiopharmaceutical Administration Event UID") | 1 | M |  |
7 | > | CONTAINS | CODE | EV (113505, DCM, "Intravenous Extravasation Symptoms") | 1-n | U |  | DCID 10043 “Intravenous Extravasation Symptoms”
8 | > | CONTAINS | NUM | EV (113506, DCM, "Estimated Extravasation Activity") | 1 | U |  | UNITS = EV(%, UCUM, "percent")
9 | > | CONTAINS | DATETIME | EV (123003, DCM, "Radiopharmaceutical Start DateTime") | 1 | M |  |
10 | > | CONTAINS | DATETIME | EV (123004, DCM, "Radiopharmaceutical Stop DateTime") | 1 | U |  |
11 | > | CONTAINS | NUM | EV (113507, DCM, "Administered activity") | 1 | M |  | UNITS = EV (MBq, UCUM, "MBq")
12 | > | CONTAINS | NUM | EV (123005, DCM, "Radiopharmaceutical Volume") | 1 | U |  | UNITS = EV (cm3, UCUM, "cm3")
13 | > | CONTAINS | NUM | EV (113508, DCM, "Pre-Administration Measured Activity") | 1 | U |  | UNITS = EV (MBq, UCUM, "MBq")
14 | >> | HAS OBS CONTEXT | CODE | EV (113540, DCM, "Activity Measurement Device") | 1 | U |  | DCID 10041 “Source of Radioisotope Activity Information”
15 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
16 | > | CONTAINS | NUM | EV (113509, DCM, "Post-Administration Measured Activity") | 1 | U |  | UNITS = EV (MBq, UCUM, "MBq")
17 | >> | HAS OBS CONTEXT | CODE | EV (113540, DCM, "Activity Measurement Device") | 1 | U |  | DCID 10041 “Source of Radioisotope Activity Information”
18 | >> | HAS OBS CONTEXT | INCLUDE | DTID 1002 “Observer Context” | 1-n | U |  |
19 | > | CONTAINS | INCLUDE | DTID 10023 “Organ Dose” | 1-n | U |  |
20 | > | CONTAINS | CODE | EV (G-C340, SRT, "Route of administration") | 1 | M |  | BCID 11 “Route of Administration”
21 | >> | HAS PROPERTIES | CODE | EV (G-C581, SRT, "Site of") | 1 | MC | IF Row 22 equals (G-D101, SRT, "Intravenous route") or (G-D103, SRT, "Intramuscular route") | DCID 3746 “Percutaneous Entry Site”
22 | >>> | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IF Row 23 has laterality | DCID 244 “Laterality”
23 | > | HAS OBS CONTEXT | INCLUDE | DTID 1020 “Person Participant” | 1-n | M |  | $PersonProcedureRole = EV (113851, DCM, "Irradiation Administering")
24 | > | CONTAINS | CODE | EV (121147, DCM, "Billing Code(s) ") | 1-n | U |  |
25 | > | CONTAINS | CODE | EV (113510, DCM, "Drug Product Identifier") | 1-n | U |  |
26 | > | CONTAINS | TEXT | EV (111529, DCM, "Brand Name") | 1 | U |  |
27 | > | CONTAINS | TEXT | EV (113511, DCM, "Radiopharmaceutical Dispense Unit Identifier") | 1 | U |  |
28 | >> | CONTAINS | TEXT | EV (113512, DCM, "Radiopharmaceutical Lot Identifier") | 1-n | U |  |
29 | >> | CONTAINS | TEXT | EV (113513, DCM, "Reagent Vial Identifier") | 1-n | U |  |
30 | >> | CONTAINS | TEXT | EV (113514, DCM, "Radionuclide Identifier") | 1-n | U |  |
31 | > | CONTAINS | TEXT | EV (113516, DCM, "Prescription Identifier") | 1 | U |  |
32 | > | CONTAINS | TEXT | EV (121106, DCM, "Comment") | 1 | U |  |

Content Item Descriptions

Row 4 | The value of Half-life that was used for computing the decay of the administered radiopharmaceutical. It is not intended for use by the receiver for any further computation.
Row 5 | Activity per unit mass of the radiopharmaceutical at Radiopharmaceutical Start Time
Row 6 | Unique identification of a single radiopharmaceutical administration event.
Row 8 | The estimated percentage of administered activity lost at the injection site. The estimation includes extravasation, paravenous administration and leakage at the injection site.
Row 9 | The time the radiopharmaceutical was administered to the patient for imaging purposes.
Row 11 | Total amount of radioactivity administered to the patient at Radiopharmaceutical Start Time. It is a computed field from the TID 10022 Pre-Administration Measured Activity Row 13, TID 10022 Post-Administration Measured Activity Row 17, Radionuclide Half Life Row 4 and Radiopharmaceutical Start Time Row 9. Does not include estimated extravasation activity.
Rows 13, 16 | Observation DateTime (0040,A032) shall be used to record when the measurement was taken.
Row 23 | Identifies the person administering the product.
Row 24 | The billing codes for the preparation and administration of the radiopharmaceutical. It does not include performance and interpretation of the imaging.
Row 25 | Registered drug establishment code for the product. A coding scheme example is NDC, WHO-DDE or RxNorm. Multiple entries can be used for equivalent drug product codes.
Row 27 | The human readable identification of the specific radiopharmaceutical quantity (dose) administered to the patient.
Row 28 | Identifies the vial, batch or lot number from which the individual radiopharmaceutical quantity (dose) was produced. Row 27 the Radiopharmaceutical Identifier records the identification for each individual dose.
Row 29 | Identifies the lot or unit serial number for the reagent component for the radiopharmaceutical identified in row 27.
Row 30 | Identifies the lot or unit serial number for the radionuclide component for the radiopharmaceutical identified in row 27.

### TID 10023 Organ Dose

This template conveys the information about the dose to a single organ.

Type:

  Extensible

Order:

  Significant

Table TID 10023. Organ Dose

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (113517, DCM, "Organ Dose Information") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | M |  | DCID 10044 “Radiosensitive Organs”
3 | > | HAS CONCEPT MOD | CODE | EV (G-C171, SRT, "Laterality") | 1 | MC | IFF anatomy has laterality | DCID 244 “Laterality”
4 | > | CONTAINS | NUM | EV (G-D701, SRT, "Mass") | 1 | U |  | UNITS = EV (g, UCUM, "grams")
5 | >> | HAS CONCEPT MOD | TEXT | EV (G-C036, SRT, "Measurement Method") | 1 | M |  |
6 | > | CONTAINS | NUM | EV (113518, DCM, "Organ Dose") | 1 | M |  | UNITS = EV (mGy, UCUM, "mGy")
7 | >> | HAS PROPERTIES | CODE | EV (121406, DCM, "Reference Authority") | 1 | MC | XOR Row 8 | BCID 10040 “Radiopharmaceutical Organ Dose Reference Authority”
8 | >> | HAS PROPERTIES | TEXT | EV (121406, DCM, "Reference Authority") | 1 | MC | XOR Row 7 |

Content Item Descriptions

Row 3 | For paired organs,use (G-A102, SRT, "Right and Left") to report the estimated absorbed dose for both organs.
Row 4 | The estimated mass of organ in grams used when calculating the organ dose.
Row 5 | Method used to obtain the estimate. This could include a method that does not involve performing a measurement (e.g., Standard Organ Mass Tables).
Row 6 | Organ dose (in units of mGy). Organ is specified by row 2.

### TID 10024 Radiopharmaceutical Administration Patient Characteristics

This template describes the characteristics of the patient that are specific to the current clinical presentation (visit). The characteristics noted may affect the activity received, and how dose is calculated for the patient. Patient Characteristic concepts in this template, which may replicate attributes in the Patient Study Module, are included here as possible targets of by-reference relationships from other content items in the SR tree.

Type:

  Extensible

Order:

  Significant

Table TID 10024. Radiopharmaceutical Administration Patient Characteristics

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | EV (121118, DCM, "Patient Characteristics") |  |  |  |
2 | > | CONTAINS | CODE | EV (109054, DCM, "Patient state") | 1-n | U |  | DCID 10045 “Radiopharmaceutical Patient State”
3 | > | CONTAINS | NUM | EV (121033, DCM, "Subject Age") | 1 | U |  | UNITS = DCID 7456 “Units of Measure for Age”
4 | > | CONTAINS | CODE | EV (121032, DCM, "Subject Sex") | 1 | U |  | DCID 7455 “Sex”
5 | > | CONTAINS | NUM | EV (8302-2, LN, "Patient Height") | 1 | U |  | UNITS = EV (cm, UCUM, "cm")
6 | > | CONTAINS | NUM | EV (29463-7, LN, "Patient Weight") | 1 | U |  | UNITS = EV (kg, UCUM, "kg").
7 | > | CONTAINS | NUM | EV (8277-6, LN, "Body Surface Area") | 1 | U |  | UNITS = EV (m2, UCUM, "m^2")
8 | >> | INFERRED FROM | CODE | EV (8278-4, LN, "Body Surface Area Formula") | 1 | U |  | BCID 3663 “Body Surface Area Equations”
9 | > | CONTAINS | NUM | EV (F-01860, SRT, "Body Mass Index") | 1 | U |  | UNITS = EV (kg/m2, UCUM, "kg/m^2")
10 | >> | INFERRED FROM | CODE | EV (121420, DCM, "Equation") | 1 | U |  | DT (122265, DCM, "BMI = Wt/Ht^2")
11 | > | CONTAINS | NUM | EV (14749-6, LN, "Glucose") | 1 | U |  | UNITS = EV ("mmol/l", UCUM, "mmol/l")
12 | > | CONTAINS | NUM | EV(113550, DCM, "Fasting Duration") | 1 | U |  | UNITS = DT (h, UCUM, "hours")
13 | > | CONTAINS | NUM | EV(113551, DCM, "Hydration Volume") | 1 | U |  | UNITS = DT (ml, UCUM, "ml")
14 | > | CONTAINS | TEXT | EV (113552, DCM, "Recent Physical Activity") | 1 | U |  |
15 | > | CONTAINS | NUM | EV (2160-0, LN, "Serum Creatinine") | 1 | U |  | UNITS = DT (mg/dl, UCUM, "mg/dl")
16 | > | CONTAINS | NUM | EV (F-70210, SRT, "Glomerular Filtration Rate") | 1-n | U |  | UNITS = DT (ml/min{1.73_m2}, UCUM, "ml/min/1.73m2")
17 | >> | HAS CONCEPT MOD | CODE | EV (G-C036, SRT, "Measurement Method") | 1 | U |  | DCID 10047 “GFR Measurement Methods”
18 | >> | HAS CONCEPT MOD | CODE | EV (121050, DCM, "Equivalent meaning of concept name") | 1 | M |  | DCID 10046 “GFR Measurements”

Content Item Descriptions

Row 3 | Defaults to value of Patient's Age (0010,1010) in Patient Study Module
Row 5 | Patient height may differ from Patient's Size (0010,1020). Row 4 is the height value used for any height based protocols. Observation DateTime (0040,A032) may be used to record when the measurement was taken.
Row 6 | Patient weight may differ from Patient's Weight (0010,1030). Row 5 is the weight value used for any weight based protocols. Observation DateTime (0040,A032) shall be used to record when the measurement was taken.
Row 11 | Patient's Blood Glucose level. Observation DateTime (0040,A032) shall be used to record when the measurement was taken.
Row 15 | Serum Creatinine level. Observation DateTime (0040,A032) shall be used to record when the measurement was taken.
Row 16 | Glomerular Filtration Rate Observation DateTime (0040,A032) shall be used to record when the measurement was taken. The formatting of the UCUM units is aligned with LOINC. See http://unitsofmeasure.org/trac/ticket/98

# B DCMR Context Groups (Normative)

This Annex specifies the content of Context Groups required by DICOM IODs.

### Note

Section 7.1 of this Part defines the fields of Context Group tables.

## CID 2 Anatomic Modifier

Type:

  Extensible

Version:

  20020904

Table CID 2. Anatomic Modifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A100 | Right
SRT | G-A101 | Left
SRT | G-A102 | Bilateral
SRT | G-A103 | Unilateral
SRT | G-A104 | Lateral
SRT | G-A105 | Anterior
SRT | G-A106 | Posterior
SRT | G-A107 | Cephalic
SRT | G-A108 | Caudal
SRT | G-A109 | Medial
SRT | G-A110 | Central
SRT | G-A111 | Peripheral
SRT | G-A112 | External
SRT | G-A113 | Internal
SRT | G-A114 | Intermediate
SRT | G-A115 | Inferior
SRT | G-A116 | Superior
SRT | G-A117 | Transverse
SRT | G-A118 | Proximal
SRT | G-A119 | Distal
SRT | G-A120 | Postaxial
SRT | G-A121 | Preaxial
SRT | G-A122 | Apical
SRT | G-A123 | Basal
SRT | G-A127 | Afferent
SRT | G-A128 | Efferent
SRT | G-A138 | Coronal
SRT | G-A139 | Superficial
SRT | G-A140 | Deep
SRT | G-A142 | Horizontal
SRT | G-A143 | Longitudinal
SRT | G-A144 | Vertical
SRT | G-A145 | Sagittal
SRT | G-A147 | Axial
SRT | G-A151 | Extra-articular
SRT | G-A168 | Surface
SRT | G-A169 | Gutter
SRT | G-A170 | Hilar
SRT | G-A171 | Capsular
SRT | G-A172 | Subcapsular
SRT | G-A174 | Edge
SRT | G-A180 | Anterolateral
SRT | G-A182 | Posterolateral
SRT | G-A15A | Intra-articular
SRT | G-A428 | Marginal

## CID 4 Anatomic Region

Type:

  Extensible

Version:

  20110124

Table CID 4. Anatomic Region

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 4030 “CT, MR and PET Anatomy Imaged”
Include CID 4040 “Endoscopy Anatomic Regions”
Include CID 4042 “XA/XRF Anatomy Imaged”
SRT | T-32100 | Atrium
SRT | T-D8100 | Axilla
SRT | T-D2100 | Back
SRT | T-D6500 | Broad ligament
SRT | T-D1206 | Buccal region of face
SRT | T-D2600 | Buttock
SRT | T-72100 | Calyx
SRT | T-D1206 | Cheek
SRT | T-AA200 | Cornea
SRT | T-AB000 | Ear
SRT | T-41000 | Endo-arterial
SRT | T-32000 | Endo-cardiac
SRT | T-56000 | Endo-esophageal
SRT | T-21300 | Endo-nasal
SRT | T-23050 | Endo-nasopharyngeal
SRT | T-59600 | Endo-rectal
SRT | T-71000 | Endo-renal
SRT | T-73000 | Endo-ureteric
SRT | T-75000 | Endo-urethral
SRT | T-82000 | Endo-vaginal
SRT | T-40000 | Endo-vascular
SRT | T-48000 | Endo-venous
SRT | T-74250 | Endo-vesical
SRT | T-D4200 | Epigastric region
SRT | T-AA810 | Eyelid
SRT | T-D1200 | Face
SRT | T-D2310 | Flank
SRT | T-15200 | Fontanel of skull
SRT | T-D2600 | Gluteal region
SRT | T-15710 | Hip joint
SRT | T-D4240 | Hypogastric region
SRT | T-D1212 | Hypoglossal
SRT | T-55300 | Hypopharynx
SRT | T-D4010 | Intra-abdominal
SRT | G-A15A | Intra-articular
SRT | T-D1400 | Intracranial
SRT | T-56000 | Intra-esophageal
SRT | T-D6221 | Intra-pelvic
SRT | T-D3000 | Intra-thoracic
SRT | T-D4211 | Left hypochondriac region
SRT | T-D7020 | Left inguinal region
SRT | T-D4140 | Left lower quadrant of abdomen
SRT | T-D2340 | Left lumbar region
SRT | T-D4130 | Left upper quadrant of abdomen
SRT | T-04003 | Lower inner quadrant of breast
SRT | T-04005 | Lower outer quadrant of breast
SRT | T-D2300 | Lumbar region
SRT | T-28000 | Lung
SRT | T-51000 | Mouth
SRT | T-21000 | Nose
SRT | T-D4450 | Omental bursa
SRT | T-D4600 | Omentum
SRT | T-87000 | Ovary
SRT | T-65010 | Pancreatic duct
SRT | T-D3136 | Parasternal
SRT | T-91000 | Penis
SRT | T-D2700 | Perineum
SRT | T-D9310 | Popliteal fossa
SRT | T-72000 | Renal pelvis
SRT | T-D4900 | Retroperitoneum
SRT | T-D4212 | Right hypochondriac region
SRT | T-D7010 | Right inguinal region
SRT | T-D4120 | Right lower quadrant of abdomen
SRT | T-D2342 | Right lumbar region
SRT | T-D4110 | Right upper quadrant of abdomen
SRT | T-D1160 | Scalp
SRT | T-AA110 | Sclera
SRT | T-98000 | Scrotum
SRT | T-A7010 | Spinal cord
SRT | T-D4210 | Subcostal
SRT | T-D1603 | Submandibular area
SRT | T-D3213 | Subxiphoid
SRT | T-D1620 | Supraclavicular region of neck
SRT | T-D4240 | Suprapubic region
SRT | T-11218 | Suprasternal notch
SRT | T-D9100 | Thigh
SRT | T-D3000 | Thorax
SRT | T-53000 | Tongue
SRT | T-D4230 | Umbilical region
SRT | T-04002 | Upper inner quadrant of breast
SRT | T-04004 | Upper outer quadrant of breast
SRT | T-82000 | Vagina
SRT | A-04140 | Vascular graft
SRT | T-32400 | Ventricle
SRT | T-81000 | Vulva
SRT | T-15460 | Wrist joint

## CID 5 Transducer Approach

Type:

  Extensible

Version:

  20061024

Table CID 5. Transducer Approach

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A100 | Right
SRT | G-A101 | Left
SRT | G-A104 | Lateral
SRT | G-A105 | Anterior
SRT | G-A106 | Posterior
SRT | G-A108 | Caudal
SRT | G-A109 | Medial
SRT | G-A110 | Central
SRT | G-A111 | Peripheral
SRT | G-A112 | External
SRT | G-A113 | Internal
SRT | G-A115 | Inferior
SRT | G-A116 | Superior
SRT | G-A117 | Transverse
SRT | G-A118 | Proximal
SRT | G-A119 | Distal
SRT | G-A122 | Apical
SRT | G-A168 | Surface
SRT | G-A599 | Ascending
SRT | G-A600 | Descending
SRT | T-03000 | Subcutaneous tissue
SRT | T-A1120 | Dura mater
SRT | T-A1280 | Pia mater
SRT | A-2C600 | External prosthesis for sonographic procedure [Stand-off]
SRT | A-2C602 | Water bag prosthesis for imaging procedure
SRT | A-2C604 | Saline bag prosthesis for imaging procedure
SRT | A-2C606 | Gel prosthesis for imaging procedure
SRT | G-A107 | Cranial
SRT | G-A10A | Midline
SRT | G-A188 | Mid-longitudinal
SRT | G-A189 | Parasagittal
SRT | R-42142 | Intraluminal
SRT | G-A171 | Capsular
SRT | T-D0048 | Lumen
SRT | G-4022 | Contact with
SRT | T-D0062 | Parenchyma

### Note

In a prior version of this Context Group, the codes G-A11A, G-A11B, G-A12A, G-A16A, G-A16B, G-A16C, and G-A16D were specified for various concepts. The use of some of those codes conflicts with their assignment to other concepts in SNOMED, and their use in this context is deprecated. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 6 Transducer Orientation

Type:

  Extensible

Version:

  20040322

Table CID 6. Transducer Orientation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A138 | Coronal
SRT | G-A143 | Longitudinal
SRT | G-A145 | Sagittal
SRT | G-A189 | Parasagittal
SRT | G-A472 | Oblique
SRT | G-A185 | Long axis
SRT | G-A13B | Off axis
SRT | G-A186 | Short axis
SRT | G-A191 | Five chamber
SRT | G-A19B | Two chamber
SRT | G-A19C | Four chamber
SRT | G-A117 | Transverse

### Note

In a prior version of this table, the code G-A11B was specified for the concept Parasagittal. The use of this code conflicts with its assignment to another concept in SNOMED, and its use in this context is deprecated. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 7 Ultrasound Beam Path

Type:

  Extensible

Version:

  20020904

Table CID 7. Ultrasound Beam Path

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A1A9 | Trans-hepatic
SRT | G-A1B2 | Trans-gastric
SRT | G-A1A5 | Trans-pleural
SRT | G-A1B3 | Trans-mural
SRT | G-A1A8 | Trans-orbital
SRT | G-A1A6 | Trans-pancreatic
SRT | G-A1A4 | Trans-renal
SRT | G-D032 | Trans-temporal
SRT | G-A1A2 | Trans-thecal
SRT | G-A1A1 | Trans-vesical
SRT | G-A1A3 | Trans-splenic
SRT | G-D033 | Trans-esophageal
SRT | G-D001 | Trans-abdominal
SRT | G-D002 | Trans-vaginal (endovaginal)

## CID 8 Angiographic Interventional Devices

Type:

  Extensible

Version:

  20020904

Table CID 8. Angiographic Interventional Devices

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-25500 | Stent
SRT | A-26800 | Catheter
SRT | A-81080 | Laser
SRT | C-20005 | Glue
SRT | A-25600 | Atherectomy device
SRT | A-25614 | Embolization ball
SRT | A-26912 | Percutaneous transluminal angioplasty balloon
SRT | A-25612 | Embolization coil
SRT | A-25612 | Gianturco coil
SRT | A-27322 | Detachable balloon
SRT | A-26A06 | Fixed object
SRT | A-26A08 | Grid
SRT | A-26802 | Guiding catheter
SRT | A-25616 | Embolization particulate
SRT | A-25610 | Rotational atherectomy device
SRT | A-10141 | Measuring ruler
DCM | 122485 | Sphere

## CID 9 Image Guided Therapeutic Procedures

Type:

  Extensible

Version:

  20020904

Table CID 9. Image Guided Therapeutic Procedures

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-39780 | Vasoconstriction
SRT | F-39800 | Vasodilatation
SRT | P1-03100 | Biopsy
SRT | P1-03176 | Removal of foreign body
SRT | P1-05035 | Intra-arterial infusion of thrombolytic agent
SRT | P1-05052 | Irrigation following insertion of catheter
SRT | P1-05535 | Catheterization
SRT | P1-30350 | Atherectomy
SRT | P1-30351 | Atherectomy by rotary cutter
SRT | P1-30352 | Atherectomy by laser
SRT | P1-30530 | Selective embolization of artery
SRT | P5-31500 | Percutaneous transluminal balloon angioplasty
SRT | P5-39010 | Transcatheter therapy for embolization
SRT | P5-39050 | Percutaneous retrieval of intravascular foreign body
SRT | P1-00018 | Failed attempted procedure
SRT | P1-05550 | Stent placement
SRT | P1-05536 | Catheter manipulation
SRT | P1-05537 | Catheter replacement
SRT | P1-05538 | Occlusion of catheter
SRT | P1-05539 | Removal of catheter
SRT | P5-39015 | Transcatheter deployment of detachable balloon
SRT | P5-39191 | Percutaneous insertion of intravascular filter
SRT | P1-86100 | Amniocentesis
SRT | P5-B8310 | Ultrasonic guidance for amniocentesis
SRT | P1-86520 | Amnioinfusion [injection of amnion]
SRT | P1-86180 | Intrauterine cordocentesis
SRT | P1-28160 | Thoracentesis
SRT | P1-86E70 | Breech Version [Obstetrical Version]
SRT | P1-86101 | Decompression amniocentesis [decompression of amnion]
SRT | P2-68060 | Intrauterine transfusion
SRT | P1-86C50 | Fetocide (selective reduction)
SRT | P1-93506 | Prostaglandin injection

## CID 10 Interventional Drug

Type:

  Extensible

Version:

  20020904

Table CID 10. Interventional Drug

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-21005 | Ethanol
SRT | C-22947 | Methylene blue
SRT | C-51000 | Antihistamine
SRT | C-67770 | Atropine
SRT | C-72000 | Diuretic
SRT | C-80110 | Antiarrhythmic drug
SRT | C-80120 | Inotropic agent
SRT | C-80123 | Cardiotonic drug
SRT | C-80125 | Cardiac depressant drug
SRT | C-80130 | Cardiac adrenergic blocking agent
SRT | C-80131 | Alpha-adrenergic blocking agent
SRT | C-80135 | beta-Adrenergic blocking agent
SRT | C-80330 | Digoxin
SRT | C-80400 | Lidocaine
SRT | C-80401 | Lidocaine hydrochloride
SRT | C-80430 | Nifedipine
SRT | C-80450 | Propranolol
SRT | C-80460 | Quinidine
SRT | C-80490 | Verapamil
SRT | C-81100 | Hypotensive agent
SRT | C-81120 | Centrally acting hypotensive agent
SRT | C-81560 | Nitroglycerin
SRT | C-A2010 | Glucagon preparation
SRT | C-A6500 | Anticoagulant
SRT | C-A6530 | Warfarin
SRT | C-A6540 | Heparin
SRT | C-A6700 | Anti-heparin agent
SRT | C-A6710 | Protamine sulfate
SRT | C-A6900 | Coagulant
SRT | C-A6920 | Injectable fibrinogen
SRT | C-A7000 | Hemostatic agent
SRT | C-A7001 | Astringent drug
SRT | C-A7021 | Antihemophilic factor preparation
SRT | C-A7040 | Thrombin preparation
SRT | C-A7042 | Thromboplastin preparation
SRT | C-A7220 | Dextran
SRT | C-A7400 | Thrombolytic agent
SRT | C-A7420 | Streptokinase preparation
SRT | C-A7430 | Urokinase preparation
SRT | C-A7440 | Injectable fibrinolysin
SRT | C-C2318 | Priscoline hydrocholoride ampuls
SRT | F-B2110 | Epinephrine

## CID 11 Route of Administration

Type:

  Extensible

Version:

  20100608

Table CID 11. Route of Administration

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-D101 | Intravenous route
SRT | G-D102 | Intra-arterial route
SRT | G-D103 | Intramuscular route
SRT | G-D104 | Subcutaneous route
SRT | G-D105 | Intracutaneous route
SRT | G-D106 | Intraperitoneal route
SRT | G-D107 | Intramedullary route
SRT | G-D108 | Intrathecal route
SRT | G-D109 | Intra-articular route
SRT | G-D111 | Intraepithelial route
SRT | G-D112 | Topical route
SRT | G-D140 | Oral route
SRT | G-D142 | Transluminal route
SRT | G-D144 | Intraluminal route
SRT | G-D146 | Extraluminal route
SRT | G-D150 | By inhalation
SRT | G-D160 | Per rectum
SRT | G-D164 | Vaginal route
SRT | G-D17C | Intracoronary route
SRT | G-D173 | Intracardiac route
SRT | R-F2C86 | Intraventricular route - cardiac

## CID 12 Radiographic Contrast Agent

Type:

  Extensible

Version:

  20110609

Table CID 12. Radiographic Contrast Agent

Coding Scheme Designator | Code Value | Code Meaning | Trade Name (Informative) (From http://www.accessdata.fda.gov/scripts/cder/drugsatfda/index.cfm)
--- | --- | --- | ---
SRT | A-80230 | Air |
SRT | C-10110 | Oxygen |
SRT | C-10120 | Water |
SRT | C-10520 | Carbon dioxide |
SRT | C-12217 | Barium Sulfate |
SRT | C-17800 | Gadolinium |
SRT | C-B0300 | Contrast agent |
SRT | C-B0310 | Radiopaque medium |
SRT | C-B0312 | Non radiopaque medium |
SRT | C-B0315 | Bunamiodyl |
SRT | C-B0316 | Chloriodized oil |
SRT | C-B0317 | Diatrizoate | Angiovist™(Berlex), Cardiografin™(Bracco), Cystografin™(Bracco), Gastrografin™(Bracco), Gastrovist™(Berlex), Hypaque™(GE), MD-nn™(Mallinckrodt), Reno-nn™(Bracco), Renografin™(Bracco), Renovist™(Bracco), Sinografin™(Bracco), Urovist™(Berlex)
SRT | C-B0318 | Iodipamide | Cholographin™(Bracco), Sinografin™(Bracco)
SRT | C-B0319 | Iodized oil |
SRT | C-B0323 | Iodoalphionic acid |
SRT | C-B0324 | Meglumine iodipamide | Cholographin Meglumine™(Bracco)
SRT | C-B0325 | Sodium iodipamide | Cholographin Sodium™(Bracco)
SRT | C-B0326 | Iodamide meglumine | Renovue™(Bracco)
SRT | C-B0327 | Iodopyracet |
SRT | C-B0328 | Iopanoic acid | Telepaque™(GE)
SRT | C-B0331 | Iophendylate | Pantopaque™(Alcon)
SRT | C-B0333 | Iophenoxic acid |
SRT | C-B0335 | Ipodate | Bilivist™(Berlex), Oragrafin™(Bracco)
SRT | C-B0337 | Propyliodone | Dionosil™(GSK)
SRT | C-B0338 | Sodium acetrizoate | Salpix™(Ortho)
SRT | C-B0341 | Iodophthalein |
SRT | C-B0342 | Sodium diprotrizoate |
SRT | C-B0344 | Sodium iodomethamate |
SRT | C-B0345 | Meglumine diatrizoate | Angiovist™(Berlex), Cardiografin™(Bracco), Cystografin™(Bracco), Gastrografin™(Bracco), Gastrovist™(Berlex), Hypaque™(GE), MD-nn™(Mallinckrodt), Reno-nn™(Bracco), Renografin™(Bracco), Renovist™(Bracco), Sinografin™(Bracco), Urovist™(Berlex)
SRT | C-B0347 | Sodium diatrizoate | Angiovist™(Berlex), Gastrografin™(Bracco), Gastrovist™(Berlex), Hypaque™(GE), MD-nn™(Mallinckrodt), Renografin™(Bracco), Renovist™(Bracco), Urovist™(Berlex)
SRT | C-B0348 | Metrizamide | Amipaque™(GE)
SRT | C-B0349 | Sodium tyropanate |
SRT | C-B0301 | Ionic iodinated contrast agent |
SRT | C-B0302 | Non-ionic iodinated contrast agent |
SRT | C-B0322 | Iohexol | Omnipaque™(GE)
SRT | C-B03BC | Iodixanol | Visipaque™(GE)
SRT | C-B03C3 | Gadodiamide | Omniscan™(GE)
SRT | C-B05A3 | Mangafodipir trisodium | Teslascan™(GE)
SRT | C-B038B | Iothalamate | Conray™(Mallinckrodt), Cysto-Conray™(Mallinckrodt), Vascoray™(Mallinckrodt)
SRT | C-B0339 | Ioxaglate | Hexbrix™(Mallinckrodt)
SRT | C-B03C9 | Metrizoate | Isopaque™(GE)
SRT | C-B03AA | Dimeglumine gadopentetate | Magnevist™(Berlex)
SRT | C-B0329 | Iopamidol | Isovue™(Bracco)
SRT | C-B0332 | Ioversol | Optiray™(Mallinckrodt)
SRT | C-B03H2 | Iopromide | Ultravist or Imeron
SRT | C-B0303 | Ioxilan | Imagenil

### Note

The codes drawn from SNOMED are recommended to be those from the concept hierarchy of "radiographic contrast media" in the hierarchy "pharmaceutical/biological product", and secondarily from the hierarchy "substance".

## CID 13 Radiographic Contrast Agent Ingredient

Type:

  Extensible

Version:

  20051101

Table CID 13. Radiographic Contrast Agent Ingredient

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-11400 | Iodine
SRT | C-17800 | Gadolinium
SRT | C-10520 | Carbon Dioxide
SRT | C-12200 | Barium
SRT | C-17200 | Xenon
SRT | A-80230 | Air
SRT | C-10110 | Oxygen
SRT | C-10120 | Water
SRT | C-130F9 | Iron

## CID 18 Isotopes in Radiopharmaceuticals

Type:

  Extensible

Version:

  20141110

Table CID 18. Isotopes in Radiopharmaceuticals

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-105A2 | ^14^Carbon
SRT | C-111A1 | ^18^Fluorine
SRT | C-155A1 | ^22^Sodium
SRT | C-155A2 | ^24^Sodium
SRT | C-106A1 | ^32^Phosphorus
SRT | C-135A2 | ^42^Potassium
SRT | C-135A3 | ^43^Potassium
SRT | C-129A2 | ^51^Chromium
SRT | C-144A3 | ^57^Cobalt
SRT | C-144A4 | ^58^Cobalt
SRT | C-130A3 | ^59^Iron
SRT | C-144A6 | ^60^Cobalt
SRT | C-127A2 | ^64^Copper
SRT | C-127A3 | ^67^Copper
SRT | C-131A2 | ^67^Gallium
SRT | C-116A3 | ^75^Selenium
SRT | C-173A5 | ^81m^Krypton
SRT | C-173A7 | ^85^Krypton
SRT | C-158A3 | ^85^Strontium
SRT | C-158A5 | ^87m^Strontium
SRT | C-158A6 | ^89^Strontium
SRT | C-162A7 | ^90^Yttrium
SRT | C-180A2 | ^97^Ruthenium
SRT | C-163A8 | ^99m^Technetium
SRT | C-145A4 | ^111^Indium
SRT | C-145A5 | ^113m^Indium
SRT | C-114A4 | ^123^Iodine
SRT | C-114A6 | ^125^Iodine
SRT | C-172A5 | ^127^Xenon
SRT | C-114B1 | ^131^Iodine
SRT | C-122A5 | ^133^Barium
SRT | C-172A8 | ^133^Xenon
SRT | C-178A8 | ^153^Gadolinium
SRT | C-B1134 | ^153^Samarium
SRT | C-181A3 | ^169^Ytterbium
SRT | C-101ED | ^177^Lutetium
SRT | C-156A6 | ^178^Tantalum
SRT | C-11906 | ^186^Rhenium
SRT | C-1018D | ^188^Rhenium
DCM | 126604 | ^191m^Iridium
SRT | C-146A9 | ^198^Gold
SRT | C-146B1 | ^199^Gold
SRT | C-138A9 | ^201^Thallium
SRT | C-132A8 | ^203^Lead
SRT | C-136A2 | ^223^Radium

### Note

The use of this Context Group in the Radionuclide Code Sequence (0054,0300) of the “Nuclear Medicine Image IOD” in PS3.3 requires a Coding Scheme Designator value of "99SDM".

## CID 19 Patient Orientation

Type:

  Extensible

Version:

  20020904

Table CID 19. Patient Orientation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-10440 | erect
SRT | F-10450 | recumbent
SRT | F-10460 | semi-erect

### Note

The use of this Context Group in the Patient Orientation Code Sequence (0054,0410) of the “Nuclear Medicine Image IOD” in PS3.3 and the “Positron Emission Tomography Image IOD” in PS3.3 requires a Coding Scheme Designator value of "99SDM".

## CID 20 Patient Orientation Modifier

Type:

  Extensible

Version:

  20070524

Table CID 20. Patient Orientation Modifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-10310 | prone
SRT | F-10316 | semi-prone
SRT | F-10318 | lateral decubitus
SRT | F-10320 | standing
SRT | F-10326 | anatomical
SRT | F-10330 | kneeling
SRT | F-10336 | knee-chest
SRT | F-10340 | supine
SRT | F-10346 | lithotomy
SRT | F-10348 | Trendelenburg
SRT | F-10349 | inverse Trendelenburg
SRT | F-10380 | frog
SRT | F-10390 | stooped-over
SRT | F-103A0 | sitting
SRT | F-10410 | curled-up
SRT | F-10317 | right lateral decubitus
SRT | F-10319 | left lateral decubitus
SRT | R-40799 | lordotic

### Note

The use of this Context Group in the Patient Orientation Modifier Code Sequence (0054,0412) of the “Nuclear Medicine Image IOD” in PS3.3 and the “Positron Emission Tomography Image IOD” in PS3.3 requires a Coding Scheme Designator value of "99SDM".

## CID 21 Patient Equipment Relationship

Type:

  Extensible

Version:

  20040322

Table CID 21. Patient Equipment Relationship

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-10516 | oblique
SRT | F-10470 | headfirst
SRT | F-10480 | feet-first
SRT | R-10515 | transverse

### Note

1. The use of this Context Group in the Patient Orientation Modifier Code Sequence (0054,0412) of the “Nuclear Medicine Image IOD” in PS3.3 and the “Positron Emission Tomography Image IOD” in PS3.3 requires a Coding Scheme Designator value of "99SDM".

2. In a prior version of this Context Group, the codes G-5190 and G-5191 were specified for the concepts "headfirst" and "feet-first". The use of these codes is deprecated as they are not actually in SNOMED. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

3. For devices that do not have a conventional gantry geometry, the vendor of that device may describe in the Conformance Statement how the codes are interpreted with respect to the device geometry.

## CID 23 Cranio-Caudad Angulation

Type:

  Extensible

Version:

  20020904

Table CID 23. Cranio-Caudad Angulation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A107 | Cephalic
SRT | G-A108 | Caudal

## CID 25 Radiopharmaceuticals

Type:

  Extensible

Version:

  20110224

Table CID 25. Radiopharmaceuticals

Coding Scheme Designator | Code Value | Code Meaning | Trade name (Informative)
--- | --- | --- | ---
SRT | C-B1302 | Carbon^14^ D-xylose |
SRT | C-B1300 | Carbon^14^ triolein |
SRT | C-B1304 | Cholyl-carbon^14^ glycine |
SRT | C-B1140 | Chromic phosphate P^32^ |
SRT | C-B1012 | Chromium^51^ albumin |
SRT | C-B1013 | Chromium^51^ chloride |
SRT | C-B1051 | Colloidal gold Au^198^ |
SRT | C-B1063 | Colloidal Indium^111^ |
SRT | C-B1017 | Copper^64^ acetate |
SRT | C-B1016 | Copper^64^ versenate |
SRT | C-B1018 | Copper^67^ ceruloplasmin |
SRT | C-B1021 | Cyanocobalamin Co^57^ |
SRT | C-B1022 | Cyanocobalamin Co^58^ |
SRT | C-B1023 | Cyanocobalamin Co^60^ |
SRT | C-B1000 | Diagnostic radioisotope |
SRT | C-B1092 | Diiodofluorecein I^131^ |
SRT | C-B1062 | Disodium indium^111^ |
SRT | C-B1122 | Ferrous chloride Fe^59^ |
SRT | C-B1121 | Ferrous citrate Fe^59^ |
SRT | C-B1123 | Ferrous sulfate Fe^59^ |
SRT | C-B1082 | Fibrinogen I^123^ |
SRT | C-B1031 | Fluorodeoxyglucose F^18^ |
SRT | C-B1041 | Gallium^67^ citrate |
SRT | C-145AB | Indium^111 Capromab Pendetide | Prostascint
SRT | C-14512 | Indium^111 Chloride | Zevalin
SRT | C-145AA | Indium^111 Pentetreotide | Octreoscan
SRT | C-B1061 | Indium^111^ pentetate |
SRT | C-B1066 | Indium^111^ red cell label |
SRT | C-B1067 | Indium^111^ transferrin |
SRT | C-B1065 | Indium^111^-Fe(OH) >3< |
SRT | C-B1135 | Indium^111^oxyquinoline |
SRT | C-B1068 | Indium^113m^ bleomycin |
SRT | C-B1069 | Indium^113m^ chloride |
SRT | C-B1072 | Indium^113m^ oxoquinoline platelet label |
SRT | C-B1073 | Indium^113m^ oxoquinoline RBC label |
SRT | C-B1071 | Indium^113m^ oxoquinoline WBC label |
SRT | C-B1070 | Indium^113m^ pentetate |
SRT | C-B1084 | Iodinated I^125^ albumin |
SRT | C-B1100 | Iodinated I^125^ human serum albumin |
SRT | C-B1094 | Iodinated I^125^ levothyroxine |
SRT | C-B1093 | Iodinated I^125^ oleic acid and triolein |
SRT | C-B1096 | Iodinated I^125^ povidone |
SRT | C-B1097 | Iodinated I^125^ Rose Bengal |
SRT | C-B1098 | Iodinated I^125^ sealed source |
SRT | C-B1099 | Iodinated I^125^ sodium iodine |
SRT | C-B1090 | Iodinated I^131^ aggregated albumin |
SRT | C-B1089 | Iodinated I^131^ albumin |
SRT | C-B1111 | Iodinated I^131^ gamma globulin |
SRT | C-114AB | Iodine^123 15-(4-Iodophenyl) -3(R,S) -Methylpentadecanoic Acid BMIPP | Cardiodine
SRT | C-B110E | Iodine^123 3-Iodobenzylguanidine MIBG |
SRT | C-B112D | Iodine^131 3-Iodobenzylguanidine MIBG |
SRT | C-114B6 | Iodine^131 Methylnorcholestenol | Adosterol
SRT | C-B1109 | Iodine^131^ polyvinylpyrrolidone |
SRT | C-B1087 | Iodocholesterol I^131^ |
SRT | C-B1095 | Iodohippurate I^123^ sodium |
SRT | C-B1105 | Iodohippurate I^125^ sodium |
SRT | C-B1091 | Iodohippurate I^131^ sodium |
SRT | C-B1108 | Iofetamine I^123^ hydrochloride |
SRT | C-B1088 | Iothalamate sodium I^125^ |
SRT | C-B1124 | Iron Fe^59^ labeled dextran |
SRT | C-173A5 | Krypton^81m |
SRT | C-B1083 | Oleic acid I^125^ |
SRT | C-B1251 | Pentetate calcium trisodium Yb^169^ |
SRT | C-B1151 | Potassium carbonate K^42^ |
SRT | C-B1152 | Potassium chloride K^42^ |
SRT | C-B1150 | Potassium chloride K^43^ |
SRT | C-B1085 | Rose Bengal sodium I^131^ |
SRT | C-B1172 | Selenium^75^ HCAT |
SRT | C-B1171 | Selenomethionione Se^75^ |
SRT | C-B1176 | Sodium chloride Na^22^ |
SRT | C-B1175 | Sodium chloride Na^24^ |
SRT | C-B1011 | Sodium chromate Cr^51^ |
SRT | C-B1032 | Sodium fluoride F^18^ |
SRT | C-B1081 | Sodium iodide I^123^ |
SRT | C-B1086 | Sodium iodide I^131^ |
SRT | C-B1206 | Sodium pertechnetate Tc^99m^ |
SRT | C-B1142 | Sodium phosphate P^32^ |
SRT | C-B1180 | Strontium chloride Sr^85^ |
SRT | C-B1181 | Strontium chloride Sr^87^ |
SRT | C-B1182 | Strontium nitrate Sr^85^ |
SRT | C-B1183 | Strontium nitrate Sr^87^ |
SRT | C-B1205 | Technetium Tc^99c^ albumin microspheres |
SRT | C-B1200 | Technetium Tc^99m^ aggregated albumin |
SRT | C-B1204 | Technetium Tc^99m^ albumin colloid |
SRT | C-B1133 | Technetium Tc^99m^ depreotide |
SRT | C-B1207 | Technetium Tc^99m^ disofenin |
SRT | C-B1223 | Technetium Tc^99m^ exametazine |
SRT | C-B1210 | Technetium Tc^99m^ iron ascorbate |
SRT | C-B1209 | Technetium Tc^99m^ lidofenin |
SRT | C-B1208 | Technetium Tc^99m^ mebrofenin |
SRT | C-B1212 | Technetium Tc^99m^ medronate |
SRT | C-B1203 | Technetium Tc^99m^ microaggregated albumin |
SRT | C-B1225 | Technetium Tc^99m^ N-substituted iminodiacetate |
SRT | C-B1213 | Technetium Tc^99m^ oxidronate |
SRT | C-B1214 | Technetium Tc^99m^ pentetate |
SRT | C-B1215 | Technetium Tc^99m^ pyro and polyphosphates |
SRT | C-B1216 | Technetium Tc^99m^ serum albumin |
SRT | C-163AB | Technetium Tc^99m^ sestamibi |
SRT | C-B1220 | Technetium Tc^99m^ sodium glucoheptonate |
SRT | C-B1211 | Technetium Tc^99m^ stannous etidronate |
SRT | C-B1221 | Technetium Tc^99m^ succimer |
SRT | C-B1222 | Technetium Tc^99m^ sulfur colloid |
SRT | C-B1224 | Technetium Tc^99m^ tagged red cells |
SRT | C-163AC | Technetium Tc^99m^Teboroxime |
SRT | C-163AD | Technetium Tc^99m^Tetrofosmin |
SRT | C-163BD | Technetium^99m Dimercaptosuccinic Acid DMSA | Kidneyscinti
SRT | C-163B6 | Technetium^99m Galactosyl Human Serum Albumin Diethylenetriamine GSA | Asialoscinti
SRT | C-163B7 | Technetium^99m Hydroxymethylene diphosphonate HMDP |
SRT | C-163B9 | Technetium^99m labeled carbon | Technegas
SRT | C-163B8 | Technetium^99m Mercaptoacetyl triglycine MAG3 | MAGscinti
SRT | C-163BA | Technetium^99m N-pyridoxyl-5-methyltryptophan | Hepatimage
SRT | C-163BB | Technetium^99m Phytate |
SRT | C-163BC | Technetium^99m Stannous Colloid |
SRT | C-B1231 | Thallous chloride Tl^201^ |
SRT | C-B1010 | Therapeutic radioisotope |
SRT | C-B1251 | Yb^169^-DTPA - pentetate |

## CID 26 Nuclear Medicine Projections

Type:

  Extensible

Version:

  20040322

Table CID 26. Nuclear Medicine Projections

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A138 | Coronal
SRT | G-A145 | Sagittal
SRT | G-A147 | Axial
SRT | G-5206 | Right anterior oblique
SRT | G-5207 | Left anterior oblique
SRT | G-5208 | Right posterior oblique
SRT | G-5209 | Left posterior oblique
SRT | G-5210 | Oblique axial
SRT | G-5212 | Sagittal-oblique axial
SRT | G-5220 | Medial-lateral
SRT | G-5221 | Lateral-medial
SRT | G-5222 | Right lateral projection
SRT | G-5223 | Left lateral projection
SRT | G-5224 | Medio-lateral oblique
SRT | G-5225 | Latero-medial oblique
SRT | G-A117 | Transverse
SRT | G-A104 | Lateral
Include CID 27 “Basic Cardiac Views”
SRT | G-5215 | Anterior projection
SRT | G-5216 | Posterior projection

### Note

1. In a prior version of this table, the code G-A117 was specified for the concept Transaxial, and R-11300 was specified for the concept Transverse. Since these concepts are synonymous in nuclear projections, and since SNOMED assigns G-A117 to the concept Transverse, the use of R-11300 is deprecated. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

2. The following Code Values were formerly included in CID 26 “Nuclear Medicine Projections” and are retired:

   SRT | G-5200 | Antero-posterior
   SRT | G-5201 | Postero-anterior
   SRT | G-5203 | Frontal oblique
   SRT | G-5204 | Antero-posterior Oblique
   SRT | G-5205 | Postero-anterior Oblique
   SRT | G-5211 | Frontal-oblique axial
   SRT | G-5213 | Submento-vertex axial
   SRT | G-5214 | Oblique submento-vertex
   SRT | G-5226 | Right to left oblique
   SRT | G-5227 | Left to right oblique

## CID 27 Basic Cardiac Views

Type:

  Extensible

Version:

  20120822

Table CID 27. Basic Cardiac Views

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A186 | Short Axis
SRT | G-A18A | Vertical Long Axis
SRT | G-A18B | Horizontal Long Axis

## CID 29 Acquisition Modality

This Context Group includes codes that may be used to identify an image or waveform acquisition modality, as used in Attribute Modality (0008,0060) of a Modality Worklist Scheduled Procedure Step or a Composite SOP Instance (see PS3.3). It generally corresponds to a class of diagnostic equipment, or to a specific acquisition function or technique in a device. This Context Group may be used as the value set for HL7 v2 Table 0259 (see HL7 v2.6 Chapter 8 Section 8.8.8.47).

### Note

This Context Group is not the complete set of codes that may appear in the Attribute Modality (0008,0060); these are only the codes associated with orderable acquisition processes (not post-processing).

Type:

  Extensible

Version:

  20121129

Table CID 29. Acquisition Modality

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | AR | Autorefraction
DCM | BMD | Bone Mineral Densitometry
DCM | BDUS | Ultrasound Bone Densitometry
DCM | EPS | Cardiac Electrophysiology
DCM | CR | Computed Radiography
DCM | CT | Computed Tomography
DCM | DX | Digital Radiography
DCM | ECG | Electrocardiography
DCM | ES | Endoscopy
DCM | XC | External-camera Photography
DCM | GM | General Microscopy
DCM | HD | Hemodynamic Waveform
DCM | IO | Intra-oral Radiography
DCM | IVOCT | Intravascular Optical Coherence Tomography
DCM | IVUS | Intravascular Ultrasound
DCM | KER | Keratometry
DCM | LEN | Lensometry
DCM | MR | Magnetic Resonance
DCM | MG | Mammography
DCM | NM | Nuclear Medicine
DCM | OAM | Ophthalmic Axial Measurements
DCM | OCT | Optical Coherence Tomography
DCM | OPM | Ophthalmic Mapping
DCM | OP | Ophthalmic Photography
DCM | OPR | Ophthalmic Refraction
DCM | OPT | Ophthalmic Tomography
DCM | OPV | Ophthalmic Visual Field
DCM | OSS | Optical Surface Scanner
DCM | PX | Panoramic X-Ray
DCM | PT | Positron emission tomography
DCM | RF | Radiofluoroscopy
DCM | RG | Radiographic imaging
DCM | SM | Slide Microscopy
DCM | SRF | Subjective Refraction
DCM | US | Ultrasound
DCM | VA | Visual Acuity
DCM | XA | X-Ray Angiography

## CID 30 DICOM Devices

This Context Group includes codes that may be used to identify a class of equipment that uses DICOM.

Type:

  Extensible

Version:

  20030108

Table CID 30. DICOM Devices

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 29 “Acquisition Modality”
DCM | ARCHIVE | Archive
DCM | COMP | Computation Server
DCM | CAD | Computer Assisted Detection/Diagnosis
DCM | DSS | Department System Scheduler
DCM | FILMD | Film Digitizer
DCM | MCD | Media Creation Device
DCM | PRINT | Hard Copy Print Server
DCM | CAPTURE | Image Capture
DCM | LOG | Procedure Logging
DCM | RT | Radiation Therapy Device
DCM | WSD | Workstation

## CID 31 Abstract Priors

This Context Group includes codes that may be used to identify imaging procedures that may be referred to as priors for the purpose of image set selection in Hanging Protocols.

Type:

  Extensible

Version:

  20060126

Table CID 31. Abstract Priors

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-7292 | On admission
SRT | R-400B2 | Intraoperative
SRT | R-41FD9 | Pre-admission
SRT | R-411C0 | Pre-dose
SRT | R-404DA | Post-dose
SRT | R-413C5 | Pre-operative
SRT | R-413B7 | Post-operative
DCM | 109120 | On admission to unit
DCM | 109121 | On discharge
DCM | 109122 | On discharge from unit
DCM | 109123 | Pre-intervention
DCM | 109124 | Post-intervention
DCM | 109125 | At last appointment

## CID 42 Numeric Value Qualifier

Type:

  Extensible

Version:

  20020114

Table CID 42. Numeric Value Qualifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 114000 | Not a number
DCM | 114001 | Negative Infinity
DCM | 114002 | Positive Infinity
DCM | 114003 | Divide by zero
DCM | 114004 | Underflow
DCM | 114005 | Overflow
DCM | 114006 | Measurement failure
DCM | 114007 | Measurement not attempted
DCM | 114008 | Calculation failure
DCM | 114009 | Value out of range
DCM | 114010 | Value unknown
DCM | 114011 | Value indeterminate

## CID 50 Instance Availability Status

Type:

  Non-Extensible

Version:

  20090616

Table CID 50. Instance Availability Status

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | NEARLINE | Nearline
DCM | OFFLINE | Offline
DCM | ONLINE | Online
DCM | UNAVAILABLE | Unavailable

## CID 82 Units of Measurement

Not defined as a table of codes per se, but rather constructed from UCUM. See Section 7.2.2.

## CID 83 Units for Real World Value Mapping

Type:

  Extensible

Version:

  20080123

Table CID 83. Units for Real World Value Mapping

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 84 “PET Units”
UCUM | [hnsf'U] | Hounsfield unit

## CID 84 PET Units

Type:

  Extensible

Version:

  20141110

Table CID 84. PET Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 85 “SUV Units”
UCUM | {counts} | Counts
UCUM | {counts}/s | Counts per second
UCUM | {propcounts} | Proportional to counts
UCUM | {propcounts}/s | Proportional to counts per second
UCUM | cm2 | Centimeter**2
UCUM | % | Percent
UCUM | Bq/ml | Becquerels/milliliter
UCUM | mg/min/ml | Milligrams/minute/milliliter
UCUM | umol/min/ml | Micromole/minute/milliliter
UCUM | ml/min/g | Milliliter/minute/gram
UCUM | ml/g | Milliliter/gram
UCUM | /cm | /Centimeter
UCUM | umol/ml | Micromole/milliliter

## CID 85 SUV Units

Type:

  Extensible

Version:

  20141110

Table CID 85. SUV Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | {SUVbw}g/ml | Standardized Uptake Value body weight
UCUM | {SUVlbm}g/ml | Standardized Uptake Value lean body mass
UCUM | {SUVbsa}cm2/ml | Standardized Uptake Value body surface area
UCUM | {SUVibw}g/ml | Standardized Uptake Value ideal body weight

### Note

The formulas for the determination of SUVbw, SUVbsa, SUVlbm and SUVibw are defined in Sugawara et al. Reevaluation of the Standardized Uptake Value for FDG: Variations with Body Weight and Methods for Correction.Radiology, 1999 at http://radiology.rsna.org/content/213/2/521. The patient size correction factors are summarized here, where weight is in kg and height is in cm:

SUVbw: males & females: weight

SUVlbm: males: 1.10 * weight - 120 * (weight/height) ^2

females: 1.07 * weight - 148 * (weight/height) ^2

SUVbsa: males & females: weight^ 0.425 * height^0.725 * 0.007184

SUVibw: males: 48.0 + 1.06 * (height - 152)

females: 45.5 + 0.91 * (height - 152)

## CID 91 Functional Condition Present During Acquisition

Type:

  Extensible

Version:

  20100625

Table CID 91. Functional Condition Present During Acquisition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3271 “Hemodynamic Physiological Challenges”
SRT | F-F7100 | Phonation
SRT | F-12300 | Weight bearing
SRT | F-72230 | Voiding
DCM | 109134 | Prior to voiding
DCM | 109135 | Post voiding

## CID 92 Joint Position During Acquisition

Type:

  Extensible

Version:

  20100625

Table CID 92. Joint Position During Acquisition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109136 | Neutral musculoskeletal position
SRT | F-10110 | Flexion
SRT | F-10100 | Extension
SRT | F-10120 | Abduction
SRT | F-10130 | Adduction
SRT | F-10210 | Internal rotation
SRT | F-10220 | External rotation
SRT | F-10226 | Supination
SRT | F-10216 | Pronation
SRT | F-10240 | Torsion

## CID 93 Joint Positioning Method

Type:

  Extensible

Version:

  20100625

Table CID 93. Joint Positioning Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-13060 | Passive movement
SRT | P0-05083 | Manipulation of joint

## CID 94 Physical Force Applied During Acquisition

Type:

  Extensible

Version:

  20100625

Table CID 94. Physical Force Applied During Acquisition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P0-02160 | Traction - action
SRT | P0-021B2 | Compression - action
SRT | P0-021AB | Rotation - action

## CID 100 Quantitative Diagnostic Imaging Procedures

Type:

  Extensible

Version:

  20141110

Table CID 100. Quantitative Diagnostic Imaging Procedures

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P5-09051 | Magnetic resonance imaging guidance
DCM | 126020 | Multiparametric MRI
DCM | 126021 | Multiparametric MRI of prostate
DCM | 126022 | Multiparametric MRI of whole body
SRT | P5-0907F | Dynamic magnetic resonance imaging of knee
SRT | P5-70694 | Dynamic magnetic resonance imaging of pelvis
LN | 44139-4 | PET whole body
SRT | P5-080FF | PET/CT FDG imaging of whole body
SRT | P5-08118 | PET/CT MET imaging of whole body
RADLEX | RPID96 | CT HEAD PERFUSION WITH IV CONTRAST
RADLEX | RPID5258 | NM HEAD PERFUSION BRAIN SPECT
RADLEX | RPID5427 | NM HEAD PERFUSION BRAIN PET-CT AV-45

## CID 220 Level of Significance

Type:

  Extensible

Version:

  20030327

Table CID 220. Level of Significance

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-00333 | Most significant
SRT | R-0030C | Highly significant
SRT | R-10045 | Significant
SRT | R-00345 | Not significant
SRT | R-10046 | Significance Undetermined

## CID 221 Measurement Range Concepts

Type:

  Extensible

Version:

  20030327

Table CID 221. Measurement Range Concepts

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 226 “Population Statistical Descriptors”
Include CID 227 “Sample Statistical Descriptors”

## CID 222 Normality Codes

Type:

  Extensible

Version:

  20030327

Table CID 222. Normality Codes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A460 | Normal
SRT | R-42037 | Abnormal
SRT | R-002C4 | Abnormally High
SRT | R-002C5 | Abnormally Low
SRT | G-A385 | Normality Undetermined

## CID 223 Normal Range Values

Type:

  Extensible

Version:

  20030327

Table CID 223. Normal Range Values

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-0038B | Normal Range Upper Limit
SRT | R-10041 | Normal Range Lower Limit

## CID 224 Selection Method

Type:

  Extensible

Version:

  20030327

Table CID 224. Selection Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121410 | User chosen value
DCM | 121411 | Most recent value chosen
DCM | 121412 | Mean value chosen

## CID 225 Measurement Uncertainty Concepts

Type:

  Extensible

Version:

  20030327

Table CID 225. Measurement Uncertainty Concepts

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-00363 | +/-, range of measurement uncertainty
SRT | R-00364 | +, range of upper measurement uncertainty
SRT | R-00362 | -, range of lower measurement uncertainty

## CID 226 Population Statistical Descriptors

Type:

  Extensible

Version:

  20121101

Table CID 226. Population Statistical Descriptors

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-00337 | 95th Percentile Value of population
SRT | R-00338 | 90th Percentile Value of population
SRT | R-00346 | 1 Sigma Upper Value of population
SRT | R-00387 | 2 Sigma Upper Value of population
SRT | R-00317 | Mean Value of population
SRT | R-00319 | Median Value of population
SRT | R-00377 | 10th Percentile Value of population
SRT | R-00397 | 5th Percentile Value of population
SRT | R-00347 | 1 Sigma Lower Value of population
SRT | R-00388 | 2 Sigma Lower Value of population
DCM | 121414 | Standard deviation of population
DCM | 121417 | 2 Sigma deviation of population

### Note

The SNOMED meaning for R-00317 is "Mean - numeric estimation technique", but in the context of its use here, a more specific meaning has been used.

## CID 227 Sample Statistical Descriptors

Type:

  Extensible

Version:

  20030327

Table CID 227. Sample Statistical Descriptors

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121415 | Percentile Ranking of measurement
DCM | 121416 | Z-Score of measurement

## CID 228 Equation or Table

Type:

  Extensible

Version:

  20030327

Table CID 228. Equation or Table

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121420 | Equation
DCM | 121421 | Equation Citation
DCM | 121424 | Table of Values
DCM | 121422 | Table of Values Citation
DCM | 121423 | Method Citation

## CID 230 Yes-No

Type:

  Non-Extensible

Version:

  20060613

Table CID 230. Yes-No

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-0038D | Yes
SRT | R-00339 | No
SRT | R-0038A | Undetermined

## CID 240 Present-Absent

Type:

  Non-Extensible

Version:

  20050110

Table CID 240. Present-Absent

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A203 | Present
SRT | R-4089B | Absent
SRT | R-40271 | Presence Undetermined

## CID 242 Normal-Abnormal

This Context Group is a subset of CID 222 “Normality Codes”.

Type:

  Extensible

Version:

  20030327

Table CID 242. Normal-Abnormal

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A460 | Normal
SRT | R-42037 | Abnormal
SRT | G-A385 | Normality Undetermined

## CID 244 Laterality

Type:

  Non-Extensible

Version:

  20030108

Table CID 244. Laterality

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A100 | Right | 24028007 | C0205090
SRT | G-A101 | Left | 7771000 | C0205091
SRT | G-A102 | Right and left | 51440002 | C0238767
SRT | G-A103 | Unilateral | 66459002 | C0205092

## CID 250 Positive-Negative

Type:

  Extensible

Version:

  20040112

Table CID 250. Positive-Negative

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A200 | Positive
SRT | R-40759 | Negative

## CID 251 Severity of Complication

Type:

  Extensible

Version:

  20040112

Table CID 251. Severity of Complication

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-404F9 | Major
SRT | R-404FC | Minor

## CID 252 S-M-L Size Descriptor

CID 6118 is a superset of this Context Group.

Type:

  Extensible

Version:

  20080927

Table CID 252. S-M-L Size Descriptor

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-404A8 | Small
SRT | R-404A9 | Medium
SRT | R-404AA | Large

## CID 270 Observer Type

Type:

  Non-Extensible

Version:

  20040920

Table CID 270. Observer Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121006 | Person
DCM | 121007 | Device

## CID 271 Observation Subject Class

Type:

  Non-Extensible

Version:

  20071102

Table CID 271. Observation Subject Class

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121025 | Patient
DCM | 121026 | Fetus
DCM | 121027 | Specimen
DCM | 121192 | Device Subject

## CID 400 Audit Event ID

Type:

  Extensible

Version:

  20100826

Table CID 400. Audit Event Id

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110100 | Application Activity
DCM | 110101 | Audit Log Used
DCM | 110102 | Begin Transferring DICOM Instances
DCM | 110103 | DICOM Instances Accessed
DCM | 110104 | DICOM Instances Transferred
DCM | 110105 | DICOM Study Deleted
DCM | 110106 | Export
DCM | 110107 | Import
DCM | 110108 | Network Entry
DCM | 110112 | Query
DCM | 110113 | Security Alert
DCM | 110114 | User Authentication

## CID 401 Audit Event Type Code

Type:

  Extensible

Version:

  20100826

Table CID 401. Audit Event Type Code

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110120 | Application Start
DCM | 110121 | Application Stop
DCM | 110122 | Login
DCM | 110123 | Logout
DCM | 110124 | Attach
DCM | 110125 | Detach
DCM | 110126 | Node Authentication
DCM | 110127 | Emergency Override Started
DCM | 110128 | Network Configuration
DCM | 110129 | Security Configuration
DCM | 110130 | Hardware Configuration
DCM | 110131 | Software Configuration
DCM | 110132 | Use of Restricted Function
DCM | 110133 | Audit Recording Stopped
DCM | 110134 | Audit Recording Started
DCM | 110135 | Object Security Attributes Changed
DCM | 110136 | Security Roles Changed
DCM | 110137 | User Security Attributes Changed

## CID 402 Audit Active Participant Role ID Code

Type:

  Extensible

Version:

  20100826

Table CID 402. Audit Active Participant Role ID Code

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110150 | Application
DCM | 110151 | Application Launcher
DCM | 110152 | Destination Role ID
DCM | 110153 | Source Role ID
DCM | 110154 | Destination Media
DCM | 110155 | Source Media

## CID 403 Security Alert Type Code

Type:

  Extensible

Version:

  20100826

Table CID 403. Security Alert Type Code

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110126 | Node Authentication
DCM | 110127 | Emergency Override Started
DCM | 110128 | Network Configuration
DCM | 110129 | Security Configuration
DCM | 110130 | Hardware Configuration
DCM | 110131 | Software Configuration
DCM | 110132 | Use of Restricted Function
DCM | 110133 | Audit Recording Stopped
DCM | 110134 | Audit Recording Started
DCM | 110135 | Object Security Attributes Changed
DCM | 110136 | Security Roles Changed
DCM | 110137 | User Security Attributes Changed
DCM | 110138 | Emergency Override Stopped
DCM | 110139 | Remote Service Operation Started
DCM | 110140 | Remote Service Operation Stopped
DCM | 110141 | Local Service Operation Started
DCM | 110142 | Local Service Operation Stopped

## CID 404 Audit Participant Object ID Type Code

Type:

  Extensible

Version:

  20100621

Table CID 404. Audit Participant Object ID Type Code

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110180 | Study Instance UID
DCM | 110181 | SOP Class UID
DCM | 110182 | Node ID

## CID 405 Media Type Code

Type:

  Extensible

Version:

  20100824

Table CID 405. Media Type Code

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110030 | USB Disk Emulation
DCM | 110031 | Email
DCM | 110032 | CD
DCM | 110033 | DVD
DCM | 110034 | Compact Flash
DCM | 110035 | Multi-media Card
DCM | 110036 | Secure Digital Card
DCM | 110037 | URI
DCM | 110010 | Film
DCM | 110038 | Paper Document

## CID 3000 Audio Channel Source

Type:

  Extensible

Version:

  20040326

Table CID 3000. Audio Channel Source

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109110 | Voice
DCM | 109111 | Operator's narrative
DCM | 109112 | Ambient room environment
DCM | 109113 | Doppler audio
DCM | 109114 | Phonocardiogram
DCM | 109115 | Physiological audio signal

## CID 3001 ECG Leads

This Context Group comprises the ECG lead identifiers of ISO/IEEE 11073-10101, including human and canine leads. The terms included in the table below may not constitute the complete list; see the ISO/IEEE Standard.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10101 available through http://standards.ieee.org.

Type:

  Extensible

Version:

  20130613

Table CID 3001. ECG Leads

Coding Scheme | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:71 | Lead A (Nehb - Anterior) | MDC_ECG_LEAD_A
MDC | 2:75 | Auxiliary unipolar lead 1 | MDC_ECG_LEAD_A1
MDC | 2:76 | Auxiliary unipolar lead 2 | MDC_ECG_LEAD_A2
MDC | 2:77 | Auxiliary unipolar lead 3 | MDC_ECG_LEAD_A3
MDC | 2:78 | Auxiliary unipolar lead 4 | MDC_ECG_LEAD_A4
MDC | 2:127 | Auxiliary bipolar lead 1 | MDC_ECG_LEAD_AB1
MDC | 2:128 | Auxiliary bipolar lead 2 | MDC_ECG_LEAD_AB2
MDC | 2:129 | Auxiliary bipolar lead 3 | MDC_ECG_LEAD_AB3
MDC | 2:130 | Auxiliary bipolar lead 4 | MDC_ECG_LEAD_AB4
MDC | 2:133 | EASI Lead AI | MDC_ECG_LEAD_AI
MDC | 2:132 | EASI Lead AS | MDC_ECG_LEAD_AS
MDC | 2:64 | aVF, augmented voltage, foot | MDC_ECG_LEAD_AVF
MDC | 2:63 | aVL, augmented voltage, left | MDC_ECG_LEAD_AVL
MDC | 2:62 | aVR, augmented voltage, right | MDC_ECG_LEAD_AVR
MDC | 2:65 | −aVR | MDC_ECG_LEAD_AVRneg
MDC | 2:86 | Chest lead | MDC_ECG_LEAD_C
MDC | 2:124 | negative: low right scapula Lead | MDC_ECG_LEAD_CB5
MDC | 2:98 | Chest lead (symmetric placement) | MDC_ECG_LEAD_CC
MDC | 2:99 | Chest lead per V1 and V1R placement | MDC_ECG_LEAD_CC1
MDC | 2:100 | Chest lead per V2 and V2R placement | MDC_ECG_LEAD_CC2
MDC | 2:101 | Chest lead per V3 and V3R placement | MDC_ECG_LEAD_CC3
MDC | 2:102 | Chest lead per V4 and V4R placement | MDC_ECG_LEAD_CC4
MDC | 2:19 | Chest lead per V5 and V5R placement | MDC_ECG_LEAD_CC5
MDC | 2:103 | Chest lead per V6 and V6R placement | MDC_ECG_LEAD_CC6
MDC | 2:104 | Chest lead per V7 and V8R placement | MDC_ECG_LEAD_CC7
MDC | 2:122 | Lead CH5 | MDC_ECG_LEAD_CH5
MDC | 2:105 | Chest-manubrium lead | MDC_ECG_LEAD_CM
MDC | 2:106 | Chest-manubrium lead per V1 placement | MDC_ECG_LEAD_CM1
MDC | 2:107 | Chest-manubrium lead per V2 placement | MDC_ECG_LEAD_CM2
MDC | 2:108 | Chest-manubrium lead per V3 placement | MDC_ECG_LEAD_CM3
MDC | 2:109 | Chest-manubrium lead per V4 placement | MDC_ECG_LEAD_CM4
MDC | 2:20 | Chest-manubrium lead per V5 placement | MDC_ECG_LEAD_CM5
MDC | 2:110 | Chest-manubrium lead per V6 placement | MDC_ECG_LEAD_CM6
MDC | 2:121 | Chest-manubrium lead per V7 placement | MDC_ECG_LEAD_CM7
MDC | 2:125 | Lead CR5 | MDC_ECG_LEAD_CR5
MDC | 2:123 | negative: right infraclavicular fossa | MDC_ECG_LEAD_CS5
MDC | 2:148 | Canine, fifth right intercostal space near edge of sternum | MDC_ECG_LEAD_CV5RL
MDC | 2:149 | Canine, sixth left intercostal space near edge of sternum | MDC_ECG_LEAD_CV6LL
MDC | 2:150 | Canine, sixth left intercostal space at costochondral junction | MDC_ECG_LEAD_CV6LU
MDC | 2:70 | Lead D (Nehb - Dorsal) | MDC_ECG_LEAD_D
MDC | 2:114 | Derived Lead aVF | MDC_ECG_LEAD_dAVF
MDC | 2:113 | Derived Lead aVL | MDC_ECG_LEAD_dAVL
MDC | 2:112 | Derived Lead aVR | MDC_ECG_LEAD_dAVR
MDC | 2:73 | Defibrillator lead: anterior-lateral | MDC_ECG_LEAD_DEFIB
MDC | 2:31 | Derived Lead I | MDC_ECG_LEAD_dI
MDC | 2:32 | Derived Lead II | MDC_ECG_LEAD_dII
MDC | 2:111 | Derived Lead III | MDC_ECG_LEAD_dIII
MDC | 2:33 | Derived Lead V1 | MDC_ECG_LEAD_dV1
MDC | 2:34 | Derived Lead V2 | MDC_ECG_LEAD_dV2
MDC | 2:35 | Derived Lead V3 | MDC_ECG_LEAD_dV3
MDC | 2:36 | Derived Lead V4 | MDC_ECG_LEAD_dV4
MDC | 2:37 | Derived Lead V5 | MDC_ECG_LEAD_dV5
MDC | 2:38 | Derived Lead V6 | MDC_ECG_LEAD_dV6
MDC | 2:131 | EASI Lead ES | MDC_ECG_LEAD_ES
MDC | 2:74 | External pacing lead: anterior-posterior | MDC_ECG_LEAD_EXTERN
MDC | 2:27 | Frank Lead A | MDC_ECG_LEAD_fA
MDC | 2:26 | Frank Lead C | MDC_ECG_LEAD_fC
MDC | 2:25 | Frank Lead E | MDC_ECG_LEAD_fE
MDC | 2:29 | Frank Lead F | MDC_ECG_LEAD_fF
MDC | 2:30 | Frank Lead H | MDC_ECG_LEAD_fH
MDC | 2:24 | Frank Lead I | MDC_ECG_LEAD_fI
MDC | 2:28 | Frank Lead M | MDC_ECG_LEAD_fM
MDC | 2:1 | Lead I | MDC_ECG_LEAD_I
MDC | 2:2 | Lead II | MDC_ECG_LEAD_II
MDC | 2:61 | Lead III | MDC_ECG_LEAD_III
MDC | 2:72 | Lead J (Nehb - Inferior) | MDC_ECG_LEAD_J
MDC | 2:21 | Left Arm Lead | MDC_ECG_LEAD_LA
MDC | 2:23 | Left Leg Lead | MDC_ECG_LEAD_LL
MDC | 2:91 | Modified chest lead (left arm indifferent) | MDC_ECG_LEAD_MCL
MDC | 2:92 | Modified chest lead per V1 placement | MDC_ECG_LEAD_MCL1
MDC | 2:93 | Modified chest lead per V2 placement | MDC_ECG_LEAD_MCL2
MDC | 2:94 | Modified chest lead per V3 placement | MDC_ECG_LEAD_MCL3
MDC | 2:95 | Modified chest lead per V4 placement | MDC_ECG_LEAD_MCL4
MDC | 2:96 | Modified chest lead per V5 placement | MDC_ECG_LEAD_MCL5
MDC | 2:97 | Modified chest lead per V6 placement | MDC_ECG_LEAD_MCL6
MDC | 2:126 | Modified limb lead | MDC_ECG_LEAD_ML
MDC | 2:22 | Right Arm Lead | MDC_ECG_LEAD_RA
MDC | 2:147 | Right Leg Lead | MDC_ECG_LEAD_RL
MDC | 2:134 | EASI upper sternum lead | MDC_ECG_LEAD_S
MDC | 2:87 | Precordial lead | MDC_ECG_LEAD_V
MDC | 2:3 | Lead V1 | MDC_ECG_LEAD_V1
MDC | 2:151 | Canine, over dorsal spinous process of 7th thoracic vertebra | MDC_ECG_LEAD_V10
MDC | 2:4 | Lead V2 | MDC_ECG_LEAD_V2
MDC | 2:10 | Lead V2R | MDC_ECG_LEAD_V2R
MDC | 2:5 | Lead V3 | MDC_ECG_LEAD_V3
MDC | 2:11 | Lead V3R | MDC_ECG_LEAD_V3R
MDC | 2:6 | Lead V4 | MDC_ECG_LEAD_V4
MDC | 2:12 | Lead V4R | MDC_ECG_LEAD_V4R
MDC | 2:7 | Lead V5 | MDC_ECG_LEAD_V5
MDC | 2:13 | Lead V5R | MDC_ECG_LEAD_V5R
MDC | 2:8 | Lead V6 | MDC_ECG_LEAD_V6
MDC | 2:14 | Lead V6R | MDC_ECG_LEAD_V6R
MDC | 2:9 | Lead V7 | MDC_ECG_LEAD_V7
MDC | 2:15 | Lead V7R | MDC_ECG_LEAD_V7R
MDC | 2:66 | Lead V8 | MDC_ECG_LEAD_V8
MDC | 2:68 | Lead V8R | MDC_ECG_LEAD_V8R
MDC | 2:67 | Lead V9 | MDC_ECG_LEAD_V9
MDC | 2:69 | Lead V9R | MDC_ECG_LEAD_V9R
MDC | 2:90 | Lead VF, nonaugmented voltage, vector of LL | MDC_ECG_LEAD_VF
MDC | 2:89 | Lead VL, nonaugmented voltage, vector of LA | MDC_ECG_LEAD_VL
MDC | 2:88 | Lead VR, nonaugmented voltage, vector of RA | MDC_ECG_LEAD_VR
MDC | 2:16 | Lead X | MDC_ECG_LEAD_X
MDC | 2:17 | Lead Y | MDC_ECG_LEAD_Y
MDC | 2:18 | Lead Z | MDC_ECG_LEAD_Z
MDC | 2:0 | Unspecified lead | MDC_ECG_LEAD_CONFIG

### Note

1. A prior version of this context group used codes from the SCP-ECG vocabulary.

2. In a prior version of this table, the code 2:26 was specified for the concept Chest lead and the code 2:19 was specified for the concept Chest-manubrium lead per V5 placement.

## CID 3003 Hemodynamic Waveform Sources

Type:

  Extensible

Version:

  20020904

Table CID 3003. Hemodynamic Waveform Sources

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-DB22 | Aortic pressure waveform
SRT | G-DB31 | Aortic valve pullback pressure waveform
SRT | G-DB24 | Arterial pressure waveform
SRT | G-DB23 | Central venous pressure waveform
SRT | G-DB33 | Dye dilution cardiac output waveform
SRT | G-DB20 | Femoral artery pressure waveform
SRT | G-DB12 | Hemodynamic flow waveform
SRT | G-DB34 | Hemodynamic impedance waveform
SRT | G-DB13 | Hemodynamic oxygen saturation waveform
SRT | G-DB11 | Hemodynamic pressure waveform
SRT | G-DB10 | Hemodynamic waveform
SRT | G-DB19 | Left atrium pressure waveform
SRT | G-DB16 | Left ventricle pressure waveform
SRT | G-DB28 | Mitral valve pullback pressure waveform
SRT | G-DB25 | Pulmonary artery oxygen saturation waveform
SRT | G-DB21 | Pulmonary artery pressure waveform
SRT | G-DB27 | Pulmonary artery wedge pressure waveform
SRT | G-DB26 | Pulmonary capillary wedge pressure waveform
SRT | G-DB30 | Pulmonary valve pullback pressure waveform
SRT | G-DB14 | Respiration impedance waveform
SRT | G-DB18 | Right atrium pressure waveform
SRT | G-DB17 | Right ventricle pressure waveform
SRT | G-DB15 | Temperature waveform
SRT | G-DB32 | Thermal cardiac output waveform
SRT | G-DB29 | Tricuspid valve pullback pressure waveform

## CID 3004 Arterial Pulse Waveform

Type:

  Extensible

Version:

  20090409

Table CID 3004. Arterial Pulse Waveform

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109116 | Arterial Pulse Waveform

## CID 3005 Respiration Waveform

Type:

  Extensible

Version:

  20090409

Table CID 3005. Respiration Waveform

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109117 | Respiration Waveform

## CID 3010 Cardiovascular Anatomic Locations

Type:

  Extensible

Version:

  20130617

Table CID 3010. Cardiovascular Anatomic Locations

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-42500 | Abdominal aorta | 7832008 | C0003484
SRT | T-48503 | Anomalous pulmonary vein | 128585006 | C0265914
SRT | T-49215 | Antecubital vein | 128553008 | C1276271
SRT | T-48440 | Anterior cardiac vein | 43967001 | C0226661
SRT | T-45530 | Anterior communicating artery | 8012006 | C0149562
SRT | T-45730 | Anterior spinal artery | 17388009 | C0149603
SRT | T-47700 | Anterior tibial artery | 68053000 | C0085816
SRT | T-42000 | Aorta | 15825003 | C0003483
SRT | T-42300 | Aortic arch | 57034009 | C0003489
SRT | D3-81922 | Aortic fistula | 128551005 | C1290392
SRT | T-32602 | Apex of left ventricle | 128564006 | C0580781
SRT | T-32502 | Apex of right ventricle | 128565007 | C0445242
SRT | T-41000 | Artery | 51114001 | C0003842
SRT | T-42100 | Ascending aorta | 54247002 | C0003956
SRT | T-47100 | Axillary Artery | 67937003 | C0004455
SRT | T-49110 | Axillary vein | 68705008 | C0004456
SRT | T-48340 | Azygos vein | 72107004 | C0004526
SRT | A-00203 | Baffle | 128981007 | C1289790
SRT | T-45800 | Basilar artery | 59011009 | C0004811
SRT | T-D00AB | Body conduit | 91830000 | C1735317
SRT | T-49424 | Boyd's perforating vein | 128548003 | C1267522
SRT | T-47160 | Brachial artery | 17137000 | C0006087
SRT | T-49350 | Brachial vein | 20115005 | C0226812
SRT | T-45010 | Carotid Artery | 69105007 | C0007272
SRT | T-49240 | Cephalic vein | 20699002 | C0226802
SRT | T-45510 | Cerebral artery | 88556005 | C0007770
SRT | D4-31320 | Common atrium | 69524004 | C0018817
SRT | T-45100 | Common carotid artery | 32062004 | C0162859
SRT | T-46710 | Common iliac artery | 73634005 | C1261084
SRT | T-48920 | Common iliac vein | 46027005 | C0226758
SRT | D4-31120 | Common ventricle | 45503006 | C0152424
SRT | D4-32504 | Congenital coronary artery fistula to left atrium | 128555001 | C1290487
SRT | D4-32506 | Congenital coronary artery fistula to left ventricle | 128556000 | C1290488
SRT | D4-32509 | Congenital coronary artery fistula to right atrium | 128557009 | C1290489
SRT | D4-32510 | Congenital coronary artery fistula to right ventricle | 128558004 | C1290490
SRT | D3-40208 | Congenital pulmonary arteriovenous fistula | 111289009 | C0155675
SRT | T-43000 | Coronary artery | 41801008 | C0205042
SRT | T-48410 | Coronary sinus | 90219004 | C0456944
SRT | T-42400 | Descending aorta | 32672002 | C3163626
SRT | T-49429 | Dodd's perforating vein | 128554002 | C1267525
SRT | T-45200 | External carotid artery | 22286001 | C0007275
SRT | T-46910 | External iliac artery | 113269004 | C0226398
SRT | T-48930 | External iliac vein | 63507001 | C0226761
SRT | T-45240 | Facial artery | 23074001 | C0226109
SRT | T-47400 | Femoral artery | 7657000 | C0015801
SRT | T-49410 | Femoral vein | 83419000 | C0015809
SRT | T-48820 | Gastric vein | 110568007 | C0750610
SRT | T-47490 | Genicular artery | 128559007 | C0447108
SRT | T-48420 | Great cardiac vein | 5928000 | C0226659
SRT | T-46420 | Hepatic artery | 76015000 | C0019145
SRT | T-48720 | Hepatic vein | 8993003 | C0019155
SRT | T-4942A | Hunterian perforating vein | 128560002 | C1267526
SRT | T-46700 | Iliac artery | 10293006 | C0020887
SRT | T-48470 | Inferior cardiac vein | 60477001 | C1267462
SRT | T-48540 | Inferior left pulmonary vein | 51249003 | C0226686
SRT | T-46520 | Inferior mesenteric artery | 33795007 | C0162860
SRT | T-48520 | Inferior right pulmonary vein | 113273001 | C0226676
SRT | T-48710 | Inferior vena cava | 64131007 | C0042458
SRT | T-46010 | Innominate artery | 12691009 | C0006094
SRT | T-48620 | Innominate vein | 8887007 | C0006095
SRT | T-45300 | Internal carotid artery | 86117002 | C0007276
SRT | T-48170 | Internal jugular vein | 12123001 | C0226550
SRT | T-46740 | Internal iliac artery | 90024005 | C0226364
SRT | T-46200 | Internal mammary artery | 69327007 | C0226276
SRT | D4-31052 | Juxtaposed atrial appendage | 128563000 | C1290478
SRT | T-45410 | Lacrimal artery | 59749000 | C0226171
SRT | T-45416 | Lacrimal artery of right eye | 128979005 | C0923299
SRT | T-32300 | Left atrium | 82471001 | C0225860
SRT | T-32310 | Left auricular appendage | 33626005 | C0225861
SRT | T-47420 | Left femoral artery | 113270003 | C0226448
SRT | T-44400 | Left pulmonary artery | 50408007 | C0226069
SRT | T-32600 | Left ventricle | 87878005 | C0225897
SRT | T-32640 | Left ventricle inflow | 70238003 | C0225911
SRT | D4-31022 | Left ventricle outflow chamber | 128570000 | C1290476
SRT | T-32650 | Left ventricle outflow tract | 13418002 | C0225912
SRT | T-45230 | Lingual artery | 113264009 | C0226104
SRT | T-46960 | Lumbar artery | 34635009 | C0226408
SRT | T-46500 | Mesenteric artery | 86570000 | C0025465
SRT | T-4884A | Mesenteric vein | 128583004 | C0025473
SRT | T-45250 | Occipital artery | 31145008 | C0226117
SRT | T-48214 | Occipital vein | 32114007 | C0226579
SRT | T-45400 | Ophthalmic artery | 53549008 | C0029078
SRT | D4-32012 | Patent ductus arteriosus | 83330001 | C0013274
SRT | T-47630 | Peroneal artery | 8821006 | C0226476
SRT | T-47500 | Popliteal artery | 43899006 | C0032649
SRT | T-48810 | Portal vein | 32764006 | C0032718
SRT | T-45320 | Posterior communication artery | 43119007 | C0149559
SRT | T-49535 | Posterior medial tributary | 128569001 | C1267527
SRT | T-47600 | Posterior tibial artery | 13363002 | C0086835
SRT | T-F7001 | Primitive aorta | 14944004 | C0231136
SRT | T-F7040 | Primitive pulmonary artery | 91707000 | C0231157
SRT | T-44000 | Pulmonary artery | 81040000 | C0034052
SRT | D4-33142 | Pulmonary artery conduit | 128584005 | C1290491
SRT | T-32190 | Pulmonary chamber of cor triatriatum | 128586007 | C1267246
SRT | T-48581 | Pulmonary vein | 122972007 | C0034090
SRT | D4-33512 | Pulmonary vein confluence | 128566008 | C1290492
SRT | D4-33514 | Pulmonary venous atrium | 128567004 | C1290493
SRT | T-47300 | Radial artery | 45631007 | C0162857
SRT | T-46600 | Renal artery | 2841007 | C0035065
SRT | T-48740 | Renal vein | 56400007 | C0035092
SRT | T-32200 | Right atrium | 73829009 | C0225844
SRT | T-32210 | Right auricular appendage | 68300000 | C0225845
SRT | T-47410 | Right femoral artery | 69833005 | C0226447
SRT | T-44200 | Right pulmonary artery | 78480002 | C0226054
SRT | T-32500 | Right ventricle | 53085002 | C0225883
SRT | T-32540 | Right ventricle inflow | 8017000 | C0225891
SRT | D4-31032 | Right ventricle outflow chamber | 128572008 | C1290477
SRT | T-32550 | Right ventricle outflow tract | 44627009 | C0225892
SRT | T-D930A | Saphenofemoral junction | 128587003 | C0447132
SRT | T-49530 | Saphenous vein | 60734001 | C0036186
SRT | T-46460 | Splenic artery | 22083002 | C0037996
SRT | T-48890 | Splenic vein | 35819009 | C0038001
SRT | T-46100 | Subclavian artery | 36765005 | C0038530
SRT | T-48330 | Subclavian vein | 9454009 | C0038532
SRT | T-45270 | Superficial temporal artery | 15672000 | C0226130
SRT | T-48530 | Superior left pulmonary vein | 43863001 | C0226682
SRT | T-46510 | Superior mesenteric artery | 42258001 | C0162861
SRT | T-48510 | Superior right pulmonary vein | 8629005 | C0226671
SRT | T-45210 | Superior thyroid artery | 72021004 | C0226093
SRT | T-48610 | Superior vena cava | 48345005 | C0042459
SRT | T-44007 | Systemic collateral artery to lung | 128589000 | C0345096
SRT | D4-33516 | Systemic venous atrium | 128568009 | C1290494
SRT | T-42070 | Thoracic aorta | 113262008 | C1522460
SRT | D4-31400 | Truncus arteriosus communis | 61959006 | C0041207
SRT | T-46400 | Truncus coeliacus | 57850000 | C0007569
SRT | T-47200 | Ulnar artery | 44984001 | C0162858
SRT | T-F1810 | Umbilical artery | 50536004 | C0041632
SRT | T-48817 | Umbilical vein | 61660003 | C1267489
SRT | T-48000 | Vein | 29092000 | C0042449
SRT | T-48003 | Venous network | 34340008 | C0226503
SRT | T-45700 | Vertebral artery | 85234005 | C0042559

## CID 3011 Electrophysiology Anatomic Locations

Type:

  Extensible

Version:

  20130617

Table CID 3011. Electrophysiology Anatomic Locations

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-32850 | Accessory atrioventricular bundle | 8225009 | C0006383
SRT | T-32602 | Apex of left ventricle | 128564006 | C0580781
SRT | T-32502 | Apex of right ventricle | 128565007 | C0445242
SRT | T-32830 | Atrioventricular bundle | 345000 | C0006382
SRT | T-32820 | Atrioventricular node | 25943004 | C0004247
SRT | T-32400 | Common ventricle | 21814001 | C0018827
SRT | T-48410 | Coronary sinus | 90219004 | C0456944
SRT | T-39010 | Epicardium | 6871001 | C0225968
SRT | T-48420 | Great cardiac vein | 5928000 | C0226659
SRT | G-DE02 | High right atrium | 128591008 | C0456955
SRT | T-48540 | Inferior left pulmonary vein | 51249003 | C0226686
SRT | T-48520 | Inferior right pulmonary vein | 113273001 | C0226676
SRT | G-DE04 | Lateral high right atrium | 128592001 | C1264751
SRT | T-32833 | Left anterior division of left branch of left atrioventricular bundle | 84654008 | C0225918
SRT | T-32300 | Left Atrium | 82471001 | C0225860
SRT | T-32310 | Left auricular appendage | 33626005 | C0225861
SRT | T-32832 | Left branch of atrioventricular bundle | 74031005 | C0459156
SRT | T-32834 | Left posterior division of left branch of left atrioventricular bundle | 91085002 | C0225919
SRT | T-32600 | Left ventricle | 87878005 | C0225897
SRT | T-32640 | Left ventricle inflow | 70238003 | C0225911
SRT | T-32650 | Left ventricle outflow tract | 13418002 | C0225912
SRT | G-DE08 | Low right atrium | 128594000 | C0456956
SRT | G-DE06 | Mid right atrium | 128593006 | C0225856
SRT | T-48430 | Middle cardiac vein | 73580002 | C0226660
SRT | T-35310 | Mitral ring | 65197004 | C0225947
SRT | T-48411 | Ostium of coronary sinus | 71271007 | C0226656
SRT | T-48581 | Pulmonary vein | 122972007 | C0034090
SRT | T-35210 | Pulmonic ring | 90318009 | C0225935
SRT | T-32840 | Purkinje fibers | 13050003 | C0034144
SRT | T-35120 | Right atrioventricular ostium | 90561006 | C0225927
SRT | T-32200 | Right Atrium | 73829009 | C0225844
SRT | T-32210 | Right auricular appendage | 68300000 | C0225845
SRT | T-32831 | Right branch of Atrioventricular bundle | 57383004 | C0225916
SRT | T-32500 | Right ventricle | 53085002 | C0225883
SRT | T-32540 | Right ventricle inflow | 8017000 | C0225891
SRT | T-32550 | Right ventricle outflow tract | 44627009 | C0225892
SRT | T-32810 | Sino-atrial node | 88210001 | C0037189
SRT | T-48530 | Superior left pulmonary vein | 43863001 | C0226682
SRT | T-48510 | Superior right pulmonary vein | 8629005 | C0226671
SRT | T-32202 | Tendon of Todaro | 128595004 | C0456939
SRT | T-35110 | Tricuspid ring | 113259005 | C0225926

### Note

In a prior version of this Context Group the code T-48500 rather than T-48581 was defined for the concept Pulmonary Vein; this was inconsistent with the DICOM approach of selecting the "structure of" rather than "entire" concept. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 3014 Coronary Artery Segments

Type:

  Extensible

Version:

  20130403

Table CID 3014. Coronary Artery Segments

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID | Equivalent BARI Code
--- | --- | --- | --- | --- | ---
SRT | T-43117 | 1st Diagonal Coronary Artery | 91750005 | C0524430 | 15
SRT | T-4312B | 1st Left Posterolateral Coronary Artery | 91757008 | C0524437 | 24
SRT | T-43128 | 1st Marginal Coronary Artery | 91754001 | C0524434 | 20
SRT | T-43213 | 1st Right posterolateral Coronary Artery | 91761002 | C0524441 | 6
SRT | T-43002 | 1st Septal Coronary Artery | 244251006 | C0447058 | 17
SRT | T-43118 | 2nd Diagonal Coronary Artery | 91751009 | C0524431 | 16
SRT | T-4312C | 2nd Left Posterolateral Coronary Artery | 91758003 | C0524438 | 25
SRT | T-43129 | 2nd Marginal Coronary Artery | 91755000 | C0524435 | 21
SRT | T-43214 | 2nd Right posterolateral Coronary Artery | 91762009 | C0524442 | 7
SRT | T-43119 | 3rd diagonal Coronary Artery | 91752002 | C0524432 | 29
SRT | T-4312D | 3rd Left Posterolateral Coronary Artery | 91759006 | C0524439 | 26
SRT | T-4312A | 3rd Marginal Coronary Artery | 91756004 | C0524436 | 22
SRT | T-43215 | 3rd Right posterolateral Coronary Artery | 91763004 | C0524443 | 8
SRT | T-43230 | Marginal Coronary Artery | 22765000 | C0226050 | 10
SRT | T-43124 | AV groove continuation of Circumflex Artery | 75902001 | C0226041 | 23
SRT | T-43122 | Distal Circumflex Coronary Artery | 6511003 | C0226039 | 19A
SRT | T-43112 | Distal Left Anterior Descending Coronary Artery | 36672000 | C0226034 | 14
SRT | T-43202 | Distal Right Coronary Artery | 41879009 | C0226044 | 3
BARI | 15A | 1st Diagonal Coronary Artery Laterals |  |  | 15A
BARI | 20A | 1st Marginal Coronary Artery Laterals |  |  | 20A
BARI | 16A | 2nd Diagonal Coronary Artery Laterals |  |  | 16A
BARI | 21A | 2nd Marginal Coronary Artery Laterals |  |  | 21A
BARI | 29A | 3rd Diagonal Coronary Artery Laterals |  |  | 29A
BARI | 22A | 3rd Marginal Coronary Artery Laterals |  |  | 22A
BARI | 28A | Ramus Laterals |  |  | 28A
SRT | T-43107 | Left Main Coronary Artery | 3227004 | C0226031 | 11
SRT | T-43105 | Left Main Coronary Artery Ostium | 76862008 | C0226030 | 11A
SRT | T-43126 | Left Posterior Descending Artery | 56322004 | C0278433 | 27
SRT | T-43127 | Mid Circumflex Coronary Artery | 91753007 | C0524433 | 19
SRT | T-43115 | Mid Left Anterior Descending Coronary Artery | 91748002 | C0524428 | 13
SRT | T-D6515 | Mid Right Coronary Artery | 450960006 | C3472627 | 2
SRT | T-43210 | Posterior Descending Right Coronary Artery | 53655008 | C0226047 | 4
BARI | 9 | Posterior descending septal perforators |  |  | 9
SRT | T-43121 | Proximal Circumflex Coronary Artery | 52433000 | C0226038 | 18
SRT | T-43111 | Proximal Left Anterior Descending Coronary Artery | 68787002 | C0226033 | 12
SRT | T-43201 | Proximal Right Coronary Artery | 91083009 | C0226043 | 1
SRT | T-43003 | Intermediate Artery (Ramus) | 244252004 | C0447059 | 28
SRT | T-43205 | Right Coronary Artery Ostium | 56789007 | C0226045 | 1A
SRT | T-43212 | Right posterior AV Coronary Artery | 12800002 | C0226048 | 5

### Note

In prior editions, this Context Group included BARI [1992] codes as the primary set. These have been replaced with equivalent SNOMED codes for the major artery segments (see PS3.16-2011).

## CID 3015 Coronary Arteries

Type:

  Non-Extensible

Version:

  20130403

Table CID 3015. Coronary Arteries

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 3014 “Coronary Artery Segments”
SRT | T-43110 | Left Anterior Descending Coronary Artery | 59438005 | C0226032
SRT | T-43200 | Right Coronary Artery | 13647002 | C1261316
SRT | T-43120 | Circumflex Coronary Artery | 57396003 | C0226037
SRT | T-43125 | Left Posterolateral Circumflex Coronary Artery | 57823005 | C0278432
SRT | T-4312E | Left Posterior Descending Circumflex Coronary Artery | 91760001 | C0524440
SRT | T-41065 | Coronary Artery Graft | 264293000 | C0440761

## CID 3016 Major Coronary Arteries

Type:

  Non-Extensible

Version:

  20110818

Table CID 3016. Major Coronary Arteries

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-43110 | Left Anterior Descending Coronary Artery | 59438005 | C0226032
SRT | T-43200 | Right Coronary Artery | 13647002 | C1261316
SRT | T-43120 | Circumflex Coronary Artery | 57396003 | C0226037
SRT | T-43107 | Left Main Coronary Artery | 3227004 | C0226031

## CID 3019 Cardiovascular Anatomic Location Modifiers

Type:

  Extensible

Version:

  20110124

Table CID 3019. Cardiovascular Anatomic Location Modifiers

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | G-A105 | Anterior | 36592002 | C0205094
SRT | G-D873 | Arterial graft to cited segment | 128950003 | C1264698
SRT | G-A110 | Central | 26216008 | C0205099
SRT | G-A119 | Distal | 46053002 | C0205108
SRT | G-D870 | Graft to cited segment, body | 128947001 | C1264695
SRT | G-D872 | Graft to cited segment, distal anastomosis | 128948006 | C1264697
SRT | G-D871 | Graft to cited segment, proximal anastomosis | 128949003 | C1264696
SRT | G-A115 | Inferior | 36885000 | C0205104
SRT | G-A104 | Lateral | 49370004 | C0205093
SRT | G-A101 | Left | 7771000 | C0205091
SRT | T-3215A | Ostium | 128946005 | C0444567
SRT | G-A106 | Posterior | 82583001 | C0205095
SRT | G-A118 | Proximal | 40415009 | C0205107
SRT | G-A100 | Right | 24028007 | C0205090
SRT | G-A116 | Superior | 43014004 | C1282910
SRT | G-D874 | Venous graft to cited segment | 128951004 | C1264699
SRT | T-40003 | Entire Vessel | 361097006 | C1283786
DCM | 122101 | Aneurysm on cited vessel |  |
DCM | 122102 | Graft to cited segment, proximal section |  |
DCM | 122103 | Graft to cited segment, mid section |  |
DCM | 122104 | Graft to cited segment, distal section |  |

## CID 3082 Cardiology Units of Measurement (Retired)

This Context Group was a subset of CID 82 “Units of Measurement”, and is retired. See PS3.16-2011.

## CID 3083 Units of Radioactivity

Type:

  Extensible

Version:

  20080927

Table CID 3083. Units of Radioactivity

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | Bq | becquerel
UCUM | MBq | megabecquerel
UCUM | mCi | millicurie

## CID 3090 Time Synchronization Channel Types

Type:

  Extensible

Version:

  20020904

Table CID 3090. Time Synchronization Channel Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109001 | Digital timecode (NOS)
DCM | 109002 | ECG-based gating signal, processed
DCM | 109003 | IRIG-B timecode
DCM | 109004 | X-Ray Fluoroscopy On Signal
DCM | 109005 | X-Ray On Trigger

## CID 3101 Cardiac Procedural State Values

Type:

  Extensible

Version:

  20091021

Table CID 3101. Cardiac Procedural State Values

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01604 | Resting State
SRT | F-05019 | Cardiac Stress State
DCM | 109092 | Reinjection State
DCM | 109093 | Redistribution State
DCM | 109094 | Delayed Redistribution State

## CID 3102 Rest-Stress

Type:

  Extensible

Version:

  20080927

Table CID 3102. Rest-Stress

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01604 | Resting State
DCM | 109091 | Cardiac Stress State

## CID 3104 Cardiac Synchronization Technique

This Context Group corresponds to the Enumerated Values of Cardiac Synchronization Technique (0018,9037) (see PS3.3).

Type:

  Non-Extensible

Version:

  20111028

Table CID 3104. Cardiac Synchronization Technique

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
DCM | 109080 | Real time acquisition
DCM | 109081 | Prospective gating
DCM | 109082 | Retrospective gating
DCM | 109083 | Paced

## CID 3106 PET Cardiology Protocols

Type:

  Extensible

Version:

  20080927

Table CID 3106. PET Cardiology Protocols

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122791 | PET Myocardial Perfusion, Rest only
DCM | 122792 | PET Myocardial Perfusion, Stress only
DCM | 122793 | PET Myocardial Perfusion, Rest and Stress
DCM | 122795 | PET Myocardial Viability, Rest only
DCM | 122796 | PET Myocardial Viability, Stress only
DCM | 122797 | PET Myocardial Viability, Rest and Stress

## CID 3107 PET Cardiology Radiopharmaceuticals

Type:

  Extensible

Version:

  20080927

Table CID 3107. PET Cardiology Radiopharmaceuticals

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-B1031 | Fluorodeoxyglucose F^18^
SRT | C-107A1 | ^13^Nitrogen
SRT | C-159A2 | ^82^Rubidium

## CID 3108 NM/PET Procedures

Type:

  Extensible

Version:

  20140419

Table CID 3108. NM/PET Procedures

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P5-D30F8 | Nuclear medicine cardiovascular study
SRT | P5-0A006 | PET heart study
SRT | P5-D6000 | Radioisotope study of endocrine system
SRT | P5-D6500 | Radioisotope study of hematopoietic system
SRT | P5-D5000 | Radioisotope study of gastrointestinal system
SRT | P5-D0063 | Radionuclide study for localization of inflammatory disease
SRT | P5-D10F8 | Nuclear medicine diagnostic procedure on musculoskeletal system
SRT | P5-D90F8 | Nuclear medicine diagnostic procedure on nervous system
SRT | P5-D0040 | Radionuclide localization of tumor
SRT | P5-D2000 | Radioisotope study of respiratory system
SRT | P5-D7000 | Radioisotope study of genitourinary system
SRT | P5-0A001 | PET brain study
SRT | P5-0A00D | PET breast study
SRT | P5-0A00A | PET study for localization of tumor

## CID 3110 Nuclear Cardiology Protocols

Type:

  Extensible

Version:

  20080927

Table CID 3110. Nuclear Cardiology Protocols

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P5-D300B | Stress thallium procedure
DCM | 122781 | Rest thallium/stress technetium procedure
DCM | 122782 | Rest technetium/stress technetium 1 day procedure
DCM | 122783 | Rest technetium/stress technetium 2 day procedure
DCM | 122784 | Stress technetium/rest technetium 1 day procedure
DCM | 122785 | NM Myocardial Viability procedure

## CID 3111 Nuclear Cardiology Radiopharmaceuticals

Type:

  Extensible

Version:

  20080927

Table CID 3111. Nuclear Cardiology Radiopharmaceuticals

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-B1130 | Thallium-201
SRT | C-B10A2 | Tc-99m sestamibi
SRT | C-B10A4 | Tc-99m tetrofosmin

## CID 3112 Attenuation Correction

Type:

  Extensible

Version:

  20080927

Table CID 3112. Attenuation Correction

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122726 | Algorithmic Attenuation Correction
DCM | 122727 | NM Transmission Attenuation Correction
DCM | 122728 | CT-based Attenuation Correction
DCM | 122729 | No Attenuation Correction

## CID 3113 Types of Perfusion Defects

Type:

  Extensible

Version:

  20080927

Table CID 3113. Types of Perfusion Defects

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-3014D | Reversible myocardial perfusion defect
SRT | F-3014F | Fixed myocardial perfusion defect
SRT | F-3014E | Partially Reversible myocardial perfusion defect
DCM | 122748 | False Positive defect finding

## CID 3114 Study Quality

Type:

  Extensible

Version:

  20080927

Table CID 3114. Study Quality

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122740 | Excellent image quality
DCM | 122741 | Good image quality
DCM | 122742 | Poor image quality
DCM | 111235 | Unusable - Quality renders image unusable

## CID 3115 Stress Imaging Quality Issues

Type:

  Extensible

Version:

  20080927

Table CID 3115. Stress Imaging Quality Issues

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111210 | Motion blur
DCM | 122743 | Body habitus attenuation
DCM | 122744 | Breast attenuation
DCM | 122745 | Diaphragmatic attenuation
SRT | F-04FD3 | Subdiaphragmatic uptake

## CID 3116 NM Extracardiac Findings

Type:

  Extensible

Version:

  20080927

Table CID 3116. NM Extracardiac Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-04FA0 | Normal extracardiac uptake
SRT | F-04FB8 | Increased lung uptake
SRT | F-04FE3 | Abnormal extracardiac uptake

## CID 3117 Attenuation Correction Methods

Type:

  Extensible

Version:

  20080927

Table CID 3117. Attenuation Correction Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122720 | OSEM algorithm
DCM | 122721 | Chang method

## CID 3118 Level of Risk

Type:

  Extensible

Version:

  20080927

Table CID 3118. Level of Risk

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-4044 | Normal risk
SRT | G-4041 | Low risk
SRT | G-4045 | Low to moderate risk
SRT | G-4042 | Moderate risk
SRT | G-4046 | Moderate to high risk
SRT | G-4043 | High risk
SRT | G-A648 | Uncertain risk

## CID 3119 LV Function

Type:

  Extensible

Version:

  20080927

Table CID 3119. LV Function

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A460 | Normal
SRT | F-300FA | Impaired left ventricular function

## CID 3120 Perfusion Findings

Type:

  Extensible

Version:

  20080927

Table CID 3120. Perfusion Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-30172 | Myocardial perfusion normal
SRT | G-A466 | Equivocal
SRT | R-42037 | Abnormal

## CID 3121 Perfusion Morphology

Type:

  Extensible

Version:

  20080927

Table CID 3121. Perfusion Morphology

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D3-1070D | Myocardial ischemia
SRT | D3-15000 | Myocardial Infarction
SRT | D3-10711 | Mixed Ischemia and Infarction

## CID 3122 Ventricular Enlargement

Type:

  Extensible

Version:

  20080927

Table CID 3122. Ventricular Enlargement

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-00343 | Normal size cardiac chamber
SRT | R-0032A | Mildly enlarged cardiac chamber
SRT | R-00331 | Moderately enlarged cardiac chamber
SRT | R-00316 | Markedly enlarged cardiac chamber

## CID 3200 Stress Test Procedure

Type:

  Extensible

Version:

  20080927

Table CID 3200. Stress Test Procedure

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | P0-006E4 | Exercise stress test | 165079009
SRT | P2-31107 | Pharmacologic stress test | 424064009
SRT | P2-31011 | Pharmacologic and exercise stress test | 428813002
SRT | P2-3110B | Paced stress test | 428685003

## CID 3201 Indications for Stress Test

Type:

  Extensible

Version:

  20080927

Table CID 3201. Indications for Stress Test

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-37000 | Chest Pain
SRT | R-413C5 | Pre-operative
SRT | D3-13040 | Coronary Artery Disease
SRT | D3-16000 | Heart failure
SRT | F-03C97 | Heart disease risk factors
SRT | F-201B3 | Dyspnea
SRT | R-00357 | Post PTCA
SRT | G-03A5 | History of CABG
SRT | F-00103 | Abnormal exercise tolerance test
SRT | F-38002 | Abnormal ECG
SRT | D3-30000 | Arrhythmia
SRT | D3-13012 | Angina pectoris
SRT | D3-02000 | Hypertension
SRT | F-37150 | Palpitations
SRT | D3-31290 | Supraventricular tachycardia
SRT | D3-00006 | Syncope
SRT | G-03AA | History of Myocardial Infarction
SRT | D3-33120 | Left bundle branch block
SRT | D3-10800 | Valvular heart disease
SRT | P7-00044 | Occupational requirement

## CID 3202 Chest Pain

Type:

  Extensible

Version:

  20080927

Table CID 3202. Chest Pain

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D3-13020 | Stable Angina
SRT | D3-12700 | Unstable Angina
SRT | R-0038F | Atypical Angina
SRT | F-37015 | Noncardiac Chest Pain
SRT | F-A265A | Chest pain not present
SRT | D3-13037 | Typical Angina
DCM | 122799 | Anginal Equivalent

## CID 3203 Exerciser Device

Type:

  Extensible

Version:

  20080927

Table CID 3203. Exerciser Device

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-17230 | Bicycle ergometer
SRT | A-17222 | Treadmill
SRT | A-1002A | Arm ergometer

## CID 3204 Stress Agents

Type:

  Extensible

Version:

  20080927

Table CID 3204. Stress Agents

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent | Trade Name (Informative)
--- | --- | --- | --- | ---
SRT | C-81590 | Dipyridamole | 66859009 | Persantine
SRT | C-68030 | Dobutamine | 26523005 |
SRT | C-80349 | Adenosine | 108502004 |
SRT | C-67770 | Atropine | 73949004 |
SRT | C-80012 | Adenosine A2 receptor agonist | 432062000 | Regadenoson

## CID 3205 Indications for Pharmacological Stress Test

Type:

  Extensible

Version:

  20080927

Table CID 3205. Indications for Pharmacological Stress Test

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D3-33120 | Left bundle branch block
SRT | R-0039E | Patient has pacemaker
SRT | DA-26000 | Paralytic syndrome
SRT | F-A4580 | Ataxia or incoordination
SRT | D3-8005B | Peripheral vascular disease
SRT | D2-50000 | Pulmonary disease
SRT | F-18002 | Gait problem
SRT | F-A0846 | Transient limb paralysis
SRT | F-01380 | Asthenia (debility)
SRT | F-029F7 | Cachexia
SRT | DD-13000 | Fracture of lower limb
SRT | DD-33500 | Open wound of lower limb
SRT | G-02BD | Lower limb amputation
SRT | G-0202 | Request by Physician
SRT | S-20570 | Dependence on enabling machine or device
SRT | G-044D | Recent Myocardial infarction
SRT | F-33019 | Cannot reach target heart rate
DCM | 122764 | Patient weight exceeds equipment limit

## CID 3206 Non-invasive Cardiac Imaging Procedures

Type:

  Extensible

Version:

  20080927

Table CID 3206. Non-Invasive Cardiac Imaging Procedures

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P5-D30F8 | Nuclear medicine cardiovascular study
SRT | P5-D3304 | Cardiac blood pool imaging (nuclear)
SRT | P5-0A006 | PET heart study
SRT | P5-0A100 | SPECT
SRT | P5-B3000 | Echocardiography
SRT | P5-09011 | Cardiac MRI

## CID 3207 Stress Test Procedure Phases

Type:

  Extensible

Version:

  20081031

Table CID 3207. Stress Test Procedure Phases

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01604 | Resting State
SRT | F-05019 | Cardiac stress state
SRT | F-05028 | Peak cardiac stress state
SRT | F-05018 | Cardiac stress recovery state
SRT | F-25040 | Hyperventilation

## CID 3208 Summary Codes Exercise ECG

Type:

  Extensible

Version:

  20080927

Table CID 3208. Summary Codes Exercise ECG

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-00101 | Exercise ECG normal
SRT | F-00103 | Exercise ECG abnormal
SRT | F-201B6 | Exercise ECG equivocal
SRT | R-4135B | Not performed

## CID 3209 Summary Codes Stress Imaging

Type:

  Extensible

Version:

  20080927

Table CID 3209. Summary Codes Stress Imaging

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-04AB2 | Imaging result normal
SRT | F-04AB3 | Imaging result abnormal
SRT | F-04A13 | Imaging result equivocal
SRT | R-4135B | Not performed

## CID 3210 Speed of Response

Type:

  Extensible

Version:

  20080927

Table CID 3210. Speed of Response

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A460 | normal
SRT | R-40AA8 | accentuated
SRT | R-40AA7 | blunted

## CID 3211 BP Response

Type:

  Extensible

Version:

  20080927

Table CID 3211. BP Response

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A460 | normal
SRT | D3-04000 | Hypotensive
SRT | D3-02000 | Hypertensive
SRT | R-40AA7 | blunted

## CID 3212 Treadmill Speed

Type:

  Extensible

Version:

  20080927

Table CID 3212. Treadmill Speed

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | km/h | km/h
UCUM | [mi_i]/h | mph

## CID 3213 Stress Hemodynamic Findings

Type:

  Extensible

Version:

  20080927

Table CID 3213. Stress Hemodynamic Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D3-0400A | Exertional hypotension
SRT | D3-0200B | Exertional hypertension
SRT | F-380B2 | Chronotropic incompetence

## CID 3215 Perfusion Finding Method

Type:

  Extensible

Version:

  20080927

Table CID 3215. Perfusion Finding Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-41D8B | ECG analysis
SRT | P3-41910 | Image analysis

## CID 3217 Comparison Finding

Type:

  Extensible

Version:

  20080927

Table CID 3217. Comparison Finding

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122775 | Agreement with prior findings
DCM | 122776 | Disagreement with prior findings

## CID 3220 Stress Symptoms

Type:

  Extensible

Version:

  20080927

Table CID 3220. Stress Symptoms

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-201B3 | Dyspnea
SRT | F-18010 | Claudication
SRT | D3-00006 | Syncope
SRT | D0-30017 | Flushing
SRT | F-52760 | Nausea
SRT | F-06017 | Dizziness
SRT | F-01360 | Fatigue
SRT | F-37000 | Chest pain
SRT | F-37006 | Chest discomfort
Include CID 3202 “Chest Pain”

## CID 3221 Stress Test Termination Reasons

Type:

  Extensible

Version:

  20080927

Table CID 3221. Stress Test Termination Reasons

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-37000 | Chest pain
SRT | F-38002 | Abnormal ECG
SRT | F-01360 | Fatigue
SRT | F-201B3 | Dyspnea
SRT | R-214DD | Patient Refused exercise test
SRT | F-021E1 | Target Heart Rate Achieved
SRT | D3-04001 | Hypotensive episode
SRT | D3-02004 | Hypertensive episode
SRT | D3-30000 | Arrhythmia
SRT | F-18010 | Claudication
SRT | R-4038D | End of Protocol
SRT | D3-00006 | Syncope

## CID 3227 QTc Measurements

This Context Group include both global and per lead corrected QT measurements specified in the ISO/IEEE 11073-10102 MDC nomenclature. Note that the MDC code for the per lead measurement is a base code for post-coordination with separately encoded lead identifiers. MDC also defines pre-coordinated codes that include both the measurement and the lead, which may be used in the same context as this Context Group; see the ISO/IEEE Standard.

While this Context Group includes distinct codes for the various QT correction algorithms, templates using this Context Group may allow post-coordination using the QTc algorithm codes of CID 3678 “Qt Correction Algorithms”.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

Table CID 3227. QTc Measurements

Coding Scheme | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:15876 | QTc interval global | MDC_ECG_TIME_PD_QTC
MDC | 2: 33792 | QTc interval per lead | MDC_ECG_TIME_PD_QTC_<lead>
MDC | 2:15880 | QTc global using Bazett formula | MDC_ECG_TIME_PD_QTC_BAZETT
MDC | 2:15880 | QTc global using Framingham formula | MDC_ECG_TIME_PD_QTC_FRAMINGHAM
MDC | 2:15892 | QTc global using Fredericia formula | MDC_ECG_TIME_PD_QTC_FREDERICA
MDC | 2:15892 | QTc global using Hodges formula | MDC_ECG_TIME_PD_QTC_HODGES
MDC | 2:34048 | QTc per lead using Bazett formula | MDC_ECG_TIME_PD_QTcB_<lead>
MDC | 2:34304 | QTc per lead using Fredericia formula | MDC_ECG_TIME_PD_QTcF_<lead>

## CID 3228 ECG Timing Measurements

This Context Group include both global and per lead ECG measurements specified in the ISO/IEEE 11073-10102 MDC nomenclature. Note that the MDC codes for "per lead" measurements are base codes for post-coordination with separately encoded lead identifiers. MDC also defines pre-coordinated codes that include both the measurement and the lead, which may be used in the same context as this Context Group; see the ISO/IEEE Standard.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20080927

Table CID 3228. ECG Timing Measurements

Coding Scheme | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:15872 | PR interval global | MDC_ECG_TIME_PD_PR
MDC | 2:16160 | QT interval global | MDC_ECG_TIME_PD_QT
MDC | 2:16156 | QRS duration global | MDC_ECG_TIME_PD_QRS
MDC | 2:16184 | P duration global | MDC_ECG_TIME_PD_P
MDC | 2:16140 | PP interval global | MDC_ECG_TIME_PD_PP
MDC | 2:16168 | RR interval global | MDC_ECG_TIME_PD_RR
MDC | 2:7168 | PR interval per lead | MDC_ECG_TIME_PD_PR_<lead>
MDC | 2:8192 | QT interval per lead | MDC_ECG_TIME_PD_QT_<lead>
MDC | 2:7936 | QRS duration per lead | MDC_ECG_TIME_PD_QRS_<lead>
MDC | 2:6656 | P duration per lead | MDC_ECG_TIME_PD_P_<lead>
MDC | 2:32768 | PP interval per lead | MDC_ECG_TIME_PD_PP_<lead>
MDC | 2:33024 | RR interval per lead | MDC_ECG_TIME_PD_RR_<lead>

## CID 3229 ECG Axis Measurements

This Context Group comprises the ECG axis measurements of ISO/IEEE 11073-10102. The terms included in the table below may not constitute the complete list; see the ISO/IEEE Standard.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20080927

Table CID 3229. ECG Axis Measurements

Coding Scheme | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:16132 | QRS axis | MDC_ECG_ANGLE_QRS_FRONT
MDC | 2:16128 | P Axis | MDC_ECG_ANGLE_P_FRONT
MDC | 2:16136 | T axis | MDC_ECG_ANGLE_T_FRONT

## CID 3230 ECG Findings

### Note

MDC Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20080927

Table CID 3230. ECG Findings

Coding Scheme | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
SRT | F-000B7 | Normal | MDC_ECG_BEAT_NORMAL
SRT | D3-30A03 | Atrial premature contraction | MDC_ECG_BEAT_ATR_P_C
SRT | D3-31740 | Ventricular premature contraction | MDC_ECG_BEAT_V_P_C
SRT | D3-31520 | Atrial Fibrillation | MDC_ECG_RHY_ATR_FIB
SRT | D3-31290 | Supraventricular Tachycardia | MDC_ECG_RHY_SV_TACHY
SRT | D3-31710 | Non-sustained ventricular tachycardia | MDC_ECG_RHY_V_TACHY_PAROX
SRT | D3-31700 | Ventricular tachycardia | MDC_ECG_RHY_V_TACHY
SRT | D3-31720 | Ventricular fibrillation | MDC_ECG_RHY_V_FIB
SRT | D3-33000 | Intraventricular conduction disturbance | MDC_ECG_BEAT_BLK_IVCD
SRT | D3-33120 | Left bundle branch block | MDC_ECG_BEAT_LBB_BLK_COMP
SRT | D3-33110 | Right bundle branch block | MDC_ECG_BEAT_RBB_BLK_COMP
SRT | D3-33122 | Incomplete Left bundle branch block | MDC_ECG_BEAT_LBB_BLK_INCOMP
SRT | D3-33112 | Incomplete Right bundle branch block | MDC_ECG_BEAT_RBB_BLK_INCOMP
SRT | D3-33200 | Bifascicular Block | MDC_ECG_BEAT_BLK_BIFASC
SRT | D3-33140 | Left anterior fascicular block | MDC_ECG_BEAT_BLK_ANT_L_HEMI
SRT | D3-33150 | Left posterior fascicular block | MDC_ECG_BEAT_BLK_POS_L_HEMI
SRT | D3-30001 | First degree Atrioventricular block | MDC_ECG_RHY_AV_HEART_BLK_DEG_1
SRT | R-F81AE | Second degree Atrioventricular block | MDC_ECG_RHY_AV_HEART_BLK_DEG_2
SRT | D3-32102 | Third degree Atrioventricular block | MDC_ECG_RHY_AV_HEART_BLK_DEG_3
SRT | D3-31351 | Ventricular pre-excitation | MDC_ECG_BEAT_PREX
SRT | F-38278 | ST depression |
SRT | F-38277 | ST elevation |
SRT | F-380B3 | Early repolarization |
SRT | F-38794 | Nonspecific ST-T abnormality |
SRT | F-38793 | Secondary ST-T abnormality |

## CID 3231 ST Segment Findings

Type:

  Extensible

Version:

  20080927

Table CID 3231. ST Segment Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-000C3 | ST Interval Normal
DCM | 122750 | Non-diagnostic - low heart rate
DCM | 122751 | Non-diagnostic - resting ST abnormalities
DCM | 122752 | Non-diagnostic - ventricular pacing or LBBB
SRT | G-A205 | Weakly positive
SRT | G-A200 | Positive
DCM | 122755 | Strongly positive
DCM | 122756 | Strongly positive - ST elevation

## CID 3232 ST Segment Location

Type:

  Extensible

Version:

  20091021

Table CID 3232. ST Segment Location

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-3260A | Left ventricle anterior segment
SRT | T-3260C | Left ventricle inferior segment
SRT | T-3260D | Left ventricle lateral segment
SRT | T-3260B | Left ventricle septal segment
SRT | T-32602 | Left ventricle apical segment

## CID 3233 ST Segment Morphology

Type:

  Extensible

Version:

  20080927

Table CID 3233. ST Segment Morphology

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122757 | ST Depression - Horizontal
DCM | 122758 | ST Depression - Upsloping
DCM | 122759 | ST Depression - Downsloping
SRT | F-38277 | ST Elevation
SRT | F-38278 | ST Depression

## CID 3234 Ectopic Beat Morphology

Type:

  Extensible

Version:

  20080927

Table CID 3234. Ectopic Beat Morphology

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D3-31700 | Ventricular tachycardia
SRT | F-33750 | Ventricular bigeminy
SRT | D3-31744 | Multifocal PVCs
SRT | D3-31742 | Unifocal PVCs
SRT | D3-31704 | Ventricular tachycardia, polymorphic

## CID 3235 Perfusion Comparison Findings

Type:

  Extensible

Version:

  20080927

Table CID 3235. Perfusion Comparison Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | R-4075C | No change | 260388006
SRT | R-215D9 | New ischemia | 428927006
SRT | R-215DE | Less ischemia | 429232006
SRT | R-215D5 | Resolution of ischemia | 428824000
SRT | R-215E1 | More ischemia | 429477006
SRT | R-215E0 | New infarction | 429391004

## CID 3236 Tolerance Comparison Findings

Type:

  Extensible

Version:

  20080827

Table CID 3236. Tolerance Comparison Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | R-4075C | No change | 260388006
SRT | F-00454 | Decreased tolerance | 102460003
SRT | F-00453 | Increased tolerance | 102459008

## CID 3237 Wall Motion Comparison Findings

Type:

  Extensible

Version:

  20080927

Table CID 3237. Wall Motion Comparison Findings

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | R-4075C | No change | 260388006
SRT | R-215DC | New wall motion abnormality | 429058004
SRT | R-215D6 | Improvement of wall motion | 428825004

## CID 3238 Stress Scoring Scales

Type:

  Extensible

Version:

  20080927

Table CID 3238. Stress Scoring Scales

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-E002 | Duke treadmill score
DCM | 122770 | Ratio of achieved to predicted maximal oxygen consumption
DCM | 122771 | Ratio of achieved to predicted functional capacity
DCM | 122772 | Aerobic index
DCM | 122773 | ST/HR Index

## CID 3239 Perceived Exertion Scales

Type:

  Extensible

Version:

  20080927

Table CID 3239. Perceived Exertion Scales

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122734 | Borg RPE Scale
DCM | 122735 | Borg CR10 Scale

## CID 3240 Electrophysiology Measurement Functions and Techniques

Type:

  Extensible

Version:

  20020904

Table CID 3240. Electrophysiology Measurement Functions and Techniques

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109006 | Differential signal
DCM | 109007 | His bundle electrogram
DCM | 109008 | Monopole signal
DCM | 109009 | Pacing (electrical) stimulus, voltage
DCM | 109010 | Radio frequency ablation, power
DCM | 109011 | Voltage measurement by basket catheter
DCM | 109012 | Voltage measurement by mapping catheter
DCM | 109013 | Voltage measurement, NOS

## CID 3241 Hemodynamic Measurement Techniques

Type:

  Extensible

Version:

  20020904

Table CID 3241. Hemodynamic Measurement Techniques

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | PA-50038 | Averaged hemodynamic measurement method
SRT | PA-50035 | Composite hemodynamic measurement method
SRT | PA-50034 | Computed hemodynamic measurement method
SRT | PA-5003B | Conductance catheter method
SRT | PA-5003C | Doppler catheter method
SRT | PA-50031 | Dual catheter method
SRT | PA-50039 | Fluid filled catheter method
SRT | PA-5003D | Fiberoptic catheter method
SRT | PA-5003E | Hall catheter method
SRT | PA-50033 | Pullback method
SRT | PA-50032 | Pulmonary capillary wedge method
SRT | PA-50036 | Static catheter method
SRT | PA-5003F | Thermistor catheter method
SRT | PA-5003A | Tip manometer method
SRT | PA-50037 | Wedge method

## CID 3250 Catheterization Procedure Phase

Type:

  Extensible

Version:

  20030327

Table CID 3250. Catheterization Procedure Phase

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-7299 | Cardiac catheterization bailout phase
SRT | G-7293 | Cardiac catheterization baseline phase
SRT | G-7294 | Cardiac catheterization image acquisition phase
SRT | G-7295 | Cardiac catheterization intervention phase
SRT | G-729B | Cardiac catheterization post contrast phase
SRT | G-7298 | Cardiac catheterization post-intervention phase
SRT | G-7296 | Cardiac catheterization pre-intervention phase
SRT | G-929D | Cardiac catheterization test/challenging phase
SRT | G-7297 | Cardiac catheterization therapy phase
SRT | P1-3160A | Catheterization of both left and right heart with graft
SRT | P1-3160B | Catheterization of both left and right heart without graft
SRT | P1-31604 | Catheterization of left heart
SRT | P1-31602 | Catheterization of right heart
SRT | P1-31612 | Transseptal catheterization
SRT | P2-71317 | Drug Infusion Challenge
SRT | P2-71310 | Exercise challenge
SRT | F-01604 | Resting State

## CID 3254 Electrophysiology Procedure Phase

Type:

  Extensible

Version:

  20020904

Table CID 3254. Electrophysiology Procedure Phase

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-729D | Atrial Effective Refractory Period, evaluation of
SRT | G-7304 | Carotid Sinus Massage procedure phase
SRT | G-7306 | Electrophysiology Mapping phase
SRT | G-729A | Electrophysiology procedure baseline phase
SRT | G-7408 | Post-ablation phase
SRT | G-7305 | Post-defibrillation procedure phase
SRT | G-729F | Radiofrequency Ablation procedure phase
SRT | G-729C | Sinus Node Recovery Time, evaluation of
SRT | G-729E | Ventricular Effective Refractory Period, evaluation of

## CID 3261 Stress Protocols

Type:

  Extensible

Version:

  20081027

Table CID 3261. Stress Protocols

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | P2-7131C | Balke protocol | 129097005
SRT | P2-7131A | Bruce protocol | 129095002
SRT | P2-7131D | Ellestad protocol | 129098000
SRT | P2-7131B | Modified Bruce protocol | 129096001
SRT | P2-713A1 | Modified Naughton protocol | 129102008
SRT | P2-713A0 | Naughton protocol | 129101001
SRT | P2-7131F | Pepper protocol | 129100000
SRT | P2-7131E | Ramp protocol | 129099008
SRT | P2-31010 | Exercise stress ECG test | 46136006
SRT | P2-31102 | Stress test using Bicycle Ergometer | 26046004
SRT | P2-31107 | Pharmacologic Stress protocol | 424064009
SRT | P2-3110A | Dipyridamole Stress protocol | 422685009
SRT | P2-31109 | Adenosine Stress protocol | 424444005
SRT | P2-31108 | Dobutamine Stress protocol | 424225000
SRT | P2-31011 | Pharmacologic and exercise stress test | 428813002
SRT | P2-3110B | Stress test using cardiac pacing | 428685003

## CID 3262 ECG Patient State Values

Type:

  Extensible

Version:

  20020904

Table CID 3262. ECG Patient State Values

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01602 | Baseline state
SRT | F-01606 | Exercise state
SRT | F-01608 | Post-exercise state
SRT | F-01604 | Resting state
SRT | F-10340 | Supine body position

## CID 3263 Electrode Placement Values

This Context Group comprises the ECG lead placement system identifiers of ISO/IEEE 11073-10102. The terms included in the table below may not constitute the complete list; see the ISO/IEEE Standard.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20080927

Table CID 3263. Electrode Placement Values

Coding Scheme | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 10:11264 | Unspecified 12-lead system | MDC_ECG_LDSYS_12LD_UNSPECIFIED
MDC | 10:11265 | Standard 12-lead positions, electrodes placed individually | MDC_ECG_LDSYS_12LD_STD
MDC | 10:11266 | Mason-Likar lead positions, electrodes placed individually | MDC_ECG_LDSYS_12LD_MASON_LIKAR
MDC | 10:11267 | Mason-Likar lead positions, V1-V6 in electrode pad | MDC_ECG_LDSYS_12LD_VPAD
MDC | 10:11268 | 12-lead electrode pad | MDC_ECG_LDSYS_12LD_PAD
MDC | 10:11269 | 12-lead derived from Frank XYZ leads | MDC_ECG_LDSYS_12LD_FROM_FRANK
MDC | 10:11270 | 12-lead derived from non-standard leads | MDC_ECG_LDSYS_12LD_NON_STANDARD
MDC | 10:11271 | 12-lead for bicycle exercise testing, limb leads on back of patient | MDC_ECG_LDSYS_12LD_BICYCLE
MDC | 10:11272 | Standard 12-lead positions one intercostal space higher | MDC_ECG_LDSYS_12LD_RAISED_INTERCOSTAL
MDC | 10:11273 | Unspecified XYZ lead system | MDC_ECG_LDSYS_XYZ_UNSPECIFIED
MDC | 10:11274 | Frank XYZ lead system | MDC_ECG_LDSYS_XYZ_FRANK
MDC | 10:11275 | McFee-Parungao XYZ lead system | MDC_ECG_LDSYS_XYZ_MCFEE_PARUNAGO
MDC | 10:11276 | Cube XYZ lead system | MDC_ECG_LDSYS_XYZ_CUBE
MDC | 10:11277 | Bipolar uncorrected XYZ lead system | MDC_ECG_LDSYS_XYZ_BIPOLAR
MDC | 10:11278 | Pseudo-orthogonal XYZ lead system | MDC_ECG_LDSYS_XYZ_PSEUDO_ORTH
MDC | 10:11279 | XYZ leads derived from standard 12-lead | MDC_ECG_LDSYS_XYZ_FROM_12LD
MDC | 10:11280 | NEHB lead system | MDC_ECG_LDSYS_3LD_NEHB
MDC | 10:11281 | 3-lead system, CC5-CM5-ML | MDC_ECG_LDSYS_3LD_CC5_CM5_ML
MDC | 10:11282 | 3-lead system, CC5-CM5-CH5 | MDC_ECG_LDSYS_3LD_CM5_CC5_CH5
MDC | 10:11283 | 12-lead from Frank leads XYZ leads by Dower transformation | MDC_ECG_LDSYS_12LD_FROM_DOWER
MDC | 10:11284 | 12-lead from EASI leads (ES, AS, AI) by Dower/EASI transformation | MDC_ECG_LDSYS_12LD_FROM_EASI
MDC | 10:11285 | 12-lead from Limb Leads (I, II) and one or more V leads | MDC_ECG_LDSYS_12LD_FROM_LIMB
MDC | 10:11286 | Standard 12-lead and XYZ | MDC_ECG_LDSYS_12LD_STD_AND_XYZ
MDC | 10:11287 | Standard 12-lead and NEHB | MDC_ECG_LDSYS_12LD_STD_AND_NEHB
MDC | 10:11288 | Standard 12-lead and CC5-CM5-ML | MDC_ECG_LDSYS_12LD_STD_AND_CC5_CM5_ML
MDC | 10:11289 | Standard 12-lead and CM5-CC5-CH5 | MDC_ECG_LDSYS_12LD_STD_AND_CM5_CC5_CH5
MDC | 10:11290 | Standard 12-lead with extra leads to the right and/or left sides | MDC_ECG_LDSYS_12LD_STD_EXTD
MDC | 10:11291 | Standard 12-lead extended to the right by V5R, V4R, V3R | MDC_ECG_LDSYS_12LD_STD_EXTD_RIGHT
MDC | 10:11292 | Standard 12-lead extended to the left by V7, V8, V9 | MDC_ECG_LDSYS_12LD_STD_EXTD_LEFT

### Note

A prior version of this context group used codes from the SCP-ECG vocabulary.

## CID 3264 XYZ Electrode Placement Values (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3271 Hemodynamic Physiological Challenges

Type:

  Extensible

Version:

  20100625

Table CID 3271. Hemodynamic Physiological Challenges

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P2-71317 | Drug infusion
SRT | P2-71310 | Exercise challenge
SRT | P2-71306 | Handgrip
SRT | P2-71302 | Head up
SRT | P2-71314 | Held inspiration
SRT | P2-71316 | Held ventilation
SRT | P2-71304 | Leg up
SRT | P2-71308 | Negative lower body pressure
SRT | P2-35000 | Pacing
SRT | P2-71318 | Post volume challenge
SRT | P2-71312 | Vagal stimulation
SRT | R-40928 | Valsalva maneuver

## CID 3335 ECG Annotations

This Context Group comprises the nomenclature of ISO/IEEE 11073-10102, limited to the hierarchies under Reference IDs MDC_ECG_WAVEC, MDC_ECG_WAVEP, MDC_ECG_BEAT, and MDC_ECG_NOISE.

The base terms from those hierarchies are included in the table below for reference. Note that these base terms are pre-coordinated with a variety of concept discriminators, and the code values for those pre-coordinated terms are arithmetically derived from the code values of the base terms. For the complete current list of terms and discriminator values, see the ISO/IEEE Standard. All pre-coordinated terms (annotation plus discriminators) within the identified hierarchies are part of this Context Group.

### Note

1. This Context Group is used in the Concept Name Code Sequence of the Waveform Annotation Sequence (0040,B020). See PS3.3.

2. A prior version of this context group used codes from the SCP-ECG coding system.

3. Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20130613

Table CID 3335. ECG Annotations

Coding Scheme | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 10:256 | P wave | MDC_ECG_WAVC_PWAVE
MDC | 10:320 | P' wave (second deflection in P wave) | MDC_ECG_WAVC_PPWAVE
MDC | 10:384 | P'' wave (third deflection in P wave) | MDC_ECG_WAVC_PPPWAVE
MDC | 10:448 | Q wave | MDC_ECG_WAVC_QWAVE
MDC | 10:512 | QS wave | MDC_ECG_WAVC_QSWAVE
MDC | 10:576 | R wave | MDC_ECG_WAVC_RWAVE
MDC | 10:640 | R' wave (second deflection in R Wave) | MDC_ECG_WAVC_RRWAVE
MDC | 10:704 | R'' wave (third deflection in R Wave) | MDC_ECG_WAVC_RRRWAVE
MDC | 10:768 | Notch | MDC_ECG_WAVC_NOTCH
MDC | 10:832 | S wave | MDC_ECG_WAVC_SWAVE
MDC | 10:896 | S' wave (second deflection in S Wave) | MDC_ECG_WAVC_SSWAVE
MDC | 10:960 | S'' wave (third deflection in S Wave) | MDC_ECG_WAVC_SSSWAVE
MDC | 10:1024 | T wave | MDC_ECG_WAVC_TWAVE
MDC | 10:1088 | T' wave (second deflection in T Wave) | MDC_ECG_WAVC_TTWAVE
MDC | 10:1152 | U wave | MDC_ECG_WAVC_UWAVE
MDC | 10:1216 | Delta wave | MDC_ECG_WAVC_DELTA
MDC | 10:1280 | Isoelectric region from global QRS onset to specific lead onset | MDC_ECG_WAVC_IWAVE
MDC | 10:1344 | Isoelectric region from specific lead QRS Offset to global offset | MDC_ECG_WAVC_KWAVE
MDC | 10:1408 | Osborne wave | MDC_ECG_WAVC_JWAVE
MDC | 10:1472 | Entire Beat (Pon to Toff, excluding U) | MDC_ECG_WAVC_PQRSTWAVE
MDC | 10:1536 | Entire Beat (Qon to Toff, excluding P and U) | MDC_ECG_WAVC_QRSTWAVE
MDC | 10:1600 | Entire QRS (excluding P, T and U) | MDC_ECG_WAVC_QRSWAVE
MDC | 10:1664 | TU fused wave | MDC_ECG_WAVC_TUWAVE
MDC | 10:1728 | Ventricular flutter wave | MDC_ECG_WAVC_VFLWAVE
MDC | 10:1792 | Atrial flutter wave | MDC_ECG_WAVC_AFLWAVE
MDC | 10:1856 | Isoelectric point or segment | MDC_ECG_WAVC_ISO
MDC | 10:1920 | PR Segment | MDC_ECG_WAVC_PRSEG
MDC | 10:1984 | ST Segment | MDC_ECG_WAVC_STSEG
MDC | 10:2048 | J-point | MDC_ECG_WAVC_STJ
MDC | 10:2112 | ST measurement point | MDC_ECG_WAVC_STM
MDC | 10:2176 | Isolated QRS-like artifact | MDC_ECG_WAVC_ARFCT
MDC | 10:2240 | Calibration pulse (individual pulse) | MDC_ECG_WAVC_CALP
MDC | 10:2304 | ST change | MDC_ECG_WAVC_STCH
MDC | 10:2368 | T-wave change | MDC_ECG_WAVC_TCH
MDC | 10:2432 | Ventricular Activation Time | MDC_ECG_WAVC_VAT
MDC | 10:4096 | Antibradycardia pace spike | MDC_ECG_WAVP_PACE
MDC | 10:4352 | atrium Antibradycardia pace spike | MDC_ECG_WAVP_PACE_ATR
MDC | 10:4608 | right atrium Antibradycardia pace spike | MDC_ECG_WAVP_PACE_ATR_R
MDC | 10:4864 | left atrium Antibradycardia pace spike | MDC_ECG_WAVP_PACE_ATR_L
MDC | 10:5120 | ventricular Antibradycardia pace spike | MDC_ECG_WAVP_PACE_V
MDC | 10:5376 | right ventricle Antibradycardia pace spike | MDC_ECG_WAVP_PACE_V_R
MDC | 10:5632 | left ventricle Antibradycardia pace spike | MDC_ECG_WAVP_PACE_V_L
MDC | 10:5888 | transthoracic Antibradycardia pace spike | MDC_ECG_WAVP_PACE_EXT
MDC | 10:6144 | Antitachycardia pace spike | MDC_ECG_WAVP_ATPACE
MDC | 10:6400 | atrium Antitachycardia pace spike | MDC_ECG_WAVP_ATPACE_ATR
MDC | 10:6656 | ventricle Antitachycardia pace spike | MDC_ECG_WAVP_ATPACE_V
MDC | 10:6912 | transthoracic Antitachycardia pace spike | MDC_ECG_WAVP_ATPACE_EXT
MDC | 10:7168 | Cardioversion spike | MDC_ECG_WAVP_CDVS
MDC | 10:7424 | atrium Cardioversion spike | MDC_ECG_WAVP_CDVS_ATR
MDC | 10:7680 | ventricle Cardioversion spike | MDC_ECG_WAVP_CDVS_V
MDC | 10:7936 | transthoracic Cardioversion spike | MDC_ECG_WAVP_CDVS_EXT
MDC | 10:8192 | Defibrillation spike | MDC_ECG_WAVP_DEFIB
MDC | 10:8448 | atrium Defibrillation spike | MDC_ECG_WAVP_DEFIB_ATR
MDC | 10:8704 | ventricle Defibrillation spike | MDC_ECG_WAVP_DEFIB_V
MDC | 10:8960 | transthoracic Defibrillation spike | MDC_ECG_WAVP_DEFIB_EXT
MDC | 10:8192 | Heart beat | MDC_ECG_BEAT
MDC | 10:8208 | Normal beat (sinus beat, normal conduction) | MDC_ECG_BEAT_NORMAL
MDC | 10:8224 | Abnormal beat | MDC_ECG_BEAT_ABNORMAL
MDC | 10:8240 | Dominant beat | MDC_ECG_BEAT_DOMINANT
MDC | 10:8256 | Supraventricular premature contraction | MDC_ECG_BEAT_SV_P_C
MDC | 10:8272 | Atrial premature contraction (beat) | MDC_ECG_BEAT_ATR_P_C
MDC | 10:8288 | Junctional (nodal) premature contraction | MDC_ECG_BEAT_JUNC_P_C
MDC | 10:8304 | Aberrated atrial premature beat (Ashman beat) | MDC_ECG_BEAT_ATR_P_C_ABERR
MDC | 10:8320 | Non-conducted p-wave (blocked) | MDC_ECG_BEAT_ATR_PWAVE_BLK
MDC | 10:8336 | Ventricular premature contraction beat | MDC_ECG_BEAT_V_P_C
MDC | 10:8352 | Fusion of ventricular and normal beat | MDC_ECG_BEAT_V_P_C_FUSION
MDC | 10:8368 | R-on-T premature ventricular beat | MDC_ECG_BEAT_V_P_C_RonT
MDC | 10:8384 | Supraventricular escape beat | MDC_ECG_BEAT_SV_ESC
MDC | 10:8400 | Atrial escape beat | MDC_ECG_BEAT_ATR_ESC
MDC | 10:8416 | Junctional (nodal) escape beat | MDC_ECG_BEAT_JUNC_ESC
MDC | 10:8432 | Ventricular escape beat | MDC_ECG_BEAT_V_ESC
MDC | 10:8448 | Bundle branch block beat | MDC_ECG_BEAT_BB_BLK
MDC | 10:8464 | Left bundle branch block beat | MDC_ECG_BEAT_LBB_BLK_COMP
MDC | 10:8480 | Incomplete left bundle branch block beat | MDC_ECG_BEAT_LBB_BLK_INCOMP
MDC | 10:8496 | Right bundle branch block beat | MDC_ECG_BEAT_RBB_BLK_COMP
MDC | 10:8512 | Incomplete right bundle branch block beat | MDC_ECG_BEAT_RBB_BLK_INCOMP
MDC | 10:8528 | Left anterior fascicular block beat | MDC_ECG_BEAT_BLK_ANT_L_HEMI
MDC | 10:8544 | Left posterior fascicular block beat | MDC_ECG_BEAT_BLK_POS_L_HEMI
MDC | 10:8560 | bifascicular block beat | MDC_ECG_BEAT_BLK_BIFASC
MDC | 10:8576 | trifascicular block beat | MDC_ECG_BEAT_BLK_TRIFASC
MDC | 10:8592 | bilateral bundle-branch block beat | MDC_ECG_BEAT_BLK_BILAT
MDC | 10:8608 | intraventricular conduction disturbance | MDC_ECG_BEAT_BLK_IVCD
MDC | 10:8624 | pre-excitation | MDC_ECG_BEAT_PREX
MDC | 10:8640 | Wolf-Parkinson-White syndrome | MDC_ECG_BEAT_WPW_UNK
MDC | 10:8656 | Wolf-Parkinson type A | MDC_ECG_BEAT_WPW_A
MDC | 10:8672 | Wolf-Parkinson type B | MDC_ECG_BEAT_WPW_B
MDC | 10:8688 | Lown-Ganong-Levine syndrome | MDC_ECG_BEAT_LGL
MDC | 10:8704 | Paced beat | MDC_ECG_BEAT_PACED
MDC | 10:8720 | Pacemaker Fusion beat | MDC_ECG_BEAT_PACED_FUS
MDC | 10:8736 | Unclassifiable beat | MDC_ECG_BEAT_UNKNOWN
MDC | 10:8752 | Pacemaker Learning beat | MDC_ECG_BEAT_LEARN
MDC | 10:11200 | No Noise | MDC_ECG_NOISE_CLEAN
MDC | 10:11216 | Moderate Noise, beats can be detected but cannot be classified | MDC_ECG_NOISE_MODERATE
MDC | 10:11232 | Severe Noise, beats cannot be detected or classified | MDC_ECG_NOISE_SEVERE
MDC | 10:11248 | No ECG signal is available | MDC_ECG_NOISE_NOSIGNAL

### Note

In a prior version of this table, the code 10:608 was specified for the concept R wave.

## CID 3337 Hemodynamic Annotations

Type:

  Extensible

Version:

  20050322

Table CID 3337. Hemodynamic Annotations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109014 | 35% of thermal/dye dilution CO
DCM | 109015 | 70% of thermal/dye dilution CO
DCM | 109016 | A wave peak pressure
DCM | 109017 | A wave pressure, average
DCM | 109018 | Beat detected (accepted)
DCM | 109019 | Beat detected (rejected)
SRT | F-00E22 | Average diastolic blood pressure
SRT | F-00E1F | Minimum diastolic blood pressure
SRT | F-32011 | End diastole
DCM | 109023 | End of expiration
DCM | 109024 | End of inspiration
DCM | 109070 | End of systole
DCM | 109071 | Indicator mean transit time
DCM | 109025 | Max dp/dt
DCM | 109026 | Max neg dp/dt
SRT | F-31150 | Mean blood pressure
DCM | 109028 | Peak of thermal cardiac output bolus
DCM | 109029 | Start of expiration
DCM | 109030 | Start of inspiration
DCM | 109031 | Start of thermal CO
SRT | F-00E14 | Average systolic blood pressure
SRT | F-00E11 | Maximum systolic blood pressure
DCM | 109072 | Tau
DCM | 109073 | V max myocardial
DCM | 109034 | V wave peak pressure
DCM | 109035 | V wave pressure, average
DCM | 109036 | Valve close
DCM | 109037 | Valve open

## CID 3339 Electrophysiology Annotations

Type:

  Extensible

Version:

  20020904

Table CID 3339. Electrophysiology Annotations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109038 | Ablation off
DCM | 109039 | Ablation on
DCM | 109040 | HIS bundle wave
DCM | 109041 | P wave
DCM | 109042 | Q wave
DCM | 109043 | R wave
DCM | 109044 | S wave
DCM | 109045 | Start of atrial contraction
DCM | 109046 | Start of atrial contraction (subsequent)
DCM | 109047 | Stimulation at rate 1 interval
DCM | 109048 | Stimulation at rate 2 interval
DCM | 109049 | Stimulation at rate 3 interval
DCM | 109050 | Stimulation at rate 4 interval
DCM | 109051 | T wave
DCM | 109052 | V wave
DCM | 109053 | V wave of next beat

## CID 3400 Procedure Log Titles

Type:

  Extensible

Version:

  20030327

Table CID 3400. Procedure Log Titles

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121120 | Cath Lab Procedure Log

## CID 3401 Types of Log Notes

Type:

  Extensible

Version:

  20030327

Table CID 3401. Types of Log Notes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121171 | Tech Note
DCM | 121172 | Nursing Note
DCM | 121173 | Physician Note
DCM | 121174 | Procedure Note
DCM | 121123 | Patient Status or Event

## CID 3402 Patient Status and Events

Type:

  Extensible

Version:

  20090615

Table CID 3402. Patient Status and Events

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122001 | Patient called to procedure room
DCM | 122002 | Patient admitted to procedure room
DCM | 122003 | Patient given pre-procedure instruction
DCM | 122004 | Patient informed consent given
DCM | 122005 | Patient advance directive given
DCM | 122006 | Nil Per Os (NPO) status confirmed
DCM | 122007 | Patient assisted to table
DCM | 122008 | Patient prepped and draped
DCM | 122009 | Patient connected to continuous monitoring
DCM | 122010 | Patient transferred to holding area
DCM | 122011 | Patient transferred to surgery
DCM | 122012 | Patient transferred to CCU
DCM | 122020 | Patient disoriented
DCM | 122021 | Patient reports nausea
DCM | 122022 | Patient reports discomfort
DCM | 122023 | Patient reports chest pain
DCM | 122024 | Patient reports no pain
DCM | 122025 | Patient alert
DCM | 122026 | Patient restless
DCM | 122027 | Patient sedated
DCM | 122028 | Patient asleep
DCM | 122029 | Patient unresponsive
DCM | 122030 | Patient has respiratory difficulty
DCM | 122031 | Patient coughed
DCM | 122032 | Patient disconnected from continuous monitoring
DCM | 122033 | Hemostasis achieved
DCM | 122034 | Hemostasis not achieved - oozing
DCM | 122035 | Hemostasis not achieved - actively bleeding
DCM | 122036 | Patient given post-procedure instruction
DCM | 122038 | Patient pronounced dead
DCM | 122039 | Patient transferred to morgue
DCM | 122037 | Patient discharged from department

## CID 3403 Percutaneous Entry

Type:

  Extensible

Version:

  20030327

Table CID 3403. Percutaneous Entry

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3746 “Percutaneous Entry Site”
Include CID 3747 “Percutaneous Closure”

## CID 3404 Staff Actions

Type:

  Extensible

Version:

  20030327

Table CID 3404. Staff Actions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122041 | Personnel Arrived
DCM | 122042 | Personnel Departed
DCM | 122043 | Page Sent To
DCM | 122044 | Consultation With
DCM | 122045 | Office called

## CID 3405 Procedure Action Values

Type:

  Extensible

Version:

  20030327

Table CID 3405. Procedure Action Values

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P1-30350 | Atherectomy
SRT | P1-30351 | Atherectomy by rotary cutter
SRT | P1-30352 | Atherectomy by laser
SRT | P1-30530 | Selective embolization of artery
SRT | P5-31500 | Percutaneous transluminal balloon angioplasty
SRT | P5-39010 | Transcatheter therapy for embolization
SRT | P5-39050 | Percutaneous retrieval of intravascular foreign body
SRT | P1-05550 | Stent placement
SRT | P5-39015 | Transcatheter deployment of detachable balloon
SRT | P5-39191 | Percutaneous insertion of intravascular filter
Include CID 3250 “Catheterization Procedure Phase”
Include CID 3406 “Non-coronary Transcatheter Interventions”
Include CID 3428 “Imaging Procedures”

## CID 3406 Non-coronary Transcatheter Interventions

Type:

  Extensible

Version:

  20030327

Table CID 3406. Non-Coronary Transcatheter Interventions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122053 | Valvular Intervention
DCM | 122054 | Aortic Intervention
DCM | 122055 | Septal Defect Intervention
DCM | 122056 | Vascular Intervention
DCM | 122057 | Myocardial biopsy

## CID 3407 Purpose of Reference to Object

Type:

  Extensible

Version:

  20030327

Table CID 3407. Purpose of Reference to Object

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122072 | Pre-procedure log
DCM | 122073 | Analysis or measurements for current procedure
DCM | 122075 | Prior report for current patient

## CID 3408 Actions With Consumables

Type:

  Extensible

Version:

  20030327

Table CID 3408. Actions With Consumables

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122076 | Consumable taken from inventory
DCM | 122077 | Consumable returned to inventory
DCM | 122078 | Remaining consumable disposed
DCM | 122079 | Consumable unusable

## CID 3409 Administration of Drugs/Contrast

Type:

  Extensible

Version:

  20030327

Table CID 3409. Administration of Drugs/Contrast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122081 | Drug start
DCM | 122082 | Drug end
DCM | 122083 | Drug administered
DCM | 122084 | Contrast start
DCM | 122085 | Contrast end
DCM | 122086 | Contrast administered
DCM | 122087 | Infusate start
DCM | 122088 | Infusate end

## CID 3410 Numeric Parameters of Drugs/Contrast

Type:

  Extensible

Version:

  20070124

Table CID 3410. Numeric Parameters of Drugs/Contrast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122091 | Volume administered
DCM | 122092 | Undiluted dose administered
DCM | 122093 | Concentration
DCM | 122094 | Rate of administration
DCM | 122095 | Duration of administration
DCM | 122096 | Volume unadministered or discarded
DCM | 121382 | Quantity administered
DCM | 121383 | Mass administered

## CID 3411 Intracoronary Devices

Type:

  Extensible

Version:

  20110609

Table CID 3411. Intracoronary Devices

Coding Scheme Designator | Code Value | Code Meaning | NCDR Equivalent | Trade Name (Informative)
--- | --- | --- | --- | ---
SRT | A-26912 | Percutaneous Transluminal Angioplasty Balloon | 113-1 |
SRT | R-002F0 | Cutting Balloon Angioplasty (CBA) Device | 113-2 |
SRT | A-25500 | Stent | 113-3 |
SRT | R-002FD | Directional Coronary Atherectomy (DCA) Device | 113-4 |
SRT | A-25610 | Rotational Atherectomy Device | 113-5 | Rotablator™
SRT | R-0036F | Saline Thrombectomy | 113-6 | AngioJet™
SRT | A-26920 | Transluminal Extraction Catheter (TEC) | 113-7 |
SRT | A-81080 | Laser | 113-8 |
SRT | R-00312 | Intravascular Ultrasound (IVUS) Device | 113-9 |
SRT | R-00310 | Intracoronary Doppler guide wire | 113-10 | Flowire™
SRT | R-00311 | Intracoronary pressure guide wire | 113-11 |
SRT | A-040ED | Brachytherapy Device |  |
SRT | R-00361 | Radiofrequency Ablation Device |  |
SRT | A-00D87 | Intravascular Optical Coherence Tomography Device |  |
SRT | A-00927 | Guide Wire |  |
SRT | A-26802 | Guiding Catheter |  |

## CID 3412 Intervention Actions and Status

Type:

  Extensible

Version:

  20030327

Table CID 3412. Intervention Actions and Status

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122301 | Guidewire crossing lesion unsuccessful
DCM | 122302 | Guidewire crossing lesion successful
DCM | 122303 | Angioplasty balloon inflated
DCM | 122304 | Angioplasty balloon deflated
DCM | 122305 | Device deployed
DCM | 122306 | Stent re-expanded
DCM | 122307 | Object removed
DCM | 122308 | Radiation applied
DCM | 122309 | Radiation removed
DCM | 122310 | Interventional device placement unsuccessful
DCM | 122311 | Interventional device placed
DCM | 122312 | Intervention performed
DCM | 122313 | Interventional device withdrawn

## CID 3413 Adverse Outcomes

Type:

  Extensible

Version:

  20030327

Table CID 3413. Adverse Outcomes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122167 | Death During Catheterization
Include CID 3754 “Vascular Complications”
Include CID 3755 “Cath Complications”

## CID 3414 Procedure Urgency

Type:

  Extensible

Version:

  20030327

Table CID 3414. Procedure Urgency

Coding Scheme Designator | Code Value | Code Meaning | NCDR Equivalents
--- | --- | --- | ---
SRT | G-D210 | Elective Procedure | 21-1, 78-1, 92-1
SRT | G-D216 | Urgent Procedure | 21-2, 78-2, 92-2
SRT | G-D209 | Emergent Procedure | 21-3, 78-3, 92-3
SRT | R-41C8D | Salvage Procedure | 21-4, 78-4, 92-4

## CID 3415 Cardiac Rhythms

This Context Group comprises the ECG rhythm annotations of ISO/IEEE 11073-10102. The terms included in the table below may not constitute the complete list; see the ISO/IEEE Standard.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20080927

Table CID 3415. Cardiac Rhythms

Coding Scheme | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 10:9216 | Sinus Rhythm | MDC_ECG_RHY_SINUS_RHY
MDC | 10:9232 | Normal Sinus Rhythm | MDC_ECG_RHY_SINUS_NORMAL_RHY
MDC | 10:9248 | Sinus Bradycardia | MDC_ECG_RHY_SINUS_BRADY
MDC | 10:9264 | Sinus Tachycardia | MDC_ECG_RHY_SINUS_TACHY
MDC | 10:9280 | Sinus Arrhythmia | MDC_ECG_RHY_SINUS_ARRHY
MDC | 10:9296 | Respiratory Sinus Arrhythmia | MDC_ECG_RHY_RESP_ARRHY
MDC | 10:9312 | Non-Respiratory Sinus Arrhythmia | MDC_ECG_RHY_NON_RESP_ARRHY
MDC | 10:9328 | Wandering Sinus Pacemaker within the sinus node | MDC_ECG_RHY_WANDP_ARRHY
MDC | 10:9344 | Wandering Pacemaker between the sinus node and the A-V node | MDC_ECG_RHY_WANDPAV_ARRHY
MDC | 10:9360 | Atrial Ectopic Rhythm | MDC_ECG_RHY_ATR_ECT_RHY
MDC | 10:9376 | Atrial Bigeminy | MDC_ECG_RHY_ATR_BIGEM
MDC | 10:9392 | Atrial Tachycardia | MDC_ECG_RHY_ATR_TACHY
MDC | 10:9408 | Paroxsysmal Atrial Tachycardia | MDC_ECG_RHY_ATR_TACHY_PAROX
MDC | 10:9424 | Multifocal Atrial Tachycardia | MDC_ECG_RHY_ATR_TACHY_MF
MDC | 10:9440 | Automatic Atrial Tachycardia | MDC_ECG_RHY_ATR_TACHY_AUTO
MDC | 10:9456 | Atrial flutter | MDC_ECG_RHY_ATR_FLUT
MDC | 10:9472 | Atrial fibrillation | MDC_ECG_RHY_ATR_FIB
MDC | 10:9488 | Supraventricular (atrial or junctional) Ectopic Rhythm | MDC_ECG_RHY_SV_ECT_RHY
MDC | 10:9504 | Supraventricular Tachycardia (atrial or junctional) | MDC_ECG_RHY_SV_TACHY
MDC | 10:9520 | Supraventricular Paroxsysmal Tachycardia | MDC_ECG_RHY_SV_TACHY_PAROX
MDC | 10:9536 | AV junctional (nodal) rhythm | MDC_ECG_RHY_JUNC_RHY
MDC | 10:9552 | AV junctional (nodal) escape rhythm | MDC_ECG_RHY_JUNC_ESC_BEATS
MDC | 10:9568 | Accelerated AV junctional (nodal) rhythm | MDC_ECG_RHY_JUNC_ACCEL
MDC | 10:9584 | Junctional Tachycardia | MDC_ECG_RHY_JUNC_TACHY
MDC | 10:9600 | AV reciprocating tachycardia | MDC_ECG_RHY_AV_TACHY_RECIP
MDC | 10:9616 | Reentrant AV nodal tachycardia | MDC_ECG_RHY_AV_TACHY_REENTRANT
MDC | 10:9632 | First Degree AV Block | MDC_ECG_RHY_AV_HEART_BLK_DEG_1
MDC | 10:9648 | Second Degree AV Block | MDC_ECG_RHY_AV_HEART_BLK_DEG_2
MDC | 10:9664 | Second Degree AV Block Type I (Wenckebach, or Mobitz Type I) | MDC_ECG_RHY_AV_HEART_BLK_DEG_2_I
MDC | 10:9680 | Second Degree AV Block Type II (Mobitz Type II) | MDC_ECG_RHY_AV_HEART_BLK_DEG_2_II
MDC | 10:9696 | Third Degree AV Block (complete AV block) | MDC_ECG_RHY_AV_HEART_BLK_DEG_3
MDC | 10:9712 | AV Dissociation | MDC_ECG_RHY_AV_DISSOC
MDC | 10:9728 | AV dissociation with interference | MDC_ECG_RHY_AV_DISSOC_INT
MDC | 10:9744 | Isorhythmic AV dissociation | MDC_ECG_RHY_AV_DISSOC_ISO
MDC | 10:9760 | Complete AV dissociation | MDC_ECG_RHY_AV_DISSOC_COMP
MDC | 10:9776 | First Degree SA Block | MDC_ECG_RHY_SA_HEART_BLK_DEG_1
MDC | 10:9792 | Second Degree SA Block Type I (Wenckebach) | MDC_ECG_RHY_SA_HEART_BLK_DEG_2_I
MDC | 10:9808 | Second Degree SA Block Type II | MDC_ECG_RHY_SA_HEART_BLK_DEG_2_II
MDC | 10:9824 | Third Degree SA Block (complete SA block) | MDC_ECG_RHY_SA_HEART_BLK_DEG_3
MDC | 10:9840 | Ventricular rhythm | MDC_ECG_RHY_V_RHY
MDC | 10:9856 | Idioventricular (ventricular escape) rhythm | MDC_ECG_RHY_V_IDIO_RHY
MDC | 10:9872 | Ventricular Parasystole | MDC_ECG_RHY_V_PARA
MDC | 10:9888 | Accelerated idioventricular rhythm | MDC_ECG_RHY_V_AIVR
MDC | 10:9904 | Slow Ventricular Tachycardia (Idioventricular Tachycardia) | MDC_ECG_RHY_V_IDIO_TACHY
MDC | 10:9920 | Ventricular Bigeminy | MDC_ECG_RHY_V_BIGEM
MDC | 10:9936 | Ventricular Trigeminy | MDC_ECG_RHY_V_TRIGEM
MDC | 10:9952 | Ventricular Couplet | MDC_ECG_RHY_V_P_C_CPLT
MDC | 10:9968 | Ventricular Run | MDC_ECG_RHY_V_P_C_RUN
MDC | 10:9984 | Ventricular Tachycardia (nonparoxysmal) | MDC_ECG_RHY_V_TACHY
MDC | 10:10000 | Ventricular Flutter | MDC_ECG_RHY_V_FLUT
MDC | 10:10016 | Ventricular Fibrillation | MDC_ECG_RHY_V_FIB
MDC | 10:10032 | Nonsustained Ventricular Tachycardia (paroxysmal) | MDC_ECG_RHY_V_TACHY_PAROX
MDC | 10:10048 | Sustained Monomorphic Ventricular Tachycardia | MDC_ECG_RHY_V_TACHY_MONO
MDC | 10:10064 | Polymorphic Ventricular Tachycardia | MDC_ECG_RHY_V_TACHY_POLY
MDC | 10:10080 | Torsades de Pointes Ventricular Tachycardia | MDC_ECG_RHY_V_TACHY_TDP
MDC | 10:10096 | pre-excitation | MDC_ECG_RHY_PREX
MDC | 10:10112 | Wolf-Parkinson-White syndrome | MDC_ECG_RHY_WPW_UNK
MDC | 10:10128 | Wolf-Parkinson type A | MDC_ECG_RHY_WPW_A
MDC | 10:10144 | Wolf-Parkinson type B | MDC_ECG_RHY_WPW_B
MDC | 10:10160 | Lown-Ganong-Levine syndrome | MDC_ECG_RHY_LGL
MDC | 10:10336 | Asystole | MDC_ECG_RHY_ASYSTOLE
MDC | 10:10352 | Irregular rhythm | MDC_ECG_RHY_IRREG
MDC | 10:10368 | Low Heart Rate Variability | MDC_ECG_RHY_LHRV
MDC | 10:10416 | T-wave alternans | MDC_ECG_RHY_TALT
MDC | 10:10432 | Bradycardia | MDC_ECG_RHY_BRADY
MDC | 10:10448 | Calibration signal (sustained) | MDC_ECG_RHY_CALS
MDC | 10:10176 | Atrial Demand Mode Pacing | MDC_ECG_RHY_EPADM
MDC | 10:10240 | Ventricular Demand Mode Pacing | MDC_ECG_RHY_EPVDM
MDC | 10:10304 | Anti-Tachycardia Pacing | MDC_ECG_RHY_EPAVT

### Note

A prior version of this context group used codes from the SCP-ECG vocabulary.

## CID 3416 Respiration Rhythms

Type:

  Extensible

Version:

  20030327

Table CID 3416. Respiration Rhythms

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-21301 | normal respiratory rhythm
SRT | F-21303 | irregular breathing
SRT | F-20130 | gasping respiration
SRT | F-21334 | abnormal respiratory rhythm
SRT | F-21331 | respiration intermittent

## CID 3418 Lesion Risk

Type:

  Extensible

Version:

  20030327

Table CID 3418. Lesion Risk

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 111-1 | Low Risk Lesion
NCDR | 2.0b | 111-2 | Moderate Risk Lesion
NCDR | 2.0b | 111-3 | High Risk Lesion

## CID 3419 Findings Titles

Type:

  Extensible

Version:

  20030327

Table CID 3419. Findings Titles

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121071 | Finding
DCM | 121073 | Impression
DCM | 121075 | Recommendation

## CID 3421 Procedure Action

Type:

  Extensible

Version:

  20030327

Table CID 3421. Procedure Action

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121130 | Start Procedure Action
DCM | 121131 | End Procedure Action
DCM | 121132 | Suspend Procedure Action
DCM | 121133 | Resume Procedure Action

## CID 3422 Device Use Actions

Type:

  Extensible

Version:

  20030327

Table CID 3422. Device Use Actions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-002F8 | Device inserted into sheath
SRT | R-002F7 | Device at site of interest
SRT | R-002FB | Device withdrawn / removed
SRT | R-002F6 | Device applied to patient
SRT | R-002FA | Device used
SRT | R-10042 | Device crossed septum
DCM | 122089 | Device crossed lesion

## CID 3423 Numeric Device Characteristics

Type:

  Extensible

Version:

  20070124

Table CID 3423. Numeric Device Characteristics

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A22A | Length
SRT | M-02550 | Diameter
DCM | 122097 | Catheter Curve
DCM | 122098 | Transmit Frequency
SRT | G-D705 | Volume
DCM | 121208 | Inter-Marker Distance

## CID 3425 Intervention Parameters

Type:

  Extensible

Version:

  20030327

Table CID 3425. Intervention Parameters

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-002D0 | Angioplasty Inflation pressure
SRT | R-002CF | Angioplasty Inflation duration
SRT | R-0036C | Rotational Atherectomy Speed
SRT | R-002F2 | Delivered Radiation Dose
SRT | R-10043 | Ablation power
SRT | R-10044 | Ablation frequency

## CID 3426 Consumables Parameters

Type:

  Extensible

Version:

  20030327

Table CID 3426. Consumables Parameters

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121145 | Description of Material
DCM | 121148 | Unit Serial Identifier
DCM | 121149 | Lot Identifier

## CID 3427 Equipment Events

Type:

  Extensible

Version:

  20100608

Table CID 3427. Equipment Events

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110501 | Equipment failure
DCM | 122047 | Equipment brought to procedure room
DCM | 122048 | Equipment ready
DCM | 122049 | Equipment removed

## CID 3428 Imaging Procedures

Type:

  Extensible

Version:

  20030327

Table CID 3428. Imaging Procedures

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P5-009A0 | Angiography
SRT | P5-32130 | Aortography
SRT | P5-30100 | Coronary Arteriography
SRT | P5-3003A | Cardiac ventriculography
SRT | P5-30041 | Left Ventriculography
SRT | P5-3003F | Right Ventriculography
SRT | P5-30107 | Bypass graft angiography
DCM | 122058 | Arterial conduit angiography
SRT | P5-B3002 | Transesophageal echocardiography
SRT | P5-B3003 | Transthoracic echocardiography
SRT | P5-B3004 | Epicardial echocardiography
SRT | P5-B001D | Intravascular ultrasound
SRT | P5-B3006 | Intracardiac echocardiography

## CID 3429 Catheterization Devices

Type:

  Extensible

Version:

  20030327

Table CID 3429. Catheterization Devices

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-28051 | Intra-Aortic Balloon Pump (IABP)
SRT | R-00306 | Fluid filled catheter
SRT | R-00304 | Fiberoptic catheter
SRT | R-0030A | Hall catheter
SRT | R-00379 | Thermistor catheter
SRT | R-00383 | Tip manometer
SRT | A-26860 | Swann-Ganz catheter
SRT | F-9B75C | Sheath
SRT | R-10041 | Transseptal catheter
DCM | 122052 | Bioptome
Include CID 3411 “Intracoronary Devices”

## CID 3430 DateTime Qualifiers

Type:

  Extensible

Version:

  20030327

Table CID 3430. DateTime Qualifiers

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121136 | DateTime Unsynchronized
DCM | 121137 | DateTime Estimated

## CID 3440 Peripheral Pulse Locations

Type:

  Extensible

Version:

  20030327

Table CID 3440. Peripheral Pulse Locations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-47160 | Brachial Artery
SRT | T-45100 | Carotid Artery
SRT | T-47741 | Dorsalis Pedis Artery
SRT | T-47400 | Femoral Artery
SRT | T-47500 | Popliteal Artery
SRT | T-47600 | Posterior Tibial Artery
SRT | T-47300 | Radial Artery
SRT | T-47200 | Ulnar Artery

## CID 3441 Patient Assessments

Type:

  Extensible

Version:

  20030327

Table CID 3441. Patient Assessments

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 8884-9 | Cardiac Rhythm
LN | 9304-7 | Respiration Rhythm
SRT | F-046D8 | Skin condition assessment
SRT | F-043E6 | Respiration assessment
SRT | F-04317 | Patient mental state assessment

## CID 3442 Peripheral Pulse Methods

Type:

  Extensible

Version:

  20030327

Table CID 3442. Peripheral Pulse Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P2-01510 | Palpation
SRT | P1-30022 | Doppler

## CID 3446 Skin Condition

Type:

  Extensible

Version:

  20030327

Table CID 3446. Skin Condition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122271 | skin condition Warm
DCM | 122272 | skin condition Cool
DCM | 122273 | skin condition Cold
DCM | 122274 | skin condition Dry
DCM | 122275 | skin condition Clammy
DCM | 122276 | skin condition Diaphoretic
DCM | 122277 | skin condition Flush
DCM | 122278 | skin condition Mottled
DCM | 122279 | skin condition Pale

## CID 3448 Airway Assessment

Type:

  Extensible

Version:

  20030327

Table CID 3448. Airway Assessment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122281 | airway unobstructed
DCM | 122282 | airway partially obstructed
DCM | 122283 | airway severely obstructed

## CID 3451 Calibration Objects

Type:

  Extensible

Version:

  20040614

Table CID 3451. Calibration Objects

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-26800 | Catheter
SRT | A-10141 | Measuring Ruler
DCM | 122485 | Sphere

## CID 3452 Calibration Methods

Type:

  Extensible

Version:

  20040614

Table CID 3452. Calibration Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122486 | Geometric Isocenter
DCM | 122487 | Geometric Non-Isocenter
DCM | 122488 | Calibration Object Used

## CID 3453 Cardiac Volume Methods

Type:

  Extensible

Version:

  20040614

Table CID 3453. Cardiac Volume Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122558 | Area Length Kennedy
DCM | 122559 | Area Length Dodge
DCM | 122560 | Area Length Wynne
DCM | 122562 | Multiple Slices
DCM | 122563 | Boak
DCM | 122564 | TS Pyramid
DCM | 122565 | Two Chamber
DCM | 122566 | Parallelepiped

## CID 3455 Index Methods

Type:

  Extensible

Version:

  20040614

Table CID 3455. Index Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 8277-6 | BSA
DCM | 122572 | BSA^1.219

## CID 3456 Sub-segment Methods

Type:

  Extensible

Version:

  20040614

Table CID 3456. Sub-Segment Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122574 | Equidistant method
DCM | 122575 | User selected method

## CID 3458 Contour Realignment

Type:

  Extensible

Version:

  20040614

Table CID 3458. Contour Realignment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122475 | Center of Gravity
DCM | 122476 | Long Axis Based
DCM | 122477 | No Realignment

## CID 3460 Circumferential Extent

Type:

  Extensible

Version:

  20040614

Table CID 3460. Circumferential Extent

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122464 | LAD Region in RAO Projection
DCM | 122465 | RCA Region in RAO Projection

## CID 3461 Regional Extent

Type:

  Extensible

Version:

  20040614

Table CID 3461. Regional Extent

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122466 | Single LAD Region in RAO Projection
DCM | 122467 | Single RCA Region in RAO Projection
DCM | 122468 | Multiple LAD Region in RAO Projection
DCM | 122469 | Multiple RCA Region in RAO Projection
DCM | 122470 | LAD Region in LAO Projection
DCM | 122471 | RCA Region in LAO Projection
DCM | 122472 | CFX Region in LAO Projection

## CID 3462 Chamber Identification

Type:

  Extensible

Version:

  20040614

Table CID 3462. Chamber Identification

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-32600 | Left Ventricle
SRT | T-32500 | Right Ventricle
SRT | T-32300 | Left Atrium
SRT | T-32200 | Right Atrium

## CID 3463 Ventricle Identification

Type:

  Non-Extensible

Version:

  20080927

Table CID 3463. Ventricle Identification

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-32600 | Left Ventricle
SRT | T-32500 | Right Ventricle

## CID 3465 QA Reference Methods

Type:

  Extensible

Version:

  20040614

Table CID 3465. QA Reference Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122489 | Curve Fitted Reference
DCM | 122490 | Interpolated Local Reference
DCM | 122491 | Mean Local Reference

## CID 3466 Plane Identification

Type:

  Extensible

Version:

  20130806

Table CID 3466. Plane Identification

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-40985 | Right Anterior Oblique
SRT | R-10220 | Left Anterior Oblique
SRT | R-10206 | Antero-posterior
SRT | R-10236 | Left Lateral
SRT | R-101C3 | Cranial LAO
SRT | R-101C5 | Cranial RAO
SRT | R-101C4 | Caudal LAO
SRT | R-101C6 | Caudal RAO

### Note

In a prior version of this Context Group, "right anterior oblique" was assigned the code R-10218, which in SNOMED is actually "Indirect iris transillumination"; this code has been replaced with the correct code R-40985.

## CID 3467 Ejection Fraction

Type:

  Extensible

Version:

  20040614

Table CID 3467. Ejection Fraction

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 8808-8 | Left Ventricular Ejection Fraction by Angiography
LN | 8815-3 | Right Ventricular Ejection Fraction by Angiography
DCM | 122406 | Left Atrial Ejection Fraction by Angiography

## CID 3468 ED Volume

Type:

  Extensible

Version:

  20110124

Table CID 3468. ED Volume

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 8821-1 | Left Ventricular ED Volume
LN | 8822-9 | Right Ventricular ED Volume
DCM | 122407 | Left Atrial ED Volume

## CID 3469 ES Volume

Type:

  Extensible

Version:

  20040614

Table CID 3469. ES Volume

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 8823-7 | Left Ventricular ES Volume
LN | 8824-5 | Right Ventricular ES Volume
DCM | 122408 | Left Atrial ES Volume

## CID 3470 Vessel Lumen Cross-sectional Area Calculation Methods

Type:

  Extensible

Version:

  20040614

Table CID 3470. Vessel Lumen Cross-Sectional Area Calculation Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122473 | Circular method
DCM | 122474 | Densitometric method

## CID 3471 Estimated Volumes

Type:

  Extensible

Version:

  20040614

Table CID 3471. Estimated Volumes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121216 | Volume estimated from single 2D region
DCM | 121218 | Volume estimated from two non-coplanar 2D regions

## CID 3472 Cardiac Contraction Phase

Type:

  Extensible

Version:

  20071031

Table CID 3472. Cardiac Contraction Phase

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-32020 | Systolic
SRT | F-32010 | Diastolic

## CID 3480 IVUS Procedure Phases

This context group outlines the phases of a catheterization procedure in which measurements are performed.

Type:

  Extensible

Version:

  20040614

Table CID 3480. IVUS Procedure Phases

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-7298 | Cardiac catheterization post-intervention phase
SRT | G-7296 | Cardiac catheterization pre-intervention phase

## CID 3481 IVUS Distance Measurements

This context group is the set of distance measurements made in an IVUS procedure.

Type:

  Extensible

Version:

  20050110

Table CID 3481. IVUS Distance Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122330 | EEM Diameter
SRT | G-0364 | Vessel lumen diameter
SRT | R-101AD | Stent Diameter
DCM | 122331 | Plaque Plus Media Thickness
DCM | 122332 | Lumen Perimeter

## CID 3482 IVUS Area Measurements

This context group is the set of cross-sectional area measurements made in an IVUS procedure.

Type:

  Extensible

Version:

  20040614

Table CID 3482. IVUS Area Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122333 | EEM Cross-Sectional Area
SRT | G-0366 | Vessel lumen cross-sectional area
SRT | R-101AF | Stent Cross-Sectional Area
DCM | 122334 | Plaque plus Media Cross-Sectional Area
DCM | 122335 | In-Stent Neointimal Cross-Sectional Area

## CID 3483 IVUS Longitudinal Measurements

This context group is a set of measurements that are made on a longitudinal image. A longitudinal image is a perpendicular cut plane reconstructed from an IVUS pullback multi-frame image.

Type:

  Extensible

Version:

  20050110

Table CID 3483. IVUS Longitudinal Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-101B0 | Stent Length
SRT | R-101BC | Stenotic Lesion Length
DCM | 122341 | Calcium Length
DCM | 122364 | Stent Gap

## CID 3484 IVUS Indices and Ratios

This context group is the set of index and ratio calculations made in an IVUS procedure.

Type:

  Extensible

Version:

  20040614

Table CID 3484. IVUS Indices and Ratios

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122343 | Lumen Eccentricity Index
DCM | 122344 | Plaque plus Media Eccentricity Index
DCM | 122345 | Remodeling Index
DCM | 122346 | Stent Symmetry Index
DCM | 122347 | Stent Expansion Index
DCM | 122348 | Lumen Shape Index
DCM | 122350 | Lumen Diameter Ratio
DCM | 122351 | Stent Diameter Ratio
DCM | 122352 | EEM Diameter Ratio

## CID 3485 IVUS Volume Measurements

This context group is the set of volume measurements made from an IVUS procedure.

Type:

  Extensible

Version:

  20040614

Table CID 3485. IVUS Volume Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122371 | EEM Volume
DCM | 122372 | Lumen Volume
SRT | R-101B2 | Stent Volume
DCM | 122374 | In-Stent Neointimal Volume
DCM | 122375 | Native Plaque Volume
DCM | 122376 | Total Plaque Volume

## CID 3486 Vascular Measurement Sites

This context group is the set of sites where vascular measurements can be made.

Type:

  Extensible

Version:

  20051103

Table CID 3486. Vascular Measurement Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122380 | Proximal Reference
DCM | 122381 | Distal Reference
DCM | 122382 | Site of Lumen Minimum
DCM | 122687 | Site of Lumen Maximum

## CID 3487 Intravascular Volumetric Regions

This context group is the set of regions where intravascular volumetric measurements can be made.

Type:

  Extensible

Version:

  20110609

Table CID 3487. Intravascular Volumetric Regions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122383 | Stented Region
DCM | 122384 | Entire Pullback
DCM | 122385 | Proximal Stent Margin
DCM | 122386 | Distal Stent Margin
SRT | M-01000 | Lesion
SRT | R-002EF | Culprit Lesion

## CID 3488 Min/Max/Mean

This context group contains modifiers that indicate whether the measurement is a minimum, maximum or averaged value.

Type:

  Extensible

Version:

  20040614

Table CID 3488. Min/Max/Mean

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A437 | Maximum
SRT | R-404FB | Minimum
SRT | R-00317 | Mean

## CID 3489 Calcium Distribution

This context group is a set of modifiers specifying the distribution of a calcium deposit in an arc of calcium measurement.

Type:

  Extensible

Version:

  20040614

Table CID 3489. Calcium Distribution

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A139 | Superficial
SRT | G-A140 | Deep

## CID 3491 IVUS Lesion Morphologies

This context group is a set of qualitative assessments for lesion morphology.

Type:

  Extensible

Version:

  20040614

Table CID 3491. IVUS Lesion Morphologies

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3495 “IVUS Plaque Composition”
DCM | 122356 | Soft plaque
DCM | 122357 | In-Stent Neointima
SRT | D3-80027 | Arterial (True) Aneurysm
SRT | M-32390 | Pseudo Aneurysm
DCM | 122361 | False Lumen
SRT | R-4047B | Concentric
SRT | R-40416 | Eccentric
SRT | M-52103 | Plaque Ulceration
DCM | 122363 | Plaque Rupture
DCM | 122389 | Vulnerable Plaque
DCM | 122390 | Eroded Plaque

## CID 3492 Vascular Dissection Classifications

This context group is a set of dissection classifications commonly detected with IVUS or CT/MR angiography.

Type:

  Extensible

Version:

  20040614

Table CID 3492. Vascular Dissection Classifications

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-101B7 | Medial Dissection
SRT | R-101B8 | Intimal Dissection
SRT | R-101B9 | Adventitial Dissection
SRT | M-35063 | Intramural hematoma
DCM | 122388 | Intra-stent Dissection

## CID 3493 IVUS Relative Stenosis Severities

This context group is a set of stenosis severity classifications for multiple lesions within a segment. There will always be a worst stenosis (T-1), the stenosis with the smallest lumen size.

There can be multiple secondary stenoses (T-2, T-3, etc.), which are lesions meeting the definition of a stenosis, but with lumen sizes larger than the worst stenosis. Reference "American College of Cardiology Clinical Expert Consensus Document on Standards for Acquisition, Measurement and Reporting of Intravascular Ultrasound Studies (IVUS) ".

Type:

  Extensible

Version:

  20040614

Table CID 3493. IVUS Relative Stenosis Severities

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122367 | T-1 Worst
DCM | 122368 | T-2 Secondary
DCM | 122369 | T-3 Secondary
DCM | 122370 | T-4 Secondary

## CID 3494 IVUS Non Morphological Findings

Type:

  Extensible

Version:

  20040614

Table CID 3494. IVUS Non Morphological Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122360 | True Lumen
SRT | R-101B3 | Arterial Blood Stasis
SRT | R-101B5 | Incomplete Stent apposition
SRT | R-101B6 | Acquired Incomplete stent apposition

## CID 3495 IVUS Plaque Composition

This context group is a set of qualitative assessments defining the composition of plaque.

Type:

  Extensible

Version:

  20040614

Table CID 3495. IVUS Plaque Composition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | M-78260 | Fibrous Plaque
SRT | D6-34737 | Vascular Calcification
SRT | M-35001 | Thrombus
DCM | 122394 | Fibro-Lipidic Plaque
DCM | 122395 | Necrotic-Lipidic Plaque

## CID 3496 IVUS Fiducial Points

This context group is a set of fiducial points (anatomical markers). Fiducial points are used as identifiable axial landmarks in determining the location of a measurement in a vessel.

Type:

  Extensible

Version:

  20040614

Table CID 3496. IVUS Fiducial Points

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-035D | Collateral Branch of vessel
SRT | A-25500 | Stent
SRT | D6-34737 | Vascular Calcification
SRT | M-78260 | Fibrous Plaque
SRT | T-48000 | Vein
SRT | G-036A | Vessel Origin

## CID 3497 IVUS Arterial Morphology

Type:

  Extensible

Version:

  20050110

Table CID 3497. IVUS Arterial Morphology

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
SRT | T-41100 | Lumen of artery
SRT | R-102AE | External Elastic Membrane
Include CID 3495 “IVUS Plaque Composition”

## CID 3500 Pressure Units

Type:

  Non-Extensible

Version:

  20030327

Table CID 3500. Pressure Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | mm[Hg] | mmHg
UCUM | kPa | kPa

## CID 3502 Hemodynamic Resistance Units

Type:

  Non-Extensible

Version:

  20120327

Table CID 3502. Hemodynamic Resistance Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | [PRU] | P.R.U.
UCUM | [wood'U] | Wood U
UCUM | dyn.s.cm-5 | dyn.s.cm-5

### Note

P.R.U. is in units of mm[Hg].s/ml; Wood Units is in mm[Hg].min/l

## CID 3503 Indexed Hemodynamic Resistance Units

Type:

  Non-Extensible

Version:

  20120327

Table CID 3503. Indexed Hemodynamic Resistance Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | [PRU]/m2 | P.R.U./m2
UCUM | [wood'U]/m2 | Wood U/m2
UCUM | dyn.s.cm-5/m2 | dyn.s.cm-5/m2

## CID 3510 Catheter Size Units

Type:

  Extensible

Version:

  20030327

Table CID 3510. Catheter Size Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | [Ch] | french
UCUM | mm | mm

## CID 3515 Specimen Collection

Type:

  Extensible

Version:

  20030327

Table CID 3515. Specimen Collection

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P3-02000 | specimen collection
SRT | PA-20110 | collection of blood specimen for laboratory
SRT | PA-2011E | blood sampling from extracorporeal blood circuit

## CID 3520 Blood Source Type

Type:

  Extensible

Version:

  20030327

Table CID 3520. Blood Source Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-00376 | Systemic Artery Blood
SRT | T-C2007 | Mixed Venous Blood
SRT | R-0035B | Pulmonary Artery Blood
SRT | R-0035E | Pulmonary Vein Blood

## CID 3524 Blood Gas Pressures

Type:

  Extensible

Version:

  20030327

Table CID 3524. Blood Gas Pressures

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11557-6 | Blood Carbon dioxide partial pressure
LN | 2019-8 | Arterial Blood Carbon dioxide partial pressure
LN | 2021-4 | Venous Blood Carbon dioxide partial pressure
LN | 11556-8 | Blood Oxygen partial pressure
LN | 2703-7 | Arterial Oxygen partial pressure
LN | 2705-2 | Venous Oxygen partial pressure
LN | 19217-9 | Oxygen partial pressure at 50% saturation (P50)
LN | 19214-6 | Arterial Oxygen partial pressure at 50% saturation
LN | 19216-1 | Venous Oxygen partial pressure at 50% saturation

## CID 3525 Blood Gas Content

Type:

  Extensible

Version:

  20030327

Table CID 3525. Blood Gas Content

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 20565-8 | Blood Carbon dioxide content
LN | 2026-3 | Arterial Blood Carbon dioxide content
LN | 2027-1 | Venous Blood Carbon dioxide content
DCM | 122185 | Blood Oxygen content
LN | 19218-7 | Arterial Oxygen content
LN | 19220-3 | Venous Oxygen content
LN | 10232-7 | Aortic Root Oxygen content
LN | 10245-9 | Pulmonary Artery Main Oxygen content
LN | 10247-5 | Pulmonary Wedge Oxygen content

## CID 3526 Blood Gas Saturation

Type:

  Extensible

Version:

  20030327

Table CID 3526. Blood Gas Saturation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122187 | Blood Carbon dioxide saturation
LN | 20564-1 | Blood Oxygen saturation
LN | 2708-6 | Arterial Oxygen saturation
LN | 2711-0 | Venous Oxygen saturation
LN | 2709-4 | Capillary Blood Oxygen Saturation
LN | 2710-2 | Capillary Blood Oxygen Saturation, by Oximetry

## CID 3527 Blood Base Excess

Type:

  Extensible

Version:

  20030327

Table CID 3527. Blood Base Excess

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11555-0 | Blood Base Excess
LN | 1925-7 | Arterial Blood Base Excess
LN | 1927-3 | Venous Blood Base Excess

## CID 3528 Blood pH

Type:

  Extensible

Version:

  20030327

Table CID 3528. Blood pH

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11558-4 | Blood pH
LN | 2744-1 | Arterial Blood pH
LN | 2746-6 | Venous Blood pH

## CID 3529 Arterial / Venous Content

Type:

  Extensible

Version:

  20030327

Table CID 3529. Arterial / Venous Content

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 19218-7 | Arterial Content (FCa)
LN | 19220-3 | Venous Content (FCv)
DCM | 122188 | Pulmonary Arterial Content (FCpa)
DCM | 122189 | Pulmonary Venous Content (FCpv)

## CID 3530 Oxygen Administration Actions

Type:

  Extensible

Version:

  20030327

Table CID 3530. Oxygen Administration Actions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121161 | Begin oxygen administration
DCM | 121162 | End oxygen administration

## CID 3531 Oxygen Administration

Type:

  Extensible

Version:

  20030327

Table CID 3531. Oxygen Administration

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-0034A | Oxygen Administration by nasal cannula
SRT | R-00349 | Oxygen Administration by mask
DCM | 121163 | Oxygen Administration by ventilator

## CID 3550 Circulatory Support Actions

Type:

  Extensible

Version:

  20030327

Table CID 3550. Circulatory Support Actions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121157 | Begin Circulatory Support
DCM | 121158 | End Circulatory Support

## CID 3551 Ventilation Actions

Type:

  Extensible

Version:

  20030327

Table CID 3551. Ventilation Actions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121168 | Begin Ventilation
DCM | 121169 | End Ventilation

## CID 3552 Pacing Actions

Type:

  Extensible

Version:

  20030327

Table CID 3552. Pacing Actions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121166 | Begin Pacing
DCM | 121167 | End Pacing

## CID 3553 Circulatory Support

Type:

  Extensible

Version:

  20030327

Table CID 3553. Circulatory Support

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-28051 | Intra-Aortic Balloon Pump
SRT | R-00303 | External Counter-Pulsation
SRT | A-11FCD | Left Ventricular Assist Device
SRT | P2-77110 | Extra-corporeal circulation
SRT | P1-36858 | Cardiopulmonary bypass

## CID 3554 Ventilation

Type:

  Extensible

Version:

  20030327

Table CID 3554. Ventilation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-002CC | Ambu Bag
SRT | R-00359 | Pressure Support Ventilator
SRT | R-0038C | Volume Support Ventilator

## CID 3555 Pacing

Type:

  Extensible

Version:

  20030327

Table CID 3555. Pacing

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P2-35000 | Pacing
SRT | R-00315 | pacing with magnet
SRT | P2-35200 | atrial pacing
SRT | P2-35002 | ventricular pacing
SRT | R-002D9 | A-V sequential pacing
SRT | P2-35440 | temporary transcutaneous pacing

## CID 3560 Blood Pressure Methods

Type:

  Extensible

Version:

  20030327

Table CID 3560. Blood Pressure Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-00318 | Blood pressure cuff method

## CID 3600 Relative Times

Type:

  Non-Extensible

Version:

  20030327

Table CID 3600. Relative Times

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-407E0 | Before
SRT | R-407E1 | During
SRT | R-42517 | After

## CID 3602 Hemodynamic Patient State

Type:

  Extensible

Version:

  20030327

Table CID 3602. Hemodynamic Patient State

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01602 | Baseline state
SRT | F-10340 | Supine body position
SRT | F-01604 | Resting state
SRT | F-01606 | Exercise state
SRT | F-01608 | Post-exercise state

## CID 3604 Arterial Lesion Locations

Type:

  Extensible

Version:

  20030327

Table CID 3604. Arterial Lesion Locations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3015 “Coronary Arteries”
Include CID 3606 “Arterial Source Locations”

## CID 3606 Arterial Source Locations

Type:

  Extensible

Version:

  20061024

Table CID 3606. Arterial Source Locations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-42500 | Abdominal aorta
SRT | T-45530 | anterior communicating artery
SRT | T-45730 | anterior spinal artery
SRT | T-42000 | Aorta
SRT | T-42300 | Aortic Arch
SRT | D3-81922 | Aortic fistula
SRT | T-41000 | Artery
SRT | T-42100 | Ascending aorta
SRT | T-47100 | Axillary Artery
SRT | A-00203 | Baffle
SRT | T-45800 | basilar artery
SRT | T-47160 | Brachial artery
SRT | T-46010 | brachiocephalic trunk
SRT | T-45010 | Carotid Artery
SRT | T-45510 | cerebral artery
SRT | T-45100 | Common carotid artery
SRT | T-43000 | Coronary Artery
SRT | T-42400 | Descending aorta
SRT | T-45240 | facial artery
SRT | T-47400 | Femoral artery
SRT | D4-32504 | Fistula coronary to left atrium
SRT | D4-32506 | Fistula coronary to left ventricle
SRT | D4-32508 | Fistula coronary to right atrium
SRT | D4-32510 | Fistula coronary to right ventricle
SRT | T-47490 | geniculate artery
SRT | T-46420 | Hepatic artery
SRT | T-46700 | Iliac artery
SRT | T-46010 | Innominate artery
SRT | T-45300 | internal carotid artery
SRT | T-46200 | Internal mammary artery
SRT | T-45410 | lacrimal artery
SRT | T-47651 | lateral plantar artery
SRT | T-44400 | Left pulmonary artery
SRT | T-45230 | lingual artery
SRT | T-46960 | lumbar artery
SRT | T-46500 | mesenteric artery
SRT | T-47661 | medial plantar artery
SRT | T-F7001 | Neo-aorta (primitive aorta)
SRT | T-F7040 | Neonatal pulmonary artery (primitive PA)
SRT | T-45250 | occipital artery
SRT | T-45400 | ophthalmic artery
SRT | D4-32012 | patent ductus arteriosus
SRT | T-47630 | peroneal artery
SRT | T-47500 | popliteal artery
SRT | T-45320 | posterior communicating artery
SRT | R-F5517 | Pulmonary arteriovenous fistula
SRT | T-44000 | Pulmonary artery
SRT | D4-33142 | Pulmonary artery conduit
SRT | R-00360 | Pulmonary vein wedge
SRT | T-47300 | radial artery
SRT | T-46600 | Renal artery
SRT | T-47410 | Right femoral artery
SRT | T-44200 | Right pulmonary artery
SRT | T-46100 | Subclavian Artery
SRT | T-45270 | superficial temporal artery
SRT | T-45210 | superior thyroid artery
SRT | T-44007 | Systemic collateral Artery to lung
SRT | T-42070 | Thoracic aorta
SRT | T-4704C | tibial artery
SRT | D4-31400 | Truncus Arteriosus Communis
SRT | T-F1810 | Umbilical artery
SRT | T-45700 | Vertebral artery

## CID 3607 Venous Source Locations

Type:

  Extensible

Version:

  20130617

Table CID 3607. Venous Source Locations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-48503 | Anomalous pulmonary vein
SRT | T-49215 | Antecubital Vein
SRT | T-49110 | Axillary vein
SRT | T-48340 | Azygos vein
SRT | T-48052 | Basilic vein
SRT | T-49424 | Boyd perforating vein
SRT | T-49350 | Brachial vein
SRT | T-48003 | Central venous system
SRT | T-49240 | cephalic vein
SRT | T-49429 | Dodd perforating vein
SRT | T-49410 | Femoral vein
SRT | T-48820 | gastric vein
SRT | T-48720 | hepatic vein
SRT | T-4942A | Hunterian perforating vein
SRT | T-48710 | Inferior Vena cava
SRT | T-48620 | Innominate vein
SRT | T-48170 | Internal jugular vein
SRT | T-4884A | mesenteric vein
SRT | T-48810 | portal vein
SRT | T-49535 | posterior medial tributary
SRT | T-48581 | Pulmonary vein
SRT | D4-33512 | Pulmonary vein confluence
SRT | T-48740 | Renal vein
SRT | T-D930A | Saphenofemoral junction
SRT | T-49530 | Saphenous vein
SRT | T-48890 | splenic vein
SRT | T-48330 | Subclavian vein
SRT | T-48610 | Superior vena cava
SRT | T-48817 | Umbilical vein
SRT | T-48000 | Vein
SRT | R-003AA | Vena anonyma

### Note

In a prior version of this Context Group the code T-48500 rather than T-48581 was defined for the concept Pulmonary Vein; this was inconsistent with the DICOM approach of selecting the "structure of" rather than "entire" concept. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 3608 Atrial Source Locations

Type:

  Extensible

Version:

  20030327

Table CID 3608. Atrial Source Locations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-00203 | Baffle
SRT | D4-31320 | Common atrium
SRT | T-32320 | Coronary sinus
SRT | D4-31052 | Juxtaposed appendage
SRT | T-32300 | Left atrium
SRT | G-DB27 | Pulmonary artery wedge
SRT | G-DB26 | Pulmonary capillary wedge
SRT | D4-33514 | Pulmonary venous atrium
SRT | T-32190 | Pulmonary chamber in cor triatriatum
SRT | T-32200 | Right Atrium
SRT | D4-33516 | Systemic venous atrium

## CID 3609 Ventricular Source Locations

Type:

  Extensible

Version:

  20030327

Table CID 3609. Ventricular Source Locations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-32400 | Common ventricle
SRT | T-32600 | Left ventricle
SRT | T-32602 | Left ventricle apex
SRT | T-32640 | Left ventricle inflow
SRT | D4-31022 | Left ventricle outflow chamber
SRT | T-32650 | Left ventricle outflow tract
SRT | T-32500 | Right ventricle
SRT | T-32502 | Right ventricle apex
SRT | T-32540 | Right ventricle inflow
SRT | D4-31032 | Right ventricle outflow chamber
SRT | T-32550 | Right ventricle outflow tract

## CID 3610 Gradient Source Locations

Type:

  Extensible

Version:

  20030327

Table CID 3610. Gradient Source Locations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-35300 | Mitral Valve
SRT | T-35400 | Aortic Valve
SRT | T-35100 | Tricuspid valve
SRT | T-35200 | Pulmonary valve
SRT | T-44000 | Pulmonary artery
SRT | T-32650 | Left ventricle outflow tract
SRT | T-32550 | Right ventricle outflow tract
SRT | D4-31150 | Ventricular Septal defect
SRT | D4-31220 | Atrial Septal defect
SRT | D4-32014 | Coarctation of aorta

## CID 3611 Pressure Measurements

Type:

  Extensible

Version:

  20050322

Table CID 3611. Pressure Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109016 | A wave peak pressure
DCM | 122196 | C wave pressure
LN | 8462-4 | Intravascular diastolic blood pressure
SRT | F-00E22 | Average diastolic blood pressure
SRT | F-00E1F | Minimum diastolic blood pressure
DCM | 122191 | Ventricular End Diastolic pressure
DCM | 122197 | Gradient pressure, average
DCM | 122198 | Gradient pressure, peak
SRT | F-31150 | Mean blood pressure
DCM | 122199 | Pressure at dp/dt max
LN | 8480-6 | Intravascular Systolic Blood pressure
SRT | F-00E14 | Average systolic blood pressure
SRT | F-00E11 | Maximum systolic blood pressure
DCM | 109034 | V wave peak pressure
DCM | 122208 | x-descent pressure
DCM | 122209 | y-descent pressure
DCM | 122210 | z-point pressure

## CID 3612 Blood Velocity Measurements

Type:

  Extensible

Version:

  20030327

Table CID 3612. Blood Velocity Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122201 | Diastolic blood velocity, mean
DCM | 122202 | Diastolic blood velocity, peak
DCM | 122203 | Systolic blood velocity, mean
DCM | 122204 | Systolic blood velocity, peak
DCM | 122205 | Blood velocity, mean
DCM | 122206 | Blood velocity, minimum
DCM | 122207 | Blood velocity, peak

## CID 3613 Hemodynamic Time Measurements

Type:

  Extensible

Version:

  20030327

Table CID 3613. Hemodynamic Time Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122182 | R-R interval
DCM | 109072 | Tau
DCM | 122211 | Left Ventricular ejection time
DCM | 122212 | Left Ventricular filling time
DCM | 122213 | Right Ventricular ejection time
DCM | 122214 | Right Ventricular filling time
DCM | 109071 | Indicator mean transit time

## CID 3614 Valve Areas, Non-mitral

Type:

  Extensible

Version:

  20030327

Table CID 3614. Valve Areas, Non-Mitral

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-0231F | Aortic Valve Area
SRT | F-02321 | Pulmonic Valve Area
SRT | F-02322 | Tricuspid Valve Area
DCM | 122160 | Derived Non-Valve Area

## CID 3615 Valve Areas

Type:

  Extensible

Version:

  20030327

Table CID 3615. Valve Areas

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3614 “Valve Areas, Non-mitral”
SRT | F-02320 | Mitral Valve Area

## CID 3616 Hemodynamic Period Measurements

Type:

  Extensible

Version:

  20030327

Table CID 3616. Hemodynamic Period Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-002D2 | Aortic Systolic Ejection Period (SEPa)
SRT | R-0035C | Pulmonary Systolic Ejection Period (SEPp)
SRT | R-0032C | Mitral Diastolic Filling Period (DFPm)
SRT | R-003A9 | Tricuspid Diastolic Filling Period (DFPt)
SRT | R-002F5 | Derived Period, Non-Valve

## CID 3617 Valve Flows

Type:

  Extensible

Version:

  20030327

Table CID 3617. Valve Flows

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-002D3 | Aortic Valve Flow
SRT | R-0032D | Mitral Valve Flow
SRT | R-0035D | Pulmonary Valve Flow
SRT | R-00385 | Tricuspid Valve Flow
SRT | R-00394 | Derived Flow, Non-Valve

## CID 3618 Hemodynamic Flows

Type:

  Extensible

Version:

  20030327

Table CID 3618. Hemodynamic Flows

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122161 | Pulmonary Flow
DCM | 122162 | Systemic Flow

## CID 3619 Hemodynamic Resistance Measurements

Type:

  Extensible

Version:

  20030327

Table CID 3619. Hemodynamic Resistance Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-03E86 | Pulmonary Vascular Resistance
SRT | F-03E7E | Systemic Vascular Resistance
DCM | 122215 | Total Pulmonary Resistance
DCM | 122216 | Total Vascular Resistance

## CID 3620 Hemodynamic Ratios

Type:

  Extensible

Version:

  20030327

Table CID 3620. Hemodynamic Ratios

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 8581-1 | Tibial/brachial index
SRT | F-0238B | Pulmonary/Systemic Flow Ratio
DCM | 122217 | Coronary Flow reserve
DCM | 122218 | Diastolic/Systolic velocity ratio
DCM | 122219 | Hyperemic ratio
SRT | F-031A2 | Pulsatility Index
DCM | 122220 | Hemodynamic Resistance Index
Include CID 3621 “Fractional Flow Reserve”

## CID 3621 Fractional Flow Reserve

Type:

  Extensible

Version:

  20030327

Table CID 3621. Fractional Flow Reserve

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-00307 | Fractional flow reserve
SRT | R-00308 | Fractional Flow Reserve using intracoronary bolus
SRT | R-00309 | Fractional Flow Reserve using intravenous infusion

## CID 3627 Measurement Type

Type:

  Extensible

Version:

  20060613

Table CID 3627. Measurement Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | R-002E1 | Best value | 371912002
SRT | R-00317 | Mean | 373098007
SRT | R-00319 | Median | 373099004
SRT | R-0032E | Mode | 373100007
SRT | R-00355 | Point source measurement | 371913007
SRT | R-00353 | Peak to peak | 371914001
SRT | R-41D27 | Visual estimation | 258083009
SRT | R-10260 | Estimated | 414135002
SRT | R-41D2D | Calculated | 258090004
SRT | R-41D41 | Measured | 258104002

## CID 3628 Cardiac Output Methods

Type:

  Extensible

Version:

  20030327

Table CID 3628. Cardiac Output Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-002E5 | Thermal Bath
SRT | R-002E7 | Thermal Inline
SRT | R-002E6 | Dye Dilution

## CID 3629 Procedure Intent

This Context Group specifies the intent for a procedure or a procedure step, depending on the context of invocation. The intent for a procedure step may be different than that of the procedure in which it occurs.

### Note

1. For example, a surgical biopsy procedure may have "Diagnostic Intent", while the imaging procedure step within that procedure may have "Guidance Intent".

2. Collection of specimens is generally "Diagnostic Intent"; "Forensic Intent" is typically used for autopsies; "Palliative Intent" and "Adjuvant Intent" may apply to certain radiotherapy procedures.

3. In SNOMED-CT, "Staging Intent" is a subsidiary concept (refinement) of "Diagnostic Intent". The following are subsidiary concepts of "Therapeutic Intent": Adjunct, Adjuvant, Curative, Neo-adjuvant, Prophylactic, and Supportive. Prophylactic is also a subsidiary concept of Preventive intent.

Type:

  Extensible

Version:

  20120326

Table CID 3629. Procedure Intent

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-408C3 | Diagnostic Intent
SRT | R-41531 | Therapeutic Intent
SRT | R-002E9 | Combined Diagnostic and Therapeutic Procedure
DCM | 113680 | Quality Control Intent
SRT | R-408F2 | Staging intent
SRT | R-40641 | Guidance Intent
SRT | R-40644 | Forensic Intent
SRT | R-42453 | Screening Intent
SRT | R-40644 | Palliative Intent
SRT | R-41564 | Adjunct intent
SRT | R-41561 | Adjuvant intent
SRT | R-41560 | Curative intent
SRT | R-41562 | Neo-adjuvant intent
SRT | R-41563 | Supportive intent
SRT | P0-02179 | Preventive intent
SRT | P0-02180 | Prophylactic intent

## CID 3630 Cardiovascular Anatomic Locations

Type:

  Extensible

Version:

  20030327

Table CID 3630. Cardiovascular Anatomic Locations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3606 “Arterial Source Locations”
Include CID 3607 “Venous Source Locations”
Include CID 3608 “Atrial Source Locations”
Include CID 3609 “Ventricular Source Locations”
Include CID 3610 “Gradient Source Locations”

## CID 3640 Hypertension

Type:

  Extensible

Version:

  20040920

Table CID 3640. Hypertension

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D3-40300 | Pulmonary hypertension
SRT | D3-02000 | Systemic arterial hypertension

## CID 3641 Hemodynamic Assessments

Type:

  Extensible

Version:

  20040920

Table CID 3641. Hemodynamic Assessments

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-03E0D | Left Ventricular Systolic Pressure
SRT | F-03E0E | Left Ventricular End-Diastolic Pressure
SRT | F-0212C | Pulmonary Artery Pressure
SRT | F-03E86 | Pulmonary Vascular Resistance
SRT | F-31146 | Pulmonary Capillary Wedge Pressure
SRT | F-03DFE | Right Ventricular Systolic Pressure
SRT | F-03E02 | Right Ventricular End-Diastolic Pressure
SRT | F-03DE9 | Right Atrial Pressure
SRT | F-39790 | Vascular Resistance
SRT | F-31120 | Diastolic Pressure

## CID 3642 Degree Findings

Type:

  Extensible

Version:

  20040920

Table CID 3642. Degree Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A316 | Decreased
SRT | G-A373 | Elevated
SRT | G-A37A | Severely Elevated
SRT | R-40765 | Normal Range

## CID 3651 Hemodynamic Measurement Phase

This context group is a subset of CID 3250 “Catheterization Procedure Phase”.

Type:

  Extensible

Version:

  20030327

Table CID 3651. Hemodynamic Measurement Phase

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-7293 | Cardiac catheterization baseline phase
SRT | G-729B | Cardiac catheterization post contrast phase
SRT | G-7298 | Cardiac catheterization post-intervention phase
SRT | G-929D | Cardiac catheterization test/challenging phase
SRT | R-002E3 | Cardiac catheterization gradient assessment phase
SRT | P2-71317 | Drug Infusion Challenge
SRT | P2-71310 | Exercise challenge
SRT | F-01604 | Resting State

## CID 3663 Body Surface Area Equations

Type:

  Extensible

Version:

  20100609

Table CID 3663. Body Surface Area Equations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122240 | BSA = 0.003207*WT^(0.7285-0.0188 log (WT)) *HT^0.3
DCM | 122241 | BSA = 0.007184*WT^ 0.425*HT^0.725
DCM | 122242 | BSA = 0.0235*WT^0.51456*HTcm^ 0.42246
DCM | 122243 | BSA = 0.024265*WT^0.5378*HTcm^0.3964
DCM | 122244 | BSA = (HT * WT/36) ^0.5
DCM | 122245 | BSA = 1321+0.3433*WT
DCM | 122246 | BSA = 0.0004688*WT^(0.8168-0.0154*log(WT))
DCM | 122266 | BSA = 0.007358*WT^ 0.425*HT^0.725
DCM | 122267 | BSA = 0.010265*WT^ 0.423*HT^0.651
DCM | 122268 | BSA = 0.008883*WT^ 0.444*HT^0.663
DCM | 122269 | BSA = 0.038189*WT^ 0.423*HT^0.362
DCM | 122270 | BSA = 0.009568*WT^ 0.473*HT^0.655

## CID 3664 Oxygen Consumption Equations and Tables

Type:

  Extensible

Version:

  20030327

Table CID 3664. Oxygen Consumption Equations and Tables

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122247 | VO2male = BSA (138.1 - 11.49 * loge(age) + 0.378*HRf)
DCM | 122248 | VO2female = BSA (138.1 - 17.04 * loge(age) + 0.378*HRf)
DCM | 122249 | VO2 = VeSTPD * 10 * (FIO2 - FE02)
DCM | 122250 | VO2 = 152 * BSA
DCM | 122251 | VO2 = 175 * BSA
DCM | 122252 | VO2 = 176 * BSA
DCM | 122253 | Robertson & Reid table
DCM | 122254 | Fleisch table
DCM | 122255 | Boothby table

## CID 3666 P50 Equations

Type:

  Extensible

Version:

  20030327

Table CID 3666. P50 Equations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122256 | if (prem age< 3days) P50 = 19.9
DCM | 122257 | if (age < 1day) P50 = 21.6
DCM | 122258 | if (age < 30day) P50 = 24.6
DCM | 122259 | if (age < 18y) P50 = 27.2
DCM | 122260 | if (age < 40y) P50 = 27.4
DCM | 122261 | if (age > 60y) P50 = 29.3

## CID 3667 Framingham Scores

Type:

  Extensible

Version:

  20030327

Table CID 3667. Framingham Scores

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122230 | 10 Year CHD Risk
DCM | 122231 | Comparative Average10 Year CHD Risk
DCM | 122232 | Comparative Low10 Year CHD Risk

## CID 3668 Framingham Tables

Type:

  Extensible

Version:

  20030327

Table CID 3668. Framingham Tables

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122233 | LDL Cholesterol Score Sheet for Men
DCM | 122234 | LDL Cholesterol Score Sheet for Women
DCM | 122235 | Total Cholesterol Score Sheet for Men
DCM | 122236 | Total Cholesterol Score Sheet for Women

## CID 3670 ECG Procedure Types

Type:

  Extensible

Version:

  20110330

Table CID 3670. ECG Procedure Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P2-3120A | 12-Lead ECG
SRT | P2-3120E | 15-Lead ECG
SRT | P2-3120C | 18-Lead ECG

## CID 3671 Reason for ECG Exam

Type:

  Extensible

Version:

  20030327

Table CID 3671. Reason for ECG Exam

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-00300 | Emergency procedure
SRT | P1-00410 | Pre-Surgery testing
SRT | R-00348 | Outpatient procedure
SRT | R-0035A | Procedure in Cardiac Care Unit
SRT | P2-10700 | Emergency Department patient visit
SRT | R-00302 | Evaluation of murmur
SRT | R-0036E | Routine procedure

## CID 3672 Pacemakers

This Context Group includes the full set of codes for types of pacemakers specified in the NASPE/BPEG Generic Pacemaker Code (NBG). The Coding Scheme Designator (0008,0102) shall be NBG.

### Note

1. A prior version of this context group used codes from the SCP-ECG vocabulary.

2. Further information at http://www.hrsonline.org/News/ep-history/topics-in-depth/modecodehistory.cfm. For reference, the scheme is reproduced here:

   Code Position | 1 - Chamber(s) paced | 2 - Chamber(s) sensed | 3 - Response to sensing | 4 - Rate modulation | 5- Multisite pacing
   Code values | O = None A = Atrium V = Ventricle D = Dual (A+V) S = Single(A or V - Mfr designation only) | O = None A = Atrium V = Ventricle D = Dual (A+V) S = Single(A or V - Mfr designation only) | O = None T = Triggered I = Inhibited D = Dual (T+I) | O = None R = Rate modulation | O = None A = Atrium V = Ventricle D = Dual (A+V)

## CID 3673 Diagnosis (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3675 Other Filters (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3676 Lead Measurement Technique

Type:

  Extensible

Version:

  20030327

Table CID 3676. Lead Measurement Technique

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-002DA | Averaged
SRT | R-0036D | Routine
SRT | R-00319 | Median
SRT | R-0036A | Representative
SRT | R-00373 | Single Beats

## CID 3677 Summary Codes ECG

Type:

  Extensible

Version:

  20111028

Table CID 3677. Summary Codes ECG

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | F-000B7 | Normal ECG | 164854000
SRT | F-38002 | Abnormal ECG | 102594003
SRT | F-38056 | Borderline Normal ECG | 251135002
SRT | F-38095 | ECG Equivocal | 370359005
DCM | 122753 | Non-diagnostic ECG |

### Note

A prior version of this context group used codes from the SCP-ECG vocabulary.

## CID 3678 Qt Correction Algorithms

Type:

  Extensible

Version:

  20081029

Table CID 3678. QT Correction Algorithms

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122730 | Bazett QT Correction Algorithm
DCM | 122731 | Hodges QT Correction Algorithm
DCM | 122732 | Fridericia QTc Algorithm
DCM | 122733 | Framingham QTc Algorithm

### Note

A prior version of this context group used codes from the SCP-ECG vocabulary.

## CID 3679 ECG Morphology Descriptions (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3680 ECG Lead Noise Descriptions

This Context Group comprises the ECG noise annotations of ISO/IEEE 11073-10102. The terms included in the table below may not constitute the complete list; see the ISO/IEEE Standard.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

Table CID 3680. ECG Lead Noise Descriptions

Coding Scheme | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 10:11200 | No noise | MDC_ECG_NOISE_CLEAN
MDC | 10:11216 | Moderate noise | MDC_ECG_NOISE_MODERATE
MDC | 10:11232 | Severe noise | MDC_ECG_NOISE_SEVERE
MDC | 10:11248 | No signal | MDC_ECG_NOISE_NOSIGNAL

A prior version of this context group used codes from the SCP-ECG vocabulary.

## CID 3681 ECG Lead Noise Modifiers (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3682 Probability (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3683 Modifiers (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3684 Trend (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3685 Conjunctive Terms (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3686 ECG Interpretive Statements (Retired)

This Context Group is retired. See PS3.16-2009.

## CID 3687 Electrophysiology Waveform Durations

This Context Group consists of the per-lead terms under the hierarchy of Reference ID MDC_ECG_TIME_PD in the ISO/IEEE 11073-10102 nomenclature.

The base terms from that hierarchy are included in the table below for reference. The per-lead base terms are pre-coordinated with concept discriminators for specific leads, and the code values for those pre-coordinated terms are arithmetically derived from the code values of the base terms. For the complete current list of terms and discriminator values, see the ISO/IEEE Standard. All pre-coordinated terms (measurements plus discriminators) within the identified hierarchy are part of this Context Group.

### Note

1. A prior version of this context group used codes from the SCP-ECG coding system.

2. Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

Table CID 3687. Electrophysiology Waveform Durations

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:6656 | P duration, per lead | MDC_ECG_TIME_PD_P
MDC | 2:4608 | P onset to P1 duration, per lead | MDC_ECG_TIME_PD_P1
MDC | 2:4864 | P onset to P2 duration, per lead | MDC_ECG_TIME_PD_P2
MDC | 2:5120 | P onset to P3 duration, per lead | MDC_ECG_TIME_PD_P3
MDC | 2:7168 | P offset to QRS onset duration, per lead | MDC_ECG_TIME_PD_PR
MDC | 2:7680 | Q duration, per lead | MDC_ECG_TIME_PD_Q
MDC | 2:7936 | QRS duration, per lead | MDC_ECG_TIME_PD_QRS
MDC | 2:8192 | QT duration, per lead | MDC_ECG_TIME_PD_QT
MDC | 2:11264 | R1 duration, per lead | MDC_ECG_TIME_PD_R_1
MDC | 2:11520 | R2 duration, per lead | MDC_ECG_TIME_PD_R_2
MDC | 2:11776 | R3 duration, per lead | MDC_ECG_TIME_PD_R_3
MDC | 2:12032 | S1 duration, per lead | MDC_ECG_TIME_PD_S_1
MDC | 2:12288 | S2 duration, per lead | MDC_ECG_TIME_PD_S_2
MDC | 2:12544 | S3 duration, per lead | MDC_ECG_TIME_PD_S_3
MDC | 2:11008 | Ventricular activation time, per lead | MDC_ECG_TIME_PD_VENT_ACTIV
MDC | 2:32768 | PP time period, per lead | MDC_ECG_TIME_PD_PP
MDC | 2:33024 | RR time period, per lead | MDC_ECG_TIME_PD_RR
MDC | 2:33280 | PQ time period, per lead | MDC_ECG_TIME_PD_PQ
MDC | 2:33536 | PQ segment time period, per lead | MDC_ECG_TIME_PD_PQ_SEG
MDC | 2:34560 | QTU time period, per lead | MDC_ECG_TIME_PD_QTU

## CID 3688 Electrophysiology Waveform Voltages

This Context Group consists of the codes of the hierarchies under Reference IDs MDC_ECG_ELEC_POTL and MDC_ECG_AMPL of the ISO/IEEE 11073-10102 nomenclature.

The base terms from those hierarchies are included in the table below for reference. The per lead base terms are pre-coordinated with concept discriminators for specific leads, and the code values for those pre-coordinated terms are arithmetically derived from the code values of the base terms. For the complete current list of terms and discriminator values, see the ISO/IEEE Standard. All pre-coordinated terms (measurements plus discriminators) within the identified hierarchies are part of this Context Group.

### Note

1. A prior version of this context group used codes from the SCP-ECG coding system.

2. Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

Table CID 3688. Electrophysiology Waveform Voltages

Coding Scheme Designator | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:1024 | J point Amplitude, per lead | MDC_ECG_AMPL_J
MDC | 2:14848 | Amplitude at 20 ms into ST segment, per lead | MDC_ECG_ELEC_POTL_ST_20
MDC | 2:15104 | Amplitude at 40 ms into ST segment, per lead | MDC_ECG_ELEC_POTL_ST_40
MDC | 2:14336 | Amplitude at 60 ms into ST segment, per lead | MDC_ECG_ELEC_POTL_ST_60
MDC | 2:14592 | Amplitude at 80 ms into ST segment, per lead | MDC_ECG_ELEC_POTL_ST_80
MDC | 2:1280 | P maximum amplitude, per lead | MDC_ECG_AMPL_P_MAX
MDC | 2:1536 | P minimum amplitude, per lead | MDC_ECG_AMPL_P_MIN
MDC | 2:3072 | P3 amplitude, per lead | MDC_ECG_AMPL_P3
MDC | 2:1792 | Q amplitude, per lead | MDC_ECG_AMPL_Q
MDC | 2:2048 | R amplitude, per lead | MDC_ECG_AMPL_R
MDC | 2:12800 | R1 amplitude, per lead | MDC_ECG_ELEC_POTL_R_1
MDC | 2:13056 | R2 amplitude, per lead | MDC_ECG_ELEC_POTL_R_2
MDC | 2:13312 | R3 amplitude, per lead | MDC_ECG_ELEC_POTL_R_3
MDC | 2:2304 | S amplitude, per lead | MDC_ECG_AMPL_S
MDC | 2:13568 | S1 amplitude, per lead | MDC_ECG_ELEC_POTL_S_1
MDC | 2:13824 | S2 amplitude, per lead | MDC_ECG_ELEC_POTL_S_2
MDC | 2:14080 | S3 amplitude, per lead | MDC_ECG_ELEC_POTL_S_3
MDC | 2:2560 | T maximum amplitude, per lead | MDC_ECG_AMPL_T_MAX
MDC | 2:2816 | T minimum amplitude, per lead | MDC_ECG_AMPL_T_MIN
MDC | 2:768 | ST amplitude, per lead | MDC_ECG_AMPL_ST

## CID 3689 ECG Global Waveform Durations

This Context Group consists of the global terms under the hierarchy of Reference ID MDC_ECG_TIME_PD in the ISO/IEEE 11073-10102 nomenclature.

The base terms from that hierarchy are included in the table below for reference. The base terms may be pre-coordinated with concept discriminators, and the code values for those pre-coordinated terms are arithmetically derived from the code values of the base terms. For the complete current list of terms and discriminator values, see the ISO/IEEE Standard. All pre-coordinated terms (measurements plus discriminators) within the identified hierarchy are part of this Context Group.

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

Table CID 3689. ECG Global Waveform Durations

Coding Scheme | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 2:16184 | P duration, global | MDC_ECG_TIME_PD_P_GL
MDC | 2:16140 | PP time period, global | MDC_ECG_TIME_PD_PP_GL
MDC | 2:16144 | PQ time period, global | MDC_ECG_TIME_PD_PQ_GL
MDC | 2:15872 | PR time period, global | MDC_ECG_TIME_PD_PR_GL
MDC | 2:16148 | PQ segment time period, global | MDC_ECG_TIME_PD_PQ_SEG_GL
MDC | 2:16156 | QRS duration, global | MDC_ECG_TIME_PD_QRS_GL
MDC | 2:16160 | QT duration, global | MDC_ECG_TIME_PD_QT_GL
MDC | 2:16000 | RR time period, global | MDC_ECG_TIME_PD_RR_GL
MDC | 2:16004 | QTU time period, global | MDC_ECG_TIME_PD_QTU_GL

## CID 3690 ECG Control Variables Numeric

This Context Group includes the ECG control variables specified in the ISO/IEEE 11073-10102 nomenclature that take numeric values. The terms are included in the table below for reference; these may not constitute the complete current list (see the ISO/IEEE Standard).

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

Table CID 3690. ECG Control Variables Numeric

Coding Scheme | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 10:11393 | Sample rate | MDC_ECG_CTL_VBL_SAMPLE_RATE
MDC | 10:11394 | Sensitivity | MDC_ECG_CTL_VBL_SENSITIVITY
MDC | 10:11395 | Zero offset | MDC_ECG_CTL_VBL_ZERO_OFFSET
MDC | 10:11397 | Pad value | MDC_ECG_CTL_VBL_PAD_VALUE
MDC | 10:11398 | Time skew | MDC_ECG_CTL_VBL_TIME_SKEW
MDC | 10:11399 | Sample skew | MDC_ECG_CTL_VBL_SAMPLE_SKEW
MDC | 10:11400 | Time offset | MDC_ECG_CTL_VBL_TIME_OFFSET
MDC | 10:11403 | Low pass filter cutoff freq | MDC_ECG_CTL_VBL_ATTR_FILTER_CUTOFF_FREQ
MDC | 10:11408 | Notch filter frequency | MDC_ECG_CTL_VBL_ATTR_FILTER_NOTCH_FREQ
MDC | 10:11409 | Notch filter bandwidth | MDC_ECG_CTL_VBL_ATTR_FILTER_NOTCH_BANDWIDTH
MDC | 10:11418 | Interpolator SNR | MDC_ECG_CTL_VBL_INTERPOLATOR_SNR

## CID 3691 ECG Control Variables Text

This Context Group includes the ECG control variables specified in the ISO/IEEE 11073-10102 nomenclature that take text or coded values. The terms are included in the table below for reference; these may not constitute the complete current list (see the ISO/IEEE Standard).

### Note

Codes reprinted by permission of IEEE, Copyright 2004 by IEEE. ISO/IEEE 11073-10102 available through http://standards.ieee.org/.

Type:

  Extensible

Version:

  20110330

Table CID 3691. ECG Control Variables Text

Coding Scheme | Code Value | Code Meaning | ISO/IEEE 11073 MDC Equivalent Reference ID (Informative)
--- | --- | --- | ---
MDC | 10:11402 | Low pass filter | MDC_ECG_CTL_VBL_ATTR_FILTER_LOW_PASS
MDC | 10:11404 | High pass filter | MDC_ECG_CTL_VBL_ATTR_FILTER_HIGH_PASS
MDC | 10:11406 | High pass filter description | MDC_ECG_CTL_VBL_ATTR_FILTER_DESCRIPTION
MDC | 10:11407 | Notch filter | MDC_ECG_CTL_VBL_ATTR_FILTER_NOTCH
MDC | 10:11410 | Notch filter description | MDC_ECG_CTL_VBL_ATTR_FILTER_NOTCH _DESCRIPTION
MDC | 10:11412 | Baseline description | MDC_ECG_CTL_VBL_BASELINE_DESC
MDC | 10:11414 | Interpolator | MDC_ECG_CTL_VBL_INTERPOLATOR
MDC | 10:11416 | Interpolator description | MDC_ECG_CTL_VBL_INTERPOLATOR_DESC

## CID 3692 ICDs

This Context Group includes the full set of codes for types of implanted cardioverter/defibrillators (ICDs) specified in the NASPE/BPEG Defibrillator Code (NBD). The Coding Scheme Designator (0008,0102) shall be NBD.

### Note

Further information at http://www.hrsonline.org/News/ep-history/topics-in-depth/modecodehistory.cfm. For reference, the scheme is reproduced here:

Code Position | Shock chamber | Antitachycardia pacing chamber | Tachycardia detection | Antibradycardia pacing chamber
Code values | O = None A = Atrium V = Ventricle D = Dual (A+V) | O = None A = Atrium V = Ventricle D = Dual (A+V) | E = Electrogram H = Hemodynamic | O = None A = Atrium V = Ventricle D = Dual (A+V)

Short Form
ICD-S = ICD with shock capability only ICD-B = ICD with bradycardia pacing as well as shock ICD-T = ICD with tachycardia (and bradycardia) pacing as well as shock

## CID 3700 Cath Diagnosis

Type:

  Extensible

Version:

  20030327

Table CID 3700. Cath Diagnosis

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D3-13000 | Coronary artery disease
SRT | D3-15100 | Acute myocardial infarction
SRT | F-37012 | Atypical chest pain
SRT | D3-13020 | Stable Angina
SRT | D3-12400 | Atypical Angina, Variant Angina
SRT | D3-12700 | Unstable Angina, Progressive Angina
SRT | D3-13014 | Post-infarction angina
SRT | R-00368 | Recurrent angina Post-PTCA
SRT | R-00367 | Recurrent angina Post-DCA
SRT | R-00369 | Recurrent angina Post-Rotational Atherectomy
SRT | R-00366 | Recurrent angina Post-Stent
SRT | R-00365 | Recurrent angina Post-CABG
SRT | D3-16010 | Congestive heart failure
SRT | D2-61100 | Pulmonary edema
SRT | D3-00200 | cardiogenic shock
SRT | R-002CB | Acute ventricular septal rupture
SRT | D3-29010 | Mitral valve disease
SRT | D3-29011 | Mitral stenosis
SRT | D3-29012 | Mitral regurgitation
SRT | D3-29096 | Acute mitral regurgitation
SRT | D3-13021 | Silent ischemia
SRT | R-00336 | s/p MI positive stress for ischemia
SRT | D3-26000 | Myocarditis
SRT | D3-28012 | Subacute bacterial endocarditis
SRT | D3-2906A | Idiopathic hypertrophic subaortic stenosis
SRT | D3-40300 | Pulmonary hypertension
SRT | D3-29040 | Tricuspid valve disease
SRT | D3-29042 | Tricuspid regurgitation
SRT | D3-29013 | Mitral valve prolapse
SRT | D3-31700 | Ventricular tachycardia
SRT | D3-31720 | Ventricular fibrillation
SRT | D3-21000 | Congestive cardiomyopathy
SRT | D3-02500 | Hypertensive heart disease
SRT | D3-22100 | Restrictive cardiomyopathy
SRT | D3-90000 | Pericardial disease
SRT | D3-90100 | Pericardial tamponade
SRT | D3-29020 | Aortic valve disease
SRT | D3-29021 | Aortic stenosis
SRT | D3-29025 | Aortic insufficiency
SRT | D4-31220 | Atrial septal defect
SRT | D3-80016 | Aortic dissection
SRT | D3-29050 | Pulmonic valve disease
SRT | D4-31159 | Ventricular septal defect
SRT | D3-83300 | Aortic aneurysm
SRT | R-10042 | Arhythmia Evaluation
SRT | D3-31520 | Atrial fibrillation
SRT | D4-31000 | heart disease, congenital
SRT | D3-91030 | Constrictive pericarditis

## CID 3701 Cardiac Valves and Tracts

Type:

  Extensible

Version:

  20030327

Table CID 3701. Cardiac Valves and Tracts

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-35300 | Mitral Valve
SRT | T-35400 | Aortic Valve
SRT | T-35100 | Tricuspid valve
SRT | T-35200 | Pulmonary valve
SRT | T-32650 | Left ventricle outflow tract

## CID 3703 Wall Motion

Type:

  Extensible

Version:

  20111028

Table CID 3703. Wall Motion

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | R-00378 | Not Evaluated | 373121007
SRT | R-41198 | Unknown | 261665006
DCM | 122288 | Not visualized |
SRT | R-00344 | Normal wall motion | 373122000
SRT | R-0030D | Hyperkinetic region | 373123005
SRT | F-32056 | Hypokinesis | 37706002
SRT | R-00327 | Mild Hypokinesis | 371868005
SRT | R-0032F | Moderate Hypokinesis | 371869002
SRT | R-00370 | Severe Hypokinesis | 371870001
SRT | F-30004 | Akinesis | 195675009
SRT | F-32052 | Dyskinesis | 25437005

### Note

In prior editions, this Context Group included incorrect codes for "Hypokinesis" and "Mild Hypokinesis" (see PS3.16-2011).

## CID 3704 Myocardium Wall Morphology Findings

Type:

  Extensible

Version:

  20030327

Table CID 3704. Myocardium Wall Morphology Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122112 | Normal Myocardium
SRT | D3-10510 | Ventricular Aneurysm
DCM | 122113 | Scarred Myocardium
DCM | 122114 | Thinning Myocardium

## CID 3705 Chamber Size

Type:

  Extensible

Version:

  20030327

Table CID 3705. Chamber Size

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-00343 | Normal size cardiac chamber
SRT | R-002C6 | Abnormally small cardiac chamber
SRT | R-0032A | Mildly Enlarged cardiac chamber
SRT | R-00331 | Moderately Enlarged cardiac chamber
SRT | R-00316 | Markedly Enlarged cardiac chamber

## CID 3706 Overall Contractility

Type:

  Extensible

Version:

  20111028

Table CID 3706. Overall Contractility

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-00341 | Normal wall contractility
SRT | R-00398 | Hyperkinesis
SRT | F-32056 | Hypokinesis
SRT | F-30004 | Akinesis

## CID 3707 VSD Description

Type:

  Extensible

Version:

  20030327

Table CID 3707. VSD Description

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D4-31154 | Membranous
SRT | R-0033B | Non-restrictive
SRT | D4-31166 | Restrictive
SRT | R-40775 | None

## CID 3709 Aortic Root Description

Type:

  Extensible

Version:

  20030327

Table CID 3709. Aortic Root Description

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-0033C | Normal Aortic Root
SRT | R-00301 | Enlarged Aortic Root
SRT | R-002CD | Aneurysm of Aortic Root
SRT | R-002D1 | Annular Abscess of Aortic Root
SRT | R-003A1 | Post Stenotic Dilation
SRT | D3-83660 | Ruptured Sinus of Valsalva

## CID 3710 Coronary Dominance

Type:

  Extensible

Version:

  20111028

Table CID 3710. Coronary Dominance

Coding Scheme Designator | Code Value | Code Meaning | Equivalent SNOMED-CT Concept ID
--- | --- | --- | ---
SRT | D4-3252C | Left Coronary Dominance | 253729004
SRT | D4-3252B | Right Coronary Dominance | 253728007
SRT | D4-3252D | Balanced Coronary Dominance | 253730009

### Note

In prior editions, this Context Group included NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes (see PS3.16-2011).

## CID 3711 Valvular Abnormalities

Type:

  Extensible

Version:

  20030327

Table CID 3711. Valvular Abnormalities

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D3-29001 | Stenosis
SRT | F-32400 | Regurgitation
SRT | R-0030B | Calcified Heart Valve
SRT | R-0030F | Immobile Heart Valve
SRT | R-00305 | Heart Valve Flail
SRT | D3-28005 | Valvular endocarditis

## CID 3712 Vessel Descriptors

Type:

  Extensible

Version:

  20030327

Table CID 3712. Vessel Descriptors

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-00389 | Ulcerated
SRT | R-0036B | Restenotic
SRT | R-002E2 | Bifurcation
SRT | R-002EF | Culprit
SRT | R-002CE | Aneurysmal
SRT | R-002FC | Diffuse Disease
SRT | R-00314 | Luminal Irregularities
SRT | R-411C5 | Muscle Bridge
SRT | R-10050 | Stenotic
SRT | R-10051 | Ectatic
SRT | G-A264 | Calcified
SRT | M-35100 | Thrombus
SRT | R-10048 | Tortuous
SRT | R-10049 | Stented

## CID 3713 TIMI Flow Characteristics

Type:

  Extensible

Version:

  20030327

Table CID 3713. TIMI Flow Characteristics

Coding Scheme Designator | Code Value | Code Meaning | NCDR Equivalent
--- | --- | --- | ---
SRT | R-0037E | 0: No Perfusion | 106-0, 107-0
SRT | R-0037F | 1: Penetration without Perfusion | 106-1, 107-1
SRT | R-00381 | 2: Partial Perfusion | 106-2, 107-2
SRT | R-00382 | 3: Complete Perfusion | 106-3, 107-3

## CID 3714 Thrombus

Type:

  Extensible

Version:

  20030327

Table CID 3714. Thrombus

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-0033A | No Thrombus
SRT | R-00356 | Possible Thrombus
SRT | R-002F1 | Definite Thrombus
SRT | R-00371 | Severe Thrombus

## CID 3715 Lesion Margin

Type:

  Extensible

Version:

  20050614

Table CID 3715. Lesion Margin

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A545 | Smooth
SRT | G-A402 | Irregular
SRT | R-00335 | Multiple Irregularities
SRT | R-403CC | Ulcerative

## CID 3716 Severity

Type:

  Extensible

Version:

  20070827

Table CID 3716. Severity

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-40775 | None
SRT | R-404FA | Mild
SRT | R-00329 | Mild to Moderate
SRT | G-A002 | Moderate
SRT | R-00330 | Moderate to Severe
SRT | G-A003 | Severe
SRT | R-4099D | Fatal

## CID 3717 Myocardial Wall Segments

This 17-segment model of left ventricular myocardial wall segments uses the terminology specified in " AHA Scientific Statement: Standardized Myocardial Segmentation and Nomenclature for Tomographic Imaging of the Heart"(see Section 2).

Type:

  Non-Extensible

Version:

  20030327

Table CID 3717. Myocardial Wall Segments

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | T-32619 | left ventricle basal anterior segment | 264850008
SRT | R-10075 | left ventricle basal anteroseptal segment | 396482007
SRT | R-10076 | left ventricle basal inferoseptal segment | 396646008
SRT | T-32615 | left ventricle basal inferior segment | 264846001
SRT | R-10079 | left ventricle basal inferolateral segment | 396652009
SRT | R-1007A | left ventricle basal anterolateral segment | 396654005
SRT | T-32617 | left ventricle mid anterior segment | 264848000
SRT | R-10077 | left ventricle mid anteroseptal segment | 396647004
SRT | R-10078 | left ventricle mid inferoseptal segment | 396649001
SRT | T-32616 | left ventricle mid inferior segment | 264847005
SRT | R-1007B | left ventricle mid inferolateral segment | 396655006
SRT | R-1007C | left ventricle mid anterolateral segment | 396656007
SRT | T-32613 | left ventricle apical anterior segment | 264844003
SRT | T-32614 | left ventricle apical septal segment | 264845002
SRT | T-32618 | left ventricle apical inferior segment | 264849008
SRT | T-3261C | left ventricle apical lateral segment | 264853005
SRT | T-32602 | apex of left ventricle | 128564006

## CID 3718 Myocardial Wall Segments in Projection

This context group specifies the left ventricular myocardial wall segments as seen in typical right anterior oblique (RAO) and left anterior oblique (LAO) angiographic projections.

Type:

  Extensible

Version:

  20030614

Table CID 3718. Myocardial Wall Segments in Projection

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | T-32619 | left ventricle basal anterior segment | 264850008
SRT | T-32634 | myocardium of anterolateral region | 73050001
SRT | T-32636 | myocardium of apex of heart | 47962008
SRT | T-32632 | myocardium of diaphragmatic region | 72542009
SRT | T-32615 | left ventricle basal inferior segment | 264846001
SRT | T-32603 | left ventricle basal lateral segment | 277631004
SRT | T-32633 | myocardium of posterolateral region | 33272004
SRT | T-32637 | myocardium of inferolateral region | 16239001
SRT | T-32614 | left ventricle apical septal segment | 264845002
SRT | T-32601 | left ventricular basal septal segment | 277630003
SRT | R-101C0 | left ventricular posterobasal segment | 408720008

## CID 3719 Canadian Clinical Classification

Type:

  Extensible

Version:

  20070827

Table CID 3719. Canadian Clinical Classification

Coding Scheme Designator | Code Value | Code Meaning | NCDR [2.0b] Equivalent
--- | --- | --- | ---
SRT | F-A265A | Chest pain not present | 50-0
SRT | D3-12001 | Angina Class I | 50-I
SRT | D3-12002 | Angina Class II | 50-II
SRT | D3-12003 | Angina Class III | 50-IIII
SRT | D3-12004 | Angina Class IV | 50-IV

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3720 Cardiac History Dates (Retired)

This Context Group is retired. See PS3.16-2007.

## CID 3721 Cardiovascular Surgeries

Type:

  Extensible

Version:

  20081027

Table CID 3721. Cardiovascular Surgeries

Coding Scheme Designator | Code Value | Code Meaning | NCDR [2.0b] Equivalent
--- | --- | --- | ---
SRT | R-102B4 | Percutaneous coronary intervention | 40
SRT | P1-3301A | Coronary artery bypass graft | 42
SRT | P1-32000 | Operation on heart valve | 44
SRT | P1-31C03 | Ablation operation for arrhythmia |
SRT | P0-004BA | Implantation of cardiac pacemaker |
SRT | P1-3157D | Implantation of automatic cardiac defibrillator |
SRT | P1-0555A | Abdominal aortic aneurysm stenting |
SRT | P1-31D00 | Heart transplant |
SRT | P1-080B4 | Correction of congenital cardiovascular deformity |

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3722 Diabetic Therapy

Type:

  Extensible

Version:

  20141103

Table CID 3722. Diabetic Therapy

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | F-02F14 | Diabetic on Dietary Treatment | 170745003
SRT | F-02F15 | Diabetic on Oral Treatment | 170746002
SRT | F-02F16 | Diabetic on Insulin | 170747006

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3723 MI Types

Type:

  Extensible

Version:

  20070827

Table CID 3723. MI Types

Coding Scheme Designator | Code Value | Code Meaning | NCDR [2.0b] Equivalent
--- | --- | --- | ---
SRT | D3-1511A | Non ST Elevation Myocardial Infarction | 94-1
SRT | D3-15119 | ST Elevation Myocardial Infarction | 94-2

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3724 Smoking History

Type:

  Extensible

Version:

  20070827

Table CID 3724. Smoking History

Coding Scheme Designator | Code Value | Code Meaning | NCDR [2.0b] Equivalent | SNOMED-CT Concept ID Equivalent
--- | --- | --- | --- | ---
SRT | F-9321F | No History of Smoking | 38-0 | 266919005
SRT | S-32000 | Current Smoker | 38-1 | 77176002
SRT | S-32070 | Former Smoker | 38-2 | 8517006

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3726 Indications for Coronary Intervention

Type:

  Extensible

Version:

  20030327

Table CID 3726. Indications for Coronary Intervention

Coding Scheme Designator | Code Value | Code Meaning | NCDR Equivalent
--- | --- | --- | ---
DCM | 122171 | Coronary lesion > = 50% stenosis |
SRT | D3-00200 | Cardiogenic Shock | 123

## CID 3727 Indications for Catheterization

Type:

  Extensible

Version:

  20111028

Table CID 3727. Indications for Catheterization

Coding Scheme Designator | Code Value | Code Meaning | Equivalent SNOMED-CT Concept ID
--- | --- | --- | ---
SRT | D3-00200 | cardiogenic shock | 89138009
SRT | D3-10800 | valvular heart disease | 368009
SRT | D3-30000 | Arrhythmia | 44808001
SRT | D3-10030 | ischemic heart disease | 414545008
SRT | F-000FF | cardiac function test abnormal | 165076002
SRT | P1-31D00 | heart transplant | 32413006
SRT | D4-31000 | heart disease - congenital | 13213009
SRT | D3-20000 | cardiomyopathy | 85898001
SRT | D3-10000 | heart disease | 56265001

### Note

1. (D3-10000, SRT, "heart disease") should be used only when a more specific characterization of the disease is not applicable.

2. In prior editions, this Context Group included NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes (see PS3.16-2011).

## CID 3728 Cath Findings

Type:

  Extensible

Version:

  20030327

Table CID 3728. Cath Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-0033F | Normal left heart hemodynamics
SRT | R-00342 | Normal right heart hemodynamics
SRT | R-0033E | Normal left and right heart hemodynamics
SRT | R-00340 | Normal left ventricular systolic function and wall motion
SRT | R-0033D | Normal coronary arteries
SRT | R-00328 | Mild intimal coronary irregularities, no significant stenoses
SRT | R-00374 | Single vessel coronary artery disease.
SRT | R-002FE | Double vessel coronary artery disease.
SRT | R-00386 | Triple vessel coronary artery disease.
SRT | R-00334 | Multi vessel coronary artery disease.
SRT | R-00313 | Left main coronary artery disease
SRT | R-00372 | Significant coronary bypass graft disease
SRT | D3-29021 | Aortic stenosis
SRT | D3-29025 | Aortic insufficiency
SRT | D3-29011 | Mitral stenosis
SRT | D3-29012 | Mitral regurgitation
SRT | R-002F3 | Depression of left ventricular systolic function
SRT | R-002C8 | Acute mitral regurgitation from chordal rupture
SRT | R-002C7 | Acute mitral regurgitation from chordal dysfunction
SRT | R-002CA | Acute mitral regurgitation from papillary muscle rupture
SRT | R-002C9 | Acute mitral regurgitation from papillary muscle dysfunction
SRT | D3-29013 | Mitral valve prolapse
SRT | D3-2100 | Congestive cardiomyopathy
SRT | D3-23000 | Hypertrophic cardiomyopathy with obstruction
SRT | D3-20003 | Hypertrophic cardiomyopathy without obstruction
SRT | D3-02500 | Hypertensive heart disease
SRT | D3-22100 | Restrictive cardiomyopathy
SRT | D3-90100 | Pericardial tamponade
SRT | D3-91030 | Constrictive pericarditis
SRT | D3-40300 | Pulmonary hypertension
SRT | D4-31220 | Atrial septal defect
SRT | D4-31159 | Ventricular septal defect
SRT | R-002CB | Acute ventricular septal rupture
SRT | D4-31000 | heart disease - congenital

## CID 3729 Admission Status

Type:

  Extensible

Version:

  20070827

Table CID 3729. Admission Status

Coding Scheme Designator | Code Value | Code Meaning | NCDR [2.0b] Equivalent
--- | --- | --- | ---
SRT | P0-10010 | Elective | 17-1
SRT | P0-10800 | Emergency Department | 17-2
SRT | P0-10210 | Transfer | 17-3

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3730 Insurance Payor

Type:

  Extensible

Version:

  20030327

Table CID 3730. Insurance Payor

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 18-1 | Government
NCDR | 2.0b | 18-2 | Commercial
NCDR | 2.0b | 18-3 | Health Maintenance Organization
NCDR | 2.0b | 18-4 | None

## CID 3733 Primary Cause of Death

Type:

  Extensible

Version:

  20030327

Table CID 3733. Primary Cause of Death

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 25-1 | Cardiac
NCDR | 2.0b | 25-2 | Neurologic
NCDR | 2.0b | 25-3 | Renal
NCDR | 2.0b | 25-4 | Vascular
NCDR | 2.0b | 25-5 | Infection
NCDR | 2.0b | 25-6 | Pulmonary
NCDR | 2.0b | 25-7 | Valvular
NCDR | 2.0b | 25-8 | Other

## CID 3735 Acute Coronary Syndrome Time Period

Type:

  Extensible

Version:

  20030327

Table CID 3735. Acute Coronary Syndrome Time Period

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 51-1 | < = 6 hours
NCDR | 2.0b | 51-2 | between 6 hours and 24 hours
NCDR | 2.0b | 51-3 | between 24 hours and 7 days

## CID 3736 NYHA Classification

Type:

  Extensible

Version:

  20070827

Table CID 3736. NYHA Classification

Coding Scheme Designator | Code Value | Code Meaning | NCDR [2.0b] Equivalent
--- | --- | --- | ---
SRT | F-3018B | NYHA Class I | 47-I
SRT | F-3018C | NYHA Class II | 47-II
SRT | F-3018D | NYHA Class III | 47-III
SRT | F-3018E | NYHA Class IV | 47-IV

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3737 Non-invasive Test - Ischemia

Type:

  Extensible

Version:

  20030327

Table CID 3737. Non-Invasive Test - Ischemia

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 48-1 | Not Done
NCDR | 2.0b | 48-2 | Positive
NCDR | 2.0b | 48-3 | Negative
NCDR | 2.0b | 48-4 | Equivocal
NCDR | 2.0b | 48-5 | Arrhythmia

## CID 3738 Pre-Cath Angina Type

Type:

  Extensible

Version:

  20030327

Table CID 3738. Pre-Cath Angina Type

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 49-I | Atypical Chest Pain
NCDR | 2.0b | 49-II | Stable Angina
NCDR | 2.0b | 49-IIIa | Acute Coronary Syndrome: Unstable Angina
NCDR | 2.0b | 49-IIIb | Acute Coronary Syndrome: Non ST-Elevation Myocardial Infarction
NCDR | 2.0b | 49-IIIc | Acute Coronary Syndrome: ST-Elevation Myocardial Infarction

## CID 3739 Cath Procedure Type

Type:

  Extensible

Version:

  20030327

Table CID 3739. Cath Procedure Type

Coding Scheme Designator | Code Value | Code Meaning | NCDR Equivalent
--- | --- | --- | ---
SRT | P1-31602 | Catheterization of right heart | 54-1
SRT | P1-31604 | Catheterization of left heart | 54-2
SRT | P1-3160A | Catheterization of both left and right heart with graft |
SRT | P1-3160B | Catheterization of both left and right heart without graft |
DCM | 122061 | Percutaneous Coronary Intervention | 54-3

## CID 3740 Thrombolytic Administration

Type:

  Extensible

Version:

  20030327

Table CID 3740. Thrombolytic Administration

Coding Scheme Designator | Code Value | Code Meaning | NCDR Equivalent
--- | --- | --- | ---
SRT | R-0037D | Contraindicated | 57-1
SRT | R-0037C | Administered less than 3 hours before PCI | 57-2
SRT | R-0037A | Administered between 3 and 6 hours before PCI | 57-3
SRT | R-0037B | Administered between 6 hours and 7 days before PCI | 57-4

## CID 3741 Medication Administration, Lab Visit

Type:

  Extensible

Version:

  20030327

Table CID 3741. Medication Administration, Lab Visit

Coding Scheme Designator | Code Value | Code Meaning | NCDR Equivalent
--- | --- | --- | ---
SRT | R-00321 | Contraindicated | 58-1
SRT | R-0031B | Administered before lab visit | 58-2
SRT | R-0031C | Administered during lab visit | 58-3
SRT | R-0031A | Administered after lab visit | 58-4

## CID 3742 Medication Administration, PCI

Type:

  Extensible

Version:

  20030327

Table CID 3742. Medication Administration, PCI

Coding Scheme Designator | Code Value | Code Meaning | NCDR Equivalent
--- | --- | --- | ---
SRT | R-00320 | Not Administered |
SRT | R-00321 | Contraindicated | 59-1
SRT | R-0031F | Administered Prior to Percutaneous Coronary Intervention | 59-2
SRT | R-0039A | Administered During Percutaneous Coronary Intervention | 59-3
SRT | R-00399 | Administered After Percutaneous Coronary Intervention | 59-4

## CID 3743 Clopidogrel/ticlopidine Administration

Type:

  Extensible

Version:

  20030327

Table CID 3743. Clopidogrel/Ticlopidine Administration

Coding Scheme Designator | Code Value | Code Meaning | NCDR Equivalent
--- | --- | --- | ---
SRT | R-00320 | Not Administered | 60-1
SRT | R-00321 | Contraindicated | 60-2
SRT | R-0031E | Administered Less than 72 Hours before PCI | 60-3
SRT | R-00399 | Administered After Percutaneous Coronary Intervention | 60-4

## CID 3744 EF Testing Method

Type:

  Extensible

Version:

  20030327

Table CID 3744. EF Testing Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P5-3003A | Cardiac ventriculography
SRT | P5-D3300 | Radionuclide ventriculography
SRT | P5-B3081 | Adult echocardiography

## CID 3745 Calculation Method

Type:

  Extensible

Version:

  20050110

Table CID 3745. Calculation Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-10260 | Estimated
SRT | R-41D2D | Calculated

## CID 3746 Percutaneous Entry Site

This Context Group includes concepts for Percutaneous entry that are the most relevant children of SNOMED concept 297211001 "transvascular approach". Other concepts from that hierarchy may be used as local extensions to this Context Group.

Type:

  Extensible

Version:

  20110609

Table CID 3746. Percutaneous Entry Site

Coding Scheme | Code Value | Code Meaning | Equivalent SNOMED-CT Concept ID
--- | --- | --- | ---
SRT | G-D067 | Via femoral artery | 260590008
SRT | G-D1E4 | Via radial artery | 444850002
SRT | G-D05F | Via brachial artery | 260585005
SRT | G-D054 | Via artery | 103387006
SRT | G-D0C6 | Via arm vein | 261459001
SRT | G-D071 | Via femoral vein | 260601006
SRT | G-D052 | Via vein | 103386002

### Note

In prior editions, this Context Group included NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes (see PS3.16-2009).

## CID 3747 Percutaneous Closure

Type:

  Extensible

Version:

  20030327

Table CID 3747. Percutaneous Closure

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 75-0 | No closure device used at percutaneous entry
NCDR | 2.0b | 75-1 | Percutaneous entry closed by suture
NCDR | 2.0b | 75-2 | Percutaneous entry closed by sealant
NCDR | 2.0b | 75-3 | Percutaneous entry closed by other mechanism

## CID 3748 Angiographic EF Testing Method

Type:

  Extensible

Version:

  20030327

Table CID 3748. Angiographic EF Testing Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122059 | Single plane Angiography
DCM | 122060 | Bi-plane Angiography

## CID 3749 PCI Procedure Result

Type:

  Extensible

Version:

  20030327

Table CID 3749. PCI Procedure Result

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 100-1 | Successful
NCDR | 2.0b | 100-2 | Partially successful
NCDR | 2.0b | 100-3 | Unsuccessful

## CID 3750 Previously Dilated Lesion

Type:

  Extensible

Version:

  20030327

Table CID 3750. Previously Dilated Lesion

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
NCDR | 2.0b | 108-0 | not previously treated
NCDR | 2.0b | 108-1 | balloon only
NCDR | 2.0b | 108-2 | stent only
NCDR | 2.0b | 108-3 | other/any combination

## CID 3752 Guidewire Crossing

Type:

  Extensible

Version:

  20030327

Table CID 3752. Guidewire Crossing

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122301 | Guidewire crossing lesion unsuccessful
DCM | 122302 | Guidewire crossing lesion successful

## CID 3754 Vascular Complications

Type:

  Extensible

Version:

  20050110

Table CID 3754. Vascular Complications

Coding Scheme Designator | Code Value | Code Meaning | NCDR 2.0b Equivalent Code Value
--- | --- | --- | ---
SRT | M-37000 | Bleeding | 127
SRT | D3-89100 | Occlusion of artery | 128
SRT | R-102B2 | Loss of distal pulse | 129
SRT | D3-81310 | Arterial dissection | 130
SRT | M-32390 | Pseudoaneurysm | 131
SRT | M-39390 | AV Fistula | 132

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3755 Cath Complications

Type:

  Extensible

Version:

  20050110

Table CID 3755. Cath Complications

Coding Scheme Designator | Code Value | Code Meaning | NCDR 2.0b Equivalent Code Value
--- | --- | --- | ---
SRT | D3-00200 | Cardiogenic shock | 123
SRT | D3-30000 | Arrhythmia | 124
SRT | D3-8900D | Cerebrovascular Accident or Stroke | 125
SRT | D3-90100 | Cardiac tamponade | 126
SRT | DF-10781 | Contrast media adverse reaction | 133
SRT | D3-16010 | Congestive heart failure | 134
SRT | D7-11010 | Renal failure | 135
SRT | R-102B5 | Emergency Percutaneous Coronary Intervention | 136
SRT | R-102B3 | Emergency Coronary Artery Bypass | 137
SRT | D3-30800 | Cardiac arrest |

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3756 Cardiac Patient Risk Factors

Type:

  Extensible

Version:

  20100317

Table CID 3756. Cardiac Patient Risk Factors

Coding Scheme Designator | Code Value | Code Meaning | NCDR 2.0b Equivalent Code Value
--- | --- | --- | ---
SRT | G-026D | History of congestive heart failure | 30
SRT | G-023F | History of Diabetes | 31
SRT | R-102B6 | History of renal failure | 32
SRT | R-102B7 | History of chronic lung disease | 33
SRT | G-0102 | History of cerebrovascular disease | 34
SRT | D3-8005B | Peripheral vascular disease | 35
SRT | G-03AA | History of myocardial infarction | 36
SRT | G-0269 | History of Hypertension | 37
SRT | R-102B8 | History of hypercholesterolemia | 39
SRT | D3-30000 | Arrhythmia |
SRT | F-0331B | HIV Positive |
SRT | D8-60001 | Infant of Gestational Diabetic Mother (IGDM)
SRT | G-0586 | Insulin dependent mother (IDM) |

### Note

In prior editions, this Context Group included the NCDR 2.0b codes as the primary set. These have been replaced with equivalent SNOMED codes.

## CID 3757 Cardiac Diagnostic Procedures

Type:

  Extensible

Version:

  20111028

Table CID 3757. Cardiac Diagnostic Procedures

Coding Scheme | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | P5-D3304 | Cardiac blood pool imaging (nuclear) | 35621002
SRT | P5-00A25 | Cardiac cath coronary angiogram and left ventriculogram | 418903008
SRT | P1-31600 | Cardiac catheterization | 41976001
SRT | P5-00A34 | Cardiac catheterization coronary angiogram | 419416005
SRT | P5-08025 | Cardiac CT | 241547009
SRT | P5-080C2 | Cardiac CT for calcium scoring | 426005005
SRT | P5-09011 | Cardiac MRI | 241620005
SRT | P0-00CA7 | Cardiac MRI stress | 431609005
SRT | P5-00A5C | CT angiography of coronary arteries | 419545005
SRT | P5-B3000 | Echocardiography | 40701008
SRT | P5-B3050 | Exercise Stress echocardiography | 433233004
SRT | P0-006E4 | Exercise Tolerance Test | 165079009
SRT | P5-09100 | Magnetic resonance angiography | 105370006
SRT | P5-D30F8 | Nuclear medicine cardiovascular study | 108294005
SRT | P5-D0050 | Perfusion imaging (nuclear) | 35202002
SRT | P5-0A006 | PET heart study | 241439007
SRT | P2-31011 | Pharmacologic and exercise stress test | 428813002
SRT | P2-31107 | Pharmacological stress test | 424064009
SRT | P5-30045 | Radionuclide angiocardiography | 426940008
SRT | P5-D3008 | Radionuclide myocardial perfusion study | 252432008
SRT | P5-0A100 | SPECT | 105371005
SRT | P2-3110B | Stress test using cardiac pacing | 428685003
SRT | P5-B3002 | Transesophageal echocardiography | 105376000
SRT | P5-B3012 | Transthoracic echocardiography | 433236007

### Note

In a prior version of this Context Group, the code P5-B3009 was specified for Exercise stress echocardiography. That code has been retired by SNOMED, and replaced by P5-B3050. Although there is minimal possibility of misinterpretation with SOP Instances that may include the retired code, receiving applications should be aware of this change; see Annex J.

## CID 3758 Cardiovascular Family History

Type:

  Extensible

Version:

  20111028

Table CID 3758. Cardiovascular Family History

Coding Scheme | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | G-032F | Family history of cardiovascular disease | 266894000
SRT | G-0157 | Family history of diabetes mellitus | 160303001
SRT | G-011E | Family history of myocardial infarction | 266897007
SRT | G-04E3 | Family history of coronary arteriosclerosis | 430091005
SRT | R-2087E | No family history of diabetes | 160274005
SRT | R-20773 | No family history of cardiovascular disease | 160270001
SRT | F-03F6E | Family history unknown | 407559004

## CID 3760 Hypertension Therapy

Type:

  Extensible

Version:

  20070827

Table CID 3760. Hypertension Therapy

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
SRT | C-80135 | Beta blocker
SRT | C-80160 | Calcium channel blocker
SRT | C-81520 | Nitrate vasodilator
SRT | C-80150 | ACE inhibitor
SRT | C-81300 | Angiotensin II receptor antagonist
SRT | C-72000 | Diuretic

## CID 3761 Antilipemic Agents

Type:

  Extensible

Version:

  20070827

Table CID 3761. Antilipemic Agents

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
SRT | C-80609 | Anion exchange resin
SRT | C-80610 | Bile acid sequestrant
SRT | C-80680 | Fibrate
SRT | C-8060A | Fish oils
SRT | C-80800 | Statins

## CID 3762 Antiarrhythmic Agents

Type:

  Extensible

Version:

  20070827

Table CID 3762. Antiarrhythmic Agents

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
SRT | F-6181B | class I antiarrhythmic agent
SRT | F-61861 | class II antiarrhythmic agent
SRT | F-61995 | class III antiarrhythmic agent
SRT | F-618AE | class IV antiarrhythmic agent

## CID 3764 Myocardial Infarction Therapies

Type:

  Extensible

Version:

  20070827

Table CID 3764. Myocardial Infarction Therapies

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
SRT | R-102B4 | Percutaneous coronary intervention
SRT | P1-33530 | Insertion of coronary artery stent
SRT | P1-3301A | Coronary artery bypass graft
SRT | P0-00C29 | Thrombolytic therapy

## CID 3769 Concern Types

Type:

  Extensible

Version:

  20070827

Table CID 3769. Concern Types

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
SRT | F-04BA9 | Complaint
SRT | DF-00000 | Disease
SRT | R-005AE | Finding
SRT | R-005E0 | Finding reported by patient/informant
SRT | F-03E55 | Functional performance and activity
SRT | F-01000 | Problem

## CID 3770 Problem Status

Type:

  Extensible

Version:

  20070827

Table CID 3770. Problem Status

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
SRT | R-42501 | Active problem
SRT | G-A270 | Chronic
SRT | G-A397 | Intermittent
SRT | G-A39A | Recurrent
SRT | G-A47B | Suspected
SRT | R-42502 | Inactive problem
SRT | F-04B88 | Problem resolved
SRT | G-A46B | Known absent
SRT | P0-30450 | Well controlled

## CID 3772 Health Status

Type:

  Extensible

Version:

  20070827

Table CID 3772. Health Status

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
SRT | F-00001 | Alive and well
SRT | F-029D4 | In remission
SRT | R-209F6 | Symptom free
SRT | F-0600C | Chronically ill
SRT | F-06001 | Severely ill
SRT | F-00100 | Disabled
SRT | F-0351E | Severely disabled
SRT | F-04DA1 | Deceased

## CID 3773 Use Status

Type:

  Extensible

Version:

  20070827

Table CID 3773. Use Status

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
SRT | G-D316 | Ended
SRT | G-D30F | Suspended
SRT | G-D30B | In progress

## CID 3774 Social History

Type:

  Extensible

Version:

  20070827

Table CID 3774. Social History

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
SRT | F-93109 | Tobacco Smoking Behavior
SRT | F-931D4 | Drug misuse behavior
SRT | R-40C16 | Exercise
SRT | F-045CE | Nutrition
SRT | F-02573 | Alcohol consumption

## CID 3777 Implanted Devices

Type:

  Extensible

Version:

  20070827

Table CID 3777. Implanted Devices

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
SRT | A-11100 | Cardiac pacemaker
SRT | A-11206 | Implantable defibrillator
SRT | A-11FCD | Left ventricular assist device
SRT | A-28040 | Insulin pump

## CID 3778 Stages

Type:

  Extensible

Version:

  20081027

Table CID 3778. Stages

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
SRT | R-41177 | Stage 0
SRT | R-41DA8 | Stage 1
SRT | R-41DAC | Stage 2
SRT | R-41DB0 | Stage 3
SRT | R-41DB4 | Stage 4
SRT | R-4117B | Stage 5

## CID 3802 Plaque Structures

Type:

  Extensible

Version:

  20051103

Table CID 3802. Plaque Structures

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-40448 | fibrous
SRT | M-50080 | fatty degeneration
SRT | M-55420 | pathologic calcification
SRT | M-72000 | hyperplasia
SRT | G-A265 | non-calcified
SRT | G-A660 | mixed

## CID 3804 Stenosis Measurement Methods

Type:

  Extensible

Version:

  20051103

Table CID 3804. Stenosis Measurement Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122655 | NASCET
DCM | 122656 | ECST
DCM | 122650 | Area Based Method
DCM | 122651 | Diameter Based Method

## CID 3805 Stenosis Types

Type:

  Extensible

Version:

  20051103

Table CID 3805. Stenosis Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D3-81100 | arteriosclerotic vascular disease
SRT | M-01460 | compression
SRT | R-40448 | fibrous
SRT | D3-80505 | Raynaud's disease
SRT | M-300F2 | entrapment
SRT | D3-80650 | vasculitis
SRT | R-423C3 | thrombosis
SRT | M-35300 | embolism
SRT | D3-80033 | cystic adventitial disease

## CID 3806 Stenosis Shape

Type:

  Extensible

Version:

  20051103

Table CID 3806. Stenosis Shape

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-4047B | concentric
SRT | R-40416 | eccentric

## CID 3807 Volume Measurement Methods

Type:

  Extensible

Version:

  20051103

Table CID 3807. Volume Measurement Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122650 | Area Based Method
DCM | 122651 | Diameter Based Method
DCM | 122652 | Volume Based Method

## CID 3808 Aneurysm Types

Type:

  Extensible

Version:

  20051103

Table CID 3808. Aneurysm Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | M-32270 | dissecting aneurysm
SRT | D3-80017 | inflammatory aneurysm
SRT | M-32201 | ruptured aneurysm
SRT | M-24614 | berry aneurysm
SRT | M-32240 | mixed aneurysm
SRT | M-32410 | racemose aneurysm
SRT | D3-80002 | cirsoid aneurysm
SRT | M-32320 | mycotic aneurysm
SRT | M-32206 | compound aneurysm
SRT | M-32310 | miliary aneurysm
SRT | M-32340 | saccular aneurysm
SRT | M-32221 | varicose aneurysm
SRT | M-32350 | fusiform aneurysm
SRT | M-32210 | traumatic aneurysm
SRT | M-32202 | thrombosed aneurysm
SRT | M-32203 | expanding aneurysm
SRT | M-32204 | calcified aneurysm
SRT | M-32208 | multiple aneurysm
SRT | M-32360 | cylindroid aneurysm
SRT | M-32260 | serpentine aneurysm

## CID 3809 Associated Conditions

Type:

  Extensible

Version:

  20051103

Table CID 3809. Associated Conditions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D6-90600 | Marfan's Syndrome
SRT | M-10000 | Traumatic Abnormality

## CID 3810 Vascular Morphology

Type:

  Extensible

Version:

  20051103

Table CID 3810. Vascular Morphology

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | M-01470 | plaque
SRT | M-34200 | stenosis
SRT | M-32200 | aneurysm
SRT | D3-81310 | arterial dissection
SRT | A-25500 | stent
SRT | M-34000 | occlusion
SRT | M-39390 | arteriovenous fistula
SRT | M-91200 | angioma
SRT | M-32000 | dilatation
SRT | R-FAB5E | vascular coiling
SRT | M-31790 | tortuosity
SRT | M-32700 | diverticulum
SRT | M-520F8 | vascular sclerosis
SRT | D-80515 | thrombosis
SRT | M-32390 | pseudoaneurysm
SRT | M-35300 | embolism
SRT | M-74880 | fibromuscular dysplasia

## CID 3813 Stent Findings

Type:

  Extensible

Version:

  20051103

Table CID 3813. Stent Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | M-75300 | hypoplasia
SRT | M-34200 | stenosis
DCM | 122680 | endoleak
SRT | DD-661D2 | migration of implant or internal device
DCM | 122684 | stent disintegration
DCM | 122683 | stent fracture

## CID 3814 Stent Composition

Type:

  Extensible

Version:

  20051103

Table CID 3814. Stent Composition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-25502 | metal stent
SRT | A-25501 | plastic stent

## CID 3815 Source of Vascular Finding

Type:

  Extensible

Version:

  20051103

Table CID 3815. Source of Vascular Finding

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D3-80515 | thrombosis
SRT | M-35300 | embolism
SRT | M-72000 | hyperplasia
SRT | D3-80650 | vasculitis
SRT | M-8FFFF | tumor
SRT | DD-00001 | trauma
SRT | G-B102 | surgical
SRT | R-422A4 | after procedure

## CID 3817 Vascular Sclerosis Types

Type:

  Extensible

Version:

  20051103

Table CID 3817. Vascular Sclerosis Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | M-52450 | adventitial degeneration
SRT | M-52210 | arteriosclerosis with fibrinoid necrosis
SRT | M-52200 | arteriolosclerosis
SRT | M-52000 | arteriosclerosis
SRT | M-52100 | atheroma
SRT | M-52120 | atherosclerotic fibrous plaque
SRT | M-52101 | calcified atheromatous plaque
SRT | M-52102 | complicated atheromatous plaque
SRT | M-52470 | cystic medical necrosis
SRT | M-52240 | elastic vascular sclerosis
SRT | M-52130 | fatty streaks
SRT | M-52300 | fibroelastosis
SRT | M-52302 | diffuse fibroelastosis
SRT | M-52301 | focal fibroelastosis
SRT | M-52500 | phlebosclerosis
SRT | M-52103 | ulcerated atheromatous plaque
SRT | M-52400 | vascular wall degeneration

## CID 3820 Non-invasive Vascular Procedures

Type:

  Extensible

Version:

  20051103

Table CID 3820. Non-Invasive Vascular Procedures

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P5-0903A | vascular MRI
SRT | P5-09011 | cardiac MRI
SRT | P5-0807F | cardiovascular CT
SRT | P5-0802B | CT of abdominal aorta
SRT | P5-00A0D | trunk angiography
SRT | P5-009BF | peripheral angiography

## CID 3821 Papillary Muscle Included/excluded

Type:

  Extensible

Version:

  20051103

Table CID 3821. Papillary Muscle Included/Excluded

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122620 | Papillary Muscle Excluded
DCM | 122621 | Papillary Muscle Included

## CID 3823 Respiratory Status

Type:

  Extensible

Version:

  20051103

Table CID 3823. Respiratory Status

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-20010 | inspiration
SRT | F-20020 | expiration
SRT | F-20030 | autonomous breathing
SRT | R-40928 | Valsalva maneuver
DCM | 122612 | central breathing position
SRT | F-201BD | shallow breathing

## CID 3826 Heart Rhythm

Type:

  Extensible

Version:

  20051103

Table CID 3826. Heart Rhythm

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-33300 | normal sinus rhythm
SRT | D3-31500 | atrial arrhythmia
SRT | D3-31715 | ventricular arrhythmia

## CID 3827 Vessel Segments

Type:

  Extensible

Version:

  20051103

Table CID 3827. Vessel Segments

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12105 “Intracranial Cerebral Vessels”
Include CID 12106 “Intracranial Cerebral Vessels (unilateral)”
Include CID 12104 “Extracranial Arteries”
Include CID 12109 “Lower Extremity Arteries”
Include CID 12110 “Lower Extremity Veins”
Include CID 12107 “Upper Extremity Arteries”
Include CID 12108 “Upper Extremity Veins”
Include CID 12115 “Renal Vessels”
Include CID 12111 “Abdominal Arteries (lateral)”
Include CID 12112 “Abdominal Arteries (unilateral)”
Include CID 12113 “Abdominal Veins (lateral)”
Include CID 12114 “Abdominal Veins (unilateral)”
Include CID 3015 “Coronary Arteries”
Include CID 3839 “Coronary Veins”
Include CID 3840 “Pulmonary Veins”

## CID 3829 Pulmonary Arteries

Type:

  Extensible

Version:

  20110818

Table CID 3829. Pulmonary Arteries

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-44100 | Trunk of pulmonary artery
SRT | T-44010 | Suprapulmonic valve area
SRT | T-35250 | pulmonary valve sinuses
SRT | T-44400 | Left pulmonary artery
SRT | T-44200 | Right pulmonary artery

### Note

A previous version of this context group used terms with the SNOMED concept "entire" (see PS3.16-2011). The use of "structure" concepts rather than "entire" is described in Section 8.1.1.

## CID 3831 Stenosis Length

Type:

  Extensible

Version:

  20051103

Table CID 3831. Stenosis Length

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-404AC | long
SRT | R-4235F | short

## CID 3832 Stenosis Grade

Type:

  Extensible

Version:

  20051103

Table CID 3832. Stenosis Grade

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A003 | severe
SRT | G-A002 | moderate
SRT | R-404FA | mild

## CID 3833 Cardiac Ejection Fraction

Type:

  Extensible

Version:

  20051103

Table CID 3833. Cardiac Ejection Fraction

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 8810-4 | Left ventricular ejection fraction by CT
LN | 8817-9 | Right ventricular ejection fraction by CT
LN | 8811-2 | Left ventricular ejection fraction by MR
LN | 8818-7 | Right ventricular ejection fraction by MR

## CID 3835 Cardiac Volume Measurements

Type:

  Extensible

Version:

  20051103

Table CID 3835. Cardiac Volume Measurements

Coding Scheme Designator | Code Value | Code Meaning | Equivalent LOINC Code Value
--- | --- | --- | ---
Include CID 3468 “ED Volume”
Include CID 3469 “ES Volume”
SRT | F-32120 | Stroke Volume | 20562-5

## CID 3836 Time-based Perfusion Measurements

Type:

  Extensible

Version:

  20051103

Table CID 3836. Time-Based Perfusion Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122631 | Signal Earliest Peak Time
DCM | 122633 | Signal Increase Start Time
DCM | 122634 | Signal Time to Peak
DCM | 122638 | Signal Baseline Start
DCM | 122639 | Signal Baseline End

## CID 3837 Fiducial Feature

Type:

  Extensible

Version:

  20051103

Table CID 3837. Fiducial Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-3215A | Ostium
SRT | T-46600 | Renal Artery
SRT | T-42580 | Aortic Bifurcation
SRT | R-10258 | Common Iliac Bifurcation

## CID 3838 Diameter Derivation

Type:

  Extensible

Version:

  20051103

Table CID 3838. Diameter Derivation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3488 “Min/Max/Mean”
SRT | G-A117 | Transverse
DCM | 122675 | Anterior-Posterior

## CID 3839 Coronary Veins

Type:

  Extensible

Version:

  20110818

Table CID 3839. Coronary Veins

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-48340 | Azygos Vein
SRT | T-48410 | Coronary Sinus
SRT | T-48420 | Great Cardiac Vein
SRT | T-48435 | Small Cardiac Vein
SRT | T-48403 | Anterior Cardiac Vein
SRT | T-48406 | Atrial Vein
SRT | T-48407 | Atrioventricular Vein
SRT | T-48430 | Middle Cardiac Vein
SRT | T-48404 | Ventricular Vein
SRT | T-48405 | Smallest Cardiac Vein

### Note

A previous version of this context group used terms with the SNOMED concept "entire" (see PS3.16-2011). The use of "structure" concepts rather than "entire" is described in Section 8.1.1.

## CID 3840 Pulmonary Veins

Type:

  Extensible

Version:

  20110818

Table CID 3840. Pulmonary Veins

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-4858F | Pulmonary Vein
SRT | T-48502 | Left Pulmonary Vein
SRT | T-48540 | Inferior Left Pulmonary Vein
SRT | T-48530 | Superior Left Pulmonary Vein
SRT | T-48501 | Right Pulmonary Vein
SRT | T-48520 | Inferior Right Pulmonary Vein
SRT | T-48510 | Superior Right Pulmonary Vein

### Note

A previous version of this context group used terms with the SNOMED concept "entire" (see PS3.16-2011). The use of "structure" concepts rather than "entire" is described in Section 8.1.1.

## CID 3843 Myocardial Subsegment

Type:

  Extensible

Version:

  20051103

Table CID 3843. Myocardial Subsegment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-427E6 | endocardial
SRT | R-40940 | epicardial

## CID 3850 Intravascular OCT Flush Agent

Type:

  Extensible

Version:

  20110609

Table CID 3850. Intravascular OCT Flush Agent

Coding Scheme Designator | Code Value | Code Meaning | Trade Name (Informative) (From http://www.accessdata.fda.gov/scripts/cder/drugsatfda/index.cfm)
--- | --- | --- | ---
SRT | C-A7220 | Dextran |
SRT | C-70841 | Saline |
SRT | C-70434 | Lactated Ringer's |
Include CID 12 “Radiographic Contrast Agent”

## CID 4005 Partial View Section for Mammography

Type:

  Non-Extensible

Version:

  20050110

Table CID 4005. Partial View Section for Mammography

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-404CC | Anterior
SRT | R-404CE | Posterior
SRT | R-42191 | Superior
SRT | R-4094A | Inferior
SRT | R-404D5 | Medial
SRT | G-A104 | Lateral
SRT | G-A110 | Central

## CID 4009 DX Anatomy Imaged

Type:

  Extensible

Version:

  20040114

Table CID 4009. DX Anatomy Imaged

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 4031 “Common Anatomic Regions”

## CID 4010 DX View

Type:

  Extensible

Version:

  20130806

Table CID 4010. DX View

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-10202 | frontal
SRT | R-10204 | frontal oblique
SRT | R-10206 | antero-posterior
SRT | R-10208 | antero-posterior oblique
SRT | R-10210 | right posterior oblique
SRT | R-10212 | left posterior oblique
SRT | R-10214 | postero-anterior
SRT | R-10216 | postero-anterior oblique
SRT | R-40985 | right anterior oblique
SRT | R-10220 | left anterior oblique
SRT | G-A145 | sagittal
SRT | R-10224 | medial-lateral
SRT | R-40783 | lateral oblique
SRT | R-10228 | lateral-medial
SRT | R-40782 | medial oblique
SRT | R-10232 | right lateral
SRT | R-10234 | right oblique
SRT | R-10236 | left lateral
SRT | R-10238 | left oblique
SRT | R-10241 | axial
SRT | R-10242 | cranio-caudal
SRT | R-10244 | caudo-cranial
SRT | R-10246 | oblique axial
SRT | R-10248 | oblique cranio-caudal
SRT | R-10250 | oblique caudo-cranial
SRT | R-10252 | frontal-oblique axial
SRT | R-10254 | sagittal-oblique axial
SRT | R-102C1 | oblique
SRT | R-102CD | lateral
SRT | R-102C2 | tangential
SRT | R-10256 | submentovertical
SRT | R-10257 | verticosubmental
SRT | R-102C3 | plantodorsal
SRT | R-102C4 | dorsoplantar
SRT | R-102C5 | parietoacanthial
SRT | R-102C6 | acanthioparietal
SRT | R-102C7 | orbitoparietal
SRT | R-102C8 | parieto-orbital
SRT | R-10230 | latero-medial oblique
SRT | R-10226 | medio-lateral oblique
SRT | G-8300 | tissue specimen

### Note

1. In a prior version of this Context Group, Lateral Oblique was assigned the code R-10226, and Medial Oblique was assigned the code R-10230, as synonymous with Medio-Lateral Oblique and Latero-Medial Oblique, respectively. SNOMED currently distinguishes between LO and MLO, and between MO and LMO, although in most radiography contexts there is no practical distinction. Receiving applications should be aware that they may receive SOP Instances with the prior code assignments.

2. In a prior version of this Context Group, "right anterior oblique" was assigned the code R-10218, which in SNOMED is actually "Indirect iris transillumination"; this code has been replaced with the correct code R-40985.

3. In a prior version of this Context Group, a concept of "sagittal" was present with a code of R-10222, which in SNOMED is actually "Trypan blue"; this code has been replaced with the general SNOMED qualifier concept G-A145.

## CID 4011 DX View Modifier

Type:

  Extensible

Version:

  20070524

Table CID 4011. DX View Modifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-10244 | cephalad
SRT | R-10242 | caudad
SRT | R-40885 | transthoracic
SRT | R-4087B | transforamenal
SRT | G-D00B | transoral
SRT | R-40554 | transorbital
DCM | 111069 | Crosstable
SRT | R-421A4 | Mouth closed

### Note

In a prior version of this Context Group, the codes R-102C9, R-102CA, R-102CB, R-102CC, and R-102CE were specified for various concepts. Those codes are not actually in SNOMED, and their use in this context is deprecated. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 4012 Projection Eponymous Name

Type:

  Extensible

Version:

  20070524

Table CID 4012. Projection Eponymous Name

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-10261 | Albers-Schonberg
SRT | R-10262 | Alexander
SRT | R-40A88 | Apple
SRT | R-10263 | Arcelin
SRT | R-10264 | Beclere
SRT | R-10265 | Bertel
SRT | R-10266 | Blackett-Healy
SRT | R-40809 | Brewerton projection
SRT | R-10267 | Broden
SRT | R-40A89 | Burman
SRT | R-10268 | Cahoon
SRT | R-10269 | Caldwell
SRT | R-1026A | Camp-Coventry
SRT | R-1026B | Causton
SRT | R-1026C | Chamberlain
SRT | R-1026D | Chassard-Lapine
SRT | R-1026E | Chausse
SRT | R-1026F | Cleaves
SRT | R-10270 | Clements
SRT | R-10271 | Clements-Nakayama
SRT | R-40A8A | Colcher-Sussman
SRT | R-40A8B | Danelius-Miller
SRT | R-10272 | Dunlap
SRT | R-40A8F | Eraso Modification
SRT | R-10273 | Ferguson
SRT | R-40A8C | Fisk
SRT | R-10274 | Fleischner
SRT | R-40A8D | Folio
SRT | R-10275 | Friedman
SRT | R-10276 | Fuchs
SRT | R-40A8E | Garth
SRT | R-10277 | Gaynor-Hart
SRT | R-10278 | Grandy
SRT | R-10279 | Grashey
SRT | R-1027A | Haas
SRT | R-4080A | Harris Beath axial projection
SRT | R-1027B | Henschen
SRT | R-1027C | Hickey
SRT | R-40A90 | Hirtz Modification
SRT | R-1027D | Holly
SRT | R-1027E | Holmblad
SRT | R-1027F | Hough
SRT | R-10280 | Hsieh
SRT | R-10281 | Hughston
SRT | R-10282 | Isherwood
SRT | R-10283 | Judd
SRT | R-4080D | Judet projection
SRT | R-10284 | Kandel
SRT | R-10285 | Kasabach
SRT | R-10286 | Kemp Harper
SRT | R-40A91 | Kite
SRT | R-10287 | Kovacs
SRT | R-10288 | Kuchendorf
SRT | R-10289 | Kurzbauer
SRT | R-1028A | Laquerriere-Pierquin
SRT | R-1028B | Lauenstein
SRT | R-1028C | Law
SRT | R-1028D | Lawrence
SRT | R-1028E | Leonard-George
SRT | R-1028F | Lewis
SRT | R-10290 | Lilienfeld
SRT | R-10291 | Lindblom
SRT | R-10292 | Lorenz
SRT | R-10293 | Low-Beer
SRT | R-10294 | Lysholm
SRT | R-10295 | May
SRT | R-10296 | Mayer
SRT | R-10297 | Merchant
SRT | R-10298 | Miller
SRT | R-40A92 | Moore
SRT | R-4080E | Mortice projection
SRT | R-40A93 | Neer
SRT | R-10299 | Nolke
SRT | R-1029A | Norgaard
SRT | R-1029B | Ottonello
SRT | R-1029C | Pawlow
SRT | R-1029D | Pearson
SRT | R-1029E | Penner
SRT | R-1029F | Pirie
SRT | R-40A94 | Rafert
SRT | R-40A95 | Rafert-Long
SRT | R-102A0 | Rhese
SRT | R-40A96 | Robert
SRT | R-40A97 | Rosenberg
SRT | R-102A1 | Schuller
SRT | R-102A2 | Settegast
SRT | R-102A3 | Staunig
SRT | R-102A4 | Stecher
SRT | R-102A5 | Stenvers
SRT | R-40A98 | Stryker
SRT | R-102A6 | Swanson
SRT | R-102A7 | Tarrant
SRT | R-102A8 | Taylor
SRT | R-102A9 | Teufel
SRT | R-102AA | Titterington
SRT | R-102AB | Towne
SRT | R-102AC | Twining
SRT | R-102AD | Valdini
SRT | R-40816 | Van Rosen projection
SRT | R-102AE | Waters
SRT | R-102AF | West Point
SRT | R-102B0 | Wigby-Taylor
SRT | R-40A99 | Wolf
SRT | R-102B1 | Zanelli

## CID 4013 Anatomic Region for Mammography

Type:

  Non-Extensible

Version:

  20020904

Table CID 4013. Anatomic Region for Mammography

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-04000 | Breast

## CID 4014 View for Mammography

Type:

  Non-Extensible

Version:

  20090717

Table CID 4014. View for Mammography

Coding Scheme Designator | Code Value | Code Meaning | ACR MQCM 1999 Equivalent
--- | --- | --- | ---
SRT | R-10224 | medio-lateral | ML
SRT | R-10226 | medio-lateral oblique | MLO
SRT | R-10228 | latero-medial | LM
SRT | R-10230 | latero-medial oblique | LMO
SRT | R-10242 | cranio-caudal | CC
SRT | R-10244 | caudo-cranial (from below) | FB
SRT | R-102D0 | superolateral to inferomedial oblique | SIO
SRT | R-40AAA | inferomedial to superolateral oblique | ISO
SRT | R-1024A | cranio-caudal exaggerated laterally | XCCL
SRT | R-1024B | cranio-caudal exaggerated medially | XCCM
SRT | G-8310 | tissue specimen from breast |

### Note

1. In a prior version of this Context Group, Cranio-Caudal Exaggerated Laterally was assigned the code Y-X1770, and Cranio-Caudal Exaggerated Medially was assigned the code Y-X1771. Those codes are deprecated, as they are not valid SNOMED codes. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated codes, receiving applications should be aware of this change; see Annex J.

2. While SRT is the preferred Coding Scheme Designator for SNOMED, regulatory approval of mammography systems makes changes to such systems problematic. Implementers should be aware that many systems will continue to use the deprecated designator SNM3 for certain terms in this context group. It is recommended that implementations receiving Mammography Image or CAD SOP Instances support both SNM3 and SRT as equivalent Coding Scheme Designators for Attributes or Content Items that use this Context Group.

3. In a prior version of this Context Group, (R-102CF, SRT, "cranio-caudal exaggerated") was included. This is not a clinically applied view. Use of this term is deprecated, but receiving applications should be aware of its prior existence.

## CID 4015 View Modifier for Mammography

Type:

  Non-Extensible

Version:

  20090717

Table CID 4015. View Modifier for Mammography

Coding Scheme Designator | Code Value | Code Meaning | Applies only when View ACR MQCM 1999 Equivalent (CID 4014 “View for Mammography”) is: | ACR MQCM 1999 Equivalent
--- | --- | --- | --- | ---
SRT | R-102D2 | Cleavage | CC or FB | CV
SRT | R-102D1 | Axillary Tail | MLO | AT
SRT | R-102D3 | Rolled Lateral | any | ...RL
SRT | R-102D4 | Rolled Medial | any | ...RM
SRT | R-102CA | Rolled Inferior | any | ...RI
SRT | R-102C9 | Rolled Superior | any | ...RS
SRT | R-102D5 | Implant Displaced | any | ...ID
SRT | R-102D6 | Magnification | any | M...
SRT | R-102D7 | Spot Compression | any | S...
SRT | R-102C2 | Tangential | any | TAN
SRT | R-40AB3 | Nipple in profile | any | ...NP
SRT | P2-00161 | Anterior compression | any | ...AC
SRT | R-40ABE | Infra-mammary fold | any | ...IMF
SRT | R-40AB2 | Axillary tissue | any | ...AX

### Note

1. Some applications and View Modifier ACR MQCM 1999 equivalents have been extended by DICOM to incorporate additional known clinical use cases. The View Modifier ACR MQCM 1999 equivalent indicates its use as a prefix (shown by trailing "…") or suffix (shown by preceding "…") to the View ACR MQCM 1999 equivalent, or replacement for the View ACR MQCM 1999 equivalent.

2. While SRT is the preferred Coding Scheme Designator for SNOMED, regulatory approval of mammography systems makes changes to such systems problematic. Implementers should be aware that many systems will continue to use the deprecated designator SNM3 for certain terms in this context group. It is recommended that implementations receiving Mammography Image or CAD SOP Instances support both SNM3 and SRT as equivalent Coding Scheme Designators for Attributes or Content Items that use this Context Group.

## CID 4016 Anatomic Region for Intra-oral Radiography

Type:

  Non-Extensible

Version:

  20020904

Table CID 4016. Anatomic Region for Intra-Oral Radiography

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-D1213 | Jaw region
SRT | T-11170 | Maxilla
SRT | T-11180 | Mandible
SRT | T-54000 | Teeth, gums and supporting structures

### Note

In a prior version of this table, the code T-D1217 was specified for the concept "Maxilla and mandible". The use of this code conflicts with its assignment to another concept in SNOMED, and its use in this context is deprecated. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 4017 Anatomic Region Modifier for Intra-oral Radiography

Type:

  Non-Extensible

Version:

  20020904

Table CID 4017. Anatomic Region Modifier for Intra-Oral Radiography

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-51005 | Anterior 1
SRT | T-51006 | Anterior 2
SRT | T-51007 | Anterior 3
SRT | T-51008 | Premolar 1
SRT | T-51009 | Premolar 2
SRT | T-5100A | Molar 1
SRT | T-5100B | Molar 2
SRT | T-5100C | Molar 3
SRT | T-5100D | Occlusal

## CID 4018 Primary Anatomic Structure for Intra-oral Radiography (permanent Dentition - Designation of Teeth)

Type:

  Non-Extensible

Version:

  20020904

Table CID 4018. Primary Anatomic Structure for Intra-Oral Radiography(Permanent Dentition - Designation of Teeth)

Coding Scheme Designator | Code Value | Code Meaning | ISO 3950 Designation of Quadrant | ISO 3950 Designation of Tooth
--- | --- | --- | --- | ---
SRT | T-54210 | Maxillary right third molar tooth | 1 | 8
SRT | T-54220 | Maxillary right second molar tooth | 1 | 7
SRT | T-54230 | Maxillary right first molar tooth | 1 | 6
SRT | T-54240 | Maxillary right second premolar tooth | 1 | 5
SRT | T-54250 | Maxillary right first premolar tooth | 1 | 4
SRT | T-54260 | Maxillary right canine tooth | 1 | 3
SRT | T-54270 | Maxillary right lateral incisor tooth | 1 | 2
SRT | T-54280 | Maxillary right central incisor tooth | 1 | 1
SRT | T-54290 | Maxillary left central incisor tooth | 2 | 1
SRT | T-54300 | Maxillary left lateral incisor tooth | 2 | 2
SRT | T-54310 | Maxillary left canine tooth | 2 | 3
SRT | T-54320 | Maxillary left first premolar tooth | 2 | 4
SRT | T-54330 | Maxillary left second premolar tooth | 2 | 5
SRT | T-54340 | Maxillary left first molar tooth | 2 | 6
SRT | T-54350 | Maxillary left second molar tooth | 2 | 7
SRT | T-54360 | Maxillary left third molar tooth | 2 | 8
SRT | T-54370 | Mandibular left third molar tooth | 3 | 8
SRT | T-54380 | Mandibular left second molar tooth | 3 | 7
SRT | T-54390 | Mandibular left first molar tooth | 3 | 6
SRT | T-54400 | Mandibular left second premolar tooth | 3 | 5
SRT | T-54410 | Mandibular left first premolar tooth | 3 | 4
SRT | T-54420 | Mandibular left canine tooth | 3 | 3
SRT | T-54430 | Mandibular left lateral tooth | 3 | 2
SRT | T-54440 | Mandibular left central incisor tooth | 3 | 1
SRT | T-54450 | Mandibular right central incisor tooth | 4 | 1
SRT | T-54460 | Mandibular right lateral incisor tooth | 4 | 2
SRT | T-54470 | Mandibular right canine tooth | 4 | 3
SRT | T-54480 | Mandibular right first premolar tooth | 4 | 4
SRT | T-54490 | Mandibular right second premolar tooth | 4 | 5
SRT | T-54500 | Mandibular right first molar tooth | 4 | 6
SRT | T-54510 | Mandibular right second molar tooth | 4 | 7
SRT | T-54520 | Mandibular right third molar tooth | 4 | 8

## CID 4019 Primary Anatomic Structure for Intra-oral Radiography (Deciduous Dentition - Designation of Teeth)

Type:

  Non-Extensible

Version:

  20020904

Table CID 4019. Primary Anatomic Structure for Intra-Oral Radiography (Deciduous Dentition - Designation of Teeth)

Coding Scheme Designator | Code Value | Code Meaning | ISO 3950 Designation of Quadrant | ISO 3950 Designation of Tooth
--- | --- | --- | --- | ---
SRT | T-54610 | Deciduous maxillary right central incisor tooth | 5 | 1
SRT | T-54620 | Deciduous maxillary right lateral incisor tooth | 5 | 2
SRT | T-54630 | Deciduous maxillary right canine tooth | 5 | 3
SRT | T-54640 | Deciduous maxillary right first molar tooth | 5 | 4
SRT | T-54650 | Deciduous maxillary right second molar tooth | 5 | 5
SRT | T-54660 | Deciduous maxillary left central incisor tooth | 6 | 1
SRT | T-54670 | Deciduous maxillary left lateral incisor tooth | 6 | 2
SRT | T-54680 | Deciduous maxillary left canine tooth | 6 | 3
SRT | T-54690 | Deciduous maxillary left first molar tooth | 6 | 4
SRT | T-54700 | Deciduous maxillary left second molar tooth | 6 | 5
SRT | T-54760 | Deciduous mandibular left central incisor tooth | 7 | 1
SRT | T-54770 | Deciduous mandibular left lateral incisor tooth | 7 | 2
SRT | T-54780 | Deciduous mandibular left canine tooth | 7 | 3
SRT | T-54790 | Deciduous mandibular left first molar tooth | 7 | 4
SRT | T-54800 | Deciduous mandibular left second molar tooth | 7 | 5
SRT | T-54710 | Deciduous mandibular right central incisor tooth | 8 | 1
SRT | T-54720 | Deciduous mandibular right lateral incisor tooth | 8 | 2
SRT | T-54730 | Deciduous mandibular right canine tooth | 8 | 3
SRT | T-54740 | Deciduous mandibular right first molar tooth | 8 | 4
SRT | T-54750 | Deciduous mandibular right second molar tooth | 8 | 5

## CID 4020 PET Radionuclide

Type:

  Extensible

Version:

  20141110

Table CID 4020. PET Radionuclide

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-105A1 | ^11^Carbon
SRT | C-107A1 | ^13^Nitrogen
SRT | C-1018C | ^14^Oxygen
SRT | C-B1038 | ^15^Oxygen
SRT | C-111A1 | ^18^Fluorine
SRT | C-155A1 | ^22^Sodium
SRT | C-135A4 | ^38^Potassium
DCM | 126605 | ^43^Scandium
DCM | 126600 | ^44^Scandium
SRT | C-166A2 | ^45^Titanium
DCM | 126601 | ^51^Manganese
SRT | C-130A1 | ^52^Iron
SRT | C-149A1 | ^52^Manganese
SRT | C-127A4 | ^60^Copper
SRT | C-127A1 | ^61^Copper
SRT | C-127A5 | ^62^Copper
SRT | C-141A1 | ^62^Zinc
SRT | C-127A2 | ^64^Copper
SRT | C-131A1 | ^66^Gallium
SRT | C-131A3 | ^68^Gallium
SRT | C-128A2 | ^68^Germanium
DCM | 126602 | ^70^Arsenic
SRT | C-115A2 | ^72^Arsenic
SRT | C-116A2 | ^73^Selenium
SRT | C-113A1 | ^75^Bromine
SRT | C-113A2 | ^76^Bromine
SRT | C-113A3 | ^77^Bromine
SRT | C-159A2 | ^82^Rubidium
SRT | C-162A3 | ^86^Yttrium
SRT | C-168A4 | ^89^Zirconium
DCM | 126603 | ^90^Niobium
SRT | C-162A7 | ^90^Yttrium
SRT | C-163AA | ^94m^Technetium
SRT | C-114A5 | ^124^Iodine
DCM | 126606 | ^152^Terbium

## CID 4021 PET Radiopharmaceutical

Type:

  Extensible

Version:

  20141110

Table CID 4021. PET Radiopharmaceutical

Coding Scheme Designator | Code Value | Code Meaning | Other Names
--- | --- | --- | ---
DCM | 126713 | 2FA F^18^ | FA-85380
SRT | C-B1043 | Acetate C^11^ |
SRT | C-B103C | Ammonia N^13^ |
DCM | 126700 | ATSM Cu^60^ |
DCM | 126701 | ATSM Cu^61^ |
DCM | 126702 | ATSM Cu^62^ |
SRT | C-B07DB | ATSM Cu^64^ |
DCM | 126516 | Bevacizumab ^89^Zr | Avastin™ ^89^Zr
SRT | C-B07DC | Butanol O^15^ |
SRT | C-B103B | Carbon dioxide O^15^ |
SRT | C-B1045 | Carbon monoxide C^11^ |
SRT | C-B103A | Carbon monoxide O^15^ |
SRT | C-B103F | Carfentanil C^11^ |
DCM | 126513 | Cetuximab ^89^Zr | Erbitux™ ^89^Zr
DCM | 126703 | Choline C^11^ |
DCM | 126517 | cG250-F(ab')(2) ^89^Zr |
DCM | 126715 | CLR1404 I^124^ |
DCM | 126716 | CLR1404 I^131^ |
DCM | 126515 | cU36 ^89^Zr |
DCM | 126520 | Df-CD45 ^89^Zr |
DCM | 126519 | E4G10 ^89^Zr |
UMLS | C2713594 | Edotreotide Ga^68^ | DOTATOC, SMT487
SRT | C-B07DD | EDTA Ga^68^ |
DCM | 126704 | Fallypride C^11^ |
DCM | 126705 | Fallypride F^18^ |
DCM | 126706 | FLB 457 C^11^ |
DCM | 126503 | Flubatine F^18^ | NCFHEB
DCM | 126501 | Florbetaben F^18^ | NeuroCeq™
SRT | C-E0269 | Florbetapir F^18^ | AV-45, Amyvid™
DCM | 126712 | Flubatine F^18^ | NCFHEB
SRT | C-E0265 | Fluciclatide F^18^ |
SRT | C-E026A | Fluciclovine F^18^ |
SRT | C-B07DE | Flumazenil C^11^ |
SRT | C-B07DF | Flumazenil F^18^ |
UMLS | C1831937 | Fluoroestradiol (FES) F^18^ |
UMLS | C1541539 | Fluoroetanidazole F^18^ | EF5
SRT | C-B07E0 | Fluorethyltyrosin F^18^ |
SRT | C-E0273 | Fluorocholine F^18^ |
SRT | C-B1031 | Fluorodeoxyglucose F^18^ |
SRT | C-B07E1 | Fluoromisonidazole F^18^ | FMISO
SRT | C-B07E2 | Fluoromethane F^18^ |
UMLS | C2934038 | Fluoropropyl-dihydrotetrabenazine (DTBZ) F^18^ | AV-133
SRT | C-B07E3 | Fluorouracil F^18^ |
SRT | C-B07E4 | Fluorobenzothiazole F^18^ |
SRT | C-B1034 | Fluoro-L-dopa F^18^ |
DCM | 126707 | Fluorotriopride F^18^ |
SRT | C-E0267 | Flutemetamol F^18^ | Vizamyl™
SRT | C-B1046 | Germanium Ge^68^ |
SRT | C-B103D | Glutamate N^13^ |
DCM | 126709 | Glutamine C^11^ |
DCM | 126710 | Glutamine C^14^ |
DCM | 126711 | Glutamine F^18^ |
UMLS | C2981788 | ISO-1 F^18^ |
DCM | 126514 | J591 ^89^Zr |
SRT | C-B07E5 | Mespiperone C^11^ |
SRT | C-B103E | Methionine C^11^ |
SRT | C-B07E6 | Monoclonal antibody I^124^ |
DCM | 126510 | Monoclonal Antibody (mAb) ^64^Cu |
DCM | 126511 | Monoclonal Antibody (mAb) ^89^Zr |
DCM | 126714 | Nifene F^18^ |
SRT | C-B1038 | Oxygen O^15^ |
SRT | C-B1039 | Oxygen-water O^15^ |
SRT | C-B1044 | Palmitate C^11^ |
DCM | 126500 | Pittsburgh compound B C^11^ | PIB
SRT | C-B07E7 | PTSM Cu^62^ |
DCM | 126518 | R1507 ^89^Zr |
SRT | C-B1042 | Raclopride C^11^ |
SRT | C-B1037 | Rubidium chloride Rb^82^ |
SRT | C-B1032 | Sodium fluoride F^18^ |
SRT | C-B07E8 | Sodium iodide I^124^ |
SRT | C-B1047 | Sodium Na^22^ |
SRT | C-B1033 | Spiperone F^18^ |
DCM | 126502 | T807 F^18^ | AV-1451
SRT | C-B1036 | Thymidine (FLT) F^18^ |
DCM | 126512 | Trastuzumab ^89^Zr | Herceptin™ ^89^Zr
UMLS | C1742831 | tyrosine-3-octreotate Ga^68^ | DOTATATE

## CID 4028 Craniofacial Anatomic Regions

Type:

  Extensible

Version:

  20130617

Table CID 4028. Craniofacial Anatomic Regions

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-D006D | Bone structure of head and/or neck | 312779009 | C0730130
SRT | T-11501 | Cervical spine | 122494005 | C0728985
SRT | T-11156 | Ethmoid bone | 52374004 | C0015027
SRT | T-AB100 | External ear | 28347008 | C0013453
SRT | T-AA770 | Eyeball | 79652003 | C0229242
SRT | T-D0801 | Eye region | 371398005 | C0015392
SRT | T-11196 | Facial bones | 91397008 | C0015455
SRT | T-11110 | Frontal bone | 74872008 | C0016732
SRT | T-D1100 | Head | 69536005 | C0018670
SRT | T-D1000 | Head and Neck | 774007 | C0460004
SRT | T-11190 | Hyoid bone | 21387005 | C0020417
SRT | T-AB700 | Inner ear | 22945000 | C0022889
SRT | T-AB959 | Internal Auditory Canal | 361078006 | C1283773
SRT | T-D1213 | Jaw region | 661005 | C0022359
SRT | T-1115A | Lacrimal bone | 6229007 | C0222733
SRT | T-24100 | Larynx | 4596009 | C0023078
SRT | T-52000 | Lip | 48477009 | C0023759
SRT | T-11180 | Mandible | 91609006 | C0024687
SRT | T-11133 | Mastoid bone | 59066005 | C0446908
SRT | T-11170 | Maxilla | 70925003 | C0024947
SRT | T-AB300 | Middle ear | 25342003 | C0013455
SRT | T-13100 | Muscle of head | 22688005 | C0224097
SRT | T-13300 | Muscle of neck | 81727001 | C0027532
SRT | T-11149 | Nasal bone | 74386004 | C0027422
SRT | T-D1600 | Neck | 45048000 | C0027530
SRT | T-11102 | Optic canal | 55024004 | C0450102
SRT | T-D14AE | Orbital structure | 363654007 | C0029180
SRT | T-11160 | Palatine bone | 51283005 | C0222734
SRT | T-22000 | Paranasal sinus | 2095001 | C0030471
SRT | T-11120 | Parietal bone | 24924006 | C0030558
SRT | T-61007 | Salivary gland | 385294005 | C0036098
SRT | T-11100 | Skull | 89546000 | C0037303
SRT | T-51120 | Soft palate | 49460000 | C0030219
SRT | T-11150 | Sphenoid bone | 73117003 | C0037884
SRT | T-61300 | Submandibular gland | 54019009 | C0038556
SRT | T-11130 | Temporal bone | 60911003 | C0039484
SRT | T-15290 | Temporomandibular joint | 53620006 | C0039493
SRT | T-53000 | Tongue | 21974007 | C0040408
SRT | T-54010 | Tooth | 38199008 | C0040426
SRT | T-25000 | Trachea | 44567001 | C0040578
SRT | T-11011 | Vertebral column and cranium | 110517009 | C1266914
SRT | T-11166 | Zygoma | 13881006 | C0043539

## CID 4030 CT, MR and PET Anatomy Imaged

Type:

  Extensible

Version:

  20120822

Table CID 4030. CT, MR and PET Anatomy Imaged

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 4031 “Common Anatomic Regions”
SRT | T-42500 | Abdominal aorta | 7832008 | C0003484
SRT | T-41070 | Abdominal aorta and its branches | 360524005 | C1283352
SRT | T-B3000 | Adrenal gland | 23451007 | C0001625
SRT | T-42300 | Aortic arch | 57034009 | C0003489
SRT | T-A0100 | Brain | 12738006 | C0006104
SRT | T-45010 | Carotid Artery | 69105007 | C0007272
SRT | T-A6000 | Cerebellum | 113305005 | C0007765
SRT | T-45526 | Circle of Willis | 362047009 | C1284333
SRT | T-43000 | Coronary artery | 41801008 | C0205042
SRT | T-A0193 | Cranial venous system | 244391007 | C1280180
SRT | T-41040 | Iliac arterial system | 314322004 | C1282078
SRT | T-71000 | Kidney | 64033007 | C0022646
SRT | T-62000 | Liver | 10200004 | C0023884
SRT | T-65000 | Pancreas | 15776009 | C0030274
SRT | T-B7000 | Parathyroid | 111002 | C0030518
SRT | T-44000 | Pulmonary artery | 81040000 | C0034052
SRT | T-46600 | Renal artery | 2841007 | C0035065
SRT | T-C3000 | Spleen | 78961009 | C0037993
SRT | T-94000 | Testis | 40689003 | C0039597
SRT | T-42070 | Thoracic aorta | 113262008 | C1522460
SRT | T-C8000 | Thymus | 9875009 | C0040113
SRT | T-B6000 | Thyroid | 69748006 | C0040132
SRT | T-83000 | Uterus | 35039007 | C0042149

## CID 4031 Common Anatomic Regions

Type:

  Extensible

Version:

  20130617

Table CID 4031. Common Anatomic Regions

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-D4000 | Abdomen | 113345001 | C0000726
SRT | R-FAB57 | Abdomen and Pelvis | 416949008 | C1508499
SRT | T-15420 | Acromioclavicular joint | 85856004 | C0001208
SRT | T-15750 | Ankle joint | 70258002 | C0003087
SRT | T-59900 | Anus | 53505006 | C0003461
SRT | T-280A0 | Apex of Lung | 86598002 | C0225703
SRT | T-60610 | Bile duct | 28273000 | C0005400
SRT | T-74000 | Bladder | 89837001 | C0005682
SRT | T-12700 | Bone of lower limb | 72001000 | C0448188
SRT | T-D0821 | Bone of upper limb | 371195002 | C0003793
SRT | T-04000 | Breast | 76752008 | C0006141
SRT | T-26000 | Bronchus | 955009 | C0006255
SRT | T-12770 | Calcaneus | 80144004 | C0006655
SRT | T-11501 | Cervical spine | 122494005 | C0728985
SRT | T-D00F7 | Cervico-thoracic spine | 297171002 | C0729373
SRT | T-D3000 | Chest | 51185008 | C0817096
SRT | R-FAB55 | Chest and Abdomen | 416550000 | C1442171
SRT | R-FAB56 | Chest, Abdomen and Pelvis | 416775004 | C1562547
SRT | T-12310 | Clavicle | 51299004 | C0008913
SRT | T-11BF0 | Coccyx | 64688005 | C0009194
SRT | T-59300 | Colon | 71854001 | C0009368
SRT | T-58200 | Duodenum | 38848004 | C0013303
SRT | T-15430 | Elbow joint | 16953009 | C0013770
SRT | T-D0010 | Entire body | 38266002 | C0229960
SRT | T-56000 | Esophagus | 32849002 | C0014876
SRT | T-DD163 | Esophagus, stomach and duodenum | 110861005 | C1268410
SRT | T-D0300 | Extremity | 66019005 | C0015385
SRT | T-AA000 | Eye | 81745001 | C0015392
SRT | T-D0801 | Eye region | 371398005 | C0015392
SRT | T-11196 | Facial bones | 91397008 | C0015455
SRT | T-12710 | Femur | 71341001 | C0015811
T-12750 | Fibula | FIBULA
SRT | T-D8800 | Finger | 7569003 | C0016129
SRT | T-D9700 | Foot | 56459004 | C0016504
SRT | T-D8500 | Forearm | 14975008 | C0016536
SRT | T-63000 | Gallbladder | 28231008 | C0016976
SRT | T-D8700 | Hand | 85562004 | C0018563
SRT | T-D1100 | Head | 69536005 | C0018670
SRT | T-D1000 | Head and Neck | 774007 | C0460004
SRT | T-32000 | Heart | 80891009 | C0018787
SRT | T-15710 | Hip joint | 29836001 | C0019558
SRT | T-12410 | Humerus | 85050009 | C0020164
SRT | T-58600 | Ileum | 34516001 | C0020885
SRT | T-12340 | Ilium | 22356005 | C0020889
SRT | T-AB959 | Internal Auditory Canal | 361078006 | C1283773
SRT | T-D1213 | Jaw region | 661005 | C0022359
SRT | T-58400 | Jejunum | 21306003 | C0022378
SRT | T-D9200 | Knee | 72696002 | C1456798
SRT | T-59000 | Large intestine | 14742008 | C0021851
SRT | T-24100 | Larynx | 4596009 | C0023078
SRT | T-D9400 | Lower leg | 30021000 | C1140621
SRT | T-D9000 | Lower limb | 61685007 | C0023216
SRT | T-11503 | Lumbar spine | 122496007 | C0024091
SRT | T-D00F9 | Lumbo-sacral spine | 297173004 | C0574025
SRT | T-11180 | Mandible | 91609006 | C0024687
SRT | T-11133 | Mastoid bone | 59066005 | C0446908
SRT | T-11170 | Maxilla | 70925003 | C0024947
SRT | T-D3300 | Mediastinum | 72410000 | C0025066
SRT | T-14668 | Muscle of lower limb | 102292000 | C0584890
SRT | T-13600 | Muscle of upper limb | 30608006 | C0559498
SRT | T-11149 | Nasal bone | 74386004 | C0027422
SRT | T-D1600 | Neck | 45048000 | C0027530
SRT | R-FAB52 | Neck and Chest | 417437006 | C1562459
SRT | R-FAB53 | Neck, Chest and Abdomen | 416152001 | C1562378
SRT | R-FAB54 | Neck, Chest, Abdomen and Pelvis | 416319003 | C1562776
SRT | T-11102 | Optic canal | 55024004 | C0450102
SRT | T-D14AE | Orbital structure | 363654007 | C0029180
SRT | T-65600 | Pancreatic duct and bile duct systems | 110621006 | C1267614
SRT | T-22000 | Paranasal sinus | 2095001 | C0030471
SRT | T-61100 | Parotid gland | 45289007 | C0030580
SRT | T-12730 | Patella | 64234005 | C0030647
SRT | T-D6000 | Pelvis | 12921003 | C0030797
SRT | R-FAB58 | Pelvis and lower extremities | 416631005 | C1562943
DCM | 113681 | Phantom |  | C0282611
SRT | T-9200B | Prostate | 181422007 | C1278980
SRT | T-59600 | Rectum | 34402009 | C0034896
SRT | T-11300 | Rib | 113197003 | C0035561
SRT | T-15680 | Sacroiliac joint | 39723000 | C0036036
SRT | T-11AD0 | Sacrum | 54735007 | C0036037
SRT | T-12280 | Scapula | 79601000 | C0036277
SRT | T-D1460 | Sella turcica | 42575006 | C0036609
SRT | T-12980 | Sesamoid bones of foot | 58742003 | C0278418
SRT | T-D2220 | Shoulder | 16982005 | C0037004
SRT | T-11100 | Skull | 89546000 | C0037303
SRT | T-58000 | Small intestine | 30315005 | C0021852
SRT | T-D0146 | Spine | 280717001 | C0037949
SRT | T-15610 | Sternoclavicular joint | 7844006 | C0038291
SRT | T-11210 | Sternum | 56873002 | C0038293
SRT | T-57000 | Stomach | 69695003 | C0038351
SRT | T-61300 | Submandibular gland | 54019009 | C0038556
SRT | T-15770 | Tarsal joint | 27949001 | C0039318
SRT | T-15290 | Temporomandibular joint | 53620006 | C0039493
SRT | T-D9100 | Thigh | 68367000 | C0039866
SRT | T-11502 | Thoracic spine | 122495006 | C0581269
SRT | T-D00F8 | Thoraco-lumbar spine | 297172009 | C0729374
SRT | T-D8810 | Thumb | 76505004 | C0040067
SRT | T-D9800 | Toe | 29707007 | C0040357
SRT | T-25000 | Trachea | 44567001 | C0040578
SRT | T-D8200 | Upper arm | 40983000 | C0446516
SRT | T-D8000 | Upper limb | 53120007 | C0016555
SRT | T-70010 | Upper urinary tract | 17571003 | C0729865
SRT | T-73000 | Ureter | 87953007 | C0041951
SRT | T-75000 | Urethra | 13648007 | C0041967
SRT | T-88920 | Uterus and fallopian tubes | 110639002 | C1267676
SRT | T-11011 | Vertebral column and cranium | 110517009 | C1266914
SRT | T-15460 | Wrist joint | 74670003 | C1322271
SRT | T-11166 | Zygoma | 13881006 | C0043539

### Note

1. In a prior version of this table, the code T-D1217 was specified for the concept "Maxilla and mandible". The use of this code conflicts with its assignment to another concept in SNOMED, and its use in this context is deprecated. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

2. In a prior version of this table, the code T-D8300 was used for T-15430, T-12402 for T-D8500, T-15710 for T-D2500, T-73800 for T-73000, and T-11167 for T-11166. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 4032 MR Spectroscopy Metabolites

Type:

  Extensible

Version:

  20040322

Table CID 4032. MR Spectroscopy Metabolites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 4033 “MR Proton Spectroscopy Metabolites”

## CID 4033 MR Proton Spectroscopy Metabolites

Type:

  Extensible

Version:

  20070122

Table CID 4033. MR Proton Spectroscopy Metabolites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-6175A | N-acetylaspartate
SRT | F-61080 | Citrate
SRT | F-61620 | Choline
SRT | F-61380 | Creatine
DCM | 113094 | Creatine and Choline
SRT | F-61760 | Lactate
SRT | F-63600 | Lipid
DCM | 113095 | Lipid and Lactate
DCM | 113080 | Glutamate and glutamine
SRT | F-64210 | Glutamine
SRT | F-64460 | Tuarine
SRT | F-61A90 | Inositol
DCM | 113081 | Choline/Creatine Ratio
DCM | 113082 | N-acetylaspartate/Creatine Ratio
DCM | 113083 | N-acetylaspartate/Choline Ratio
DCM | 113096 | Creatine+Choline/Citrate Ratio

### Note

For the purpose of this context group, where possible, the resonance peak in the spectrum corresponding to a particular metabolite is described using the concept from SNOMED for the substance corresponding to the metabolite. E.g., the code used for "lipid" is the code for "lipid (substance) ", as this concept is effectively post-coordinated by its use in the Metabolite Map Code Sequence (0018,9083) to mean "lipid resonance peaks in MR spectroscopy".

## CID 4040 Endoscopy Anatomic Regions

Type:

  Extensible

Version:

  20071101

Table CID 4040. Endoscopy Anatomic Regions

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-D4000 | Abdomen | 113345001 | C0000726
SRT | T-59490 | Anus, rectum and sigmoid colon | 110612005 | C1267595
SRT | T-60610 | Bile duct | 28273000 | C0005400
SRT | T-74000 | Bladder | 89837001 | C0005682
SRT | T-DD123 | Bladder and urethra | 110837003 | C1268386
SRT | T-26000 | Bronchus | 955009 | C0006255
SRT | T-83200 | Cervix | 71252005 | C0007874
SRT | T-D3000 | Chest | 51185008 | C0817096
SRT | T-DD163 | Esophagus, stomach and duodenum | 110861005 | C1268410
SRT | T-AB200 | External auditory canal | 84301002 | C0013444
SRT | T-63000 | Gallbladder | 28231008 | C0016976
SRT | T-D7000 | Inguinal region | 26893007 | C0018246
SRT | T-15001 | Joint | 39352004 | C0022417
SRT | T-71000 | Kidney | 64033007 | C0022646
SRT | T-D9200 | Knee | 72696002 | C1456798
SRT | T-59000 | Large intestine | 14742008 | C0021851
SRT | T-24100 | Larynx | 4596009 | C0023078
SRT | T-40230 | Lumen of blood vessel | 91747007 | C0524424
SRT | T-D3300 | Mediastinum | 72410000 | C0025066
SRT | T-2300C | Naso pharynx | 360955006 | C1283682
SRT | T-22000 | Paranasal sinus | 2095001 | C0030471
SRT | T-55000 | Pharynx | 54066008 | C0031354
SRT | T-20101 | Pharynx and larynx | 312535008 | C0729889
SRT | T-59600 | Rectum | 34402009 | C0034896
SRT | T-D2220 | Shoulder | 16982005 | C0037004
SRT | T-59470 | Sigmoid colon | 60184004 | C0227391
SRT | T-D0146 | Spine | 280717001 | C0037949
SRT | T-DD006 | Trachea and bronchus | 110726009 | C1268276
SRT | T-70010 | Upper urinary tract | 17571003 | C0729865
SRT | T-73000 | Ureter | 87953007 | C0041951
SRT | T-88920 | Uterus and fallopian tubes | 110639002 | C1267676

### Note

1. See Annex I for examples of the relationship between anatomic regions and type of endoscopy performed.

2. In a prior version of this table, the code T-55002 was used for T-55000, and T-73800 for T-73000. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 4042 XA/XRF Anatomy Imaged

Type:

  Extensible

Version:

  20050822

Table CID 4042. XA/XRF Anatomy Imaged

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3010 “Cardiovascular Anatomic Locations”
Include CID 4031 “Common Anatomic Regions”

## CID 4050 Drug Or Contrast Agent Characteristics

Type:

  Extensible

Version:

  20070124

Table CID 4050. Drug or Contrast Agent Characteristics

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-C52F | Active Ingredient
DCM | 121380 | Active Ingredient Undiluted Concentration
DCM | 121381 | Contrast/Bolus Ingredient Opaque
SRT | G-D705 | Volume

## CID 4051 General Devices

Type:

  Extensible

Version:

  20061023

Table CID 4051. General Devices

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 8 “Angiographic Interventional Devices”
Include CID 3451 “Calibration Objects”
Include CID 4052 “Phantom Devices”

## CID 4052 Phantom Devices

Type:

  Extensible

Version:

  20061023

Table CID 4052. Phantom Devices

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113681 | Phantom
DCM | 113682 | ACR Accreditation Phantom - CT
DCM | 113683 | ACR Accreditation Phantom - MR
DCM | 113684 | ACR Accreditation Phantom - Mammography
DCM | 113685 | ACR Accreditation Phantom - Stereotactic Breast Biopsy
DCM | 113686 | ACR Accreditation Phantom - ECT
DCM | 113687 | ACR Accreditation Phantom - PET
DCM | 113688 | ACR Accreditation Phantom - ECT/PET
DCM | 113689 | ACR Accreditation Phantom - PET Faceplate
DCM | 113690 | IEC Head Dosimetry Phantom
DCM | 113691 | IEC Body Dosimetry Phantom
DCM | 113692 | NEMA XR21-2000 Phantom

## CID 4100 T1 Measurement Methods

Type:

  Extensible

Version:

  20141110

Table CID 4100. T1 Measurement Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126350 | T1 by Multiple Flip Angles
DCM | 126351 | T1 by Inversion Recovery
DCM | 126352 | T1 by Fixed Value

### Note

## CID 4101 Tracer Kinetic Models

Type:

  Extensible

Version:

  20141110

Table CID 4101. Tracer Kinetic Models

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126340 | Standard Tofts Model
DCM | 126341 | Extended Tofts Model
DCM | 126343 | First Pass Leakage Profile (FPLP)
DCM | 126344 | Shutter-Speed Model (SSM)

### Note

## CID 4102 Perfusion Measurement Methods

Type:

  Extensible

Version:

  20141110

Table CID 4102. Perfusion Measurement Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126300 | Perfusion analysis by Stable Xenon CT technique
DCM | 126301 | Perfusion analysis by IV Iodinated Contrast CT technique
DCM | 126302 | Perfusion analysis by Arterial Spin Labeling MR technique
DCM | 126303 | Perfusion analysis by Susceptibility MR technique

### Note

## CID 4103 Arterial Input Function Measurement Methods

Type:

  Extensible

Version:

  20141110

Table CID 4103. Arterial Input Function Measurement Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126360 | AIF Ignored
DCM | 126361 | Population Averaged AIF
DCM | 126362 | User-defined AIF ROI
DCM | 126363 | Automatically Detected AIF ROI
DCM | 126364 | Blind Estimation of AIF

### Note

The anatomic location relevant to the application of any AIF method is not pre-coordinated in concepts in this Context Group. Typically these would be described by the Finding Site of any related measurements in the appropriate template.

## CID 4104 Bolus Arrival Time Derivation Methods

Type:

  Extensible

Version:

  20141110

Table CID 4104. Bolus Arrival Time Derivation Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126373 | Temporal Derivative Exceeds Threshold
DCM | 126370 | Time of Peak Concentration
DCM | 126372 | Time of Leading Half-Peak Concentration

### Note

## CID 4105 Perfusion Analysis Methods

Type:

  Extensible

Version:

  20141110

Table CID 4105. Perfusion Analysis Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126310 | Least Mean Square (LMS) deconvolution
DCM | 126311 | Singular Value Decomposition (SVD) deconvolution

### Note

## CID 4106 Quantitative Methods used for Perfusion And Tracer Kinetic Models

Type:

  Extensible

Version:

  20141110

Table CID 4106. Quantitative Methods used for Perfusion And Tracer Kinetic Models

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 4100 “T1 Measurement Methods”
Include CID 4101 “Tracer Kinetic Models”
Include CID 4102 “Perfusion Measurement Methods”
Include CID 4103 “Arterial Input Function Measurement Methods”
Include CID 4104 “Bolus Arrival Time Derivation Methods”
Include CID 4105 “Perfusion Analysis Methods”
DCM | 126342 | Model-free concentration-time quantitification

### Note

1. Concepts from this context group may be used in measurement templates to describe the measurement method of measurement on an ROI.

   E.g., NUM (126312, DCM, "Ktrans") = 0.0185 /min; (G-C036, SRT, "Measurement Method") = (126341, DCM, "Extended Tofts Model")

## CID 4107 Tracer Kinetic Model Parameters

Type:

  Extensible

Version:

  20141110

Table CID 4107. Tracer Kinetic Model Parameters

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126312 | Ktrans
DCM | 126313 | kep
DCM | 126314 | ve
DCM | 126330 | tau_m
DCM | 126331 | vp

### Note

## CID 4108 Perfusion Model Parameters

Type:

  Extensible

Version:

  20141110

Table CID 4108. Perfusion Model Parameters

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113055 | Regional Cerebral Blood Flow
DCM | 126390 | Regional Blood Flow
DCM | 113056 | Regional Cerebral Blood Volume
DCM | 126391 | Regional Blood Volume
DCM | 113052 | Mean Transit Time
DCM | 113069 | Time To Peak
DCM | 126392 | Oxygen Extraction Fraction

### Note

## CID 4109 Model-Independent Dynamic Contrast Analysis Parameters

Type:

  Extensible

Version:

  20141110

Table CID 4109. Model-Independent Dynamic Contrast Analysis Parameters

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126320 | IAUC
DCM | 126321 | IAUC60
DCM | 126322 | IAUC90
DCM | 126370 | Time of Peak Concentration
DCM | 126372 | Time of Leading Half-Peak Concentration
DCM | 126371 | Bolus Arrival Time
DCM | 113069 | Time To Peak
DCM | 126374 | Temporal Derivative Threshold
DCM | 126375 | Maximum Slope
DCM | 126376 | Maximum Difference
DCM | 126377 | Tracer Concentration

### Note

## CID 4110 Tracer Kinetic Modeling Covariates

Type:

  Extensible

Version:

  20141110

Table CID 4110. Tracer Kinetic Modeling Covariates

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 20570-8 | Hematocrit

## CID 4111 Contrast Characteristics

Type:

  Extensible

Version:

  20141110

Table CID 4111. Contrast Characteristics

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126380 | Contrast Relaxivity

## CID 4200 Ophthalmic Imaging Agent

Type:

  Extensible

Version:

  20040921

Table CID 4200. Ophthalmic Imaging Agent

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-B02CC | Fluorescein
SRT | C-B0156 | Indocyanine green
SRT | C-B0295 | Rose Bengal
SRT | C-22853 | Trypan blue
SRT | C-B02C5 | Methylene blue

## CID 4201 Patient Eye Movement Command

Type:

  Extensible

Version:

  20040921

Table CID 4201. Patient Eye Movement Command

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-1022D | Primary gaze
SRT | R-404BF | Upward gaze
SRT | R-404B9 | Left upgaze
SRT | R-404BC | Left gaze
SRT | R-404B7 | Left downgaze
SRT | R-404B6 | Downgaze
SRT | R-404B8 | Right downgaze
SRT | R-404BD | Right gaze
SRT | R-404BA | Right upgaze
SRT | R-10227 | Convergent gaze

## CID 4202 Ophthalmic Photography Acquisition Device

Type:

  Extensible

Version:

  20100607

Table CID 4202. Ophthalmic Photography Acquisition Device

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-1021A | Fundus Camera
SRT | A-2B201 | Slit Lamp Biomicroscope
SRT | R-1021B | External Camera
SRT | R-1021C | Specular Microscope
SRT | A-2B210 | Operating Microscope
SRT | A-00E8A | Scanning Laser Ophthalmoscope
SRT | R-1021D | Indirect Ophthalmoscope
SRT | R-1021E | Direct Ophthalmoscope
SRT | R-1021F | Ophthalmic Endoscope
SRT | A-00FCA | Keratoscope
SRT | A-00FF4 | Pupillograph

## CID 4203 Ophthalmic Photography Illumination

Type:

  Extensible

Version:

  20100607

Table CID 4203. Ophthalmic Photography Illumination

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-1020E | Dual diffuse direct illumination
SRT | R-1020F | Fine slit beam direct illumination
SRT | R-10211 | Broad tangential direct illumination
SRT | R-10213 | Indirect sclerotic scatter illumination
SRT | R-10215 | Indirect retroillumination from the iris
SRT | R-10217 | Indirect retroillumination from the retina
SRT | R-10218 | Indirect iris transillumination
SRT | 111625 | Diffuse direct illumination
DCM | 111627 | Scotopic light
DCM | 111628 | Mesopic light
DCM | 111629 | Photopic light
DCM | 111630 | Dynamic light

Reference: From the OPS web site: http://www.opsweb.org/Op-Photo/SlitLamp/SL/SlitLamp.htm

## CID 4204 Ophthalmic Filter

Type:

  Extensible

Version:

  20110112

Table CID 4204. Ophthalmic Filter

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-010E2 | Green optical filter
SRT | A-010DF | Red optical filter
SRT | A-010DA | Blue optical filter
SRT | A-010E0 | Yellow-green optical filter
SRT | A-010D8 | Blue-green optical filter
SRT | A-010DC | Infrared optical filter
SRT | A-010E1 | Polarizing optical filter
DCM | 111609 | No filter

## CID 4205 Ophthalmic Lens

Type:

  Extensible

Version:

  20040921

Table CID 4205. Ophthalmic Lens

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-10219 | Indirect ophthalmoscopy lens
SRT | R-10239 | Concave contact fundus lens
SRT | R-1023A | Concave noncontact fundus lens
SRT | R-1023B | Contact fundus lens
SRT | A-00FAD | Goniolens
SRT | R-1023D | Convex noncontact fundus lens
SRT | R-1023E | Noncontact fundus lens
SRT | R-1023C | Convex contact fundus lens

## CID 4206 Ophthalmic Channel Description

Type:

  Extensible

Version:

  20040921

Table CID 4206. Ophthalmic Channel Description

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A12F | Blue
SRT | R-102C0 | Full Spectrum
SRT | G-A11E | Green
SRT | R-102BE | Infrared
SRT | G-A11A | Red
SRT | G-A132 | Red free
SRT | R-102BF | Ultraviolet

## CID 4207 Ophthalmic Image Position

Type:

  Extensible

Version:

  20110825

Table CID 4207. Ophthalmic Image Position

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-10229 | Diabetic Retinopathy Study field 1
SRT | R-1022A | Diabetic Retinopathy Study field 2
SRT | R-1022B | Diabetic Retinopathy Study field 3
SRT | R-1022C | Diabetic Retinopathy Study field 4
SRT | R-1022E | Diabetic Retinopathy Study field 5
SRT | R-1022F | Diabetic Retinopathy Study field 6
SRT | R-10231 | Diabetic Retinopathy Study field 7
DCM | 111621 | Field 1 for Joslin3 field
DCM | 111622 | Field 2 for Joslin 3 field
DCM | 111623 | Field 3 for Joslin 3 field
DCM | 111900 | Macula centered
DCM | 111901 | Disc centered
DCM | 111902 | Lesion centered
DCM | 111903 | Disc-macula centered
DCM | 111904 | Mid-peripheral-superior
DCM | 111905 | Mid-peripheral-superior temporal
DCM | 111906 | Mid-peripheral-temporal
DCM | 111907 | Mid-peripheral-inferior temporal
DCM | 111908 | Mid-peripheral-inferior
DCM | 111909 | Mid-peripheral-inferior nasal
DCM | 111910 | Mid-peripheral-nasal
DCM | 111911 | Mid-peripheral-superior nasal
DCM | 111912 | Peripheral-superior
DCM | 111913 | Peripheral-superior temporal
DCM | 111914 | Peripheral-temporal
DCM | 111915 | Peripheral-inferior temporal
DCM | 111916 | Peripheral-inferior
DCM | 111917 | Peripheral-inferior nasal
DCM | 111918 | Peripheral-nasal
DCM | 111919 | Peripheral-superior nasal

## CID 4208 Mydriatic Agent

Type:

  Extensible

Version:

  20040921

Table CID 4208. Mydriatic Agent

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-677B9 | Atropine
SRT | C-677C0 | Homatropine
SRT | C-97520 | Cyclopentolate
SRT | C-68165 | Phenylephrine
SRT | C-97580 | Tropicamide

## CID 4209 Ophthalmic Anatomic Structure Imaged

Type:

  Extensible

Version:

  20040921

Table CID 4209. Ophthalmic Anatomic Structure Imaged

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-AA050 | Anterior chamber of eye
SRT | T-AA180 | Both eyes
SRT | T-AA310 | Choroid of eye
SRT | T-AA400 | Ciliary body
SRT | T-AA860 | Conjunctiva
SRT | T-AA200 | Cornea
SRT | T-AA000 | Eye
SRT | T-AA810 | Eyelid
SRT | T-AA621 | Fovea centralis
SRT | T-AA500 | Iris
SRT | T-AA862 | Lacrimal caruncle
SRT | T-AA910 | Lacrimal gland
SRT | T-AA940 | Lacrimal sac
SRT | T-AA700 | Lens
SRT | T-AA830 | Lower Eyelid
SRT | T-45400 | Ophthalmic artery
SRT | T-AA630 | Optic nerve head
SRT | T-AA610 | Retina
SRT | T-AA110 | Sclera
SRT | T-AA820 | Upper Eyelid

## CID 4210 Ophthalmic Tomography Acquisition Device

Type:

  Extensible

Version:

  20131014

Table CID 4210. Ophthalmic Tomography Acquisition Device

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-00FBE | Optical Coherence Tomography Scanner
SRT | R-FAB5A | Retinal Thickness Analyzer
SRT | A-00E8B | Confocal Scanning Laser Ophthalmoscope
DCM | 111626 | Scheimpflug Camera
SRT | A-00E8C | Scanning Laser Polarimeter
DCM | 111945 | Elevation-based corneal tomographer
DCM | 111946 | Reflection-based corneal topographer
DCM | 111947 | Interferometry-based corneal tomographer

## CID 4211 Ophthalmic OCT Anatomic Structure Imaged

Type:

  Extensible

Version:

  20071016

Table CID 4211. Ophthalmic OCT Anatomic Structure Imaged

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-AA050 | Anterior chamber of eye
SRT | T-AA310 | Choroid of eye
SRT | T-AA400 | Ciliary body
SRT | T-AA860 | Conjunctiva
SRT | T-AA200 | Cornea
SRT | T-AA500 | Iris
SRT | T-AA700 | Lens
SRT | T-AA630 | Optic nerve head
SRT | T-AA610 | Retina
SRT | T-AA110 | Sclera
SRT | T-AA079 | Vitreous
SRT | T-AA220 | Corneal epithelium
SRT | T-AA260 | Corneal endothelium

## CID 4214 Ophthalmic Horizontal Directions

Type:

  Extensible

Version:

  20080124

Table CID 4214. Ophthalmic Horizontal Directions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-C028 | Inward
SRT | R-404C7 | Outward

## CID 4215 Ophthalmic Vertical Directions

Type:

  Extensible

Version:

  20080124

Table CID 4215. Ophthalmic Vertical Directions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-404BE | Up
SRT | R-404B3 | Down

## CID 4216 Ophthalmic Visual Acuity Type

Type:

  Extensible

Version:

  20080124

Table CID 4216. Ophthalmic Visual Acuity Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111685 | Autorefraction Visual Acuity
DCM | 111686 | Habitual Visual Acuity
DCM | 111687 | Prescription Visual Acuity
SRT | F-04ECE | Potential Acuity Meter Visual Acuity
SRT | F-04D54 | Best Corrected Visual Acuity
SRT | F-04D53 | Uncorrected Visual Acuity
SRT | F-04D55 | Pinhole Visual Acuity
SRT | F-04ECF | Brightness Acuity Testing Visual Acuity

## CID 4220 Visual Fixation Quality During Acquisition

Type:

  Extensible

Version:

  20090917

Table CID 4220. Visual Fixation Quality During Acquisition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A555 | Steady
SRT | G-A556 | Not Steady
SRT | G-A385 | Indeterminate

## CID 4221 Visual Fixation Quality Problem

Type:

  Extensible

Version:

  20090917

Table CID 4221. Visual Fixation Quality Problem

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110518 | Patient Movement
SRT | F-02FA4 | Eccentric Fixation
DCM | 110519 | Operator Error
DCM | 110501 | Equipment failure

## CID 4222 Ophthalmic Macular Grid Problem

Type:

  Extensible

Version:

  20090917

Table CID 4222. Ophthalmic Macular Grid Problem

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 4221 “Visual Fixation Quality Problem”
SRT | F-0123A | Constricted Pupil
SRT | DA-73402 | Lens Opacity
SRT | DA-75300 | Corneal Opacity
SRT | DA-7931D | Vitreous Opacity
SRT | R-20839 | Poor Visual Fixation
SRT | DA-76000 | Eyelid Disease
DCM | 111695 | Interfering Tears or Drops
SRT | DA-74100 | Refractive Error
DCM | 111209 | Patient Positioning Problem
SRT | F-F1722 | Dry Eyes Problem

## CID 4230 Ophthalmic Ultrasound Axial Measurements Type

Type:

  Extensible

Version:

  20100623

Table CID 4230. Ophthalmic Ultrasound Axial Measurements Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111750 | Ultrasound Contact
DCM | 111751 | Ultrasound Immersion

## CID 4231 Lens Status

Type:

  Extensible

Version:

  20100623

Table CID 4231. Lens Status

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | DA-73410 | Aphakic
SRT | R-2073F | Phakic
SRT | A-040F7 | Phakic IOL
SRT | F-02087 | Piggyback IOL
SRT | DA-73460 | Pseudophakia

## CID 4232 Vitreous Status

Type:

  Extensible

Version:

  20100623

Table CID 4232. Vitreous Status

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-035F3 | Gas in vitreous cavity
SRT | DA-7930D | Post-Vitrectomy
SRT | F-035FD | Silicone Oil
SRT | T-AA092 | Vitreous Only

## CID 4233 Ophthalmic Axial Length Measurements Segment Names

Type:

  Extensible

Version:

  20100623

Table CID 4233. Ophthalmic Axial Length Measurements Segment Names

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-AA200 | Cornea
SRT | T-AA050 | Anterior Chamber
DCM | 111778 | Single or Anterior Lens
DCM | 111779 | Posterior Lens
SRT | T-AA079 | Vitreous Cavity

## CID 4234 Refractive Surgery Types

Type:

  Extensible

Version:

  20100623

Table CID 4234. Refractive Surgery Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P1-A3102 | RK
SRT | P1-A3835 | PRK
SRT | P0-0526F | LASIK
SRT | P1-A3846 | LASEK

## CID 4235 Keratometry Descriptors

Type:

  Extensible

Version:

  20100623

Table CID 4235. Keratometry Descriptors

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111753 | Manual Keratometry
DCM | 111754 | Auto Keratometry
DCM | 111755 | Simulated Keratometry
DCM | 111756 | Equivalent K-reading

## CID 4236 IOL Calculation Formula

Type:

  Extensible

Version:

  20100623

Table CID 4236. IOL Calculation Formula

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111760 | Haigis
DCM | 111761 | Haigis-L
DCM | 111762 | Holladay 1
DCM | 111763 | Holladay 2
DCM | 111764 | Hoffer Q
DCM | 111765 | Olsen
DCM | 111766 | SRKII
DCM | 111767 | SRK-T

## CID 4237 Lens Constant Type

Type:

  Extensible

Version:

  20100623

Table CID 4237. Lens Constant Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-048FA | A-Constant
DCM | 111768 | ACD Constant
DCM | 111769 | Haigis a0
DCM | 111770 | Haigis a1
DCM | 111771 | Haigis a2
DCM | 111772 | Hoffer pACD Constant
DCM | 111773 | Surgeon Factor

## CID 4238 Refractive Error Types

Type:

  Extensible

Version:

  20100623

Table CID 4238. Refractive Error Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | DA-74120 | Myopia
SRT | DA-74110 | Hyperopia

## CID 4239 Anterior Chamber Depth Definition

Type:

  Extensible

Version:

  20100623

Table CID 4239. Anterior Chamber Depth Definition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111776 | Front Of Cornea To Front Of Lens
DCM | 111777 | Back Of Cornea To Front Of Lens

## CID 4240 Ophthalmic Measurement Or Calculation Data Source

Type:

  Extensible

Version:

  20100623

Table CID 4240. Ophthalmic Measurement or Calculation Data Source

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111780 | Measurement From This Device
DCM | 113857 | Manual Entry
DCM | 111781 | External Data Source
DCM | 111782 | Axial Measurements SOP Instance
DCM | 111783 | Refractive Measurements SOP Instance

## CID 4241 Ophthalmic Axial Length Selection Method

Type:

  Extensible

Version:

  20100623

Table CID 4241. Ophthalmic Axial Length Selection Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121412 | Mean value chosen
DCM | 121410 | User chosen value

## CID 4243 Ophthalmic Axial Length Quality Metric Type

Type:

  Extensible

Version:

  20100623

Table CID 4243. Ophthalmic Axial Length Quality Metric Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111786 | Standard Deviation of measurements used
DCM | 111787 | Signal to Noise Ratio

## CID 4244 Ophthalmic Agent Concentration Units

Type:

  Extensible

Version:

  20100623

Table CID 4244. Ophthalmic Agent Concentration Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | % | Percent
UCUM | mg/ml | mg/ml

## CID 4250 Visual Field Static Perimetry Test Patterns

Type:

  Extensible

Version:

  20100827

Table CID 4250. Visual Field Static Perimetry Test Patterns

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111800 | Visual Field 24-2 Test Pattern
DCM | 111801 | Visual Field 10-2 Test Pattern
DCM | 111802 | Visual Field 30-2 Test Pattern
DCM | 111803 | Visual Field 60-4 Test Pattern
DCM | 111804 | Visual Field Macula Test Pattern
DCM | 111805 | Visual Field Central 40 Point Test Pattern
DCM | 111806 | Visual Field Central 76 Point Test Pattern
DCM | 111807 | Visual Field Peripheral 60 Point Test Pattern
DCM | 111808 | Visual Field Full Field 81 Point Test Pattern
DCM | 111809 | Visual Field Full Field 120 Point Test Pattern
DCM | 111810 | Visual Field G Test Pattern
DCM | 111811 | Visual Field M Test Pattern
DCM | 111812 | Visual Field 07 Test Pattern
DCM | 111813 | Visual Field LVC Test Pattern
DCM | 111814 | Visual Field Central Test Pattern

## CID 4251 Visual Field Static Perimetry Test Strategies

Type:

  Extensible

Version:

  20100827

Table CID 4251. Visual Field Static Perimetry Test Strategies

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111815 | Visual Field SITA-Standard Test Strategy
DCM | 111816 | Visual Field SITA-SWAP Test Strategy
DCM | 111817 | Visual Field SITA-Fast Test Strategy
DCM | 111818 | Visual Field Full Threshold Test Strategy
DCM | 111819 | Visual Field FastPac Test Strategy
DCM | 111820 | Visual Field Full From Prior Test Strategy
DCM | 111821 | Visual Field Optima Test Strategy
DCM | 111822 | Visual Field Two-Zone Test Strategy
DCM | 111823 | Visual Field Three-Zone Test Strategy
DCM | 111824 | Visual Field Quantify-Defects Test Strategy
DCM | 111825 | Visual Field TOP Test Strategy
DCM | 111826 | Visual Field Dynamic Test Strategy
DCM | 111827 | Visual Field Normal Test Strategy
DCM | 111828 | Visual Field 1-LT Test Strategy
DCM | 111829 | Visual Field 2-LT Test Strategy
DCM | 111830 | Visual Field LVS Test Strategy
DCM | 111831 | Visual Field GATE Test Strategy
DCM | 111832 | Visual Field GATEi Test Strategy
DCM | 111833 | Visual Field 2LT-Dynamic Test Strategy
DCM | 111834 | Visual Field 2LT-Normal Test Strategy
DCM | 111835 | Visual Field Fast Threshold Test Strategy
DCM | 111836 | Visual Field CLIP Test Strategy
DCM | 111837 | Visual Field CLASS Strategy

## CID 4252 Visual Field Static Perimetry Screening Test Modes

Type:

  Extensible

Version:

  20100827

Table CID 4252. Visual Field Static Perimetry Screening Test Modes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111838 | Age corrected
DCM | 111839 | Threshold related
DCM | 111840 | Single luminance
DCM | 111841 | Foveal sensitivity related
DCM | 111842 | Related to non macular sensitivity
DCM | 121410 | User chosen value

## CID 4253 Visual Field Static Perimetry Fixation Strategy

Type:

  Extensible

Version:

  20100827

Table CID 4253. Visual Field Static Perimetry Fixation Strategy

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111843 | Automated Optical
DCM | 111844 | Blind Spot Monitoring
DCM | 111845 | Macular Fixation Testing
DCM | 111846 | Observation by Examiner
SRT | R-40775 | None

## CID 4254 Visual Field Static Perimetry Test Analysis Results

Type:

  Extensible

Version:

  20100827

Table CID 4254. Visual Field Static Perimetry Test Analysis Results

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111847 | Outside normal limits
DCM | 111848 | Borderline
DCM | 111849 | Abnormally high sensitivity
DCM | 111850 | General reduction in sensitivity
DCM | 111851 | Borderline and general reduction in sensitivity
SRT | M-00101 | Within normal limits

## CID 4255 Visual Field Illumination Color

Type:

  Extensible

Version:

  20100827

Table CID 4255. Visual Field Illumination Color

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A11D | Yellow
SRT | G-A12B | White
SRT | G-A11A | Red
SRT | G-A12F | Blue
SRT | G-A11E | Green

## CID 4256 Visual Field Procedure Modifier

Type:

  Extensible

Version:

  20100827

Table CID 4256. Visual Field Procedure Modifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-42453 | Screening
SRT | R-408C3 | Diagnostic

## CID 4257 Visual Field Global Index Name

Type:

  Extensible

Version:

  20100827

Table CID 4257. Visual Field Global Index Name

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111852 | Visual Field Index
DCM | 111853 | Visual Field Loss Due to Diffuse Defect
DCM | 111854 | Visual Field Loss Due to Local Defect
DCM | 111855 | Glaucoma Hemifield Test Analysis
DCM | 111856 | Optical Fixation Measurements

## CID 4260 Ophthalmic Mapping Units for Real World Value Mapping

Type:

  Extensible

Version:

  20110825

Table CID 4260. Ophthalmic Mapping Units for Real World Value Mapping

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | um | micrometer

## CID 4261 Ophthalmic Mapping Acquisition Method

Type:

  Extensible

Version:

  20110825

Table CID 4261. Ophthalmic Mapping Acquisition Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111920 | Time domain
DCM | 111921 | Spectral domain
DCM | 111922 | No corneal compensation
DCM | 111923 | Corneal birefringence compensation
DCM | 111924 | Retinal topography

## CID 4262 Retinal Thickness Definition

Type:

  Extensible

Version:

  20110825

Table CID 4262. Retinal Thickness Definition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111925 | Retinal nerve fiber layer thickness
DCM | 111926 | Ganglion cell complex thickness
DCM | 111927 | Total retinal thickness (ILM to IS-OS)
DCM | 111928 | Total retinal thickness (ILM to RPE)
DCM | 111929 | Total retinal thickness (ILM to BM)

## CID 4263 Ophthalmic Thickness Map Value Type

Type:

  Extensible

Version:

  20110825

Table CID 4263. Ophthalmic Thickness Map Value Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111930 | Absolute ophthalmic thickness
DCM | 111931 | Thickness deviation category from normative data
DCM | 111932 | Thickness deviation from normative data

## CID 4264 Ophthalmic Map Purposes of Reference

Type:

  Extensible

Version:

  20110825

Table CID 4264. Ophthalmic Map Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121311 | Localizer
DCM | 121322 | Source image for image processing operation
DCM | 111933 | Related ophthalmic thickness map

## CID 4265 Ophthalmic Thickness Deviation Categories

Type:

  Extensible

Version:

  20110825

Table CID 4265. Ophthalmic Thickness Deviation Categories

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111935 | p>5%
DCM | 111936 | p<5%
DCM | 111937 | p<2%
DCM | 111938 | p<1%
DCM | 111939 | p<0.5%

## CID 4266 Ophthalmic Anatomic Structure Reference Point

Type:

  Extensible

Version:

  20131014

Table CID 4266. Ophthalmic Anatomic Structure Reference Point

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | M-01000 | Lesion
SRT | T-AA621 | Fovea centralis
SRT | T-AA630 | Optic nerve head
DCM | 111934 | Disc-Fovea
SRT | T-AA215 | Entire Cornea

## CID 4267 Corneal Topography Mapping Units for Real World Value Mapping

Type:

  Extensible

Version:

  20131014

Table CID 4267. Corneal Topography Mapping Units for Real World Value Mapping

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | um | micrometer
UCUM | diop | diopters
UCUM | mm | mm

## CID 4268 Corneal Topography Map Value Type

Type:

  Extensible

Version:

  20131014

Table CID 4268. Corneal Topography Map Value Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111940 | Corneal axial power map
DCM | 111941 | Corneal instantaneous power map
DCM | 111942 | Corneal refractive power map
DCM | 111943 | Corneal elevation map
DCM | 111944 | Corneal wavefront map

## CID 5000 Languages

Context Group ID 5000 comprises the language tag coding scheme of RFC 3066. The Coding Scheme Designator (0008,0102) shall be RFC3066.

### Note

1. The RFC 3066 coding scheme is constructed from a primary subtag component encoded using the language codes of ISO 639, plus two codes for extensions for languages not represented in ISO 639. The code optionally includes a second subtag component encoded using the two letter country codes of ISO 3166, or a language code extension registered by the Internet Assigned Names Authority.

2. RFC 3066 may be obtained at http://www.ietf.org/rfc/rfc3066.txt. RFC 3066 obsoletes RFC 1766.

3. ISO 639 may be obtained at http://www.loc.gov/standards/iso639-2/langhome.html.

4. The two letter country codes of ISO 3166 may be obtained at http://www.iso.ch/iso/en/prods-services/iso3166ma/02iso-3166-code-lists/index.html

5. IANA language tag registrations may be obtained at http://www.iana.org/assignments/language-tags

6. In previous editions of the Standard, this Context Group formerly included the three letter language codes of ISO 639-2/B, using Coding Scheme Designator ISO639_2, and several IANA-registered language code extensions, using Coding Scheme Designator IANARFC1766. RFC 3066 identifies a preference for the ISO 639-1 two letter codes to the ISO 639-2 three letter codes, and the ISO 639-2/T (terminology) subset to the ISO 639-2/B (bibliographic) subset.

7. In previous editions of the Standard, this Context Group provided only language identifiers, with national or regional variant identified in a separate attribute or Content Item.

## CID 5001 Countries

Context Group ID 5001 comprises the two letter country code scheme of ISO 3166. The Coding Scheme Designator (0008,0102) shall be ISO3166_1.

### Note

The two letter country codes of ISO 3166 may be obtained at http://www.iso.ch/iso/en/prods-services/iso3166ma/02iso-3166-code-lists/index.html

## CID 5002 Organizations

Context Group ID 5002 comprises the object identifier scheme of ISO 8824-1 and ISO 9834-1, when applied to organizational identifiers (see Section 8.2). The Coding Scheme Designator (0008,0102) shall be ISO_OID.

## CID 6000 Overall Breast Composition

### Note

In future extensions, Overall Breast Composition terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

Table CID 6000. Overall Breast Composition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6001 “Overall Breast Composition From BI-RADS®”

## CID 6001 Overall Breast Composition From BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E77)

Type:

  Extensible

Version:

  20020904

Table CID 6001. Overall Breast Composition From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01711 | Almost entirely fat
SRT | F-01712 | Scattered fibroglandular densities
SRT | F-01713 | Heterogeneously dense
SRT | F-01714 | Extremely dense

## CID 6002 Change Since Last Mammogram Or Prior Surgery

### Note

In future extensions, Change Since Last Mammogram or Prior Surgery terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

Table CID 6002. Change Since Last Mammogram or Prior Surgery

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6003 “Change Since Last Mammogram Or Prior Surgery From BI-RADS®”

## CID 6003 Change Since Last Mammogram Or Prior Surgery From BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E79)

Type:

  Extensible

Version:

  20020904

Table CID 6003. Change Since Last Mammogram or Prior Surgery From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01721 | New finding
SRT | F-01722 | Finding partially removed
SRT | F-01723 | No significant changes in the finding
SRT | M-02520 | Increase in size
SRT | M-02530 | Decrease in size
SRT | F-01726 | Increase in number of calcifications
SRT | F-01727 | Decrease in number of calcifications
SRT | F-01728 | Less defined
SRT | F-01729 | More defined
SRT | F-0172A | Removal of implant since previous mammogram
SRT | F-0172B | Implant revised since previous mammogram

## CID 6004 Mammography Characteristics of Shape

### Note

In future extensions, Mammography Characteristics of Shape terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

Table CID 6004. Mammography Characteristics of Shape

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6005 “Characteristics of Shape From BI-RADS®”

## CID 6005 Characteristics of Shape From BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E80)

Type:

  Extensible

Version:

  20020904

Table CID 6005. Characteristics of Shape From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | M-02100 | Round shape
SRT | M-02120 | Ovoid shape (Oval)
SRT | G-A640 | Lobular
SRT | G-A402 | Irregular

## CID 6006 Mammography Characteristics of Margin

### Note

In future extensions, Mammography Characteristics of Margin terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

Table CID 6006. Mammography Characteristics of Margin

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6007 “Characteristics of Margin From BI-RADS®”

## CID 6007 Characteristics of Margin From BI-RADS®

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6007. Characteristics of Margin From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01741 | Circumscribed lesion
SRT | F-01742 | Microlobulated lesion
SRT | F-01743 | Obscured lesion
SRT | F-01744 | Indistinct lesion
SRT | F-01745 | Spiculated lesion
DCM | 111343 | Angular margins

## CID 6008 Density Modifier

### Note

In future extensions, Density Modifier terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

Table CID 6008. Density Modifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6009 “Density Modifier From BI-RADS®”

## CID 6009 Density Modifier From BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E82)

Type:

  Extensible

Version:

  20020904

Table CID 6009. Density Modifier From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01751 | High density lesion
SRT | F-01752 | Equal density (isodense) lesion
SRT | F-01753 | Low density (not containing fat) lesion
SRT | F-01754 | Fat containing (radiolucent) lesion

## CID 6010 Mammography Calcification Types

### Note

In future extensions, Mammography Calcification Types terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

Table CID 6010. Mammography Calcification Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6011 “Calcification Types From BI-RADS®”

## CID 6011 Calcification Types From BI-RADS®

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6011. Calcification Types From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01761 | Coarse (popcorn-like) calcification
SRT | F-01762 | Dystrophic calcification
SRT | F-01763 | Eggshell calcification
SRT | F-01764 | Large rod-like calcification
SRT | F-01765 | Milk of calcium calcification
SRT | F-01766 | Lucent-centered calcification
SRT | F-01767 | Punctate calcification
SRT | F-01768 | Round shaped calcification
SRT | F-01769 | Calcified skin of breast
SRT | F-0176A | Calcified suture material
SRT | F-0176B | Vascular calcification
SRT | F-0176C | Amorphous calcification
SRT | F-0176D | Fine, linear (casting) calcification
SRT | F-0176E | Fine linear, branching (casting) calcification
SRT | F-0176F | Heterogeneous calcification
DCM | 111344 | Fine pleomorphic calcification
SRT | D7-90435 | Microcalcifications of the breast
DCM | 111345 | Macrocalcifications

## CID 6012 Calcification Distribution Modifier

### Note

In future extensions, Calcification Distribution Modifier terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

Table CID 6012. Calcification Distribution Modifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6013 “Calcification Distribution Modifier From BI-RADS®”

## CID 6013 Calcification Distribution Modifier From BI-RADS®

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6013. Calcification Distribution Modifier From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01770 | Diffuse calcification distribution
SRT | F-01771 | Linear calcification distribution
SRT | F-01772 | Grouped calcification distribution
SRT | F-01773 | Regional calcification distribution
SRT | F-01774 | Segmental calcification distribution
DCM | 111346 | Calcifications within a mass
DCM | 111347 | Calcifications outside of a mass

## CID 6014 Mammography Single Image Finding

### Note

In future extensions, Mammography Single Image Finding terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

Table CID 6014. Mammography Single Image Finding

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6015 “Single Image Finding From BI-RADS®”
DCM | 111099 | Selected region
DCM | 111100 | Breast geometry
DCM | 111101 | Image Quality
DCM | 111102 | Non-lesion
SRT | T-04100 | Nipple

## CID 6015 Single Image Finding From BI-RADS®

### Note

Collected from BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6015. Single Image Finding From BI-RADS®

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
SRT |  | F-01796 | Mammography breast density
SRT |  | F-01776 | Individual Calcification
SRT |  | F-01775 | Calcification Cluster
SRT |  | F-01795 | Architectural distortion of breast
SRT |  | F-01797 | Tubular density
SRT |  | T-C4351 | Intra-mammary lymph node
SRT |  | F-01798 | Trabecular thickening of breast
SRT |  | F-01710 | Breast composition
SRT |  | F-01799 | Skin retraction of breast
SRT |  | F-0179A | Skin thickening of breast
BI | 3.0 | I.E.6 | Axillary adenopathy
SRT |  | D0-00050 | Skin lesion
DCM |  | 111111 | Cooper's ligament changes
SRT |  | M-36300 | Edema
DCM |  | 111112 | Mass in the skin
DCM |  | 111113 | Mass on the skin
SRT |  | T-C4710 | Axillary lymph node

## CID 6016 Mammography Composite Feature

### Note

In future extensions, Mammography Composite Feature terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20050110

Table CID 6016. Mammography Composite Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111459 | Mass with calcifications
Include CID 6014 “Mammography Single Image Finding”
Include CID 6017 “Composite Feature From BI-RADS®”

## CID 6017 Composite Feature From BI-RADS®

### Note

Collected from BI-RADS® Third Edition

Type:

  Extensible

Version:

  20020904

Table CID 6017. Composite Feature From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01791 | Mammographic breast mass
SRT | F-01792 | Focal asymmetric breast tissue
SRT | F-01793 | Asymmetric breast tissue

## CID 6018 Clockface Location or Region

### Note

In future extensions, Clockface Location or Region terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

Table CID 6018. Clockface Location or Region

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6019 “Clockface Location Or Region From BI-RADS®”
SRT | T-D3050 | Chest wall

## CID 6019 Clockface Location Or Region From BI-RADS®

### Note

From BI-RADS® 3.1, with Addendum 3.1 (National Mammography Database, E96)

Type:

  Extensible

Version:

  20020904

Table CID 6019. Clockface Location or Region From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01781 | 1 o'clock position
SRT | F-01782 | 2 o'clock position
SRT | F-01783 | 3 o'clock position
SRT | F-01784 | 4 o'clock position
SRT | F-01785 | 5 o'clock position
SRT | F-01786 | 6 o'clock position
SRT | F-01787 | 7 o'clock position
SRT | F-01788 | 8 o'clock position
SRT | F-01789 | 9 o'clock position
SRT | F-0178A | 10 o'clock position
SRT | F-0178B | 11 o'clock position
SRT | F-0178C | 12 o'clock position
SRT | F-0178D | Subareolar region
SRT | F-0178E | Axillary tail region
SRT | F-0178F | Central region of breast
SRT | F-01794 | Axilla region

## CID 6020 Quadrant Location

### Note

In future extensions, Quadrant Location terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

Table CID 6020. Quadrant Location

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6021 “Quadrant Location From BI-RADS®”

## CID 6021 Quadrant Location From BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E97)

Type:

  Extensible

Version:

  20020904

Table CID 6021. Quadrant Location From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-04004 | Upper outer quadrant of breast
SRT | T-04002 | Upper inner quadrant of breast
SRT | T-04005 | Lower outer quadrant of breast
SRT | T-04003 | Lower inner quadrant of breast

## CID 6022 Side

### Note

In future extensions, Side terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Non-Extensible

Version:

  20020904

Table CID 6022. Side

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6023 “Side From BI-RADS®”

## CID 6023 Side From BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E98)

Type:

  Non-Extensible

Version:

  20020904

Table CID 6023. Side From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-04030 | Left breast
SRT | T-04020 | Right breast
SRT | T-04080 | Both breasts

## CID 6024 Depth

### Note

In future extensions, Depth terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

Table CID 6024. Depth

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6025 “Depth From BI-RADS®”

## CID 6025 Depth From BI-RADS®

### Note

From BI-RADS® Third Edition (National Mammography Database, E99)

Type:

  Extensible

Version:

  20020904

Table CID 6025. Depth From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A105 | Anterior
SRT | G-A109 | Middle
SRT | G-A106 | Posterior

## CID 6026 Mammography Assessment

### Note

In future extensions, Mammography Assessment terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20050822

Table CID 6026. Mammography Assessment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6027 “Assessment From BI-RADS®”
DCM | 111120 | Post Procedure Mammograms for Marker Placement

## CID 6027 Assessment From BI-RADS®

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6027. Assessment From BI-RADS®

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning
--- | --- | --- | ---
BI | 3.0 | II.AC.a | 0 - Need additional imaging evaluation
BI | 3.0 | II.AC.b.1 | 1 - Negative
BI | 3.0 | II.AC.b.2 | 2 - Benign Finding
BI | 3.0 | II.AC.b.3 | 3 - Probably Benign Finding - short interval follow-up
BI | 3.0 | II.AC.b.4 | 4 - Suspicious abnormality, biopsy should be considered
BI | 4.0 | MA.II.A.5.4A | 4A - Low suspicion
BI | 4.0 | MA.II.A.5.4B | 4B - Intermediate suspicion
BI | 4.0 | MA.II.A.5.4C | 4C - Moderate suspicion
BI | 3.0 | II.AC.b.5 | 5 - Highly suggestive of malignancy, take appropriate action
BI | 4.0 | MA.II.A.5.6 | 6 - Known biopsy proven malignancy

## CID 6028 Mammography Recommended Follow-up

### Note

In future extensions, Mammography Recommended Follow-up terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

Table CID 6028. Mammography Recommended Follow-Up

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6029 “Recommended Follow-up From BI-RADS®”
DCM | 111121 | Follow-up post biopsy as directed by clinician
SRT | P0-006F1 | Nuclear medicine procedure
SRT | P0-009B4 | Evaluation procedure
DCM | 111410 | Surgical consult

## CID 6029 Recommended Follow-up From BI-RADS®

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6029. Recommended Follow-Up From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111135 | Additional projections
SRT | R-102D6 | Magnification views
SRT | R-102D7 | Spot compression
DCM | 111136 | Spot magnification view(s)
SRT | P5-B0099 | Ultrasound procedure
DCM | 111138 | Old films for comparison
SRT | P5-40060 | Mammary ductogram
DCM | 111140 | Normal interval follow-up
DCM | 111141 | Any decision to biopsy should be based on clinical assessment
DCM | 111142 | Follow-up at short interval (1-11 months)
DCM | 111143 | Biopsy should be considered
DCM | 111144 | Needle localization and biopsy
DCM | 111145 | Histology using core biopsy
DCM | 111146 | Suggestive of malignancy - take appropriate action
DCM | 111147 | Cytologic analysis
DCM | 111148 | Biopsy should be strongly considered
DCM | 111149 | Highly suggestive of malignancy - take appropriate action
DCM | 111122 | Known biopsy proven malignancy - take appropriate action
SRT | P5-0900D | MRI of breast

## CID 6030 Mammography Pathology Codes

### Note

In future extensions, Mammography Pathology Codes terms that are not derived from BI-RADS® should be added to this context group.

Type:

  Extensible

Version:

  20020904

Table CID 6030. Mammography Pathology Codes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6031 “Benign Pathology Codes From BI-RADS®”
Include CID 6032 “High Risk Lesions Pathology Codes From BI-RADS®”
Include CID 6033 “Malignant Pathology Codes From BI-RADS®”

## CID 6031 Benign Pathology Codes From BI-RADS®

### Note

From BI-RADS® Third Edition, with Addendum 3.1 (National Mammography Database,F110)

Type:

  Extensible

Version:

  20020904

Table CID 6031. Benign Pathology Codes From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | M-41610 | Abscess
SRT | M-74200 | Adenosis
SRT | M-81400 | Adenoma
SRT | M-83240 | Adenolipoma
SRT | M-73310 | Apocrine Metaplasia
SRT | M-89830 | Adenomyoepithelioma
SRT | M-55160 | Amyloid (tumor)
DCM | 111251 | Normal axillary node
SRT | M-88610 | Angiolipoma
DCM | 111252 | Axillary node with calcifications
SRT | M-76100 | Angiomatosis
DCM | 111253 | Axillary node hyperplasia
SRT | F-8A063 | Asynchronous involution of breast
SRT | D7-90360 | Cyst of breast
DCM | 111255 | Benign cyst with blood
DCM | 111256 | Benign Calcifications
SRT | M-92200 | Chondroma
SRT | M-85040 | Intracystic papilloma
DCM | 111258 | Ductal adenoma
SRT | D7-90370 | Mammary duct ectasia
DCM | 111259 | Diabetic fibrous mastopathy
SRT | M-72170 | Ductal hyperplasia, Usual
SRT | M-88211 | Extra abdominal desmoid
SRT | D4-48014 | Ectopic (accessory) breast tissue
SRT | M-33410 | Epidermal inclusion cyst
SRT | M-36300 | Edema
SRT | M-90100 | Fibroadenoma
DCM | 111263 | Fibroadenomatoid hyperplasia
DCM | 111264 | Fibroadenolipoma
SRT | M-44140 | Foreign body (reaction)
SRT | D7-90310 | Fibrocystic disease of breast
SRT | M-78266 | Focal fibrosis
SRT | M-78800 | Fibromatosis
SRT | D7-90434 | Fat necrosis of breast
SRT | D7-90364 | Galactocele
SRT | M-95800 | Granular cell tumor
SRT | M-90160 | Giant fibroadenoma
SRT | D7-90420 | Gynecomastia
SRT | M-75500 | Hamartoma
SRT | M-91200 | Hemangioma
SRT | D3-F0620 | Hemangioma of subcutaneous tissue
SRT | M-91220 | Hemangioma - venous
SRT | M-35060 | Hematoma
SRT | M-72000 | Hyperplasia, usual
SRT | D7-90452 | Infarction of breast
SRT | M-40000 | Inflammation
SRT | T-C4351 | Intra-mammary lymph node
SRT | M-85030 | Intraductal papilloma
SRT | M-90300 | Juvenile fibroadenoma
DCM | 111277 | Juvenile papillomatosis
SRT | M-82040 | Lactating adenoma
SRT | M-88500 | Lipoma of the breast
DCM | 111279 | Lactational change
SRT | D7-90428 | Breast lobular hyperplasia
SRT | M-88900 | Leiomyoma
SRT | T-C4000 | Lymph node
DCM | 111281 | Large duct papilloma
SRT | D3-87780 | Thrombophlebitis of breast (Mondor's disease)
SRT | M-88250 | Myofibroblastoma
DCM | 111284 | Microglandular adenosis
DCM | 111285 | Multiple Intraductal Papillomas
DCM | 111286 | No abnormality
DCM | 111287 | Normal breast tissue
SRT | M-95400 | Neurofibroma
SRT | M-95401 | Neurofibromatosis
SRT | D7-F0810 | Benign neoplasm of nipple of female breast (Nipple adenoma)
DCM | 111290 | Oil cyst (fat necrosis cyst)
SRT | M-80500 | Papilloma
SRT | M-89400 | Pleomorphic adenoma
DCM | 111291 | Post reduction mammoplasty
DCM | 111292 | Pseudoangiomatous stromal hyperplasia
SRT | M-78731 | Radial scar
SRT | M-74220 | Sclerosing adenosis
SRT | M-36050 | Seroma
DCM | 111296 | Silicone granuloma
SRT | M-78060 | Scar tissue
SRT | M-82110 | Tubular adenoma
DCM | 111298 | Virginal hyperplasia

## CID 6032 High Risk Lesions Pathology Codes From BI-RADS®

### Note

From BI-RADS® Third Edition, with Addendum 3.1 (National Mammography Database,F110)

Type:

  Extensible

Version:

  20020904

Table CID 6032. High Risk Lesions Pathology Codes From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | M-72175 | Atypical intraductal hyperplasia
SRT | M-72105 | Atypical lobular hyperplasia
SRT | D7-F0A02 | Lobular carcinoma in situ of breast
DCM | 111299 | Peripheral duct papillomas
SRT | M-90201 | Phyllodes tumor

## CID 6033 Malignant Pathology Codes From BI-RADS®

### Note

From BI-RADS® Third Edition, with Addendum 3.1 (National Mammography Database,F110)

Type:

  Extensible

Version:

  20020904

Table CID 6033. Malignant Pathology Codes From BI-RADS®

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | M-82003 | Adenoid cystic carcinoma
DCM | 111300 | Axillary node with lymphoma
DCM | 111301 | Axillary nodal metastases
SRT | M-84013 | Apocrine adenocarcinoma
SRT | M-91203 | Angiosarcoma
DCM | 111307 | Basal cell carcinoma of nipple
DCM | 111303 | Blood vessel (vascular) invasion
SRT | M-84803 | Mucinous adenocarcinoma (Colloid carcinoma)
DCM | 111304 | Carcinoma in children
SRT | M-92203 | Chondrosarcoma
DCM | 111305 | Carcinoma in ectopic breast
DCM | 111306 | Carcinoma with endocrine differentiation
SRT | M-85012 | Comedocarcinoma (intraductal)
SRT | D7-F0902 | Carcinoma in situ of male breast
SRT | M-85733 | Carcinoma with metaplasia
DCM | 111309 | Cartilaginous and osseous change
DCM | 111310 | Carcinoma in pregnancy and lactation
SRT | M-89803 | Carcinosarcoma
DCM | 111312 | Intraductal comedocarcinoma with necrosis
DCM | 111341 | Intraductal carcinoma, high grade
DCM | 111313 | Intraductal carcinoma, low grade
SRT | M-85072 | Intraductal carcinoma micro-papillary
SRT | M-88103 | Fibrosarcoma
SRT | M-83153 | Glycogen-rich carcinoma
SRT | M-91501 | Hemangiopericytoma
SRT | M-96503 | Hodgkin's disease (lymphoma)
SRT | M-82013 | Invasive cribriform carcinoma
DCM | 111315 | Intracystic papillary carcinoma
SRT | M-85003 | Infiltrating duct carcinoma
DCM | 111316 | Invasive and in-situ carcinoma
SRT | M-85203 | Invasive lobular carcinoma
SRT | M-85303 | Inflammatory carcinoma
SRT | M-80503 | Papillary carcinoma (invasive)
DCM | 111318 | Leukemic infiltration
SRT | M-88903 | Leiomyosarcoma
SRT | M-88503 | Liposarcoma
SRT | M-83143 | Lipid-rich (lipid-secreting) carcinoma
DCM | 111320 | Lympathic vessel invasion
SRT | M-95903 | Lymphoma
DCM | 111322 | Occult carcinoma presenting with axillary lymph node metastases
DCM | 111323 | Metastatic cancer to the breast
DCM | 111324 | Metastatic cancer to the breast from the colon
DCM | 111325 | Metastatic cancer to the breast from the lung
DCM | 111326 | Metastatic melanoma to the breast
DCM | 111327 | Metastatic cancer to the breast from the ovary
DCM | 111328 | Metastatic sarcoma to the breast
SRT | M-85103 | Medullary carcinoma
DCM | 111329 | Multifocal intraductal carcinoma
DCM | 111330 | Metastatic disease to axillary node
SRT | M-88303 | Malignant fibrous histiocytoma
DCM | 111332 | Multifocal invasive ductal carcinoma
DCM | 111333 | Metastasis to an intramammary lymph node
DCM | 111334 | Malignant melanoma of nipple
SRT | M-95913 | Non-Hodgkin's lymphoma
SRT | D0-F035F | Neoplasm of the mammary skin
SRT | M-91803 | Osteogenic sarcoma
SRT | M-80502 | Papillary carcinoma in-situ
SRT | M-85403 | Paget's disease, mammary (of the nipple)
SRT | M-97313 | Plasmacytoma
SRT | M-90203 | Phyllodes tumor, malignant
DCM | 111338 | Recurrent malignancy
SRT | M-84903 | Signet ring cell carcinoma
DCM | 111340 | Squamous cell carcinoma of the nipple
SRT | M-78190 | Spindle cell nodule (tumor)
SRT | M-85023 | Secretory (juvenile) carcinoma of the breast
SRT | M-80703 | Squamous cell carcinoma
SRT | M-82113 | Tubular adenocarcinoma

## CID 6034 Intended Use of CAD Output

Type:

  Non-Extensible

Version:

  20020904

Table CID 6034. Intended Use of CAD Output

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111150 | Presentation Required: Rendering device is expected to present
DCM | 111151 | Presentation Optional: Rendering device may present
DCM | 111152 | Not for Presentation: Rendering device expected not to present

## CID 6035 Composite Feature Relations

Type:

  Non-Extensible

Version:

  20020904

Table CID 6035. Composite Feature Relations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111153 | Target content items are related temporally
DCM | 111154 | Target content items are related spatially
DCM | 111155 | Target content items are related contra-laterally

## CID 6036 Scope of Feature

Type:

  Non-Extensible

Version:

  20020904

Table CID 6036. Scope of Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111156 | Feature detected on the only image
DCM | 111157 | Feature detected on only one of the images
DCM | 111158 | Feature detected on multiple images
DCM | 111159 | Feature detected on images from multiple modalities

## CID 6037 Mammography Quantitative Temporal Difference Type

Type:

  Extensible

Version:

  20020904

Table CID 6037. Mammography Quantitative Temporal Difference Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-017B1 | Difference in size
SRT | F-017B2 | Difference in opacity
SRT | F-017B3 | Difference in location
SRT | F-017B4 | Difference in spatial proximity
SRT | F-017B5 | Difference in number of calcifications

## CID 6038 Mammography Qualitative Temporal Difference Type

Type:

  Extensible

Version:

  20020904

Table CID 6038. Mammography Qualitative Temporal Difference Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-017B6 | Difference in shape
SRT | F-017B7 | Difference in margin
SRT | F-017B8 | Difference in symmetry

## CID 6039 Nipple Characteristic

Type:

  Extensible

Version:

  20020904

Table CID 6039. Nipple Characteristic

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | M-02000 | Normal shape
SRT | D7-90554 | Nipple retraction

## CID 6040 Non-lesion Object Type

Type:

  Extensible

Version:

  20091029

Table CID 6040. Non-Lesion Object Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-32475 | BB shot (Lead Pellet) | 102378009 | C0522706
SRT | A-32110 | Bullet | 86122002 | C0336699
SRT | A-11101 | Cardiac Pacemaker | 118378005 | C1289799
SRT | A-26800 | Catheter | 19923001 | C0085590
SRT | A-12062 | Clip | 77720000 | C0175722
SRT | A-10044 | Collimator | 129461008 | C0454169
SRT | A-10042 | Compression paddle | 129460009 | C1268544
SRT | C-B0300 | Contrast agent | 7140000 | C0009924
SRT | A-16016 | ID Plate | 129467007 | C1268548
SRT | A-04010 | Implant | 40388003 | C0021102
SRT | A-1016B | J Wire | 129463006 | C1268545
SRT | A-00D7B | Opaque Marker | 262301009 | C0445402
DCM | 111175 | Other Marker |  |
SRT | M-78060 | Scar tissue | 12402003 | C2004491
SRT | A-13600 | Staple | 56353002 | C0524724
SRT | A-13510 | Suture material | 2760009 | C0038969
DCM | 111176 | Unspecified |  |

### Note

1. This Context Group formerly included SNOMED code J-83250, which has been replaced with A-00D7B. See Annex J.

2. The use of (111176, DCM, "Unspecified") is explicitly permitted in this context group to allow for the communication of measurements of an object of unknown type using TID 4012 “Mammography CAD Non-Lesion”.

## CID 6041 Mammography Image Quality Finding

Type:

  Extensible

Version:

  20020904

Table CID 6041. Mammography Image Quality Finding

Coding Scheme Designator | Code Value | Code Meaning | Source
--- | --- | --- | ---
DCM | 111177 | View and Laterality Marker is missing | MQSA
DCM | 111178 | View and Laterality Marker does not have both view and laterality | MQCM 1999
DCM | 111179 | View and Laterality Marker does not have approved codes | MQCM 1999
DCM | 111180 | View and Laterality Marker is not near the axilla | MQCM 1999
DCM | 111181 | View and Laterality Marker overlaps breast tissue | MQCM 1999
DCM | 111182 | View and Laterality Marker is partially obscured | MQCM 1999
DCM | 111183 | View and Laterality Marker is incorrect |
DCM | 111184 | View and Laterality Marker is off image |
DCM | 111185 | Flash is not near edge of film | MQCM 1999
DCM | 111186 | Flash is illegible, does not fit, or is lopsided | MQSA
DCM | 111187 | Flash doesn't include patient name and additional patient id | MQCM 1999
DCM | 111188 | Flash doesn't include date of examination | MQCM 1999
DCM | 111189 | Flash doesn't include facility name and location | MQSA
DCM | 111190 | Flash doesn't include technologist identification | MQCM 1999
DCM | 111191 | Flash doesn't include cassette/screen/detector identification | MQCM 1999
DCM | 111192 | Flash doesn't include mammography unit identification | MQCM 1999
DCM | 111193 | Date sticker is missing | MQCM 1999
DCM | 111194 | Technical factors missing | MQCM 1999
DCM | 111195 | Collimation too close to breast | MQCM 1999
DCM | 111196 | Inadequate compression | MQCM 1999
DCM | 111197 | MLO Insufficient pectoral muscle | MQCM 1999
DCM | 111198 | MLO No fat is visualized posterior to fibroglandular tissues | MQCM 1999
DCM | 111199 | MLO Poor separation of deep and superficial breast tissues | MQCM 1999
DCM | 111200 | MLO Evidence of motion blur | MQCM 1999
DCM | 111201 | MLO Inframammary fold is not open | MQCM 1999
DCM | 111202 | CC Not all medial tissue visualized | MQCM 1999
DCM | 111203 | CC Nipple not centered on image | MQCM 1999
DCM | 111204 | CC Posterior nipple line does not measure within 1 cm of MLO | MQCM 1999
DCM | 111205 | Nipple not in profile |
DCM | 111206 | Insufficient implant displacement incorrect | MQCM 1999
DCM | 111208 | Grid artifact(s) |
DCM | 111209 | Positioning |
DCM | 111210 | Motion blur |
DCM | 111211 | Under exposed |
DCM | 111212 | Over exposed |
DCM | 111213 | No image |
DCM | 111214 | Detector artifact(s) |
DCM | 111215 | Artifact(s) other than grid or detector artifact |
DCM | 111216 | Mechanical failure |
DCM | 111217 | Electrical failure |
DCM | 111218 | Software failure |
DCM | 111219 | Inappropriate image processing |
DCM | 111220 | Other failure |
DCM | 111221 | Unknown failure |

## CID 6042 Status of Results

Type:

  Non-Extensible

Version:

  20020904

Table CID 6042. Status of Results

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111222 | Succeeded
DCM | 111223 | Partially Succeeded
DCM | 111224 | Failed
DCM | 111225 | Not Attempted

## CID 6043 Types of Mammography CAD Analysis

Type:

  Extensible

Version:

  20020904

Table CID 6043. Types of Mammography CAD Analysis

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P5-B3402 | Spatial collocation analysis | See Note 1
SRT | P5-B3404 | Spatial proximity analysis | See Note 2
SRT | P5-B3406 | Temporal correlation |
SRT | P5-B3408 | Image quality analysis |
SRT | P5-B3410 | Focal asymmetric density analysis |
SRT | P5-B3412 | Asymmetric breast tissue analysis |
SRT | P5-B3414 | Breast composition analysis |
DCM | 111233 | Individual Impression / Recommendation Analysis |
DCM | 111234 | Overall Impression / Recommendation Analysis |

### Note

1. Spatial Collocation Analysis is used to identify features that are the same or located in the same place.

2. Spatial Proximity Analysis is used to identify features that are related spatially, such as nipple retraction associated with a spiculated mass.

## CID 6044 Types of Image Quality Assessment

Type:

  Extensible

Version:

  20020904

Table CID 6044. Types of Image Quality Assessment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111235 | Unusable - Quality renders image unusable
DCM | 111236 | Usable - Does not meet the quality control standard
DCM | 111237 | Usable - Meets the quality control standard

## CID 6045 Mammography Types of Quality Control Standard

Type:

  Extensible

Version:

  20020904

Table CID 6045. Mammography Types of Quality Control Standard

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111238 | Mammography Quality Control Manual 1999, ACR
DCM | 111239 | Title 21 CFR Section 900, Subpart B
DCM | 111240 | Institutionally defined quality control standard

## CID 6046 Units of Follow-up Interval

Type:

  Extensible

Version:

  20020904

Table CID 6046. Units of Follow-Up Interval

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | d | day
UCUM | wk | week
UCUM | mo | month
UCUM | a | year

## CID 6047 CAD Processing and Findings Summary

Type:

  Non-Extensible

Version:

  20020904

Table CID 6047. CAD Processing and Findings Summary

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111241 | All algorithms succeeded; without findings
DCM | 111242 | All algorithms succeeded; with findings
DCM | 111243 | Not all algorithms succeeded; without findings
DCM | 111244 | Not all algorithms succeeded; with findings
DCM | 111245 | No algorithms succeeded; without findings

## CID 6048 CAD Operating Point Axis Label

Type:

  Extensible

Version:

  20060612

Table CID 6048. CAD Operating Point Axis Label

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111012 | Certainty of Finding
DCM | 111047 | Probability of cancer
DCM | 111086 | False Markers per Image
DCM | 111087 | False Markers per Case
DCM | 111088 | Case Sensitivity
DCM | 111089 | Lesion Sensitivity
DCM | 111090 | Case Specificity
DCM | 111091 | Image Specificity

## CID 6050 Breast Procedure Reported

Type:

  Extensible

Version:

  20090819

Table CID 6050. Breast Procedure Reported

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111408 | Film Screen Mammography
DCM | 111409 | Digital Mammography
SRT | P5-B8500 | Ultrasonography of breast
SRT | P5-0900D | MRI of breast
SRT | P1-48011 | Pre-biopsy localization of breast lesion
SRT | P1-48145 | Fine needle aspiration of breast
SRT | P1-48142 | Diagnostic aspiration of breast cyst
SRT | P1-48304 | Core needle biopsy of breast
SRT | P1-4830F | Breast - surgical biopsy
SRT | P5-40060 | Mammary ductogram
SRT | P5-0801C | CT of breast
SRT | P5-D0042 | Radionuclide localization of tumor, limited area
SRT | P5-40030 | Specimen radiography of breast
SRT | P2-4A000 | Examination of breast
DCM | 111410 | Surgical consult
DCM | 111411 | Mammography CAD
SRT | P1-65359 | Sentinel lymph node biopsy
SRT | P5-D0061 | Radioisotope scan of lymphatic system
DCM | 111123 | Marker placement
SRT | P1-05535 | Insertion of catheter

## CID 6051 Breast Procedure Reason

### Note

Some of these terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6051. Breast Procedure Reason

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6061 “Breast Imaging Procedure Modifiers”
DCM | 111415 | Additional evaluation requested from prior study
DCM | 111416 | Follow-up at short interval from prior study
DCM | 111417 | History of breast augmentation, asymptomatic
DCM | 111418 | Review of an outside study
DCM | 111402 | Clinical finding
SRT | P1-48830 | Reduction mammoplasty
SRT | P5-C0000 | Radiation therapy
SRT | P1-48840 | Augmentation mammoplasty
DCM | 111419 | Additional evaluation requested from abnormal screening exam
SRT | P5-C0610 | Brachytherapy
DCM | 111420 | History of benign breast biopsy
DCM | 111421 | Personal history of breast cancer with breast conservation therapy
DCM | 111124 | Personal history of breast cancer with mastectomy
DCM | 111125 | Known biopsy proven malignancy
SRT | G-03D3 | Personal history of breast cancer
DCM | 111590 | Recall for technical reasons
DCM | 111591 | Recall for imaging findings
DCM | 111592 | Recall for patient symptoms/ clinical findings

## CID 6052 Breast Imaging Report Section Title

Type:

  Extensible

Version:

  20040112

Table CID 6052. Breast Imaging Report Section Title

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111423 | Physical Examination Results
DCM | 111424 | Comparison to previous exams
DCM | 121070 | Findings
DCM | 121072 | Impressions
DCM | 121074 | Recommendations
DCM | 121076 | Conclusions
DCM | 121078 | Addendum
SRT | F-01710 | Breast composition
DCM | 111413 | Overall Assessment
DCM | 121058 | Procedure reported
DCM | 111401 | Reason for procedure

## CID 6053 Breast Imaging Report Elements

Type:

  Extensible

Version:

  20040112

Table CID 6053. Breast Imaging Report Elements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111423 | Physical Examination Results
DCM | 111424 | Comparison to previous exams
DCM | 121071 | Finding
DCM | 121073 | Impression
DCM | 121075 | Recommendation
DCM | 121077 | Conclusion
SRT | F-01710 | Breast composition
DCM | 111413 | Overall Assessment
DCM | 121058 | Procedure reported
DCM | 111401 | Reason for procedure

## CID 6054 Breast Imaging Findings

Type:

  Extensible

Version:

  20050110

Table CID 6054. Breast Imaging Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-8A084 | Breast normal
SRT | F-8A057 | Calcification of breast
SRT | A-04010 | Implant
Include CID 6016 “Mammography Composite Feature”
Include CID 6057 “Ductography Findings for Breast”
Include CID 6064 “Ultrasound Findings for Breast”

## CID 6055 Breast Clinical Finding Or Indicated Problem

### Note

Some of these terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6055. Breast Clinical Finding or Indicated Problem

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-207D7 | O/E - Breast lump palpated
SRT | D7-90565 | Bloody nipple discharge
DCM | 111478 | Non-bloody discharge (from nipple)
DCM | 111479 | Difficult physical/clinical examination
SRT | D7-90010 | Disorder of breast implant
SRT | F-0179A | Skin thickening of breast
SRT | F-01799 | Skin retraction of breast
SRT | D7-90560 | Peau d'orange surface of breast
SRT | F-8A09C | Nipple problem
SRT | R-20099 | O/E - axillary lymphadenopathy
SRT | F-8A030 | Breast pain
DCM | 111480 | Cancer elsewhere
SRT | D7-90530 | Breast lump
SRT | F-8A074 | Discoloration of skin of breast
SRT | F-01760 | Radiographic calcification finding
DCM | 111126 | Image detected mass
SRT | F-03753 | Nipple discharge symptom
SRT | F-4410C | Erythema
SRT | R-20681 | O/E - lymphadenopathy NOS
SRT | DF-00577 | Disseminated malignancy of unknown primary

## CID 6056 Associated Findings for Breast

### Note

These terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20040112

Table CID 6056. Associated Findings for Breast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D7-9002A | Breast hematoma
SRT | M-78280 | Surgical scar
SRT | D7-90554 | Nipple retraction
Include CID 6015 “Single Image Finding From BI-RADS®”

## CID 6057 Ductography Findings for Breast

### Note

These terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20040112

Table CID 6057. Ductography Findings for Breast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111287 | Normal breast tissue
DCM | 111425 | Intraluminal filling defect
SRT | D7-90370 | Mammary duct ectasia
DCM | 111426 | Multiple filling defect
DCM | 111427 | Abrupt duct termination
DCM | 111428 | Extravasation
DCM | 111429 | Duct narrowing
DCM | 111430 | Cyst fill

## CID 6058 Procedure Modifiers for Breast

Type:

  Extensible

Version:

  20050822

Table CID 6058. Procedure Modifiers for Breast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6059 “Breast Implant Types”
Include CID 6060 “Breast Biopsy Techniques”
Include CID 6061 “Breast Imaging Procedure Modifiers”
Include CID 12224 “Ultrasound Image Modes”

## CID 6059 Breast Implant Types

### Note

Some of these terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20040112

Table CID 6059. Breast Implant Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-04830 | Breast implant, type not specified
SRT | A-04831 | Silicone gel implant
DCM | 111481 | Saline implant
DCM | 111482 | Polyurethane implant
DCM | 111483 | Percutaneous silicone injection
DCM | 111484 | Combination implant
DCM | 111485 | Pre-pectoral implant
DCM | 111486 | Retro-pectoral implant

## CID 6060 Breast Biopsy Techniques

### Note

Some of these terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6060. Breast Biopsy Techniques

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P1-03005 | Lumpectomy
SRT | P1-43850 | Mastectomy
SRT | P1-4834A | Quadrantectomy of breast
SRT | P5-00032 | Diagnostic radiography, stereotactic localization
SRT | P5-B0700 | Ultrasonic guidance procedure
SRT | P5-40010 | Mammography
DCM | 111487 | Mammographic (crosshair)
DCM | 111488 | Mammographic (grid)
SRT | P1-03107 | Magnetic resonance imaging guided biopsy
SRT | P1-03106 | Computed tomography guided biopsy
DCM | 111489 | Palpation guided
DCM | 111490 | Vacuum assisted

### Note

In a prior version of this Context Group, the code P1-03115 was specified for the concept "Ultrasound guided biopsy". The use of this code is too restrictive, and its use in this context is deprecated. There is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use.

## CID 6061 Breast Imaging Procedure Modifiers

Type:

  Extensible

Version:

  20090819

Table CID 6061. Breast Imaging Procedure Modifiers

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-42453 | Screening
SRT | R-408C3 | Diagnostic
DCM | 111127 | Targeted
DCM | 111128 | Survey
DCM | 122505 | Calibration
DCM | 110002 | Quality Control
DCM | 111144 | Needle localization and biopsy
DCM | 111123 | Marker placement

## CID 6062 Interventional Procedure Complications

Type:

  Extensible

Version:

  20040112

Table CID 6062. Interventional Procedure Complications

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | DD-66A67 | Hemorrhage postprocedure
DCM | 111491 | Abnormal discharge
SRT | F-0147C | Hematoma - postoperative
SRT | D0-00165 | Weal
SRT | DD-67700 | Infection as complication of medical care
SRT | F-A2632 | Persistent pain following procedure
SRT | D2-80300 | Pneumothorax
SRT | D0-00058 | Rash
SRT | M-02570 | Swelling
SRT | F-A5581 | Vasovagal attack
DCM | 111492 | No complications

## CID 6063 Interventional Procedure Results

Type:

  Extensible

Version:

  20040112

Table CID 6063. Interventional Procedure Results

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A249 | Benign
SRT | R-41DDC | High risk tumor
SRT | G-A245 | Malignant
SRT | M-09024 | Insufficient sample
SRT | F-01E06 | Indeterminate result

## CID 6064 Ultrasound Findings for Breast

### Note

These terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6064. Ultrasound Findings for Breast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01BF8 | Ultrasound scan normal
SRT | D7-90360 | Simple cyst of breast
DCM | 111460 | Complex cyst
DCM | 111461 | Intracystic lesion
SRT | D7-90370 | Mammary duct ectasia
DCM | 111462 | Solid mass
SRT | T-C4000 | Lymph node
SRT | D7-90382 | Sebaceous cyst of skin of breast
DCM | 111129 | Clustered microcysts
DCM | 111130 | Complicated cyst
SRT | M-30400 | Foreign body

## CID 6065 Instrument Approach

Type:

  Extensible

Version:

  20040112

Table CID 6065. Instrument Approach

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A109 | Medial
SRT | G-A104 | Lateral
SRT | G-A116 | Superior
SRT | G-A115 | Inferior
DCM | 111432 | Inferolateral to superomedial
DCM | 111433 | Inferomedial to superolateral
DCM | 111434 | Superolateral to inferomedial
DCM | 111435 | Superomedial to inferolateral

## CID 6066 Target Confirmation

Type:

  Extensible

Version:

  20040112

Table CID 6066. Target Confirmation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111443 | Target contained in the specimen
DCM | 111444 | Target partially obtained in the specimen
DCM | 111445 | Target not in the specimen
DCM | 111446 | Calcifications seen in the core
DCM | 111447 | Lesion completely removed
DCM | 111448 | Lesion partially removed
DCM | 111449 | Fluid obtained

## CID 6067 Fluid Color

Type:

  Extensible

Version:

  20040112

Table CID 6067. Fluid Color

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A12B | White color
SRT | G-A11D | Yellow color
DCM | 111450 | Light brown color
SRT | G-A11E | Green color
SRT | G-A12D | Gray color
DCM | 111451 | Dark red color
DCM | 111452 | Dark brown color
SRT | R-4205B | Clear
SRT | G-A12E | Brown color
DCM | 111453 | Bright red color
DCM | 111454 | Blood tinged color
SRT | G-A12C | Black color

## CID 6068 Tumor Stages From AJCC

Type:

  Extensible

Version:

  20040112

Table CID 6068. Tumor Stages From AJCC

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111494 | Stage 0
DCM | 111495 | Stage I
DCM | 111496 | Stage IIA
DCM | 111497 | Stage IIB
DCM | 111498 | Stage IIIA
DCM | 111499 | Stage IIIB
DCM | 111500 | Stage IIIC
DCM | 111501 | Stage IV

## CID 6069 Nottingham Combined Histologic Grade

Type:

  Extensible

Version:

  20040112

Table CID 6069. Nottingham Combined Histologic Grade

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-02B9B | Nottingham Combined Grade cannot be determined
SRT | G-F616 | Nottingham Combined Grade I: 3-5 points
SRT | G-F617 | Nottingham Combined Grade II: 6-7 points
SRT | G-F618 | Nottingham Combined Grade III: 8-9 points

## CID 6070 Bloom-Richardson Histologic Grade

Type:

  Extensible

Version:

  20040112

Table CID 6070. Bloom-Richardson Histologic Grade

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-F211 | Grade 1: well differentiated
SRT | G-F212 | Grade 2: moderately differentiated
SRT | G-F213 | Grade 3: poorly differentiated
SRT | R-41DC5 | Grade 4: undifferentiated

## CID 6071 Histologic Grading Method

Type:

  Extensible

Version:

  20040112

Table CID 6071. Histologic Grading Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111502 | Bloom-Richardson Grade
SRT | R-00288 | Nottingham Combined Grade

## CID 6072 Breast Implant Findings

### Note

These terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20040112

Table CID 6072. Breast Implant Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111503 | Normal implants
DCM | 111504 | Asymmetric implants
DCM | 111505 | Calcified implant
DCM | 111506 | Distorted implant
DCM | 111507 | Silicone-laden lymph nodes
DCM | 111508 | Free silicone
DCM | 111509 | Herniated implant
SRT | DD-66544 | Rupture of breast implant
DCM | 111510 | Explantation

## CID 6080 Gynecological Hormones

### Note

Some of these terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6080. Gynecological Hormones

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-B1700 | Contraceptives
SRT | C-A0900 | Estrogen product
SRT | C-A1204 | Progesterone product
SRT | C-781E0 | Tamoxifen
DCM | 111542 | Unspecified gynecological hormone
SRT | C-A0005 | Raloxifene
SRT | F-61B21 | Anastrozole

## CID 6081 Breast Cancer Risk Factors

### Note

Some of these terms were obtained from BI-RADS®

Type:

  Extensible

Version:

  20040112

Table CID 6081. Breast Cancer Risk Factors

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111550 | Personal breast cancer history
DCM | 111551 | History of endometrial cancer
DCM | 111552 | History of ovarian cancer
DCM | 111553 | History of high risk lesion on previous biopsy
DCM | 111554 | Post menopausal patient
SRT | F-84430 | Nulliparous
DCM | 111555 | Late child bearing (after 30)
DCM | 111556 | BRCA1 breast cancer gene
DCM | 111557 | BRCA2 breast cancer gene
DCM | 111558 | BRCA3 breast cancer gene
SRT | G-0325 | Family history of breast cancer
DCM | 111559 | Weak family history of breast cancer
DCM | 111560 | Intermediate family history of breast cancer
DCM | 111561 | Very strong family history of breast cancer
DCM | 111562 | Family history of prostate cancer
DCM | 111563 | Family history unknown
SRT | R-207AD | No family history of breast carcinoma

## CID 6082 Gynecological Procedures

Type:

  Extensible

Version:

  20040112

Table CID 6082. Gynecological Procedures

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P0-05CCA | Endometrial biopsy
SRT | P1-8330D | Hysterectomy
SRT | P1-03151 | Dilation and curettage

## CID 6083 Procedures for Breast

Type:

  Extensible

Version:

  20040112

Table CID 6083. Procedures for Breast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111564 | Nipple discharge cytology
Include CID 6050 “Breast Procedure Reported”
Include CID 6084 “Mammoplasty Procedures”
Include CID 6085 “Therapies for Breast”

## CID 6084 Mammoplasty Procedures

Type:

  Extensible

Version:

  20040112

Table CID 6084. Mammoplasty Procedures

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P1-48501 | Breast implantation
SRT | P1-48830 | Reduction mammoplasty
SRT | P1-48820 | Breast reconstruction
SRT | P1-48520 | Removal of breast implant

## CID 6085 Therapies for Breast

Type:

  Extensible

Version:

  20040112

Table CID 6085. Therapies for Breast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P0-0058E | Chemotherapy
SRT | P5-C0000 | Radiation therapy
SRT | P0-007AC | Hormone therapy
SRT | P1-67D40 | Bone marrow transplant

## CID 6086 Menopausal Phase

Type:

  Extensible

Version:

  20040112

Table CID 6086. Menopausal Phase

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-41FFF | Before menopause
SRT | R-422A5 | During menopause
SRT | R-410C3 | After menopause
SRT | D7-76202 | Postsurgical menopause
SRT | D7-76200 | Artificial menopause state

## CID 6087 General Risk Factors

This context group collects risk factor terms from specialized risk factor context groups into one aggregate list for general purpose use.

Type:

  Extensible

Version:

  20040112

Table CID 6087. General Risk Factors

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6081 “Breast Cancer Risk Factors”
Include CID 6088 “OB-GYN Maternal Risk Factors”

## CID 6088 OB-GYN Maternal Risk Factors

Type:

  Extensible

Version:

  20040112

Table CID 6088. OB-GYN Maternal Risk Factors

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-023F | History of - diabetes mellitus
SRT | G-0269 | History of - hypertension
SRT | G-0244 | History of - obesity
SRT | G-02D0 | History of - regular medication
SRT | G-0338 | History of substance abuse
SRT | G-0335 | History of - cardiovascular disease
DCM | 111565 | Uterine malformations
SRT | G-0304 | History of - ectopic pregnancy
DCM | 111566 | Spontaneous Abortion
DCM | 111567 | Gynecologic condition
DCM | 111568 | Gynecologic surgery
SRT | G-031E | History of - eclampsia
SRT | G-031F | History of - severe pre-eclampsia
DCM | 111569 | Previous LBW or IUGR birth
DCM | 111570 | Previous fetal malformation/syndrome
SRT | G-0305 | History of - premature delivery
DCM | 111571 | Previous RH negative or blood dyscrasia at birth
SRT | G-0319 | History of infertility
DCM | 111572 | History of multiple fetuses
SRT | D8-20100 | Multiple pregnancy
DCM | 111573 | Current pregnancy, known or suspected malformations/syndromes
DCM | 111574 | Family history, fetal malformation/syndrome

## CID 6089 Substances

Type:

  Extensible

Version:

  20040112

Table CID 6089. Substances

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-21005 | Ethyl alcohol
SRT | F-618FF | Amphetamine
SRT | F-6166C | Marijuana derivative
SRT | F-61C76 | Cocaine
SRT | F-61AC4 | Heroin
SRT | C-63A10 | Lysergic acid diethlyamide
SRT | F-6169A | Mescaline
SRT | C-6A180 | Phencyclidine
SRT | F-61A95 | Methadone
SRT | F-618D7 | Morphine
SRT | F-618FE | Methlyphenidate
SRT | C-F3310 | Chewing tobacco
SRT | C-F3302 | Cigarette smoking tobacco
SRT | F-61117 | Caffeine

## CID 6090 Relative Usage, Exposure Amount

Type:

  Extensible

Version:

  20050822

Table CID 6090. Relative Usage, Exposure Amount

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111575 | High
DCM | 111576 | Medium
DCM | 111577 | Low
DCM | 111587 | No known exposure

## CID 6091 Relative Frequency of Event Values

Type:

  Extensible

Version:

  20040112

Table CID 6091. Relative Frequency of Event Values

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-40377 | Continuous
SRT | G-7154 | Frequent
SRT | R-40365 | Mid-frequency
SRT | G-7155 | Infrequent
SRT | R-40B16 | As required
SRT | R-4112F | Single event

## CID 6092 Quantitative Concepts for Usage, Exposure

Type:

  Extensible

Version:

  20040112

Table CID 6092. Quantitative Concepts for Usage, Exposure

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-C0B7 | Dosage
DCM | 111578 | Dose frequency
DCM | 111579 | Rate of exposure
DCM | 111580 | Volume of use

## CID 6093 Qualitative Concepts for Usage, Exposure Amount

Type:

  Extensible

Version:

  20040112

Table CID 6093. Qualitative Concepts for Usage, Exposure Amount

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111581 | Relative dose amount
DCM | 111582 | Relative amount of exposure
DCM | 111583 | Relative amount of use

## CID 6094 Qualitative Concepts for Usage, Exposure Frequency

Type:

  Extensible

Version:

  20040112

Table CID 6094. Qualitative Concepts for Usage, Exposure Frequency

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111584 | Relative dose frequency
DCM | 111585 | Relative frequency of exposure
DCM | 111586 | Relative frequency of use

## CID 6095 Numeric Properties of Procedures

Type:

  Extensible

Version:

  20040112

Table CID 6095. Numeric Properties of Procedures

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111465 | Needle Gauge
DCM | 111467 | Needle Length

## CID 6096 Pregnancy Status

Type:

  Extensible

Version:

  20040112

Table CID 6096. Pregnancy Status

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-81890 | not pregnant
SRT | F-84094 | possible pregnancy
SRT | F-84000 | patient currently pregnant
SRT | R-41198 | Unknown

## CID 6097 Side of Family

Type:

  Extensible

Version:

  20040112

Table CID 6097. Side of Family

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111541 | Maternal
SRT | R-40333 | Paternal

## CID 6100 Chest Component Categories

Type:

  Extensible

Version:

  20030108

Table CID 6100. Chest Component Categories

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-28000 | Lung
DCM | 112052 | Bronchovascular
SRT | T-29000 | Pleural structure
SRT | T-D3300 | Mediastinum
SRT | T-32000 | Heart
DCM | 112053 | Osseous
SRT | T-4000E | Systemic vascular structure
SRT | R-420AE | Muscular

## CID 6101 Chest Finding Or Feature

Type:

  Extensible

Version:

  20030108

Table CID 6101. Chest Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112061 | Abnormal lines (1D)
DCM | 112033 | Abnormal opacity
DCM | 112062 | Abnormal lucency
DCM | 112063 | Abnormal calcifications
DCM | 112064 | Abnormal texture
DCM | 112005 | Radiographic anatomy
DCM | 111102 | Non-lesion
DCM | 111101 | Image quality
DCM | 111099 | Selected region

## CID 6102 Chest Finding Or Feature Modifier

Type:

  Extensible

Version:

  20030108

Table CID 6102. Chest Finding or Feature Modifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6103 “Abnormal Lines Finding Or Feature”
Include CID 6104 “Abnormal Opacity Finding Or Feature”
Include CID 6105 “Abnormal Lucency Finding Or Feature”
Include CID 6106 “Abnormal Texture Finding Or Feature”
Include CID 6109 “Radiographic Anatomy Finding Or Feature”
Include CID 6138 “Chest Non-lesion Object Type”

## CID 6103 Abnormal Lines Finding Or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6103. Abnormal Lines Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112065 | Reticulonodular pattern
DCM | 112104 | Air-fluid level
DCM | 112105 | Corona radiata
DCM | 112106 | Honeycomb pattern
DCM | 112107 | Fleischner's line(s)
DCM | 112108 | Intralobular lines
DCM | 112109 | Kerley A line
DCM | 112110 | Kerley B line
DCM | 112111 | Kerley C lines
DCM | 112112 | Parenchymal band
SRT | D2-60302 | Plate-like atelectasis
DCM | 112113 | Reticular pattern
DCM | 112114 | Septal line(s)
DCM | 112115 | Subpleural line
DCM | 112116 | Tramline shadow
DCM | 112117 | Tubular shadow

## CID 6104 Abnormal Opacity Finding Or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6104. Abnormal Opacity Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112066 | Beaded septum sign
DCM | 112067 | Nodular pattern
DCM | 112059 | Primary complex
DCM | 112068 | Pseudoplaque
DCM | 112065 | Reticulonodular pattern
DCM | 112069 | Signet-ring sign
DCM | 112004 | Abnormal interstitial pattern
SRT | F-20172 | Coin lesion
DCM | 112118 | Density
DCM | 112119 | Dependent opacity
DCM | 112120 | Ground glass opacity
DCM | 112121 | Infiltrate
SRT | M-03000 | Mass
DCM | 112122 | Micronodule
SRT | M-03010 | Nodule
DCM | 112001 | Opacity
DCM | 112123 | Phantom tumor (pseudotumor)
DCM | 112124 | Shadow
DCM | 112125 | Small irregular opacities
DCM | 112126 | Small rounded opacities
DCM | 112127 | Tree-in-bud sign
SRT | D3-40230 | Pulmonary embolism

## CID 6105 Abnormal Lucency Finding Or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6105. Abnormal Lucency Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112070 | Air bronchiologram
DCM | 112071 | Air bronchogram
DCM | 112072 | Air crescent
SRT | F-20240 | Air-trapping
DCM | 112073 | Halo sign
SRT | D2-81180 | Pneumomediastinum
SRT | D2-80300 | Pneumothorax

## CID 6106 Abnormal Texture Finding Or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6106. Abnormal Texture Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112067 | Nodular pattern
DCM | 112065 | Reticulonodular pattern
DCM | 112004 | Abnormal interstitial pattern
DCM | 112128 | Granular pattern
DCM | 112106 | Honeycomb pattern
DCM | 112129 | Miliary pattern
DCM | 112130 | Mosaic pattern
DCM | 112113 | Reticular pattern
DCM | 112125 | Small irregular opacities

## CID 6107 Width Descriptor

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6107. Width Descriptor

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-40750 | Enlarged
SRT | R-41727 | Narrow
DCM | 112077 | Vasoconstriction
DCM | 112078 | Vasodilation

## CID 6108 Chest Anatomic Structure Abnormal Distribution

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6108. Chest Anatomic Structure Abnormal Distribution

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-20240 | Air-trapping
DCM | 112079 | Architectural distortion
DCM | 112080 | Mosaic perfusion
DCM | 112060 | Oligemia
DCM | 112081 | Pleonemia

## CID 6109 Radiographic Anatomy Finding Or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6109. Radiographic Anatomy Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6110 “Lung Anatomy Finding Or Feature”
Include CID 6111 “Bronchovascular Anatomy Finding Or Feature”
Include CID 6112 “Pleura Anatomy Finding Or Feature”
Include CID 6113 “Mediastinum Anatomy Finding Or Feature”
Include CID 6114 “Osseous Anatomy Finding Or Feature”
Include CID 6116 “Muscular Anatomy”
Include CID 6117 “Vascular Anatomy”
DCM | 112082 | Interface
DCM | 112083 | Line
DCM | 112084 | Lucency

## CID 6110 Lung Anatomy Finding Or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6110. Lung Anatomy Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-28770 | Lobe of lung | 31094006 | C0225752
DCM | 112085 | Midlung window |  |
DCM | 112054 | Secondary pulmonary lobule |  |
SRT | T-280D0 | Segment of lung | 72674008 | C0225705

## CID 6111 Bronchovascular Anatomy Finding Or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6111. Bronchovascular Anatomy Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-20001 | Airway structure | 89187006 | C0458827
SRT | T-26000 | Bronchus | 955009 | C0006255
SRT | T-25201 | Carina | 28700002 | C0225594
DCM | 112086 | Carina angle |  |
DCM | 112087 | Centrilobular structures |  |
SRT | T-28080 | Hilum of lung | 46750007 | C0225701

## CID 6112 Pleura Anatomy Finding Or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6112. Pleura Anatomy Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 112088 | Anterior junction line |  |
SRT | T-D051D | Fissure of lung | 278983006 | C0458078
DCM | 112089 | Posterior junction line |  |

## CID 6113 Mediastinum Anatomy Finding Or Feature

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20130617

Table CID 6113. Mediastinum Anatomy Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-15420 | Acromioclavicular Joint | 85856004 | C0001208
SRT | T-35400 | Aortic Valve | 34202007 | C0003501
SRT | D4-31220 | Atrial Septal Defect | 70142008 | C0018817
SRT | T-32100 | Atrium | 59652004 | C0018792
SRT | T-18774 | Axillary Fascia | 368536000 | C0225236
DCM | 112090 | Azygoesophageal recess interface |  |
SRT | T-25201 | Carina | 28700002 | C0225594
SRT | T-B4000 | Carotid Body | 51345006 | C0007277
SRT | T-11240 | Costal Cartilage | 50016007 | C0222787
SRT | T-D3412 | Esophageal Hiatus | 280062008 | C0230160
SRT | T-56000 | Esophagus | 32849002 | C0014876
SRT | T-D0634 | Fascial layer | 120576005 | C1268198
SRT | T-32000 | Heart | 80891009 | C0018787
DCM | 112095 | Hiatus |  |
SRT | T-26500 | Left main bronchus | 75245000 | C0225630
SRT | T-42370 | Ligamentum arteriosum | 2160002 | C0226023
SRT | T-C4000 | Lymph node | 59441001 | C0024204
SRT | T-35300 | Mitral Valve | 91134007 | C0026264
DCM | 112091 | Paraspinal line |  |
DCM | 112092 | Posterior tracheal stripe |  |
SRT | T-35200 | Pulmonary valve | 39057004 | C0034086
SRT | T-26100 | Right main bronchus | 70074004 | C0225608
DCM | 112093 | Right tracheal stripe |  |
DCM | 112094 | Stripe |  |
SRT | T-C6510 | Thoracic Duct | 1732005 | C0039979
SRT | T-C8000 | Thymus Gland | 9875009 | C0040113
SRT | T-B6000 | Thyroid | 69748006 | C0040132
SRT | T-25000 | Trachea | 44567001 | C0040578
SRT | T-14171 | Trapezius muscle | 31764008 | C0224361
SRT | T-35100 | Tricuspid Valve | 46030003 | C0040960
SRT | T-32400 | Ventricle | 21814001 | C0018827

## CID 6114 Osseous Anatomy Finding Or Feature

Type:

  Extensible

Version:

  20030108

Table CID 6114. Osseous Anatomy Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-12310 | Clavicle | 51299004 | C0008913
SRT | T-12410 | Humerus | 85050009 | C0020164
SRT | T-11300 | Rib | 113197003 | C0035561
SRT | T-12280 | Scapula | 79601000 | C0036277
SRT | T-11500 | Spine | 44300000 | C1267072
SRT | T-11210 | Sternum | 56873002 | C0038293
SRT | T-11510 | Vertebra | 51282000 | C0549207

## CID 6115 Osseous Anatomy Modifiers

Type:

  Extensible

Version:

  20030108

Table CID 6115. Osseous Anatomy Modifiers

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-12281 | Acromion process of scapula | 31934006 | C0001209
SRT | T-11307 | Angle of rib | 14510004 | C0222812
SRT | T-11511 | Arch of vertebra | 40265002 | C0223076
SRT | T-11220 | Body of sternum | 52509009 | C0222771
SRT | T-11219 | Clavicular notch of sternum | 75319007 | C0222770
SRT | T-12282 | Coracoid process of scapula | 8931003 | C0223626
SRT | T-11308 | Costal groove | 17399006 | C0222813
SRT | T-12287 | Dorsal aspect of scapula | 51698000 | C0223631
SRT | T-1228A | Glenoid cavity of scapula | 46385009 | C1261046
SRT | T-11301 | Head of rib | 12872006 | C0222806
SRT | T-116EF | Inferior articular facet of axis | 181901007 | C0223115
SRT | T-1153F | Inferior articular process of vertebra | 317766009 | C0223083
SRT | T-11514 | Lamina of vertebra | 89340005 | C0223079
SRT | T-11211 | Manubrium of sternum | 37285002 | C0024764
SRT | T-11303 | Neck of rib | 72184008 | C0222808
SRT | T-D2236 | Pectoral girdle | 34772009 | C0427245
SRT | T-11515 | Pedicle of vertebra | 78972004 | C0223080
DCM | 112096 | Rib Scalene Tubercle |  |
DCM | 112101 | Scapular Infraspinatus Fossa |  |
DCM | 112099 | Scapular Spine |  |
DCM | 112100 | Scapular Supraspinatus Fossa |  |
SRT | T-11309 | Shaft of rib | 41601005 | C0448161
SRT | T-11512 | Spinous process of vertebra | 55678000 | C0223077
SRT | T-11221 | Sternal angle | 44612009 | C0222772
DCM | 112098 | Subscapular Fossa |  |
SRT | T-116EE | Superior articular facet of axis | 181900008 | C0223114
SRT | T-1153E | Superior articular process of vertebra | 317665004 | C0223082
SRT | T-11218 | Suprasternal notch | 26493002 | C0222769
SRT | T-11513 | Transverse process or vertebra | 73400003 | C0223078
SRT | T-11304 | Tubercle of rib | 113198008 | C0222809
SRT | T-1151F | Vertebral canal | 61853006 | C0037922
SRT | T-11531 | Vertebral foramen | 280734009 | C0459720
DCM | 112097 | Vertebral Intervertebral Notch |  |
SRT | T-11227 | Xiphoid process of sternum | 20298003 | C0043356

## CID 6116 Muscular Anatomy

Type:

  Extensible

Version:

  20130617

Table CID 6116. Muscular Anatomy

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-35020 | Chordae tendineae cordis | 102298001 | C0008484
SRT | T-13660 | Deltoid muscle | 35259002 | C0224234
SRT | T-D3400 | Diaphragm | 5798000 | C0011980
SRT | T-14020 | Erector spinae muscle | 44947003 | C0224301
SRT | T-1416B | External intercostal muscle | 244878003 | C0224353
SRT | T-14030 | Iliocostalis muscle | 57651003 | C0224302
SRT | T-13620 | Infraspinatus muscle | 72573008 | C0584882
SRT | T-14165 | Innermost intercostal muscles | 24062007 | C0224357
SRT | T-32150 | Interatrial septum | 58095006 | C0225836
SRT | T-14183 | Internal intercostal muscle | 244890001 | C0224355
SRT | T-32410 | Interventricular septum | 589001 | C0225870
SRT | T-14172 | Latissimus dorsi muscle | 15665001 | C0224362
SRT | T-14150 | Levatores costarum muscles | 73930003 | C1744586
SRT | T-14040 | Longissimus muscle | 88340001 | C0224306
SRT | T-14110 | Pectoralis major muscle | 60005003 | C0585574
SRT | T-14120 | Pectoralis minor muscle | 18686000 | C0224347
SRT | T-13450 | Scalenous anterior muscle | 50755001 | C0224173
SRT | T-14140 | Serratus anterior muscle | 18346003 | C0224349
SRT | T-14050 | Spinalis muscle | 4317002 | C0224310
SRT | T-13310 | Sternocleidomastoid muscle | 22823000 | C0224153
SRT | T-14166 | Subcostal muscle | 64658001 | C0224358
SRT | T-13650 | Subscapularis muscle | 90588001 | C0584884
SRT | T-13610 | Supraspinatus muscle | 6423006 | C0584869
SRT | T-13640 | Teres major muscle | 1193009 | C0224232
SRT | T-13630 | Teres minor muscle | 51159009 | C0224231
SRT | T-32423 | Trabeculae carnae | 118755002 | C0502348
SRT | T-141A5 | Transversus thoracis | 244933000 | C0224359
SRT | T-14171 | Trapezius muscle | 31764008 | C0224361

## CID 6117 Vascular Anatomy

Type:

  Extensible

Version:

  20130617

Table CID 6117. Vascular Anatomy

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
Include CID 3015 “Coronary Arteries”
SRT | T-42300 | Aortic arch | 57034009 | C0003489
SRT | T-42310 | Aortic isthmus | 88593004 | C0226019
DCM | 112102 | Aortic knob |  |
DCM | 112103 | Arch of the Azygos vein |  |
SRT | T-42100 | Ascending aorta | 54247002 | C0003956
SRT | T-47100 | Axillary Artery | 67937003 | C0004455
SRT | T-49110 | Axillary vein | 68705008 | C0004456
SRT | T-48340 | Azygos vein | 72107004 | C0004526
SRT | T-47160 | Brachial artery | 17137000 | C0006087
SRT | T-A9090 | Brachial plexus | 36582005 | C0006090
SRT | T-46010 | Brachiocephalic trunk | 12691009 | C0006094
SRT | T-48620 | Brachiocephalic vein | 8887007 | C0006095
SRT | T-46310 | Bronchial artery | 64468002 | C0006257
SRT | T-45100 | Common carotid artery | 32062004 | C0162859
SRT | T-46180 | Costocervical trunk | 3159004 | C0226273
SRT | T-D0765 | Descending aorta | 281130003 | C0011666
SRT | T-461A0 | Dorsal scapular artery | 91732003 | C0500583
SRT | T-4630D | Esophageal artery | 206034008 | C0226294
SRT | T-46940 | Inferior phrenic artery | 29660000 | C0226406
SRT | T-48710 | Inferior vena cava | 64131007 | C0042458
SRT | T-D305A | Intercostal artery | 281134007 | C0459917
SRT | T-48170 | Internal jugular vein | 12123001 | C0226550
SRT | T-46200 | Internal thoracic artery | 69327007 | C0226276
SRT | T-46210 | Pericardiophrenic Artery | 3924000 | C0226287
SRT | T-44000 | Pulmonary artery | 81040000 | C0034052
SRT | T-44100 | Pulmonary trunk | 45341000 | C0034052
SRT | T-48581 | Pulmonary vein | 122972007 | C0034090
SRT | T-46100 | Subclavian artery | 36765005 | C0038530
SRT | T-48330 | Subclavian vein | 9454009 | C0038532
SRT | T-46350 | Superior phrenic artery | 38991005 | C0226295
SRT | T-48610 | Superior vena cava | 48345005 | C0042459
SRT | T-46130 | Thyrocervical trunk | 6538005 | C0226263
SRT | T-45700 | Vertebral artery | 85234005 | C0042559

### Note

In a prior version of this Context Group the code T-48500 rather than T-48581 was defined for the concept Pulmonary Vein; this was inconsistent with the DICOM approach of selecting the "structure of" rather than "entire" concept. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 6118 Size Descriptor

Type:

  Extensible

Version:

  20030108

Table CID 6118. Size Descriptor

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112131 | Extremely small
DCM | 112132 | Very small
SRT | R-404A8 | Small
SRT | R-404A9 | Medium
SRT | R-404AA | Large
SRT | R-40750 | Enlarged
DCM | 112133 | Too small

## CID 6119 Chest Border Shape

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6119. Chest Border Shape

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | M-02100 | Round shape
DCM | 112134 | Elliptic
SRT | G-A402 | Irregular
DCM | 112135 | Lobulated
DCM | 112136 | Spiculated

## CID 6120 Chest Border Definition

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6120. Chest Border Definition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-40771 | Well defined
DCM | 112137 | Sharply defined
SRT | R-428E7 | Poorly defined
DCM | 112138 | Distinctly defined
DCM | 112139 | Well demarcated
DCM | 112140 | Sharply demarcated
DCM | 112141 | Poorly demarcated
DCM | 112142 | Circumscribed

## CID 6121 Chest Orientation Descriptor

Type:

  Extensible

Version:

  20030108

Table CID 6121. Chest Orientation Descriptor

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A142 | Horizontal
SRT | G-A144 | Vertical
SRT | G-A472 | Oblique

## CID 6122 Chest Content Descriptor

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6122. Chest Content Descriptor

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112143 | Air
SRT | T-D008A | Fat
DCM | 112144 | Soft tissue
DCM | 112145 | Calcium
SRT | M-30400 | Foreign material (iodized oil, mercury,talc)

## CID 6123 Chest Opacity Descriptor

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6123. Chest Opacity Descriptor

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112146 | Acinar
DCM | 112147 | Air space
DCM | 112148 | Fibronodular
DCM | 112149 | Fluffy
DCM | 112150 | Linear
DCM | 112151 | Profusion
DCM | 112152 | Silhouette sign

## CID 6124 Location in Chest

Type:

  Extensible

Version:

  20030108

Table CID 6124. Location in Chest

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6125 “General Chest Location”
Include CID 6126 “Location in Lung”
Include CID 6127 “Segment Location in Lung”

## CID 6125 General Chest Location

Type:

  Extensible

Version:

  20030108

Table CID 6125. General Chest Location

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A110 | Central
SRT | G-A111 | Peripheral
SRT | G-A122 | Apical
SRT | G-A123 | Basal

## CID 6126 Location in Lung

Type:

  Extensible

Version:

  20030108

Table CID 6126. Location in Lung

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-D3208 | Upper zone of lung
SRT | T-D3209 | Middle zone of lung
SRT | T-D320A | Lower zone of lung
SRT | T-28820 | Upper lobe of lung
SRT | T-28825 | Middle lobe of lung
SRT | T-28830 | Lower lobe of lung
DCM | 112153 | Subpleural

## CID 6127 Segment Location in Lung

Type:

  Extensible

Version:

  20030108

Table CID 6127. Segment Location in Lung

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-28230 | Anterior segment of right upper lobe
SRT | T-28630 | Anterior segment of left upper lobe
SRT | T-28220 | Posterior segment of right upper lobe

## CID 6128 Chest Distribution Descriptor

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6128. Chest Distribution Descriptor

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112154 | Bat's wing distribution
DCM | 112155 | Butterfly distribution
DCM | 112156 | Centrilobular
DCM | 112157 | Coalescent
SRT | G-A321 | Diffuse
SRT | M-020FA | Discoid
SRT | G-A324 | Disseminated
SRT | G-A351 | Focal
SRT | G-A366 | Generalized
DCM | 112158 | Lobar
SRT | G-A443 | Multifocal
SRT | G-A137 | Segmental
SRT | G-A572 | Systemic

## CID 6129 Chest Site Involvement

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6129. Chest Site Involvement

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-28000 | Lung
SRT | T-D3300 | Mediastinum
DCM | 112158 | Lobar
SRT | T-1A007 | Interstitial tissue
SRT | R-40939 | Bronchial
SRT | T-28080 | Hilum of lung
SRT | T-42000 | Aorta
SRT | T-29000 | Pleural structure
SRT | T-D3050 | Chest wall
SRT | T-D4001 | Upper abdomen

## CID 6130 Severity Descriptor

Type:

  Extensible

Version:

  20030108

Table CID 6130. Severity Descriptor

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-404FA | Mild
SRT | G-A002 | Moderate
SRT | G-A003 | Severe
SRT | G-A231 | Acute
SRT | G-A270 | Chronic
DCM | 112159 | Hyper-acute
SRT | G-A561 | Subacute

## CID 6131 Chest Texture Descriptor

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6131. Chest Texture Descriptor

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112160 | Homogeneous (uniform opacity)
DCM | 112161 | Inhomogeneous

## CID 6132 Chest Calcification Descriptor

### Note

Original source of terms is [Fraser and Pare].

Type:

  Extensible

Version:

  20030108

Table CID 6132. Chest Calcification Descriptor

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01763 | Eggshell calcification
SRT | F-01761 | Coarse (popcorn-like) calcification
DCM | 112162 | Target
SRT | G-A405 | Laminated
DCM | 112163 | Fibrocalcific
DCM | 112164 | Flocculent
SRT | R-403A7 | Nodular
SRT | F-12100 | Ossification

## CID 6133 Chest Quantitative Temporal Difference Type

Type:

  Extensible

Version:

  20100604

Table CID 6133. Chest Quantitative Temporal Difference Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-05173 | Difference in size
SRT | F-05179 | Difference in location

## CID 6134 Chest Qualitative Temporal Difference Type

Type:

  Extensible

Version:

  20090717

Table CID 6134. Chest Qualitative Temporal Difference Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-0517E | Difference in border shape
SRT | F-05166 | Difference in border definition
SRT | F-0516C | Difference in distribution
SRT | F-05170 | Difference in site involvement
SRT | F-05167 | Difference in substance
SRT | F-0516A | Difference in Texture
SRT | F-01722 | Finding partially removed
SRT | F-01723 | No significant changes in the finding
SRT | M-02520 | Increase in size
SRT | M-02530 | Decrease in size
SRT | F-01728 | Less defined
SRT | F-01729 | More defined

## CID 6135 Image Quality Finding

Type:

  Extensible

Version:

  20090402

Table CID 6135. Image Quality Finding

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111208 | Grid artifact(s)
DCM | 111209 | Positioning
DCM | 111210 | Motion blur
DCM | 111211 | Under exposed
DCM | 111212 | Over exposed
DCM | 111213 | No image
DCM | 111214 | Detector artifact(s)
DCM | 111215 | Artifact(s) other than grid or detector artifact
DCM | 111216 | Mechanical failure
DCM | 111217 | Electrical failure
DCM | 111218 | Software failure
DCM | 111219 | Inappropriate image processing
DCM | 111220 | Other failure
DCM | 111221 | Unknown failure
RADLEX | RID11327 | Beam-hardening artifact

## CID 6136 Chest Types of Quality Control Standard

Type:

  Extensible

Version:

  20030108

Table CID 6136. Chest Types of Quality Control Standard

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112035 | Performance of Pediatric and Adult Chest Radiography, ACR
DCM | 112036 | ACR Position Statement
DCM | 111240 | Institutionally defined quality control standard
DCM | 112184 | Performance of Pediatric and Adult Thoracic CT
DCM | 112185 | Performance of CT for Detection of Pulmonary Embolism in Adults
DCM | 112186 | Performance of High-Resolution CT of the Lungs in Adults

## CID 6137 Types of CAD Analysis

Type:

  Extensible

Version:

  20030108

Table CID 6137. Types of CAD Analysis

Coding Scheme Designator | Code Value | Code Meaning |
--- | --- | --- | ---
SRT | P5-B3402 | Spatial collocation analysis | See Note 1
SRT | P5-B3404 | Spatial proximity analysis | See Note 2
SRT | P5-B3406 | Temporal correlation |
SRT | P5-B3408 | Image quality analysis |

### Note

1. Spatial Co-location Analysis is used to identify features that are the same or located in the same place.

2. Spatial Proximity Analysis is used to identify different features that are related spatially.

## CID 6138 Chest Non-lesion Object Type

Type:

  Extensible

Version:

  20120822

Table CID 6138. Chest Non-Lesion Object Type

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-32110 | Bullet | 86122002 | C0336699
SRT | A-11100 | Cardiac Pacemaker | 14106009 | C0030163
SRT | A-040CB | Cardiac pacemaker lead | 360129009 | C1283151
SRT | A-26800 | Catheter | 19923001 | C0085590
DCM | 112174 | Central line |  |
SRT | A-12210 | Cervical collar | 63562005 | C0175751
DCM | 112173 | Chest tube |  |
DCM | 112178 | Coin |  |
SRT | A-25350 | Endotracheal tube | 26412008 | C0336630
SRT | A-26430 | Feeding tube | 25062003 | C2945625
DCM | 112171 | Fiducial mark |  |
SRT | A-04110 | Heart valve prosthesis | 25510005 | C0018825
SRT | A-26434 | Jejunostomy tube | 126065006 | C0879216
SRT | A-61000 | Jewelry | 80919006 | C0336902
DCM | 112175 | Kidney stent |  |
SRT | A-30360 | Needle | 79068005 | C0027551
DCM | 112177 | Nipple ring |  |
DCM | 112176 | Pancreatic stent |  |
SRT | A-12024 | Pin | 77444004 | C0175718
DCM | 112172 | Portacath |  |
SRT | A-04000 | Prosthesis | 53350007 | C0175649
SRT | M-78060 | Scar tissue | 12402003 | C2004491
SRT | A-13600 | Staple | 56353002 | C0524724
SRT | A-13510 | Suture material | 2760009 | C0038969
SRT | P1-26100 | Tracheotomy | 48387007 | C0040590
SRT | A-11C08 | Ureteric stent | 286558002 | C0183518
SRT | A-14611 | Vena cava filter | 257409000 | C0080306

### Note

The use of (111176, DCM, "Unspecified") was previously included in this context group but was removed since it does not make sense to have Chest CAD detections of an unknown type (was using TID 4015 “CAD Detections Performed” invoked by TID 4100 “Chest CAD Document Root”, TID 4102 “Chest CAD Composite Feature” and TID 4104 “Chest CAD Single Image Finding” of an unknown type).

## CID 6139 Non-lesion Modifiers

Type:

  Extensible

Version:

  20030108

Table CID 6139. Non-Lesion Modifiers

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-40819 | Internal
SRT | R-40941 | External

## CID 6140 Calculation Methods

Type:

  Extensible

Version:

  20070625

Table CID 6140. Calculation Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-10260 | Estimated
DCM | 112187 | Unspecified method of calculation
DCM | 112055 | Agatston scoring method
DCM | 112056 | Volume scoring method
DCM | 112057 | Mass scoring method
DCM | 112188 | Two-dimensional method
DCM | 112189 | Three-dimensional method

## CID 6141 Attenuation Coefficient Measurements

Type:

  Extensible

Version:

  20030108

Table CID 6141. Attenuation Coefficient Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112031 | Attenuation Coefficient
DCM | 112179 | Minimum Attenuation Coefficient
DCM | 112180 | Maximum Attenuation Coefficient
DCM | 112181 | Mean Attenuation Coefficient
DCM | 112182 | Median Attenuation Coefficient
DCM | 112183 | Standard Deviation of Attenuation Coefficient

## CID 6142 Calculated Value

Type:

  Extensible

Version:

  20070625

Table CID 6142. Calculated Value

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112017 | Cavity extent as percent of volume
DCM | 112018 | Calcification extent as percent of surface
DCM | 112019 | Calcification extent as percent of volume
DCM | 112058 | Calcium score
DCM | 112191 | Breast tissue density
DCM | 112192 | Volume of parenchymal tissue
DCM | 112193 | Volume of breast
DCM | 112194 | Mass of parenchymal tissue
DCM | 112195 | Mass of breast
DCM | 112196 | Area of Vascular Calcification
DCM | 112197 | Volume of Vascular Calcification
DCM | 112198 | Percentage of Vascular Calcification
DCM | 112199 | Mass of Vascular Calcification
DCM | 112200 | Average calcification distance in a calcification cluster
DCM | 112201 | Standard deviation distance of calcifications in a cluster

## CID 6143 Lesion Response

Type:

  Extensible

Version:

  20030108

Table CID 6143. Lesion Response

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6144 “RECIST Defined Lesion Response”

## CID 6144 RECIST Defined Lesion Response

Type:

  Extensible

Version:

  20030108

Table CID 6144. RECIST Defined Lesion Response

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112041 | Target Lesion Complete Response
DCM | 112042 | Target Lesion Partial Response
DCM | 112043 | Target Lesion Progressive Disease
DCM | 112044 | Target Lesion Stable Disease
DCM | 112045 | Non-Target Lesion Complete Response
DCM | 112046 | Non-Target Lesion Incomplete Response or Stable Disease
DCM | 112047 | Non-Target Lesion Progressive Disease

## CID 6145 Baseline Category

### Note

From RECIST

Type:

  Extensible

Version:

  20030108

Table CID 6145. Baseline Category

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112074 | Target Lesion at Baseline
DCM | 112075 | Non-Target Lesion at Baseline
DCM | 112076 | Non-Lesion at Baseline

## CID 6146 Time Point Types

Type:

  Extensible

Version:

  20141110

Table CID 6146. Time Point Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UMLS | C1442488 | Baseline
UMLS | C3539075 | Pretreatment
DCM | 126074 | Posttreatment
DCM | 126075 | Eligibility
UMLS | C1699701 | Unscheduled
UMLS | C1708760 | Nadir

### Note

1. (C1442488, UMLS, "Baseline") is (C25213, NCIt, "Baseline"). The undefined (121079, DCM, "Baseline") that is used in CID 7003 Diagnostic Imaging Report Purposes of Reference is not used in this context.

2. (C3539075, UMLS, "Pretreatment") is (C103341, NCIt, "Pretreatment").

3. (C1708760, UMLS, "Nadir") is (C43517, NCIt, "Nadir"), and is a synonymn for "lowest", though "nadir" is more commnly used in the context of therapeutic response criteria.

## CID 6147 Response Criteria

Type:

  Extensible

Version:

  20141110

Table CID 6147. Response Criteria

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112029 | WHO
DCM | 126080 | RECIST 1.0
DCM | 126081 | RECIST 1.1
NCIt | C114879 | RANO

## CID 6151 Background Echotexture

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6151. Background Echotexture

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111351 | Homogeneous fat echotexture
DCM | 111352 | Homogeneous fibroglandular echotexture
DCM | 111353 | Heterogeneous echotexture

## CID 6152 Orientation

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6152. Orientation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111355 | Parallel
DCM | 111356 | Not parallel

## CID 6153 Lesion Boundary

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6153. Lesion Boundary

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111358 | Abrupt interface
DCM | 111359 | Echogenic halo

## CID 6154 Echo Pattern

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6154. Echo Pattern

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111361 | Anechoic
DCM | 111362 | Hyperechoic
DCM | 111363 | Complex
DCM | 111364 | Hypoechoic
DCM | 111365 | Isoechoic

## CID 6155 Posterior Acoustic Features

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20060622

Table CID 6155. Posterior Acoustic Features

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111367 | No posterior acoustic features
DCM | 111368 | Posterior enhancement
DCM | 111369 | Posterior shadowing
DCM | 111370 | Combined posterior enhancement and shadowing

## CID 6157 Vascularity

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6157. Vascularity

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111373 | Vascularity not present
DCM | 111374 | Vascularity not assessed
DCM | 111375 | Vascularity present in lesion
DCM | 111376 | Vascularity present immediately adjacent to lesion
DCM | 111377 | Diffusely increased vascularity in surrounding tissue

## CID 6158 Correlation to Other Findings

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6158. Correlation to Other Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111381 | Correlates to physical exam findings
DCM | 111382 | Correlates to mammography findings
DCM | 111383 | Correlates to MRI findings
DCM | 111384 | Correlates to ultrasound findings
DCM | 111385 | Correlates to other imaging findings
DCM | 111386 | No correlation to other imaging findings
DCM | 111387 | No correlation to clinical findings

## CID 6159 Malignancy Type

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6159. Malignancy Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111389 | Invasive breast carcinoma
SRT | M-85002 | Intraductal carcinoma, non-infiltrating
DCM | 111390 | Other malignancy type

## CID 6160 Breast Primary Tumor Assessment From AJCC

Type:

  Extensible

Version:

  20050822

Table CID 6160. Breast Primary Tumor Assessment From AJCC

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-003B8 | TX: Primary tumor cannot be assessed (breast)
SRT | R-003B9 | T0: No evidence of primary tumor (breast)
SRT | R-003BB | Tis: Carcinoma in situ (breast)
SRT | R-003BC | Tis: Ductal carcinoma in situ (breast)
SRT | R-003BD | Tis: Lobular carcinoma in situ (breast)
SRT | R-003BE | Tis: Paget's disease of the nipple with no tumor
SRT | R-003BA | T1: Tumor 2 cm or less in greatest dimension (breast)
SRT | R-003BF | T1mic: Microinvasion 0.1 cm or less in greatest dimension...
SRT | R-003C0 | T1a: Tumor more than 0.1 cm but not more than 0.5 cm...
SRT | R-003C1 | T1b: Tumor more than 0.5 cm but not more than 1 cm...
SRT | R-003C2 | T1c: Tumor more than 1 cm but not more than 2 cm...
SRT | R-003C3 | T2: Tumor more than 2 cm but not more than 5 cm...
SRT | R-003C4 | T3: Tumor more than 5 cm in greatest dimension (breast)
SRT | R-003C5 | T4: Tumor of any size with direct extension to chest wall...
SRT | R-003C6 | T4a: Tumor of any size with extension to chest wall, not incl...
SRT | R-003C7 | T4b: Tumor of any size with edema (including peau d'orange) ...
SRT | R-003C8 | T4c: Tumor of any size with direct extension to chest wall...
SRT | R-003C9 | T4: Inflammatory carcinoma (breast)

## CID 6161 Clinical Regional Lymph Node Assessment for Breast

Type:

  Extensible

Version:

  20050822

Table CID 6161. Clinical Regional Lymph Node Assessment for Breast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-003CA | NX: Regional lymph nodes cannot be assessed...
SRT | R-003CB | N0: No regional lymph node metastasis histologically...
SRT | R-003D0 | N1: Metastasis in 1 to 3 axillary lymph nodes...
SRT | R-003D6 | N2: Metastasis in 4 to 9 axillary lymph nodes...
SRT | R-003D7 | N2a: Metastasis in 4 to 9 axillary lymph nodes (...2.0 mm)...
SRT | R-003D8 | N2b: Metastasis in clinically apparent internal... nodes...
SRT | G-F749 | N3: Metastasis to ipsilateral internal mammary lymph node(s)
SRT | R-003D9 | N3a: Metastasis in 10 or more axillary lymph nodes...
SRT | R-003DA | N3b: Metastasis in clinically apparent ipsilateral internal...
SRT | R-003DB | N3c: Metastasis in ipsilateral supraclavicular lymph nodes...

## CID 6162 Assessment of Metastasis for Breast

Type:

  Extensible

Version:

  20050822

Table CID 6162. Assessment of Metastasis for Breast

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-003DC | MX: Distant metastasis cannot be assessed (breast)
SRT | R-003DD | M0: No distant metastasis (breast)
SRT | R-003DE | M1: Distant metastasis (breast)

## CID 6163 Menstrual Cycle Phase

### Note

From BI-RADS®

Type:

  Extensible

Version:

  20050822

Table CID 6163. Menstrual Cycle Phase

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111392 | 1st week
DCM | 111393 | 2nd week
DCM | 111394 | 3rd week
SRT | F-840B3 | Menstruation present

## CID 6164 Time Intervals

Type:

  Extensible

Version:

  20050822

Table CID 6164. Time Intervals

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111396 | < 3 months ago
DCM | 111397 | 4 months to 1 year ago
DCM | 111398 | > 1 year ago
DCM | 111399 | Not sure

## CID 6165 Breast Linear Measurements

Type:

  Extensible

Version:

  20050822

Table CID 6165. Breast Linear Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7470 “Linear Measurements”
DCM | 121242 | Distance from nipple
DCM | 121243 | Distance from skin
DCM | 121244 | Distance from chest wall

## CID 6166 CAD Geometry Secondary Graphical Representation

Type:

  Extensible

Version:

  20060822

Table CID 6166. CAD Geometry Secondary Graphical Representation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113661 | Outline of lobulations
DCM | 113662 | Inner limits of fuzzy margin
DCM | 113663 | Outer limits of fuzzy margin
DCM | 113664 | Outline of spiculations
DCM | 113665 | Linear spiculation
DCM | 113666 | Pixelated spiculations
SRT | G-A185 | Long axis
SRT | G-A186 | Short axis
DCM | 113669 | Orthogonal location arc
DCM | 113670 | Orthogonal location arc inner margin
DCM | 113671 | Orthogonal location arc outer margin

## CID 6200 Colon Overall Assessment

Type:

  Extensible

Version:

  20090402

Table CID 6200. Colon Overall Assessment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112240 | C0 - Inadequate Study/Awaiting Prior Comparisons
DCM | 112241 | C1 - Normal Colon or Benign Lesion
DCM | 112242 | C2 - Intermediate Polyp or Indeterminate Finding
DCM | 112243 | C3 - Polyp, Possibly Advanced Adenoma
DCM | 112244 | C4 - Colonic Mass, Likely Malignant

## CID 6201 Colon Finding Or Feature

Type:

  Extensible

Version:

  20090402

Table CID 6201. Colon Finding or Feature

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111101 | Image quality
DCM | 111099 | Selected region
SRT | D5-41170 | Polyp of colon
SRT | D5-F131F | Tumor of colon
SRT | F-54005 | Rectal mass
SRT | M-32700 | Diverticulum
SRT | T-59345 | Colonic haustra
SRT | T-59666 | Feces
SRT | M-88500 | Lipoma
SRT | T-50153 | Intraluminal fluid
SRT | F-61D54 | Contrast media
SRT | T-58650 | Ileocecal valve
SRT | M-32704 | Inverted diverticulum
SRT | M-18000 | Operative Site
DCM | 111102 | Non-lesion
DCM | 112238 | Anatomic non-colon

## CID 6202 Colon Finding Or Feature Modifier

Type:

  Extensible

Version:

  20090402

Table CID 6202. Colon Finding or Feature Modifier

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 6203 “Colon Non-lesion Object Type”
Include CID 6204 “Anatomic Non-colon Findings”

## CID 6203 Colon Non-lesion Object Type

Type:

  Extensible

Version:

  20120822

Table CID 6203. Colon Non-Lesion Object Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-32110 | Bullet
SRT | A-13600 | Staple
SRT | A-13510 | Suture material
SRT | M-78060 | Scar tissue
SRT | A-26800 | Catheter
DCM | 112173 | Chest tube
SRT | A-14611 | Vena cava filter
SRT | A-04000 | Prosthesis
SRT | A-26434 | Jejunostomy tube
DCM | 112175 | Kidney stent
SRT | A-11C08 | Ureteral stent
DCM | 112176 | Pancreatic stent
SRT | A-61000 | Jewelry
DCM | 112178 | Coin
SRT | A-12024 | Pin
SRT | A-30360 | Needle
DCM | 112171 | Fiducial mark
SRT | A-120DD | Colostomy set
SRT | A-10DBC | Colostomy bag
SRT | A-1009E | Ileostomy set
SRT | A-10029 | Ileostomy bag
SRT | A-10703 | Urostomy set
SRT | A-105E3 | Urostomy bag
SRT | A-26440 | Rectal tube
SRT | A-26864 | Urethral catheter

### Note

The use of (111176, DCM, "Unspecified") was previously included in this context group but was removed since it does not make sense to have Colon CAD composite feature modifiers (TID 4125 “Colon CAD Composite Feature”) and single image finding modifiers (TID 4127 “Colon CAD Single Image Finding”) of an unknown type.

## CID 6204 Anatomic Non-colon Findings

Type:

  Extensible

Version:

  20090402

Table CID 6204. Anatomic Non-Colon Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-62000 | Liver
SRT | T-C3000 | Spleen
SRT | T-71000 | Kidney
SRT | T-B3000 | Adrenal gland
SRT | T-42000 | Aorta
SRT | T-48710 | Inferior vena cava
SRT | T-28000 | Lung
SRT | T-D016E | Bone
SRT | T-94000 | Testis
SRT | T-83000 | Uterus
SRT | T-87000 | Ovary
SRT | T-83200 | Cervix
SRT | T-92000 | Prostate
SRT | T-93000 | Seminal Vesicle
SRT | T-59600 | Rectum
SRT | T-74000 | Bladder
SRT | T-13001 | Muscle
SRT | T-40000 | Blood Vessel
SRT | T-59200 | Appendix
SRT | T-D0874 | Appendiceal stump

## CID 6205 Clockface Location for Colon

Type:

  Extensible

Version:

  20090402

Table CID 6205. Clockface Location for Colon

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-01781 | 1 o'clock position
SRT | F-01782 | 2 o'clock position
SRT | F-01783 | 3 o'clock position
SRT | F-01784 | 4 o'clock position
SRT | F-01785 | 5 o'clock position
SRT | F-01786 | 6 o'clock position
SRT | F-01787 | 7 o'clock position
SRT | F-01788 | 8 o'clock position
SRT | F-01789 | 9 o'clock position
SRT | F-0178A | 10 o'clock position
SRT | F-0178B | 11 o'clock position
SRT | F-0178C | 12 o'clock position

## CID 6206 Recumbent Patient Orientation for Colon

Type:

  Extensible

Version:

  20090402

Table CID 6206. Recumbent Patient Orientation for Colon

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-10310 | Prone
SRT | F-10340 | Supine
SRT | F-10317 | right lateral decubitus
SRT | F-10319 | left lateral decubitus

## CID 6207 Colon Quantitative Temporal Difference Type

Type:

  Extensible

Version:

  20090717

Table CID 6207. Colon Quantitative Temporal Difference Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-05173 | Difference in size
SRT | F-05179 | Difference in location
SRT | F-0516E | Difference in attenuation

## CID 6208 Colon Types of Quality Control Standard

Type:

  Extensible

Version:

  20090402

Table CID 6208. Colon Types of Quality Control Standard

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112036 | ACR Position Statement
DCM | 111240 | Institutionally defined quality control standard
DCM | 112248 | ACR Guideline, Performance of Adult CT Colonography
DCM | 112249 | ACR Standard, CT medical physics performance monitoring

## CID 6209 Colon Morphology Descriptor

Type:

  Extensible

Version:

  20090402

Table CID 6209. Colon Morphology Descriptor

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A530 | Sessile
SRT | G-A477 | Pedunculated
SRT | G-A485 | Flat
SRT | R-404F0 | Circumferential
SRT | M-38000 | Ulcer

## CID 6210 Location in Intestinal Tract

Type:

  Extensible

Version:

  20090402

Table CID 6210. Location in Intestinal Tract

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-59600 | Rectum
SRT | T-59470 | Sigmoid colon
SRT | T-59460 | Descending colon
SRT | T-59440 | Transverse colon
SRT | T-59420 | Ascending colon
SRT | T-59100 | Cecum
SRT | T-59442 | Splenic flexure of colon
SRT | T-59438 | Hepatic flexure of colon

## CID 6211 Colon CAD Material Description

Type:

  Extensible

Version:

  20090402

Table CID 6211. Colon CAD Material Description

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112144 | Soft tissue
SRT | T-D008A | Fat
SRT | A-80230 | Air
SRT | T-11034 | Bone matrix

## CID 6212 Calculated Value for Colon Findings

Type:

  Extensible

Version:

  20090402

Table CID 6212. Calculated Value for Colon Findings

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-0045B | Polyp stalk length
SRT | R-00286 | Polyp size, largest dimension
DCM | 112232 | Polyp stalk width
DCM | 112233 | Distance from anus

## CID 7000 Diagnostic Imaging Report Document Titles

Type:

  Extensible

Version:

  20140102

Table CID 7000. Diagnostic Imaging Report Document Titles

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 18745-0 | Cardiac Catheterization Report
LN | 11540-2 | CT Abdomen Report
LN | 11538-6 | CT Chest Report
LN | 11539-4 | CT Head Report
LN | 18747-6 | CT Report
LN | 18748-4 | Diagnostic Imaging Report
LN | 11522-0 | Echocardiography Report
LN | 18760-9 | Ultrasound Report
LN | 11541-0 | MRI Head Report
LN | 18755-9 | MRI Report
LN | 18756-7 | MRI Spine Report
LN | 18757-5 | Nuclear Medicine Report
LN | 17787-3 | Nuclear Medicine Thyroid Scan Report
LN | 11525-3 | Ultrasound Obstetric and Gyn Report
LN | 18758-3 | PET Scan Report
LN | 11528-7 | Radiology Report
LN | 18750-0 | Cardiac Electrophysiology Report
LN | 11524-0 | ECG Report
LN | 18752-6 | Exercise Stress Test Report
LN | 18754-2 | Holter Study Report
LN | 43468-8 | X-Ray Report
LN | 38269-7 | DEXA Skeletal System Study Report
DCM | 111400 | Breast Imaging Report
LN | 24606-6 | Mammography Screening Report
LN | 49512-7 | Fluoroscopy Study Report
LN | 47048-4 | Diagnostic Interventional Radiology Report

## CID 7001 Diagnostic Imaging Report Headings

Type:

  Extensible

Version:

  20130806

Table CID 7001. Diagnostic Imaging Report Headings

Coding Scheme Designator | Code Value | Code Meaning | Equivalent DCMR (DCM) Code
--- | --- | --- | ---
LN | 11329-0 | History | 121060
LN | 55115-0 | Request | 121062
LN | 55111-9 | Current Procedure Descriptions | 121064
LN | 55114-3 | Prior Procedure Descriptions | 121066
LN | 18834-2 | Previous Findings | 121068
LN | 18782-3 | Findings | 121070
LN | 19005-8 | Impressions | 121072
LN | 18783-1 | Recommendations | 121074
LN | 55110-1 | Conclusions | 121076
LN | 55107-7 | Addendum | 121078
LN | 18785-6 | Indications for Procedure | 121109
LN | 55108-5 | Patient Presentation | 121110
LN | 55109-3 | Complications | 121113
LN | 55112-7 | Summary | 121111
LN | 55113-5 | Key Images | 121180
LN | 73569-6 | Radiation Exposure and Protection Information | 113923
LN | 55752-0 | Clinical Information |
LN | 29549-3 | Medications Administered |
LN | 73568-8 | Communication of Critical Results |

### Note

In previous editions of the Standard, this Context Group included codes of DCMR, using Coding Scheme Designator DCM. The preferable encoding of these concepts is using the LOINC codes, however, the support of equivalent DCMR codes is recommended for backward compatibility.

## CID 7002 Diagnostic Imaging Report Elements

Type:

  Extensible

Version:

  20030327

Table CID 7002. Diagnostic Imaging Report Elements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121060 | History
DCM | 121062 | Request
DCM | 121065 | Procedure Description
DCM | 121069 | Previous Finding
DCM | 121071 | Finding
DCM | 121073 | Impression
DCM | 121075 | Recommendation
DCM | 121077 | Conclusion
SRT | DD-60002 | Complication of Procedure
DCM | 121110 | Patient Presentation
DCM | 121111 | Summary

## CID 7003 Diagnostic Imaging Report Purposes of Reference

Type:

  Extensible

Version:

  20100604

Table CID 7003. Diagnostic Imaging Report Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121079 | Baseline
DCM | 121080 | Best illustration of finding
DCM | 121112 | Source of Measurement
DCM | 121200 | Illustration of ROI

## CID 7004 Waveform Purposes of Reference

Type:

  Extensible

Version:

  20090409

Table CID 7004. Waveform Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121301 | Simultaneous Doppler
DCM | 121302 | Simultaneous Hemodynamic
DCM | 121303 | Simultaneous ECG
DCM | 121304 | Simultaneous Voice Narrative
DCM | 121305 | Simultaneous Respiratory Waveform
DCM | 121306 | Simultaneous Arterial Pulse Waveform
DCM | 121307 | Simultaneous Phonocardiographic Waveform

## CID 7005 Contributing Equipment Purposes of Reference

Type:

  Extensible

Version:

  20130617

Table CID 7005. Contributing Equipment Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109101 | Acquisition Equipment
DCM | 109102 | Processing Equipment
DCM | 109103 | Modifying Equipment
DCM | 109104 | De-identifying Equipment
DCM | 109105 | Frame Extracting Equipment
DCM | 109106 | Enhanced Multi-frame Conversion Equipment
DCM | MEDIM | Portable Media Importer Equipment
DCM | FILMD | Film Digitizer
DCM | DOCD | Document Digitizer Equipment
DCM | VIDD | Video Tape Digitizer Equipment

## CID 7006 SR Document Purposes of Reference

Type:

  Extensible

Version:

  20061023

Table CID 7006. SR Document Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121331 | Equivalent CDA Document
DCM | 121332 | Complete Rendering for Presentation
DCM | 121333 | Partial Rendering for Presentation
DCM | 121334 | Extended Rendering for Presentation
DCM | 121335 | Source Document

## CID 7007 Signature Purpose

Context Group ID 7007 comprises the signature purposes codes of ASTM E 2084-00. The Coding Scheme Designator (0008,0102) shall be "ASTM-sigpurpose". The ASTM document defines the signature purpose codes as OIDs. For the purposes of this Coding Scheme only the leaf digit is used as the Code Value (0008,0100).

### Note

ASTM E 1762 provides the full definitions for the signature purpose OIDs defined by E 2084. The recommended Code Meanings (0008,0104) are the titles of the definitions for the leaves of the OIDs. For example, the OID 1.2.840.10065.1.12.1 corresponds to the leaf "id-purpose-author", whose meaning could be encoded as "Author Signature" and whose code value is 1.

## CID 7008 Media Import

This Context Group specifies items that may be conveyed in the Billing and Materials Management Module (see PS3.3).

Type:

  Extensible

Version:

  20061024

Table CID 7008. Media Import

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110020 | Sheet Film Digitized
DCM | 110021 | Cine Film Digitized
DCM | 110022 | Video Tape Digitized
DCM | 110023 | Paper Digitized
DCM | 110024 | CD Imported
DCM | 110025 | DVD Imported
DCM | 110026 | MOD Imported
DCM | 110027 | Studies Imported
DCM | 110028 | Instances Imported

## CID 7009 Purpose of Reference to Predecessor Report

Type:

  Extensible

Version:

  20090826

Table CID 7009. Purpose of Reference to Predecessor Report

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121360 | Replaced report
DCM | 121361 | Addended report
DCM | 121362 | Preliminary report
DCM | 121363 | Partial report
DCM | 122073 | Current procedure evidence

### Note

The concepts of replaced and addended correspond to REPLACEMENT and ADDENDUM in HL7 V2.6 Chapter 9, with the exception that an EDITED value is not supported due to incompatibility with HL7 CDA.

## CID 7010 Key Object Selection Document Title

Type:

  Extensible

Version:

  20140627

Table CID 7010. Key Object Selection Document Title

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113000 | Of Interest
DCM | 113001 | Rejected for Quality Reasons
DCM | 113002 | For Referring Provider
DCM | 113003 | For Surgery
DCM | 113004 | For Teaching
DCM | 113005 | For Conference
DCM | 113006 | For Therapy
DCM | 113007 | For Patient
DCM | 113008 | For Peer Review
DCM | 113009 | For Research
DCM | 113010 | Quality Issue
DCM | 113013 | Best In Set
DCM | 113018 | For Printing
DCM | 113020 | For Report Attachment
DCM | 113021 | For Litigation
DCM | 113030 | Manifest
DCM | 113031 | Signed Manifest
DCM | 113032 | Complete Study Content
DCM | 113033 | Signed Complete Study Content
DCM | 113034 | Complete Acquisition Content
DCM | 113035 | Signed Complete Acquisition Content
DCM | 113036 | Group of Frames for Display
DCM | 113037 | Rejected for Patient Safety Reasons
DCM | 113038 | Incorrect Modality Worklist Entry
DCM | 113039 | Data Retention Policy Expired

## CID 7011 Rejected for Quality Reasons

Type:

  Extensible

Version:

  20020904

Table CID 7011. Rejected for Quality Reasons

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111207 | Image artifact(s)
DCM | 111208 | Grid artifact(s)
DCM | 111209 | Positioning
DCM | 111210 | Motion blur
DCM | 111211 | Under exposed
DCM | 111212 | Over exposed
DCM | 111213 | No image
DCM | 111214 | Detector artifact(s)
DCM | 111215 | Artifact(s) other than grid or detector artifact
DCM | 111216 | Mechanical failure
DCM | 111217 | Electrical failure
DCM | 111218 | Software failure
DCM | 111219 | Inappropriate image processing
DCM | 111220 | Other failure
DCM | 111221 | Unknown failure
DCM | 113026 | Double exposure

## CID 7012 Best in Set

Type:

  Extensible

Version:

  20020904

Table CID 7012. Best in Set

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113014 | Study
DCM | 113015 | Series
DCM | 113016 | Performed Procedure Step
DCM | 113017 | Stage-View

## CID 7020 Document Titles

Context Group ID 7020 comprises all document names (i.e., terms with Scale "DOC") within the HIPAA Attachments class of the LOINC coding scheme. The Coding Scheme Designator shall be LN.

### Note

1. A subset of this Context Group directly applicable to imaging reports is in CID 7000 “Diagnostic Imaging Report Document Titles”.

2. The LOINC coding scheme can be found at http://www.regenstrief.org/loinc.

## CID 7021 Measurement Report Document Titles

Type:

  Extensible

Version:

  20141110

Table CID 7021. Measurement Report Document Titles

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126000 | Imaging Measurement Report
DCM | 126001 | Oncology Measurement Report
DCM | 126002 | Dynamic Contrast MR Measurement Report
DCM | 126003 | PET Measurement Report

## CID 7030 Institutional Departments, Units and Services

Type:

  Extensible

Version:

  20090717

Table CID 7030. Institutional Departments, Units and Services

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-300E3 | Accident and Emergency
SRT | R-30246 | Allergy and Immunology
SRT | R-3023A | Anesthesiology
SRT | R-30247 | Audiology
SRT | R-421EB | Clinical Biochemistry
SRT | R-3027F | Breast Surgery
SRT | R-3060E | Burns Intensive Care
SRT | R-30240 | Cardiac Intensive Care
SRT | R-30282 | Cardiac Surgery
SRT | R-30248 | Cardiology
SRT | R-30280 | Cardiothoracic Surgery
SRT | R-30276 | Child and Adolescent Psychiatry
SRT | R-3023B | Clinical Oncology
SRT | R-3028E | Colorectal Surgery
SRT | R-4225D | Cytology
SRT | R-30283 | Dental Surgery
SRT | R-30250 | Dermatology
SRT | R-3061B | Diagnostic Imaging
SRT | R-3028A | Endocrine Surgery
SRT | R-30252 | Endocrinology
SRT | R-421D4 | Endoscopy
SRT | R-3028B | Gastrointestinal Surgery
SRT | R-30254 | General Medicine
SRT | R-3028F | General Surgery
SRT | R-3025A | Geriatric Medicine
SRT | R-30264 | Gynecology
SRT | R-30290 | Hand Surgery
SRT | R-3026F | Hematology
SRT | R-4223B | Hepatobiliary Surgery
SRT | R-3061D | Histopathology
SRT | R-3025B | Infectious Disease
SRT | R-3023D | Intensive Care
SRT | R-3061E | Medical Intensive Care
SRT | R-30270 | Medical Microbiology
SRT | R-3025D | Nephrology
SRT | R-305CE | Neonatal Intensive Care
SRT | R-3025E | Neurology
SRT | R-4223C | Neurosurgery
SRT | R-3025F | Nuclear Medicine
SRT | R-30265 | Obstetrics
SRT | R-30263 | Obstetrics and Gynecology
SRT | R-3025C | Ophthalmology
SRT | R-42207 | Optometry
SRT | R-30285 | Oral Surgery
SRT | R-30294 | Orthopedic Surgery
SRT | R-30289 | Otorhinolaryngology
SRT | R-3026A | Pain Management
SRT | R-30260 | Palliative Care
SRT | R-3026B | Pathology
SRT | R-30243 | Pediatric Intensive Care
SRT | R-305EA | Pediatric Medicine
SRT | R-30269 | Pediatric Oncology
SRT | R-305E9 | Pediatric Surgery
SRT | S-8000A | Primary Care Department
SRT | R-30261 | Rehabilitation
SRT | R-302A2 | Physiotherapy
SRT | R-30297 | Plastic Surgery
SRT | R-30275 | Psychiatry
SRT | R-42219 | Psychology
SRT | R-3024B | Pulmonology
SRT | R-3023C | Radiotheraphy
SRT | R-3027B | Radiology
SRT | R-30262 | Rheumatology
SRT | R-42203 | Speech and Language Therapy
SRT | R-3027D | Stroke
SRT | R-3027E | Surgery
SRT | R-305EB | Surgical Intensive Care
SRT | R-30281 | Thoracic Surgery
SRT | R-30298 | Transplant Surgery
SRT | R-30299 | Trauma Surgery
SRT | R-30616 | Tropical Medicine
SRT | R-42246 | Ultrasonography
SRT | R-3029A | Urology
SRT | R-3029B | Vascular Surgery

### Note

In SNOMED, there is often a choice of unit, department or service concepts; in DICOM, the department concept is preferred and used in this context group.

## CID 7040 Broselow-Luten Pediatric Size Categories

Type:

  Non-Extensible

Version:

  20100127

Table CID 7040. Broselow-Luten Pediatric Size Categories

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-051E3 | Broselow Luten Pink Zone (6-7 kg)
SRT | F-051DF | Broselow Luten Red Zone (8-9 kg)
SRT | F-051E4 | Broselow Luten Purple Zone (10-11 kg)
SRT | F-051E8 | Broselow Luten Yellow Zone (12-14 kg)
SRT | F-051E7 | Broselow Luten White Zone (15-18 kg)
SRT | F-051E0 | Broselow Luten Blue Zone (19-23 kg)
SRT | F-051E5 | Broselow Luten Orange Zone (24-29 kg)
SRT | F-051E6 | Broselow Luten Green Zone (30-36 kg)

## CID 7042 Calcium Scoring Patient Size Categories

Patient sizes for calibrating calcium scoring, from the Consortium for Multi-Detector CT Evaluation of Coronary Calcium.

Type:

  Non-Extensible

Version:

  20100127

Table CID 7042. Calcium Scoring Patient Size Categories

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113601 | Small: < 32.0 cm lateral thickness
DCM | 113602 | Medium: 32.0-38.0 cm lateral thickness
DCM | 113603 | Large: > 38.0 cm lateral thickness

## CID 7050 De-identification Method

Type:

  Extensible

Version:

  20110123

Table CID 7050. De-Identification Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113100 | Basic Application Confidentiality Profile
DCM | 113101 | Clean Pixel Data Option
DCM | 113102 | Clean Recognizable Visual Features Option
DCM | 113103 | Clean Graphics Option
DCM | 113104 | Clean Structured Content Option
DCM | 113105 | Clean Descriptors Option
DCM | 113106 | Retain Longitudinal Temporal Information Full Dates Option
DCM | 113107 | Retain Longitudinal Temporal Information Modified Dates Option
DCM | 113108 | Retain Patient Characteristics Option
DCM | 113109 | Retain Device Identity Option
DCM | 113110 | Retain UIDs Option
DCM | 113111 | Retain Safe Private Option

## CID 7100 RCS Registration Method Type

Type:

  Extensible

Version:

  20040115

Table CID 7100. RCS Registration Method Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125021 | Frame of Reference Identity
DCM | 125023 | Acquisition Equipment Alignment
DCM | 125025 | Visual Alignment
DCM | 125022 | Fiducial Alignment
DCM | 125024 | Image Content-based Alignment

## CID 7101 Brain Atlas Fiducials

Type:

  Extensible

Version:

  20040115

Table CID 7101. Brain Atlas Fiducials

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 125030 | Inter-Hemispheric Plane |  |
SRT | T-A2980 | Anterior Commissure | 62872008 | C0152335
SRT | T-A4904 | Posterior Commissure | 279336005 | C0152327
DCM | 125031 | Right Hemisphere Most Anterior |  |
DCM | 125032 | Right Hemisphere Most Posterior |  |
DCM | 125033 | Right Hemisphere Most Superior |  |
DCM | 125034 | Right Hemisphere Most Inferior |  |
DCM | 125035 | Left Hemisphere Most Anterior |  |
DCM | 125036 | Left Hemisphere Most Posterior |  |
DCM | 125037 | Left Hemisphere Most Superior |  |
DCM | 125038 | Left Hemisphere Most Inferior |  |

## CID 7140 Brain Structures for Volumetric Measurements

Type:

  Extensible

Version:

  20131010

Table CID 7140. Brain Structures for Volumetric Measurements

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-A3230 | Amygdala | 4958002 | C0002708
SRT | T-D0558 | Brain Stem | 119238007 | C1268144
SRT | T-A3200 | Caudate Nucleus | 11000004 | C0007461
SRT | T-A6041 | Cerebellar Cortex | 361593004 | C1284087
SRT | T-A6080 | Cerebellar White Matter | 33060004 | C0152381
SRT | T-A2020 | Cerebral Grey Matter | 40146001 | C0007776
SRT | T-A2030 | Cerebral White Matter | 68523003 | C0152295
SRT | T-D1403 | Cranial Cavity | 264452006 | C1280705
SRT | T-A1504 | Cranial Subarachnoid Space | 362312008 | C1284571
SRT | T-A1604 | Fifth Ventricle | 180933005 | C0228158
SRT | T-A1820 | Fourth Ventricle | 35918002 | C0149556
SRT | T-A3500 | Globus Pallidus | 14738005 | C0017651
SRT | T-A2570 | Hippocampus | 5366008 | C0019564
SRT | T-A1509 | Cerebellar Subarachnoid Space | 263972004 | C0446676
SRT | T-A1721 | Inferior Horn of Lateral Ventricle | 362315005 | C1284574
SRT | T-A1650 | Lateral Ventricle | 66720007 | C0152279
SRT | T-A0149 | Nucleus Accumbens | 427667007 | C0028633
SRT | T-A0190 | Intracranial structure | 128319008 | C1267697
SRT | T-A3400 | Putamen | 89278009 | C0034169
SRT | T-D0593 | Thalamus | 119406000 | C0458271
SRT | T-A1740 | Third ventricle | 49841001 | C0149555
DCM | 110700 | Ventral Diencephalon |  |
DCM | 110701 | White Matter T1 Hypointensity |  |
DCM | 110702 | White Matter T2 Hyperintensity |  |

### Note

1. (T-D1403, SRT, "Cranial Cavity") may be used to describe the volume of the entire intra-cranial space (intra-cranial volume or ICV).

2. (T-A6041, SRT, "Cerebellar Cortex") is the gray matter of the cerebellum (as distinct from (T-A6080, SRT, "Cerebellar white matter")).

3. (T-A1504, SRT, "Cranial Subarachnoid Space") may be used to describe the volume of the exterior CSF (surrounding the brain, excluding the ventricles).

4. (T-A1509, SRT, "Cerebellar Subarachnoid Space") may be used to describe the volume of the inferior intracranial CSF space (infra-tentorial).

## CID 7150 Segmentation Property Categories

Type:

  Extensible

Version:

  20140627

Table CID 7150. Segmentation Property Categories

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-D0050 | Tissue | 85756007 | C0040300
SRT | T-D000A | Anatomical Structure | 123037004 | C1268086
SRT | A-00004 | Physical object | 260787004 | C0085089
SRT | M-01000 | Morphologically Altered Structure | 49755003 | C0221198
SRT | R-42019 | Function | 246464006 | C0542341
SRT | R-42018 | Spatial and Relational Concept | 309825002 | C0587374
SRT | T-D0080 | Body Substance | 91720002 | C0504082

## CID 7151 Segmentation Property Types

Type:

  Extensible

Version:

  20130617

Table CID 7151. Segmentation Property Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 8 “Angiographic Interventional Devices”
Include CID 3010 “Cardiovascular Anatomic Locations”
Include CID 3827 “Vessel Segments”
Include CID 3829 “Pulmonary Arteries”
Include CID 4028 “Craniofacial Anatomic Regions”
Include CID 4030 “CT, MR and PET Anatomy Imaged”
Include CID 4040 “Endoscopy Anatomic Regions”
Include CID 6040 “Non-lesion Object Type”
Include CID 6109 “Radiographic Anatomy Finding Or Feature”
Include CID 6138 “Chest Non-lesion Object Type”
Include CID 7152 “Cardiac Structure Segmentation Types”
Include CID 7153 “CNS Tissue Segmentation Types”
Include CID 7154 “Abdominal Organ Segmentation Types”
Include CID 7155 “Thoracic Tissue Segmentation Types”
Include CID 7156 “Vascular Tissue Segmentation Types”
Include CID 7157 “Device Segmentation Types”
Include CID 7158 “Artifact Segmentation Types”
Include CID 7159 “Lesion Segmentation Types”
Include CID 7160 “Pelvic Organ Segmentation Types”
Include CID 7161 “Physiology Segmentation Types”
Include CID 7165 “Abstract Segmentation Types”
Include CID 7166 “Common Tissue Segmentation Types”
Include CID 7167 “Peripheral Nervous System Segmentation Types”

### Note

This set is expected to be extended as various applications further define segmentation properties.

## CID 7152 Cardiac Structure Segmentation Types

Type:

  Extensible

Version:

  20130617

Table CID 7152. Cardiac Structure Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-32000 | Heart | 80891009 | C0018787
SRT | T-42000 | Aorta | 15825003 | C0003483
SRT | T-32600 | Left Ventricle | 87878005 | C0225897
SRT | T-32500 | Right Ventricle | 53085002 | C0225883
SRT | T-39000 | Pericardium | 76848001 | C0031050
SRT | T-39050 | Pericardial cavity | 25489000 | C0225972

## CID 7153 CNS Tissue Segmentation Types

Type:

  Extensible

Version:

  20130617

Table CID 7153. CNS Tissue Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-B1100 | Adenohypophysis | 62818001 | C0032008
SRT | T-A3230 | Amygdala | 4958002 | C0002708
SRT | T-A1220 | Arachnoid | 75042008 | C0003707
FMA | 276650 | Arcuate Fasciculus |  | C2329633
SRT | T-A0100 | Brain | 12738006 | C0006104
SRT | T-A0109 | Brain cerebrospinal fluid pathway | 280371009 | C0459387
SRT | T-D0558 | Brain stem | 119238007 | C1268144
SRT | T-A1600 | Brain ventricle | 35764002 | C0007799
SRT | T-A3200 | Caudate nucleus | 11000004 | C0007461
SRT | T-A0090 | Central nervous system | 21483005 | C0927232
SRT | T-A6080 | Cerebellar white matter | 33060004 | C0152381
SRT | T-A1800 | Cerebral aqueduct | 80447000 | C0007769
SRT | T-A2020 | Cerebral cortex | 40146001 | C0007776
SRT | T-A2970 | Cerebral fornix | 87463005 | C0152334
SRT | T-A2020 | Cerebral Grey Matter | 40146001 | C0007776
SRT | T-A2030 | Cerebral White Matter | 68523003 | C0152295
SRT | T-A1000 | Cerebrospinal Fluid | 65216001 | C0007806
SRT | T-A2840 | Cingulum | 37035000 | C0228272
SRT | T-A2700 | Corpus callosum | 88442005 | C0010090
SRT | T-A3100 | Corpus striatum | 31428008 | C0010097
SRT | T-A0102 | Diencephalon | 87563008 | C0012144
SRT | T-A1120 | Dura mater | 18545000 | C0013313
SRT | T-A2594 | Entorhinal Cortex | 3937002 | C0175196
SRT | T-A1820 | Fourth ventricle | 35918002 | C0149556
SRT | T-A2200 | Frontal lobe | 83251001 | C0016733
SRT | T-A3500 | Globus pallidus | 14738005 | C0017651
SRT | T-A0096 | Gray Matter | 389081007 | C1300312
SRT | T-A2570 | Hippocampus | 5366008 | C0019564
SRT | T-A6640 | Inferior cerebellar peduncle | 67701001 | C0152393
SRT | T-A2850 | Inferior longitudinal fasciculus | 55233005 | C0228273
SRT | T-A2610 | Insula | 36169008 | C0021640
SRT | T-A7093 | Lateral corticospinal tract | 461002 | C0152402
SRT | T-A1650 | Lateral ventricle | 66720007 | C0152279
SRT | T-A0036 | Limbic lobe | 279215006 | C0458337
SRT | T-A5271 | Medial Lemniscus | 30114003 | C0228420
SRT | T-A1110 | Meninges | 1231004 | C0025285
SRT | T-A5100 | Midbrain | 61962009 | C0025462
SRT | T-A0149 | Nucleus accumbens | 427667007 | C0028633
SRT | T-B1200 | Neurohypophysis | 37512009 | C0032009
SRT | T-A2400 | Occipital lobe | 31065004 | C0028785
SRT | T-A800B | Optic chiasm | 244453006 | C0029126
SRT | T-A2880 | Optic radiation | 70105001 | C0228277
SRT | T-A8060 | Optic tract | 53238003 | C0152405
SRT | T-A2300 | Parietal lobe | 16630005 | C0030560
SRT | T-A1280 | Pia mater | 23180006 | C0031869
SRT | T-B2000 | Pineal Gland | 45793000 | C0031939
SRT | T-B1000 | Pituitary | 56329008 | C0032005
SRT | T-A4904 | Posterior cerebral commissure | 279336005 | C0152327
SRT | T-A3400 | Putamen | 89278009 | C0034169
SRT | T-D0721 | Spinal cerebrospinal fluid pathway | 280401006 | C0459413
SRT | T-A7010 | Spinal cord | 2748008 | C0037925
SRT | T-A7020 | Spinal cord gray matter | 12958003 | C0475853
SRT | T-A7070 | Spinal cord white matter | 27088001 | C0458457
SRT | T-A1500 | Subarachnoid space | 35951006 | C0038527
SRT | T-A5160 | Substantia nigra | 70007007 | C0038590
SRT | T-A6620 | Superior cerebellar peduncle | 11089000 | C0152391
SRT | T-A2820 | Superior longitudinal fasciculus | 89202009 | C0228270
SRT | T-A0103 | Telencephalon | 11628009 | C0039452
SRT | T-A2500 | Temporal lobe | 78277001 | C0039485
SRT | T-D0593 | Thalamus | 119406000 | C0458271
SRT | T-A1740 | Third ventricle | 49841001 | C0149555
SRT | T-A2830 | Uncinate fasciculus | 26230003 | C0228271
SRT | T-A0095 | White Matter | 389080008 | C1300311

## CID 7154 Abdominal Organ Segmentation Types

Type:

  Extensible

Version:

  20130617

Table CID 7154. Abdominal Organ Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-D4000 | Abdomen | 113345001 | C0000726
SRT | T-D4010 | Abdominal cavity | 52731004 | C0230168
SRT | T-14001 | Abdominal wall muscle | 195879000 | C1279385
SRT | T-B3000 | Adrenal gland | 23451007 | C0001625
SRT | T-42500 | Abdominal aorta | 7832008 | C0003484
SRT | T-60610 | Bile Duct | 28273000 | C0005400
SRT | T-48710 | Inferior vena cava | 64131007 | C0042458
SRT | T-71000 | Kidney | 64033007 | C0022646
SRT | T-62000 | Liver | 10200004 | C0023884
SRT | T-D4600 | Omentum | 27398004 | C0028977
SRT | T-D4425 | Peritoneal cavity | 83670000 | C1704247
SRT | T-D4400 | Peritioneum | 15425007 | C0031153
SRT | T-D4900 | Retroperitoneal space | 82849001 | C0035359
SRT | T-02480 | Skin of abdomen | 75093004 | C0222166
SRT | T-58000 | Small Intestine | 30315005 | C0021852
SRT | T-C3000 | Spleen | 78961009 | C0037993
SRT | T-70001 | Urinary system | 122489005 | C1508753

## CID 7155 Thoracic Tissue Segmentation Types

Type:

  Extensible

Version:

  20130617

Table CID 7155. Thoracic Tissue Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-40000 | Blood Vessel | 59820001 | C0005847
SRT | T-D0170 | Bone of thorax | 272710004 | C0448157
SRT | T-26000 | Bronchus | 955009 | C0006255
SRT | T-14122 | Chest wall muscle | 372074006 | C1269825
SRT | T-12310 | Clavicle | 51299004 | C0008913
SRT | T-D3400 | Diaphragm | 5798000 | C0011980
SRT | T-56000 | Esophagus | 32849002 | C0014876
SRT | T-28830 | Lower lobe of lung | 90572001 | C0225758
SRT | T-28000 | Lung | 39607008 | C0024109
SRT | T-D3300 | Mediastinum | 72410000 | C0025066
SRT | T-28300 | Middle lobe of right lung | 72481006 | C0225757
SRT | T-29000 | Pleura | 3120008 | C0032225
SRT | T-11300 | Rib | 113197003 | C0035561
SRT | T-02424 | Skin of chest | 74160004 | C0222149
SRT | T-11210 | Sternum | 56873002 | C0038293
SRT | T-11502 | Thoracic spine | 122495006 | C0581269
SRT | T-D3000 | Thorax | 51185008 | C0817096
SRT | T-25000 | Trachea | 44567001 | C0040578
SRT | T-C8000 | Thymus | 9875009 | C0040113
SRT | T-28820 | Upper lobe of lung | 45653009 | C0225756

## CID 7156 Vascular Tissue Segmentation Types

Type:

  Extensible

Version:

  20130617

Table CID 7156. Vascular Tissue Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-1A190 | Adventitia | 30180000 | C0225342
SRT | T-30000 | Cardiovascular system | 113257007 | C0007226
SRT | T-1A170 | Intima | 8361002 | C0162864
SRT | T-40230 | Lumen | 91747007 | C0524424
SRT | T-40210 | Media | 27208005 | C0162867
SRT | T-4105E | Systemic artery | 281159003 | C0459964
SRT | T-48081 | Systemic vein | 360592004 | C0447117
SRT | M-35001 | Thrombus | 396339007 | C0087086
SRT | M-520F8 | Vascular sclerosis | 107671003 | C0003850

## CID 7157 Device Segmentation Types

Type:

  Extensible

Version:

  20130617

Table CID 7157. Device Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-12024 | Bone Pin | 77444004 | C0175718
SRT | A-12030 | Bone Screw | 68183006 | C0005975
SRT | A-11100 | Cardiac Pacemaker | 14106009 | C0030163
SRT | A-11206 | Defibrillator | 72506001 | C0162589
SRT | A-04200 | Dental Prosthesis | 27606000 | C0162686
SRT | A-04036 | Inlay Dental Restoration | 272287005 | C0441351
SRT | A-11FCD | Left ventricular assist device | 360066001 | C0181598
SRT | A-30360 | Needle | 79068005 | C0027551
SRT | A-04034 | Radioactive implant | 19443004 | C0521196
SRT | A-25500 | Stent | 65818007 | C0038257

## CID 7158 Artifact Segmentation Types

Type:

  Extensible

Version:

  20130617

Table CID 7158. Artifact Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | A-00916 | Clothing | 272180002 | C0009072
SRT | M-30400 | Foreign body | 19227008 | C0016542
SRT | A-17350 | Table | 86407004 | C0039224

## CID 7159 Lesion Segmentation Types

Type:

  Extensible

Version:

  20130617

Table CID 7159. Lesion Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | M-41610 | Abscess | 44132006 | C0000833
SRT | M-35000 | Blood clot | 75753009 | C0302148
SRT | M-3340A | Cyst | 367643001 | C0010709
SRT | M-36300 | Edema | 79654002 | C0013604
SRT | M-35300 | Embolus | 55584005 | C1704212
SRT | M-37000 | Hemorrhage | 50960005 | C0019080
SRT | M-40000 | Inflammation | 23583003 | C0021368
SRT | M-03000 | Mass | 4147007 | C0577559
SRT | M-54000 | Necrosis | 6574001 | C0027540
SRT | M-8FFFF | Neoplasm | 108369006 | C0027651
SRT | M-03010 | Nodule | 27925004 | C0028259

## CID 7160 Pelvic Organ Segmentation Types

Type:

  Extensible

Version:

  20130617

Table CID 7160. Pelvic Organ Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-74000 | Bladder | 89837001 | C0005682
SRT | T-83200 | Cervix | 71252005 | C0007874
SRT | T-88000 | Fallopian tube | 31435000 | C0015560
SRT | T-80010 | Female external genitalia | 86969008 | C0227747
SRT | T-80020 | Female internal genitalia | 87759004 | C0227748
SRT | T-90010 | Male external genitalia | 90418005 | C0227922
SRT | T-90020 | Male internal genitalia | 38242008 | C0227923
SRT | T-87000 | Ovary | 15497006 | C0029939
SRT | T-92000 | Prostate | 41216001 | C0033572
SRT | T-59600 | Rectum | 34402009 | C0034896
SRT | T-93000 | Seminal Vesicle | 64739004 | C0036628
SRT | T-94000 | Testis | 40689003 | C0039597
SRT | T-83000 | Uterus | 35039007 | C0042149
SRT | T-82000 | Vagina | 76784001 | C0042232
SRT | T-96000 | Vas deferens | 57671007 | C0042360

## CID 7161 Physiology Segmentation Types

Type:

  Extensible

Version:

  20060822

Table CID 7161. Physiology Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | R-0039F | Perfusion | 371863001 | C1276288

## CID 7162 Surface Processing Algorithm Families

Type:

  Extensible

Version:

  20080829

Table CID 7162. Surface Processing Algorithm Families

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 123101 | Neighborhood Analysis
DCM | 123102 | Adaptive Filtering
DCM | 123103 | Edge Detection
DCM | 123104 | Morphological Operations
DCM | 123105 | Histogram Analysis
DCM | 123106 | Multi-Scale/Resolution Filtering
DCM | 123107 | Cluster Analysis
DCM | 123108 | Multispectral Processing
DCM | 123109 | Manual Processing
DCM | 123110 | Artificial Intelligence
DCM | 123111 | Deformable Models

## CID 7165 Abstract Segmentation Types

Type:

  Extensible

Version:

  20140627

Table CID 7165. Abstract Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 125040 | Background |  |
SRT | T-D0050 | Tissue | 85756007 | C0040300
SRT | F-61779 | Waste Material | 289925000 | C0043045
DCM | 125041 | Registration Input |  |

## CID 7166 Common Tissue Segmentation Types

Type:

  Extensible

Version:

  20130617

Table CID 7166. Common Tissue Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-41066 | Artery | 275989006 | C0555806
SRT | T-60650 | Bile | 70150004 | C0005388
SRT | T-C2000 | Blood | 87612001 | C0005767
SRT | T-40000 | Blood vessel | 59820001 | C0005847
SRT | F-03D38 | Body fat | 248300009 | C0344335
SRT | T-D0070 | Body fluid | 32457005 | C0005889
SRT | T-D016E | Bone | 272673000 | C0262950
SRT | T-40050 | Capillary | 20982000 | C0006901
SRT | T-D021B | Cartilage | 309312004 | C0007301
SRT | T-1A200 | Connective tissue | 21793004 | C0009780
SRT | T-59666 | Feces | 39477002 | C0015733
SRT | C-10080 | Gas | 74947009 | C0017110
SRT | T-18010 | Ligament | 52082005 | C0023685
SRT | T-C4000 | Lymph node | 59441001 | C0024204
SRT | T-C6000 | Lymphatic system | 89890002 | C0024235
SRT | T-C6010 | Lymphatic vessel | 83555006 | C0229889
SRT | T-15009 | Meniscus | 74135004 | C0224498
SRT | T-13001 | Muscle | 71616004 | C0026845
SRT | T-D0598 | Nerve | 119410002 | C1268169
SRT | T-1A310 | Organ | 2861001 | C1285092
SRT | T-01000 | Skin | 39937001 | C1123023
SRT | T-17010 | Tendon | 13024002 | C0039508
SRT | T-D0050 | Tissue | 85756007 | C0040300
SRT | T-70060 | Urine | 78014005 | C0042036
SRT | T-4806E | Vein | 181378009 | C0447146

## CID 7167 Peripheral Nervous System Segmentation Types

Type:

  Extensible

Version:

  20130617

Table CID 7167. Peripheral Nervous System Segmentation Types

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | T-A0140 | Peripheral nervous system | 3058005 | C0206417
SRT | T-A0500 | Peripheral nerve | 84782009 | C0031119
SRT | T-A8000 | Cranial nerve | 25238003 | C0010268
SRT | T-A8640 | Vagus nerve | 88882009 | C0042276
SRT | T-A9605 | Autonomic nerve | 53520000 | C0206250
SRT | T-A9630 | Sympathetic trunk | 44909008 | C0228972

## CID 7180 Abstract Multi-dimensional Image Model Component Semantics

Type:

  Extensible

Version:

  20141110

Table CID 7180. Abstract Multi-Dimensional Image Model Component Semantics

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 4033 “MR Proton Spectroscopy Metabolites”
DCM | 113063 | T1
DCM | 113065 | T2
DCM | 113064 | T2*
DCM | 113058 | Proton Density
DCM | 110800 | Spin Tagging Perfusion MR Signal Intensity
DCM | 113070 | Velocity encoded
DCM | 113067 | Temperature encoded
DCM | 110801 | Contrast Agent Angio MR Signal Intensity
DCM | 110802 | Time Of Flight Angio MR Signal Intensity
DCM | 110803 | Proton Density Weighted MR Signal Intensity
DCM | 110804 | T1 Weighted MR Signal Intensity
DCM | 110805 | T2 Weighted MR Signal Intensity
DCM | 110806 | T2* Weighted MR Signal Intensity
DCM | 113043 | Diffusion weighted
DCM | 110807 | Field Map MR Signal Intensity
DCM | 110808 | Fractional Anisotropy
DCM | 110809 | Relative Anisotropy
DCM | 113041 | Apparent Diffusion Coefficient
DCM | 110810 | Volumetric Diffusion Dxx Component
DCM | 110811 | Volumetric Diffusion Dxy Component
DCM | 110812 | Volumetric Diffusion Dxz Component
DCM | 110813 | Volumetric Diffusion Dyy Component
DCM | 110814 | Volumetric Diffusion Dyz Component
DCM | 110815 | Volumetric Diffusion Dzz Component
DCM | 110816 | T1 Weighted Dynamic Contrast Enhanced MR Signal Intensity
DCM | 110817 | T2 Weighted Dynamic Contrast Enhanced MR Signal Intensity
DCM | 110818 | T2* Weighted Dynamic Contrast Enhanced MR Signal Intensity
DCM | 110819 | Blood Oxygenation Level
DCM | 110820 | Nuclear Medicine Projection Activity
DCM | 110821 | Nuclear Medicine Tomographic Activity
DCM | 110822 | Spatial Displacement X Component
DCM | 110823 | Spatial Displacement Y Component
DCM | 110824 | Spatial Displacement Z Component
DCM | 110825 | Hemodynamic Resistance
DCM | 110826 | Indexed Hemodynamic Resistance
DCM | 112031 | Attenuation Coefficient
DCM | 110827 | Tissue Velocity
DCM | 110828 | Flow Velocity
SRT | P0-02241 | Power Doppler
DCM | 110829 | Flow Variance
DCM | 110830 | Elasticity
DCM | 110831 | Perfusion
DCM | 110832 | Speed of sound
DCM | 110833 | Ultrasound Attenuation
DCM | 113068 | Student's T-test
DCM | 113071 | Z-score
DCM | 113057 | R-Coefficient
DCM | 126220 | R2-Coefficient
DCM | 110834 | RGB R Component
DCM | 110835 | RGB G Component
DCM | 110836 | RGB B Component
DCM | 110837 | YBR FULL Y Component
DCM | 110838 | YBR FULL CB Component
DCM | 110839 | YBR FULL CR Component
DCM | 110840 | YBR PARTIAL Y Component
DCM | 110841 | YBR PARTIAL CB Component
DCM | 110842 | YBR PARTIAL CR Component
DCM | 110843 | YBR ICT Y Component
DCM | 110844 | YBR ICT CB Component
DCM | 110845 | YBR ICT CR Component
DCM | 110846 | YBR RCT Y Component
DCM | 110847 | YBR RCT CB Component
DCM | 110848 | YBR RCT CR Component
DCM | 110849 | Echogenicity
DCM | 110850 | X-Ray Attenuation
DCM | 112031 | Attenuation Coefficient
DCM | 110852 | MR signal intensity
DCM | 110853 | Binary Segmentation
DCM | 110854 | Fractional Probabilistic Segmentation
DCM | 110855 | Fractional Occupancy Segmentation
DCM | 126393 | R1
DCM | 126394 | R2
Include Section CID 4107 “Tracer Kinetic Model Parameters”
Include Section CID 4108 “Perfusion Model Parameters”
Include Section CID 4109 “Model-Independent Dynamic Contrast Analysis Parameters”
DCM | 126400 | Standardized Uptake Value
DCM | 126401 | SUVbw
DCM | 126402 | SUVlbm
DCM | 126403 | SUVbsa
DCM | 126404 | SUVibw

## CID 7181 Abstract Multi-dimensional Image Model Component Units

Type:

  Extensible

Version:

  20141110

Table CID 7181. Abstract Multi-Dimensional Image Model Component Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3500 “Pressure Units”
Include CID 3502 “Hemodynamic Resistance Units”
Include CID 3503 “Indexed Hemodynamic Resistance Units”
Include CID 7460 “Units of Linear Measurement”
Include CID 7461 “Units of Area Measurement”
Include CID 7462 “Units of Volume Measurement”
Include CID 84 “PET Units”
UCUM | 1 | no units
UCUM | {ratio} | ratio
UCUM | [hnsf'U] | Hounsfield Unit
UCUM | {counts} | Counts
UCUM | {counts}/s | Counts per second
UCUM | [arb'U] | arbitrary unit
UCUM | cm/s | centimeter/second
UCUM | mm/s | millimeter/second
UCUM | dB | decibel
UCUM | Cel | degrees Celsius
UCUM | ml/min | milliliter per minute
UCUM | ml/s | milliliter per second
UCUM | ms | millisecond
UCUM | s | second
UCUM | Hz | Hertz
UCUM | mT | milliTesla
UCUM | {Particles}/[100]g{Tissue} | number particles per 100 gram of tissue
UCUM | mm2/s | square millimeter per second
UCUM | s/mm2 | second per square millimeter
UCUM | ml/[100]g/min | milliliter per 100 gram per minute
UCUM | ml/[100]ml | milliliter per 100 milliliter
UCUM | mmol/kg{WetWeight} | millimoles per kg wet weight
UCUM | /min | /min
UCUM | /s | /s

## CID 7182 Abstract Multi-dimensional Image Model Dimension Semantics

Type:

  Extensible

Version:

  20100825

Table CID 7182. Abstract Multi-Dimensional Image Model Dimension Semantics

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110856 | Linear Displacement
DCM | 110857 | Photon Energy
DCM | 110858 | Time
DCM | 110859 | Angle

## CID 7183 Abstract Multi-dimensional Image Model Dimension Units

Type:

  Extensible

Version:

  20100825

Table CID 7183. Abstract Multi-Dimensional Image Model Dimension Units

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7460 “Units of Linear Measurement”
UCUM | ms | Millisecond
UCUM | s | Second
UCUM | deg | Degree
UCUM | rad | Radian

## CID 7184 Abstract Multi-dimensional Image Model Axis Direction

Type:

  Extensible

Version:

  20100825

Table CID 7184. Abstract Multi-Dimensional Image Model Axis Direction

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110860 | Left-Right Axis
DCM | 110861 | Head-Foot Axis
DCM | 110862 | Anterior-Posterior Axis
DCM | 110863 | Apex-Base Axis
DCM | 110864 | Anterior-Inferior Axis
DCM | 110865 | Septum-Wall Axis

## CID 7185 Abstract Multi-dimensional Image Model Axis Orientation

Type:

  Extensible

Version:

  20100825

Table CID 7185. Abstract Multi-Dimensional Image Model Axis Orientation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110866 | Right To Left
DCM | 110867 | Left To Right
DCM | 110868 | Head To Foot
DCM | 110869 | Foot To Head
DCM | 110870 | Anterior To Posterior
DCM | 110871 | Posterior To Anterior
DCM | 110872 | Apex To Base
DCM | 110873 | Base To Apex
DCM | 110874 | Anterior To Inferior
DCM | 110875 | Inferior To Anterior
DCM | 110876 | Septum To Wall
DCM | 110877 | Wall To Septum

## CID 7186 Abstract Multi-dimensional Image Model Qualitative Dimension Sample Semantics

Type:

  Extensible

Version:

  20100825

Table CID 7186. Abstract Multi-Dimensional Image Model Qualitative Dimension Sample Semantics

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 4033 “MR Proton Spectroscopy Metabolites”
DCM | 110810 | Volumetric Diffusion Dxx Component
DCM | 110811 | Volumetric Diffusion Dxy Component
DCM | 110812 | Volumetric Diffusion Dxz Component
DCM | 110813 | Volumetric Diffusion Dyy Component
DCM | 110814 | Volumetric Diffusion Dyz Component
DCM | 110815 | Volumetric Diffusion Dzz Component
DCM | 110834 | RGB R Component
DCM | 110835 | RGB G Component
DCM | 110836 | RGB B Component
DCM | 110837 | YBR FULL Y Component
DCM | 110838 | YBR FULL CB Component
DCM | 110839 | YBR FULL CR Component
DCM | 110840 | YBR PARTIAL Y Component
DCM | 110841 | YBR PARTIAL CB Component
DCM | 110842 | YBR PARTIAL CR Component
DCM | 110843 | YBR ICT Y Component
DCM | 110844 | YBR ICT CB Component
DCM | 110845 | YBR ICT CR Component
DCM | 110846 | YBR RCT Y Component
DCM | 110847 | YBR RCT CB Component
DCM | 110848 | YBR RCT CR Component

## CID 7201 Referenced Image Purposes of Reference

Type:

  Extensible

Version:

  20090409

Table CID 7201. Referenced Image Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121311 | Localizer
DCM | 121312 | Biopsy localizer
DCM | 121313 | Other partial views
DCM | 121314 | Other image of biplane pair
DCM | 121315 | Other image of stereoscopic pair
DCM | 121316 | Images related to standalone object
DCM | 121317 | Spectroscopy
DCM | 121338 | Anatomic image
DCM | 121339 | Functional image
DCM | 121340 | Spectral filtered image
DCM | 121341 | Device localizer
DCM | 121346 | Acquisition frames corresponding to volume
DCM | 121347 | Volume corresponding to spatially-related acquisition frames
DCM | 121348 | Temporal Predecessor
DCM | 121349 | Temporal Successor

## CID 7202 Source Image Purposes of Reference

Type:

  Extensible

Version:

  20110609

Table CID 7202. Source Image Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121320 | Uncompressed predecessor
DCM | 121321 | Mask image for image processing operation
DCM | 121322 | Source image for image processing operation
DCM | 121329 | Source image for montage
DCM | 121330 | Lossy compressed predecessor
DCM | 121358 | For Processing predecessor

## CID 7203 Image Derivation

Type:

  Extensible

Version:

  20141110

Table CID 7203. Image Derivation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113040 | Lossy Compression
DCM | 113041 | Apparent Diffusion Coefficient
DCM | 113042 | Pixel by pixel addition
DCM | 113043 | Diffusion weighted
DCM | 113044 | Diffusion Anisotropy
DCM | 113045 | Diffusion Attenuated
DCM | 113046 | Pixel by pixel division
DCM | 113047 | Pixel by pixel mask
DCM | 113048 | Pixel by pixel Maximum
DCM | 113049 | Pixel by pixel mean
DCM | 113050 | Metabolite Maps from spectroscopy data
DCM | 113051 | Pixel by pixel Minimum
DCM | 113052 | Mean Transit Time
DCM | 113053 | Pixel by pixel multiplication
DCM | 113054 | Negative Enhancement Integral
DCM | 113055 | Regional Cerebral Blood Flow
DCM | 113056 | Regional Cerebral Blood Volume
DCM | 113057 | R-Coefficient
DCM | 113058 | Proton Density
DCM | 113059 | Signal Change
DCM | 113060 | Signal to Noise
DCM | 113061 | Standard Deviation
DCM | 113062 | Pixel by pixel subtraction
DCM | 113063 | T1
DCM | 113064 | T2*
DCM | 113065 | T2
DCM | 113066 | Time Course of Signal
DCM | 113067 | Temperature encoded
DCM | 113068 | Student's T-Test
DCM | 113069 | Time To Peak
DCM | 113070 | Velocity encoded
DCM | 113071 | Z-Score
DCM | 113072 | Multiplanar reformatting
DCM | 113073 | Curved multiplanar reformatting
DCM | 113074 | Volume rendering
DCM | 113075 | Surface rendering
DCM | 113076 | Segmentation
DCM | 113077 | Volume editing
DCM | 113078 | Maximum intensity projection
DCM | 113079 | Minimum intensity projection
DCM | 113085 | Spatial resampling
DCM | 113086 | Edge enhancement
DCM | 113087 | Smoothing
DCM | 113088 | Gaussian blur
DCM | 113089 | Unsharp mask
DCM | 113090 | Image stitching
DCM | 113091 | Spatially-related frames extracted from the volume
DCM | 113092 | Temporally-related frames extracted from the set of volumes
DCM | 113097 | Multi-energy proportional weighting
DCM | 113093 | Polar to Rectangular Scan Conversion

## CID 7205 Purpose of Reference to Alternate Representation

Type:

  Extensible

Version:

  20040322

Table CID 7205. Purpose of Reference to Alternate Representation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121324 | Source image
DCM | 121325 | Lossy compressed image
DCM | 121326 | Alternate SOP Class instance
DCM | 121327 | Full fidelity image
DCM | 121328 | Alternate Photometric Interpretation image

## CID 7210 Related Series Purposes of Reference

Type:

  Extensible

Version:

  20140627

Table CID 7210. Related Series Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122400 | Simultaneously Acquired
DCM | 122401 | Same Anatomy
DCM | 122402 | Same Indication
DCM | 122403 | For Attenuation Correction
DCM | 121323 | Source series for image processing operation

## CID 7215 Spectroscopy Purpose of Reference

Type:

  Extensible

Version:

  20140105

Table CID 7215. Spectroscopy Purpose of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121318 | Spectroscopy Data for Water Phase Correction

## CID 7220 RT Dose Derivation

Type:

  Extensible

Version:

  20140106

Table CID 7220. RT Dose Derivation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121370 | Composed from prior doses
DCM | 121371 | Composed from prior doses and current plan

## CID 7221 RT Dose Purpose of Reference

Type:

  Extensible

Version:

  20140106

Table CID 7221. RT Dose Purpose of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121372 | Source dose for composing current dose

## CID 7222 Parametric Map Derivation Image Purpose of Reference

Type:

  Extensible

Version:

  20141110

Table CID 7222. Parametric Map Derivation Image Purpose of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121322 | Source Image for Image Processing Operation

## CID 7250 Multi-Frame Subset Type

This Context Group specifies the terms used to identify a subset of frames of a multi-frame image. It is used for encoding an equivalent of the Image SOP Instance Reference Macro (see PS3.3) in an HL7 v3 data structure (see HL7 v3 CMET, COCT_RM830120 "A_DicomCompositeObjectReference minimal").

Type:

  Extensible

Version:

  20070625

Table CID 7250. Multi-Frame Subset Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121190 | Referenced Frames
DCM | 121191 | Referenced Segment

## CID 7300 Implant Materials

Type:

  Extensible

Version:

  20101102

Table CID 7300. Implant Materials

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-61166 | Nickel Titanium
SRT | F-611FC | Gold Alloy
SRT | F-61207 | Stainless Steel Material
SRT | F-61DF9 | Polymer
SRT | F-61202 | Carbon Fiber

## CID 7301 Intervention Types

Type:

  Extensible

Version:

  20101102

Table CID 7301. Intervention Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P1-14810 | Hip joint reconstruction
SRT | P1-14505 | Hip joint implantation
SRT | P1-103D3 | Resurfacing of the femoral head
SRT | P1-189C2 | Resurfacing of the patella

## CID 7302 Implant Templates View Orientations

Type:

  Extensible

Version:

  20101102

Table CID 7302. Implant Templates View Orientations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-10206 | Antero-Posterior
SRT | R-10226 | Medio-Lateral
SRT | R-10228 | Lateral-Medial

## CID 7303 Implant Templates Modified View Orientations

Type:

  Extensible

Version:

  20101102

Table CID 7303. Implant Templates Modified View Orientations

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112300 | AP+45
DCM | 112301 | AP-45

## CID 7304 Implant Target Anatomy

Type:

  Extensible

Version:

  20130617

Table CID 7304. Implant Target Anatomy

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-15750 | Ankle Joint
SRT | T-11501 | Cervical Spine
SRT | T-D00F7 | Cervico-Thoracic Spine
SRT | T-15430 | Elbow Joint
SRT | T-11196 | Facial Bones
SRT | T-12710 | Femur
SRT | T-12711 | Head of Femur
SRT | T-D078C | Proximal Femur
SRT | T-12717 | Shaft of Femur
SRT | T-D078D | Distal Femur
SRT | T-1553D | Finger Joint
SRT | T-15710 | Hip Joint
SRT | T-D1213 | Jaw Region
SRT | T-D9200 | Knee
SRT | T-11503 | Lumbar Spine
SRT | T-D0059 | Lumbo-Sacral Spine
SRT | T-11180 | Mandible
SRT | T-11170 | Maxilla
SRT | T-12730 | Patella
SRT | T-12310 | Clavicle
SRT | T-D2220 | Shoulder
SRT | T-12410 | Humerus
SRT | T-1240F | Proximal Humerus
SRT | T-12412 | Shaft of Humerus
SRT | T-1241F | Distal Humerus
SRT | T-12420 | Radius
SRT | T-1242A | Proximal Radius
SRT | T-12423 | Shaft of Radius
SRT | T-1242B | Distal Radius
SRT | T-12430 | Ulna
SRT | T-1243A | Proximal Ulna
SRT | T-12435 | Shaft of Ulna
SRT | T-1243B | Distal Ulna
SRT | T-11100 | Skull
SRT | T-12600 | Hand
SRT | T-11502 | Thoracic Spine
SRT | T-D00FA | Thoraco-Lumbar Spine
SRT | T-15460 | Wrist Joint
SRT | T-12375 | Pelvis
SRT | T-12750 | Fibula
SRT | T-12780 | Talus
SRT | T-12770 | Calcaneus
SRT | T-12740 | Tibia
SRT | T-12746 | Shaft of Tibia
SRT | T-1274B | Distal Tibia

### Note

Consistent with other concepts in this context group that refer to specific bones or joints, the concept for Elbow has been changed from T-D8300 to T-15430 used in a prior version of this table. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 7305 Implant Planning Landmarks

Type:

  Extensible

Version:

  20101102

Table CID 7305. Implant Planning Landmarks

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7306 “Human Hip Implant Planning Landmarks”

## CID 7306 Human Hip Implant Planning Landmarks

Type:

  Extensible

Version:

  20101102

Table CID 7306. Human Hip Implant Planning Landmarks

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112302 | Anatomical axis of femur
DCM | 112303 | Acetabular Center of Rotation
DCM | 112304 | Femur Head Center of Rotation

## CID 7307 Implant Component Types

Type:

  Extensible

Version:

  20101102

Table CID 7307. Implant Component Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7308 “Human Hip Implant Types”
Include CID 7309 “Human Trauma Implant Types”

## CID 7308 Human Hip Implant Types

Type:

  Extensible

Version:

  20101102

Table CID 7308. Human Hip Implant Component Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112305 | Acetabular Cup Shell
DCM | 112306 | Acetabular Cup Insert
DCM | 112307 | Acetabular Cup Monoblock
SRT | A-04459 | Femoral Head Prosthesis
DCM | 112308 | Femoral Head Ball Component
DCM | 112309 | Femoral Head Cone Taper Component
DCM | 112310 | Femoral Stem
DCM | 112311 | Femoral Stem Distal Component
DCM | 112312 | Femoral Stem Proximal Component
DCM | 112313 | Femoral Stem Component
DCM | 112314 | Neck Component
DCM | 112315 | Monoblock Stem
DCM | 112325 | Distal Centralizer
DCM | 112316 | Prosthetic Shaft Augment
DCM | 112317 | Femoral Head Resurfacing Component

## CID 7309 Human Trauma Implant Types

Type:

  Extensible

Version:

  20101102

Table CID 7309. Human Trauma Implant Component Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-12030 | Screw
SRT | A-12010 | Bone Plate
SRT | A-12018 | DHS Plate
SRT | A-12020 | Bone Nail

## CID 7310 Implant Fixation Method

Type:

  Extensible

Version:

  20101102

Table CID 7310. Implant Fixation Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P0-02126 | Anchoring
SRT | P0-02125 | Fusion
SRT | P0-021D6 | Gluing
SRT | P1-1099B | Internal fixation using internal fixator system
SRT | P1-10999 | Internal fixation using plate
SRT | P1-10998 | Internal fixation using screw
SRT | P1-10997 | Internal fixation using staple
SRT | R-41C37 | Cemented component fixation
SRT | R-42808 | Uncemented component fixation
SRT | P1-08080 | Repair by nailing
DCM | 112318 | Pinning
DCM | 112319 | Sewing
DCM | 112320 | Bolting
DCM | 112321 | Wedging

## CID 7320 Planning Methods

Type:

  Extensible

Version:

  20101102

Table CID 7320. Planning Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 112340 | Generic 2D Planning
DCM | 112341 | Generic 3D Planning
DCM | 112342 | Generic Planning for Hip Replacement
DCM | 112343 | Generic Planning for Knee Replacement
DCM | 112344 | Müller Method Planning for Hip Replacement

## CID 7445 Device Participating Roles

Type:

  Extensible

Version:

  20120406

Table CID 7445. Device Participating Roles

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113859 | Irradiating Device
DCM | 121097 | Recording
DCM | 113942 | X-Ray Reading Device

## CID 7450 Person Roles

Type:

  Extensible

Version:

  20040112

Table CID 7450. Person Roles

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121025 | Patient
SRT | J-00552 | Healthcare professional
SRT | S-11090 | Friend
Include CID 7451 “Family Member”
Include CID 7452 “Organizational Roles”

## CID 7451 Family Member

Type:

  Extensible

Version:

  20040112

Table CID 7451. Family Member

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | S-10121 | Natural mother
SRT | S-10131 | Natural father
SRT | S-10151 | Natural sister
SRT | S-10161 | Natural brother
SRT | S-101A1 | Aunt
SRT | S-101A2 | Uncle
SRT | S-10154 | Half-sister
SRT | S-10164 | Half-brother
SRT | S-10115 | Natural grand-mother
SRT | S-10116 | Natural grand-father
SRT | S-10181 | Natural daughter
SRT | S-10191 | Natural son
SRT | S-101A9 | Female first cousin
SRT | S-101AA | Male first cousin

## CID 7452 Organizational Roles

Type:

  Extensible

Version:

  20141110

Table CID 7452. Organizational Roles

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
SRT | J-0016E | Medical Practitioner | 158965000 | C1306754
SRT | J-004E8 | Physician | 309343006 | C0031831
SRT | J-07100 | Nurse | 106292003 | C0028661
SRT | J-00187 | Radiologic Technologist | 159016003 | C0402007
SRT | J-00187 | Radiographer | 159016003 | C0402007
UMLS | C1144859 | Intern |  | C1144859
SRT | J-005E6 | Resident | 405277009 | C1320928
SRT | J-00172 | Registrar | 158971006 | C0401974
DCM | 121088 | Fellow |  |
SRT | J-005E8 | Attending | 405279007 | C1320929
SRT | J-0050A | Consultant | 309390008 | C0586911
SRT | J-0714A | Scrub nurse | 415506007 | C1531952
SRT | J-00556 | Surgeon | 304292004 | C0582175
DCM | 121092 | Sonologist |  |
UMLS | C1954848 | Sonographer |  | C1954848
UMLS | C2985483 | Radiation Physicist |  | C2985483
UMLS | C1708969 | Medical Physicist |  | C1708969

### Note

1. The distinction between a "physician" and a "surgeon" and a "medical practitioner" is subject to regional variation. In the US, "physician" is often equated with "medical practitioner", and a "surgeon" is considered to be a "physician". In the UK, a "surgeon" is a "medical practitioner" but is not a "physician". In SNOMED, "physician" and "surgeon" are distinct siblings with no direct relationship, and both are children of "medical practitioner". It is recomended that "medical practitioner" be used rather than "physician" when there is uncertainty over whether the person is or is not a "surgeon".

2. There is no distinction between a "radiographer" and a "radiologic technologist", hence the same SNOMED concept is used for both, and "radiologic technologist" is provided as a synonym for use in the US.

3. In the US, the medical practitioner not in training responsible for the care of a hospital patient is referred to as an "attending". In the UK they are referrered to as a "consultant". Though these two concepts are essentially the same, they are separate concepts in SNOMED, which defines no explicit relationship between them.

## CID 7453 Performing Roles

Type:

  Extensible

Version:

  20141110

Table CID 7453. Performing Roles

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID | UMLS Concept Unique ID
--- | --- | --- | --- | ---
DCM | 121094 | Performing |  |
UMLS | C1709880 | Referring |  | C1709880
DCM | 121096 | Requesting |  |
DCM | 121097 | Recording |  |
DCM | 121098 | Verifying |  |
DCM | 121099 | Assisting |  |
SRT | J-0714B | Circulating Nurse | 413854007 | C1531633
DCM | 121101 | Standby |  |
DCM | 113850 | Irradiation Authorizing |  |
DCM | 113851 | Irradiation Administering |  |

## CID 7454 Species

Type:

  Extensible

Version:

  20060822

Table CID 7454. Species

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | L-85B00 | homo sapiens
SRT | L-80A00 | Feline species
SRT | L-80400 | Equine species
SRT | L-80300 | Ovine species
SRT | L-80500 | Porcine species
SRT | L-80200 | Caprine species
SRT | L-80700 | Canine species
SRT | L-80100 | Bovine species

## CID 7455 Sex

This Context Group includes terms for the finding of sex of a subject for clinical purposes, such as selection of sex-based growth metrics.

Type:

  Non-Extensible

Version:

  20040112

Table CID 7455. Sex

Coding Scheme Designator | Code Value | Code Meaning | Patient's Sex (0010,0040) Equivalent
--- | --- | --- | ---
DCM | M | Male | M
DCM | F | Female | F
DCM | U | Unknown sex |
DCM | MP | Male Pseudohermaphrodite |
DCM | FP | Female Pseudohermaphrodite |
DCM | H | Hermaphrodite |
DCM | MC | Male changed to Female |
DCM | FC | Female changed to Male |
DCM | 121104 | Ambiguous sex |
DCM | 121102 | Other sex |
DCM | 121103 | Undetermined sex | O

### Note

1. These terms are distinct from the gender of a subject for administrative purposes, although the default value for clinical sex is often based on the administrative gender (e.g., see TID 1007 “Subject Context, Patient”). The administrative value "O" from Patient's Sex (0010,0040) maps by default to "undetermined" for clinical purposes.

2. This Context Group in a prior edition of the Standard included codes improperly attributed to ISO 5218.

3. These terms are derived from the terminology and codes for sex in ASTM E1633-02a "Standard Specification for Coded Values Used in the Electronic Health Record."

## CID 7456 Units of Measure for Age

Type:

  Non-Extensible

Version:

  20020904

Table CID 7456. Units of Measure for Age

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | a | year
UCUM | mo | month
UCUM | wk | week
UCUM | d | day
UCUM | h | hour
UCUM | min | minute

## CID 7460 Units of Linear Measurement

Type:

  Extensible

Version:

  20020904

Table CID 7460. Units of Linear Measurement

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | cm | centimeter
UCUM | mm | millimeter
UCUM | um | micrometer

## CID 7461 Units of Area Measurement

Type:

  Extensible

Version:

  20020904

Table CID 7461. Units of Area Measurement

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | cm2 | square centimeter
UCUM | mm2 | square millimeter
UCUM | um2 | square micrometer

## CID 7462 Units of Volume Measurement

Type:

  Extensible

Version:

  20020904

Table CID 7462. Units of Volume Measurement

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | dm3 | cubic decimeter
UCUM | cm3 | cubic centimeter
UCUM | mm3 | cubic millimeter
UCUM | um3 | cubic micrometer

### Note

A "cubic decimeter" is a "liter", just as a "cubic centimeter" is a "milliliter" (of water). Though there are specific units "l" and "ml" in UCUM, only one form is included here, since this context group is intended for use for volume measurements of a physical object derived from one or more images, rather than of fluid volume.

## CID 7464 General Region of Interest Measurement Modifiers

This context group contains modifiers of measurements of ROIs.

Type:

  Extensible

Version:

  20121101

Table CID 7464. General Region of Interest Measurement Modifiers

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3488 “Min/Max/Mean”
SRT | R-10047 | Standard Deviation
SRT | R-40507 | Total
SRT | R-00319 | Median
SRT | R-0032E | Mode
DCM | 126031 | Peak Value Within ROI
UMLS | C0681921 | Coefficient of Variance
DCM | 126051 | Skewness
DCM | 126052 | Kurtosis
UMLS | C1711260 | Variance
UMLS | C2347976 | Root Mean Square

## CID 7465 Measurements Derived From Multiple ROI Measurements

Type:

  Extensible

Version:

  20121101

Table CID 7465. Measurements Derived From Multiple ROI Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 226 “Population Statistical Descriptors”
Include CID 227 “Sample Statistical Descriptors”

## CID 7466 PET Region of Interest Measurements

Type:

  Extensible

Version:

  20141110

Table CID 7466. PET Region of Interest Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126032 | Metabolic Volume
DCM | 126033 | Total Lesion Glycolysis
DCM | 126034 | Glycolysis
DCM | 126035 | Total Lesion Proliferation
DCM | 126036 | Proliferative Activity
DCM | 126037 | Standardized Added Metabolic Activity
DCM | 126038 | Standardized Added Metabolic Activity Background
DCM | 126039 | Lesion to Background SUV Ratio
DCM | 126040 | Background for Lesion to Background SUV Ratio

## CID 7467 Grey Level Co-occurrence Matrix Measurements

Type:

  Extensible

Version:

  20141110

Table CID 7467. Grey Level Co-occurrence Matrix Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 126060 | Entropy of GLCM
DCM | 126061 | Energy of GLCM
DCM | 126062 | Homogeneity of GLCM
DCM | 126063 | Contrast of GLCM
DCM | 126064 | Dissimilarity of GLCM
DCM | 126065 | ASM of GLCM
DCM | 126066 | Correlation of GLCM

## CID 7468 Texture Measurements

Type:

  Extensible

Version:

  20141110

Table CID 7468. Texture Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7467 “Grey Level Co-occurrence Matrix Measurements”
DCM | 126050 | Fractal Dimension

## CID 7469 Generic Intensity and Size Measurements

Type:

  Extensible

Version:

  20141110

Table CID 7469. Generic Intensity and Size Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7180 “Abstract Multi-dimensional Image Model Component Semantics”
Include CID 7470 “Linear Measurements”
Include CID 7471 “Area Measurements”
Include CID 7472 “Volume Measurements”

## CID 7470 Linear Measurements

Type:

  Extensible

Version:

  20050822

Table CID 7470. Linear Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A22A | Length
DCM | 121211 | Path length
DCM | 121206 | Distance
SRT | G-A220 | Width
SRT | G-D785 | Depth
SRT | M-02550 | Diameter
SRT | G-A185 | Long Axis
SRT | G-A186 | Short Axis
SRT | G-A193 | Major Axis
SRT | G-A194 | Minor Axis
SRT | G-A195 | Perpendicular Axis
SRT | G-A196 | Radius
SRT | G-A197 | Perimeter
SRT | M-02560 | Circumference
SRT | G-A198 | Diameter of circumscribed circle
DCM | 121207 | Height

## CID 7471 Area Measurements

Type:

  Extensible

Version:

  20020904

Table CID 7471. Area Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A166 | Area
SRT | G-A16A | Area of defined region

## CID 7472 Volume Measurements

Type:

  Extensible

Version:

  20020904

Table CID 7472. Volume Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-D705 | Volume
DCM | 121216 | Volume estimated from single 2D region
DCM | 121218 | Volume estimated from two non-coplanar 2D regions
DCM | 121217 | Volume estimated from three or more non-coplanar 2D regions
DCM | 121222 | Volume of sphere
DCM | 121221 | Volume of ellipsoid
DCM | 121220 | Volume of circumscribed sphere
DCM | 121219 | Volume of bounding three dimensional region

## CID 7473 General Area Calculation Methods

Type:

  Extensible

Version:

  20070827

Table CID 7473. General Area Calculation Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122501 | Area of closed irregular polygon
DCM | 122502 | Area of a closed NURBS

## CID 7474 General Volume Calculation Methods

Type:

  Extensible

Version:

  20141110

Table CID 7474. General Volume Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122503 | Integration of sum of closed areas on contiguous slices
DCM | 126030 | Sum of segmented voxel volumes

## CID 7480 Breed

Type:

  Extensible

Version:

  20110818

Table CID 7480. Breed

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7486 “Mixed Breeds”
SRT | L-80139 | Hereford cattle superbreed
SRT | L-8C338 | Merino sheep superbreed
SRT | L-80121 | Africander cattle breed
SRT | L-80122 | Ankole cattle breed
SRT | L-80123 | Ankole-Watusi cattle breed
SRT | L-80124 | Baladicattle cattle breed
SRT | L-80125 | Belmont Red cattle breed
SRT | L-80126 | Bonsmara cattle breed
SRT | L-80127 | Damietta cattle breed
SRT | L-80128 | Horro cattle breed
SRT | L-80129 | Kuri cattle breed
SRT | L-8012A | Nguni cattle breed
SRT | L-8012B | Philippine Native cattle breed
SRT | L-8012C | Romagnola cattle breed
SRT | L-8012E | Sanhe cattle breed
SRT | L-8012F | Tswana cattle breed
SRT | L-80138 | Tuli cattle breed
SRT | L-8013A | Aliab Dinka cattle breed
SRT | L-8013B | Alur cattle breed
SRT | L-8013C | Ankina cattle breed
SRT | L-8013D | Apulian Podolian cattle breed
SRT | L-8013E | Arado cattle breed
SRT | L-8013F | Aweil Dinka cattle breed
SRT | L-8014C | Bahima cattle breed
SRT | L-8014D | Bapedi cattle breed
SRT | L-8014E | Baria (Vietnam/Madagascar) cattle breed
SRT | L-8014F | Barotse cattle breed
SRT | L-8015A | Barra do Cuanzo cattle breed
SRT | L-8015B | Bashi cattle breed
SRT | L-8015C | Basuto cattle breed
SRT | L-8015D | Batangas cattle breed
SRT | L-8015E | Bavenda cattle breed
SRT | L-8015F | Beja cattle breed
SRT | L-80161 | Calabrian cattle breed
SRT | L-80162 | Blonde-du Cap Bon cattle breed
SRT | L-80163 | Cham-Doc cattle breed
SRT | L-80164 | Chernigov cattle breed
SRT | L-80165 | Chino Santandereano cattle breed
SRT | L-80166 | Cinisara cattle breed
SRT | L-80167 | Cuprem Hybrid cattle breed
SRT | L-80168 | Dabieshan cattle breed
SRT | L-80169 | Damara cattle breed
SRT | L-8016A | Danakil cattle breed
SRT | L-8016B | Dnieper cattle breed
SRT | L-8016C | Doayo cattle breed
SRT | L-8016D | Eastern Nuer cattle breed
SRT | L-8016E | Egyptian cattle breed
SRT | L-8016F | Fogera cattle breed
SRT | L-80177 | Garfagnina cattle breed
SRT | L-80178 | Grati cattle breed
SRT | L-80179 | Gaunling cattle breed
SRT | L-8017A | Halhin Gol cattle breed
SRT | L-8017B | Holmonger cattle breed
SRT | L-8017C | Ilocos cattle breed
SRT | L-8017D | Iloilo cattle breed
SRT | L-8017E | Inkuku cattle breed
SRT | L-8017F | Iskar cattle breed
SRT | L-80180 | Istrian cattle breed
SRT | L-80181 | Javanese Ongole cattle breed
SRT | L-80182 | Javanese Zebu cattle breed
SRT | L-80183 | Jinnan cattle breed
SRT | L-80184 | Kalmyk cattle breed
SRT | L-80185 | Kaokoveld cattle breed
SRT | L-80186 | Kazakh Whitehead cattle breed
SRT | L-80187 | Kedah-Kelantan cattle breed
SRT | L-80188 | Kigezi cattle breed
SRT | L-80189 | Kisantu cattle breed
SRT | L-8018A | Kolubara cattle breed
SRT | L-8018B | Kurgan cattle breed
SRT | L-8018C | Kyoga cattle breed
SRT | L-8018D | Lucanian cattle breed
SRT | L-8018E | Maremmana cattle breed
SRT | L-8018F | Marianas cattle breed
SRT | L-80190 | Maryuti cattle breed
SRT | L-80191 | Mauritius Creole cattle breed
SRT | L-80192 | Menufi cattle breed
SRT | L-80193 | Mezzalina cattle breed
SRT | L-80194 | Modicana cattle breed
SRT | L-80195 | Moi cattle breed
SRT | L-80196 | Nama cattle breed
SRT | L-80197 | Nanyang cattle breed
SRT | L-80198 | N'Dama Sanga cattle breed
SRT | L-80199 | Nganda cattle breed
SRT | L-8019A | Nilotic Sanga cattle breed
SRT | L-8019B | Nkone cattle breed
SRT | L-8019C | North Malawi Angoni cattle breed
SRT | L-8019D | Nuer cattle breed
SRT | L-8019E | Nuras cattle breed
SRT | L-8019F | Nyoro cattle breed
SRT | L-801A0 | Ovambo cattle breed
SRT | L-801A1 | Pantelleria cattle breed
SRT | L-801A2 | Pinzhou cattle breed
SRT | L-801A3 | Porto Amboim cattle breed
SRT | L-801A4 | Posavina cattle breed
SRT | L-801A5 | Romanian Steppe cattle breed
SRT | L-801A6 | Saidi cattle breed
SRT | L-801A7 | Sardo-Modicana cattle breed
SRT | L-801A8 | Sengologa cattle breed
SRT | L-801A9 | Serere cattle breed
SRT | L-801AA | Seshaga cattle breed
SRT | L-801AB | Siberian Black Pied cattle breed
SRT | L-801AC | Socotra cattle breed
SRT | L-801AD | Southern Tswana cattle breed
SRT | L-801AE | Spreca cattle breed
SRT | L-801AF | Sunkuma cattle breed
SRT | L-801B0 | Taiwan Zebu cattle breed
SRT | L-801B1 | Thai cattle breed
SRT | L-801B2 | Thailand Fighting Zebu cattle breed
SRT | L-801B3 | Thanh-Hoa cattle breed
SRT | L-801B4 | Tibetan cattle breed
SRT | L-801B5 | Tonga cattle breed
SRT | L-801B6 | Toro cattle breed
SRT | L-801B7 | Tuni cattle breed
SRT | L-801B8 | Turkish Gray Steppe cattle breed
SRT | L-801B9 | Tuy-Hoa cattle breed
SRT | L-801BA | Ujumqin cattle breed
SRT | L-801BB | Abigar cattle breed
SRT | L-801BC | Africangnus cattle breed
SRT | L-801BD | Agerolese cattle breed
SRT | L-801BE | Albese cattle breed
SRT | L-801BF | Ukrainian Gray cattle breed
SRT | L-801C0 | Vietnamese Yellow cattle breed
SRT | L-801C1 | Watusi (USA) cattle breed
SRT | L-801C2 | Wenshan cattle breed
SRT | L-801C3 | Yakut cattle breed
SRT | L-801C4 | Yunnan Zebu cattle breed
SRT | L-801C5 | Zambia Angoni cattle breed
SRT | L-801C6 | Drakensberger cattle breed
SRT | L-801C7 | Modicana lowland cattle breed
SRT | L-801C8 | Taiwan Yellow cattle breed
SRT | L-801C9 | Menggu cattle breed
SRT | L-801CA | Albères cattlebreed
SRT | L-801CB | Alentejana cattlebreed
SRT | L-801CC | American White Park cattle breed
SRT | L-801CD | Amerifaxcattle breed
SRT | L-801CE | Anatolian Black cattle breed
SRT | L-801CF | Andalusian Black cattle breed
SRT | L-801D0 | Andalusian Gray cattle breed
SRT | L-801D1 | Angeln cattle breed
SRT | L-801D2 | Asturian Mountain cattle breed
SRT | L-801D3 | Asturian Valley cattle breed
SRT | L-801D4 | Aubrac cattle breed
SRT | L-801D5 | Aulie-Ata cattle breed
SRT | L-801D6 | Australian Lowline cattle breed
SRT | L-801D7 | Barzona cattle breed
SRT | L-801D8 | Bazadais cattle breed
SRT | L-801D9 | Beefmaker cattle breed
SRT | L-801DA | Belarus Red cattle breed
SRT | L-801DB | Belgian Blue cattle breed
SRT | L-801DC | Belgian Red cattle breed
SRT | L-801DD | Belmont Adaptaur cattle breed
SRT | L-801DE | Berrendas cattle breed
SRT | L-801DF | Blacksided Trondheim and Norland cattle breed
SRT | L-801E0 | Blanco Orejinegro cattle breed
SRT | L-801E1 | Braunvieh cattle breed
SRT | L-801E2 | British White cattle breed
SRT | L-801E3 | Cachena cattle breed
SRT | L-801E4 | Canary Island cattle breed
SRT | L-801E5 | Carinthian Blond cattle breed
SRT | L-801E6 | Caucasian cattle breed
SRT | L-801E7 | Charolais cattle breed
SRT | L-801EA | Chinese Black-and-White cattle breed
SRT | L-801EB | Corriente cattle breed
SRT | L-801EC | Costeño con Cuernos cattle breed
SRT | L-801ED | Damascus cattle breed
SRT | L-801EE | Danish Red cattle breed
SRT | L-801EF | Devon cattle breed
SRT | L-801F0 | Dølafe cattle breed
SRT | L-801F1 | Dutch Belted cattle breed
SRT | L-801F2 | Dutch Friesian cattle breed
SRT | L-801F3 | English Longhorn cattle breed
SRT | L-801F4 | Estonian Red cattle breed
SRT | L-801F5 | Evolène cattle breed
SRT | L-801F6 | Fighting Bull cattle breed
SRT | L-801F7 | Fjall cattle breed
SRT | L-801F8 | Florida Cracker/Pineywoods cattle breed
SRT | L-801F9 | Galician Blond cattle breed
SRT | L-801FA | Gascon cattle breed
SRT | L-801FB | German Red Pied cattle breed
SRT | L-801FC | Glan cattle breed
SRT | L-801FD | Gloucester cattle breed
SRT | L-801FE | Groningen Whiteheaded cattle breed
SRT | L-801FF | Hartón cattle breed
SRT | L-8031A | Bündner Oberland sheep breed
SRT | L-8031B | British Milk Sheep breed
SRT | L-8031C | Brillenschaf sheep breed
SRT | L-8031D | Brecknock Hill Cheviot sheep breed
SRT | L-8031E | Cholistani sheep breed
SRT | L-8031F | Bibrik sheep breed
SRT | L-8032A | Columbia sheep breed
SRT | L-8032B | Black Welsh Mountain Sheep breed
SRT | L-8032C | Blackhead Persian sheep breed
SRT | L-8032D | Bleu du Maine sheep breed
SRT | L-8032E | Bluefaced Leicester sheep breed
SRT | L-8032F | Bond sheep breed
SRT | L-8033A | Border Leicester sheep breed
SRT | L-8033B | Boreray sheep breed
SRT | L-8033C | Bovska sheep breed
SRT | L-8033D | Braunes Bergschaf sheep breed
SRT | L-8033E | Brazilian Somali sheep breed
SRT | L-8033F | Beulah Speckled-Face sheep breed
SRT | L-8034A | Dartmoor sheep breed
SRT | L-8034B | Fabrianese sheep breed
SRT | L-8034C | Exmoor Horn sheep breed
SRT | L-8034D | Elliottdale sheep breed
SRT | L-8034E | Drysdale sheep breed
SRT | L-8034F | Dorset Down sheep breed
SRT | L-80351 | German Blackheaded Mutton sheep breed
SRT | L-80352 | Kooka sheep breed
SRT | L-80353 | Friesian Milk Sheep breed
SRT | L-80354 | Gansu Alpine Fine-wool sheep breed
SRT | L-80355 | German Whiteheaded Mutton sheep breed
SRT | L-80356 | Graue Gehoernte Heidschnucke sheep breed
SRT | L-80357 | Han sheep breed
SRT | L-80358 | Gromark sheep breed
SRT | L-80359 | Gulf Coast Native sheep breed
SRT | L-8035A | Dorper sheep breed
SRT | L-8035B | Devon Closewool sheep breed
SRT | L-8035C | Deutsches Blaukoepfiges Fleischschaf sheep breed
SRT | L-8035D | Derbyshire Gritstone sheep breed
SRT | L-8035E | Coburger Fuchsschaf sheep breed
SRT | L-8035F | Danish Landrace sheep breed
SRT | L-80360 | Gute sheep breed
SRT | L-80361 | Hampshire sheep breed
SRT | L-80362 | Gentile di Puglia sheep breed
SRT | L-80363 | German Mountain sheep breed
SRT | L-80364 | Luzein sheep breed
SRT | L-80365 | Katahdin sheep breed
SRT | L-80366 | Leineschaf sheep breed
SRT | L-80367 | Lincoln Longwool sheep breed
SRT | L-80368 | Llanwenog sheep breed
SRT | L-80369 | Lleyn sheep breed
SRT | L-8036A | Damara sheep breed
SRT | L-8036B | Damani sheep breed
SRT | L-8036C | Dalesbred sheep breed
SRT | L-8036D | Dala sheep breed
SRT | L-8036E | Criollo sheep breed
SRT | L-8036F | Cormo sheep breed
SRT | L-80370 | Lati sheep breed
SRT | L-80371 | Lonk sheep breed
SRT | L-80372 | Langhe sheep breed
SRT | L-80373 | Manx Loaghtan sheep breed
SRT | L-80374 | Masai sheep breed
SRT | L-80375 | Merinolandschaf sheep breed
SRT | L-80376 | Lohi sheep breed
SRT | L-80377 | Ile-de-France sheep breed
SRT | L-80378 | Hasht Nagri sheep breed
SRT | L-80379 | Hazaragie sheep breed
SRT | L-8037A | Coopworth sheep breed
SRT | L-8037B | Comisana sheep breed
SRT | L-8037C | Comeback sheep breed
SRT | L-8037D | Sicilian Barbary sheep breed
SRT | L-8037E | Africana sheep breed
SRT | L-8037F | Welsh Mountain Badger Faced sheep breed
SRT | L-80380 | Hebridean sheep breed
SRT | L-80381 | Heidschnucke sheep breed
SRT | L-80382 | Herdwick sheep breed
SRT | L-80383 | Hill Radnor sheep breed
SRT | L-80384 | Icelandic sheep breed
SRT | L-80385 | Harnai sheep breed
SRT | L-80386 | Istrian Pramenka sheep breed
SRT | L-80387 | Jacob sheep breed
SRT | L-80388 | Jezerskosolcavska sheep breed
SRT | L-80389 | Kachhi sheep breed
SRT | L-8038A | Wensleydale sheep breed
SRT | L-8038B | West African Dwarf sheep breed
SRT | L-8038C | White Suffolk sheep breed
SRT | L-8038D | Whiteface Dartmoor sheep breed
SRT | L-8038E | Whiteface Woodland sheep breed
SRT | L-8038F | Xinjiang Finewool sheep breed
SRT | L-80390 | Kajli sheep breed
SRT | L-80391 | Hog Island Sheep breed
SRT | L-80392 | Biellese sheep breed
SRT | L-80393 | Chios sheep breed
SRT | L-80394 | Santa Cruz sheep breed
SRT | L-80395 | Charollais sheep breed
SRT | L-80396 | Castlemilk Moorit sheep breed
SRT | L-80397 | Campanian Barbary sheep breed
SRT | L-80398 | California Variegated Mutant sheep breed
SRT | L-80399 | California Red sheep breed
SRT | L-8039A | Sopravissana sheep breed
SRT | L-8039B | Somali sheep breed
SRT | L-8039C | Welsh Hill Speckled Face sheep breed
SRT | L-8039D | Skudde sheep breed
SRT | L-8039E | Waziri sheep breed
SRT | L-8039F | Shetland sheep breed
SRT | L-80403 | Cambridge sheep breed
SRT | L-80404 | Solognote sheep breed
SRT | L-8040A | Colombian Criollo horse breed
SRT | L-8040B | Comtois horse breed
SRT | L-8040C | Corsican horse breed
SRT | L-8040D | Costa Rican Saddle Horse horse breed
SRT | L-8040E | Costeno horse breed
SRT | L-8040F | Cuban Paso horse breed
SRT | L-80420 | Rough Fell sheep breed
SRT | L-8042D | Danish Warmblood horse breed
SRT | L-80432 | Swaledale sheep breed
SRT | L-80434 | Polypay sheep breed
SRT | L-80441 | Pagliarola sheep breed
SRT | L-80442 | Pomeranian Coarsewool sheep breed
SRT | L-80443 | Sheep, Breed Undetermined sheep breed
SRT | L-80444 | Orkney sheep breed
SRT | L-80445 | Old Norwegian sheep breed
SRT | L-80446 | Old Format Sheep breed
SRT | L-80447 | Norwegian Fur sheep breed
SRT | L-80448 | Norfolk Horn sheep breed
SRT | L-80449 | Navajo-Churro sheep breed
SRT | L-80466 | Racka sheep breed
SRT | L-80467 | Rasa Aragonesa sheep breed
SRT | L-80468 | Red Engadine sheep breed
SRT | L-80469 | Rhoenschaf sheep breed
SRT | L-80470 | Hucul horse breed
SRT | L-80471 | AraAppaloosa horse breed
SRT | L-80472 | Argentine Criollo horse breed
SRT | L-80473 | Argentine Polo Pony horse breed
SRT | L-80474 | Australian Pony horse breed
SRT | L-80475 | Auxois horse breed
SRT | L-80476 | Avelignese horse breed
SRT | L-80477 | Azerbaijan horse breed
SRT | L-80478 | Azores horse breed
SRT | L-80479 | Bali horse breed
SRT | L-8047A | Balikun horse breed
SRT | L-8047B | Waziri horse breed
SRT | L-8047C | Banker Horse horse breed
SRT | L-8047D | Bardigiano horse breed
SRT | L-8047E | Batak horse breed
SRT | L-8047F | Bavarian Warmblood horse breed
SRT | L-80480 | Belgian Ardennais horse breed
SRT | L-80481 | Belgian Halfblood horse breed
SRT | L-80482 | Belgian Warmblood horse breed
SRT | L-80483 | Bhutia horse breed
SRT | L-80484 | Black Sea Horse horse breed
SRT | L-80485 | Bosnian horse breed
SRT | L-80486 | Boulonnais horse breed
SRT | L-80487 | Brandenburg horse breed
SRT | L-80488 | Brazilian Sport Horse horse breed
SRT | L-80489 | British Appaloosa horse breed
SRT | L-8048A | British Riding Pony horse breed
SRT | L-8048B | British Spotted Pony horse breed
SRT | L-8048C | Buohai horse breed
SRT | L-8048D | Buryat horse breed
SRT | L-8048E | Calabrian horse breed
SRT | L-8048F | Camargue horse breed
SRT | L-80490 | Canadian Cutting Horse horse breed
SRT | L-80491 | Canadian Rustic Pony horse breed
SRT | L-80492 | Canadian Sport Horse horse breed
SRT | L-80493 | Canik horse breed
SRT | L-80494 | Cape Horse horse breed
SRT | L-80496 | Cerbat horse breed
SRT | L-80497 | Chakouyi horse breed
SRT | L-80498 | Chara Horse horse breed
SRT | L-80499 | Chickasaw horse breed
SRT | L-8049A | Chilote horse breed
SRT | L-8049B | Chinese Kazakh horse breed
SRT | L-8049C | Chinese Mongolian horse breed
SRT | L-8049D | Chumbivilcas horse breed
SRT | L-8049E | Chumysh horse breed
SRT | L-8049F | Cirit horse breed
SRT | L-804A1 | Irish Draft horse breed
SRT | L-804A2 | Irish Hunter horse breed
SRT | L-804A3 | Cuban Trotter horse breed
SRT | L-804A4 | Italian Heavy Draft horse breed
SRT | L-804A5 | Jabe horse breed
SRT | L-804A6 | Java horse breed
SRT | L-804A7 | Vendéen sheep breed
SRT | L-804A8 | Czech Warmblood horse breed
SRT | L-804A9 | Jinhong horse breed
SRT | L-804AA | Jinzhou horse breed
SRT | L-804AC | Danubian horse breed
SRT | L-804AD | Karachai horse breed
SRT | L-804AE | Karakacan horse breed
SRT | L-804AF | Kathiawari horse breed
SRT | L-804B1 | Ke-Er-Qin horse breed
SRT | L-804B2 | Kirgiz horse breed
SRT | L-804B3 | Kuznet horse breed
SRT | L-804B4 | Landais horse breed
SRT | L-804B5 | Lewitzer horse breed
SRT | L-804B6 | Lichuan horse breed
SRT | L-804B7 | Lijiang horse breed
SRT | L-804B8 | Llanero horse breed
SRT | L-804B9 | Lombok horse breed
SRT | L-804BA | Lundy Pony horse breed
SRT | L-804BB | Malakan horse breed
SRT | L-804BC | Malopolski horse breed
SRT | L-804BD | Datong horse breed
SRT | L-804BE | Mangalarga Paulista horse breed
SRT | L-804BF | Dulmen Pony horse breed
SRT | L-804C1 | Maremmana horse breed
SRT | L-804C2 | Marwari horse breed
SRT | L-804C3 | Megezh horse breed
SRT | L-804C4 | Megrel horse breed
SRT | L-804C5 | Merens horse breed
SRT | L-804C6 | Messara horse breed
SRT | L-804C7 | Sumba horse breed
SRT | L-804C8 | Sumbawa horse breed
SRT | L-804C9 | Swedish Ardennes horse breed
SRT | L-804CA | Dutch Tuigpaard horse breed
SRT | L-804CB | East and Southeast Anadolu horse breed
SRT | L-804CC | Thai Pony horse breed
SRT | L-804CD | Thessalonian horse breed
SRT | L-804CE | Tibetan horse breed
SRT | L-804CF | Tieling horse breed
SRT | L-804D1 | Timor horse breed
SRT | L-804D2 | Trakya horse breed
SRT | L-804D3 | Trote en Gallope horse breed
SRT | L-804D4 | Turkoman horse breed
SRT | L-804D5 | Tushin horse breed
SRT | L-804D6 | Tuva horse breed
SRT | L-804D7 | Uzunyayla horse breed
SRT | L-804D9 | Voronezh Coach Horse horse breed
SRT | L-804DA | Elegant Warmblood horse breed
SRT | L-804DB | Welsh Cob horse breed
SRT | L-804DC | Welsh Mountain Pony horse breed
SRT | L-804DE | English Hack horse breed
SRT | L-804DF | Wurttemberg horse breed
SRT | L-804E1 | Xilingol horse breed
SRT | L-804E2 | Yanqi horse breed
SRT | L-804E3 | Yemeni Horses horse breed
SRT | L-804E4 | Yili horse breed
SRT | L-804E5 | Yiwu horse breed
SRT | L-804E6 | Yunnan horse breed
SRT | L-804E7 | German Riding Pony horse breed
SRT | L-804E8 | Guanzhong horse breed
SRT | L-804E9 | Guizhou horse breed
SRT | L-804EA | Guoxia horse breed
SRT | L-804EB | Erlunchun horse breed
SRT | L-804EC | Half Saddlebred horse breed
SRT | L-804ED | Flores horse breed
SRT | L-804EE | Freiberg horse breed
SRT | L-804EF | Hessen horse breed
SRT | L-804F1 | Hinis horse breed
SRT | L-804F2 | Hirzai horse breed
SRT | L-804F3 | Hungairan Coldblood horse breed
SRT | L-804F4 | Hungarian Dun horse breed
SRT | L-804F5 | Hungarian Sport Horse horse breed
SRT | L-804F6 | International Striped Horse horse breed
SRT | L-804F7 | Irish Cob horse breed
SRT | L-804F8 | Mezen horse breed
SRT | L-804F9 | Mezohegyes Sport Horse horse breed
SRT | L-804FA | French Cob horse breed
SRT | L-804FB | French Saddle pony horse breed
SRT | L-804FC | Murakoz horse breed
SRT | L-804FE | Finnhorse Draft horse breed
SRT | L-804FF | Mecklenburg horse breed
SRT | L-80504 | Catalana chicken breed
SRT | L-80542 | Haiti Creole pig breed
SRT | L-80543 | Manor Hybrid pig breed
SRT | L-80544 | Hamline pig breed
SRT | L-80545 | Manor Ranger pig breed
SRT | L-80546 | Manor Meishan pig breed
SRT | L-80547 | Cotswold Gold pig breed
SRT | L-80548 | Cotswold Platinum pig breed
SRT | L-80549 | Cotswold 16 pig breed
SRT | L-8054A | Cotswold 29 pig breed
SRT | L-8054B | Cotswold 90 pig breed
SRT | L-8054C | Hampen pig breed
SRT | L-8054D | SPM pig breed
SRT | L-8054E | High Conformation White pig breed
SRT | L-8054F | Line 32 pig breed
SRT | L-80555 | Line 21 pig breed
SRT | L-80556 | Meatline pig breed
SRT | L-80557 | Hampline pig breed
SRT | L-80558 | Euroline pig breed
SRT | L-80559 | Norline pig breed
SRT | L-8055A | Premier pig breed
SRT | L-8055B | Tribred pig breed
SRT | L-8055C | American Essex pig breed
SRT | L-8055D | Sino-Gascony pig breed
SRT | L-8055E | Guadeloupe Creole pig breed
SRT | L-8055F | Managra pig breed
SRT | L-8056A | Canadian Landrace pig breed
SRT | L-8056B | Canadian Yorkshire pig breed
SRT | L-8056C | Minnesota #4 pig breed
SRT | L-8056D | Pineywoods pig breed
SRT | L-8056E | Catalina Island pig breed
SRT | L-8056F | Ras-n-Lansa pig breed
SRT | L-8057B | Pitman-Moore Miniature pig breed
SRT | L-8057C | Vita Vet Lab Minipig pig breed
SRT | L-8057D | Hanford Miniature pig breed
SRT | L-8057E | Black Hampshire pig breed
SRT | L-8057F | Red Hamprace pig breed
SRT | L-80583 | American Yorkshire pig breed
SRT | L-80584 | American Berkshire pig breed
SRT | L-80585 | Camborough Blue pig breed
SRT | L-80586 | Camborough 12 pig breed
SRT | L-80587 | Westrain pig breed
SRT | L-80588 | Dalland 030 pig breed
SRT | L-80589 | Razor-Back pig breed
SRT | L-8058A | Macau pig breed
SRT | L-8058B | Moura pig breed
SRT | L-8058C | Canastra pig breed
SRT | L-8058D | Pirapetinga pig breed
SRT | L-8058E | Piau pig breed
SRT | L-8058F | Nilo-Canastra pig breed
SRT | L-80595 | Canastrão pig breed
SRT | L-80596 | Canastrão, Junqueira pig breed
SRT | L-80597 | Canastrão, Capitão Chico pig breed
SRT | L-80598 | Canastrão, Zabumba pig breed
SRT | L-80599 | Canastrão, Cabano pig breed
SRT | L-8059A | Canastrão, Vermelho pig breed
SRT | L-8059B | Piau, Caruncho Piau pig breed
SRT | L-8059C | Canastrinho pig breed
SRT | L-8059D | Honduras Switch-Tail pig breed
SRT | L-8059E | Mastergilt pig breed
SRT | L-8059F | Sovereign pig breed
SRT | L-805A1 | Poltava pig breed
SRT | L-805A2 | Lipetsk pig breed
SRT | L-805A3 | Soviet Meat pig breed
SRT | L-805A4 | Central Russian pig breed
SRT | L-805A5 | Steppe Meat pig breed
SRT | L-805A6 | Kharkov pig breed
SRT | L-805A7 | Dnepropetrovsk pig breed
SRT | L-805A8 | Russian Large White pig breed
SRT | L-805A9 | Forest Mountain pig breed
SRT | L-805AA | Dnieper pig breed
SRT | L-805AB | Iberian pig breed
SRT | L-805AC | Iberian, Extremadura Red pig breed
SRT | L-805AD | Iberian, Jabugo Spotted pig breed
SRT | L-805AE | Iberian, Black Iberian pig breed
SRT | L-805AF | Philippine Native, Ilocos pig breed
SRT | L-805B1 | Philippine Native, Jalajala pig breed
SRT | L-805B2 | Mangalista pig breed
SRT | L-805B3 | Alentejana pig breed
SRT | L-805B4 | Belgian Landrace, BN pig breed
SRT | L-805B5 | French Large White pig breed
SRT | L-805B6 | Hyper Large White pig breed
SRT | L-805B7 | Tia Meslan pig breed
SRT | L-805B8 | Pen ar Lan 77 pig breed
SRT | L-805B9 | Penshire pig breed
SRT | L-805BA | Laconie pig breed
SRT | L-805BB | Murcian pig breed
SRT | L-805BC | Cavallino pig breed
SRT | L-805BD | Calabrian pig breed
SRT | L-805BE | Apulian pig breed
SRT | L-805BF | Siena Belted pig breed
SRT | L-805C1 | Calascibetta pig breed
SRT | L-805C2 | Güssing Forest Pig pig breed
SRT | L-805C3 | Swiss Edelschwein pig breed
SRT | L-805C4 | North Caucasus pig breed
SRT | L-805C5 | Don pig breed
SRT | L-805C6 | Rostov pig breed
SRT | L-805C7 | Russian Long-Eared White pig breed
SRT | L-805C8 | Russian Short-Eared White pig breed
SRT | L-805C9 | Prisheksninsk pig breed
SRT | L-805CA | Breitov pig breed
SRT | L-805CB | Livny pig breed
SRT | L-805CC | Tsivilsk pig breed
SRT | L-805CD | Urzhum pig breed
SRT | L-805CE | Minisib pig breed
SRT | L-805CF | Sakhalin White pig breed
SRT | L-805D0 | North Siberian pig breed
SRT | L-805D1 | Siberian Black Pied pig breed
SRT | L-805D2 | Kemerovo pig breed
SRT | L-805D3 | KM-1 pig breed
SRT | L-805D4 | Aksaî Black Pied pig breed
SRT | L-805D5 | Semirechensk pig breed
SRT | L-805D6 | Min pig breed
SRT | L-805D7 | Sanjiang White pig breed
SRT | L-805D8 | Basque Black Pied pig breed
SRT | L-805D9 | Corsican pig breed
SRT | L-805DA | Créole pig breed
SRT | L-805DB | Gascony pig breed
SRT | L-805DC | Limousin pig breed
SRT | L-805DD | Harbin White pig breed
SRT | L-805DE | Heilongjiang Spotted pig breed
SRT | L-805DF | Liaoning Black pig breed
SRT | L-805E1 | Huang-Huai-Hai Black, Shenxian pig breed
SRT | L-805E2 | Huang-Huai-Hai Black pig breed
SRT | L-805E3 | Bamei pig breed
SRT | L-805E4 | Hanjiang Black pig breed
SRT | L-805E5 | Ding pig breed
SRT | L-805E6 | Huai pig breed
SRT | L-805E7 | New Huai pig breed
SRT | L-805E8 | Mashen pig breed
SRT | L-805E9 | Yimeng Black pig breed
SRT | L-805EB | Hetao Lop-Ear pig breed
SRT | L-805EC | Korean Native pig breed
SRT | L-805ED | Korean Improved pig breed
SRT | L-805EE | Penbuk pig breed
SRT | L-805EF | Beijing Black pig breed
SRT | L-805F1 | Chenghua pig breed
SRT | L-805F2 | Taoyuan pig breed
SRT | L-805F3 | Taiwan Small Black pig breed
SRT | L-805F4 | Taiwan Small Red pig breed
SRT | L-805F5 | Guanling pig breed
SRT | L-805F6 | Huchuan Mountain pig breed
SRT | L-805F7 | Rongchang pig breed
SRT | L-805F8 | Wujin pig breed
SRT | L-805F9 | Dahe pig breed
SRT | L-805FA | Yanan pig breed
SRT | L-805FB | South Yunnan Short-Eared pig breed
SRT | L-805FC | Hainan, Lingao pig breed
SRT | L-805FD | Hainan, Tunchang pig breed
SRT | L-805FE | Hainan, Wenchang pig breed
SRT | L-805FF | Liang Guang Small Spotted pig breed
SRT | L-8060A | German Pasture pig breed
SRT | L-8060B | Piau, Sorocaba pig breed
SRT | L-8060C | Nilo pig breed
SRT | L-8060D | Bahia pig breed
SRT | L-8060E | Perna-Curta pig breed
SRT | L-8060F | Carunchinho pig breed
SRT | L-80613 | Mandi pig breed
SRT | L-80614 | Orehla de Colher pig breed
SRT | L-80615 | Venezuelan Black pig breed
SRT | L-80616 | Bolivian pig breed
SRT | L-80617 | Pelón pig breed
SRT | L-80618 | Mexican Wattled pig breed
SRT | L-80619 | Dalland 080 pig breed
SRT | L-8061B | Monarch pig breed
SRT | L-8061C | Bisaro pig breed
SRT | L-8061D | Black Hairless pig breed
SRT | L-8061E | Black Mangalitsa pig breed
SRT | L-80623 | Borghigiana pig breed
SRT | L-80624 | Chianina pig breed
SRT | L-80625 | Cosentina pig breed
SRT | L-80626 | Cuino pig breed
SRT | L-80627 | Friuli Black pig breed
SRT | L-80628 | Fumati pig breed
SRT | L-80629 | Galician pig breed
SRT | L-8062A | German Berkshire pig breed
SRT | L-8062B | Ghori pig breed
SRT | L-8062C | Jianli pig breed
SRT | L-8062D | Lucanian pig breed
SRT | L-8062E | Maremmana pig breed
SRT | L-8062F | Miami pig breed
SRT | L-80634 | Montmorillon pig breed
SRT | L-80635 | Old Swedish Spotted pig breed
SRT | L-80636 | Oliventina pig breed
SRT | L-80637 | Parmense pig breed
SRT | L-80638 | Romagnola pig breed
SRT | L-80639 | Siberian pig breed
SRT | L-8063A | Small White pig breed
SRT | L-8063B | Baltaret pig breed
SRT | L-8063C | Tungchang pig breed
SRT | L-8063D | Sterling pig breed
SRT | L-8063E | Vich pig breed
SRT | L-8063F | Vietnamese pig breed
SRT | L-80645 | Vitoria pig breed
SRT | L-80646 | Wai Chow pig breed
SRT | L-80647 | Yorkshire Blue and White pig breed
SRT | L-80648 | Dalland 020 pig breed
SRT | L-80649 | Wiltshire pig breed
SRT | L-8064A | Hamroc pig breed
SRT | L-8064B | DRU™ Terminals pig breed
SRT | L-8064C | Camborough 22 pig breed
SRT | L-8064D | Camborough 15 pig breed
SRT | L-8064E | PR 1050 pig breed
SRT | L-8064F | PR 1075 pig breed
SRT | L-8065A | Chryak PIC pig breed
SRT | L-8065B | Canadian Royal Blue pig breed
SRT | L-8065C | Line 500 Duroc pig breed
SRT | L-8065D | Bodmin 950 pig breed
SRT | L-8065E | Canadian Duroc pig breed
SRT | L-8065F | Canadian Hampshire pig breed
SRT | L-80664 | Ba Xuyen pig breed
SRT | L-80665 | Arapawa Island pig breed
SRT | L-80010 | Wuzhishan pig breed
SRT | L-80667 | Philippine Native pig breed
SRT | L-80668 | Sinclair Miniature pig breed
SRT | L-80669 | Saddleback pig breed
SRT | L-8066A | Yucatan Minature pig breed
SRT | L-8066B | Bantu pig breed
SRT | L-8066C | Tibetan pig breed
SRT | L-8066D | Turopolje pig breed
SRT | L-8066E | Vietnamese Pot-Bellied Pig pig breed
SRT | L-8066F | American Landrace pig breed
SRT | L-80670 | Swallow Belied Mangalitza pig breed
SRT | L-80671 | Fengjing pig breed
SRT | L-80672 | Finnish Landrace pig breed
SRT | L-80673 | Guinea Hog pig breed
SRT | L-80674 | Hezuo pig breed
SRT | L-80675 | Ossabaw Island pig breed
SRT | L-80676 | Kele pig breed
SRT | L-80677 | Krskopolje pig breed
SRT | L-80678 | Kunekune pig breed
SRT | L-80679 | Large Black-White pig breed
SRT | L-8067A | Lithuanian Native pig breed
SRT | L-8067B | Meishan pig breed
SRT | L-8067C | Jinhua pig breed
SRT | L-8067D | Ningxiang pig breed
SRT | L-8067E | Mora Romagnola pig breed
SRT | L-8067F | Mukota pig breed
SRT | L-80680 | Minzhu pig breed
SRT | L-80681 | Neijiang pig breed
SRT | L-80682 | Mulefoot pig breed
SRT | L-80683 | Normand pig breed
SRT | L-80684 | Angeln Saddleback pig breed
SRT | L-80685 | Greek Local pig breed
SRT | L-80686 | Icelandic pig breed
SRT | L-80687 | Casertana pig breed
SRT | L-80688 | Madonie-Sicilian pig breed
SRT | L-80689 | Sardinian pig breed
SRT | L-8068A | Sicilian pig breed
SRT | L-8068B | Zlotniki Spotted pig breed
SRT | L-8068C | Zlotniki White pig breed
SRT | L-8068D | Siska pig breed
SRT | L-8068E | Sumadija pig breed
SRT | L-8068F | Froxfield Pygmy pig breed
SRT | L-80690 | Danish Large White pig breed
SRT | L-80691 | Danish Duroc pig breed
SRT | L-80692 | Danish Hampshire pig breed
SRT | L-80693 | Piggham pig breed
SRT | L-80694 | New York Red pig breed
SRT | L-80695 | Finnish Yorkshire pig breed
SRT | L-80696 | Dutch Yorkshire pig breed
SRT | L-80697 | Pulawy pig breed
SRT | L-80698 | Pomeranian pig breed
SRT | L-80699 | Polish Landrace pig breed
SRT | L-8069A | Estonian Bacon pig breed
SRT | L-8069B | Latvian White pig breed
SRT | L-8069C | Lithuanian White pig breed
SRT | L-8069D | BKB-1 pig breed
SRT | L-8069E | Belorus Black Pied pig breed
SRT | L-8069F | Mirgorod pig breed
SRT | L-806A1 | Liang Guang Small Spotted, Luchuan pig breed
SRT | L-806A2 | Fujian Small pig breed
SRT | L-806A3 | North Fujian Black-and-White pig breed
SRT | L-806A4 | Fuan Spotted pig breed
SRT | L-806A5 | Putian pig breed
SRT | L-806A6 | Fuzhou Black pig breed
SRT | L-806A7 | Minbei Spotted pig breed
SRT | L-806A8 | Lantang pig breed
SRT | L-806A9 | Liang Guang Small Spotted, Guangdong Small Ear pig breed
SRT | L-806AA | Longlin pig breed
SRT | L-806AB | Yuedong Black pig breed
SRT | L-806AC | Xiang pig breed
SRT | L-806AD | Cantonese pig breed
SRT | L-806AE | Jinhua, Dongyang pig breed
SRT | L-806AF | Jinhua, Yongkang pig breed
SRT | L-806B1 | Daweizi pig breed
SRT | L-806B2 | Huazhong Two-End Black pig breed
SRT | L-806B3 | Huazhong Two-End Black, Jianli pig breed
SRT | L-806B4 | Huazhong Two-End Black, Tongcheng pig breed
SRT | L-806B5 | Huazhong Two-End Black, Satzeling pig breed
SRT | L-806B6 | Ganzhongnan Spotted pig breed
SRT | L-806B7 | Hang pig breed
SRT | L-806B8 | Leping pig breed
SRT | L-806B9 | Longyou Black pig breed
SRT | L-806BA | Wuyi Black pig breed
SRT | L-806BB | Lee-Sung pig breed
SRT | L-806BC | Lan-Yu pig breed
SRT | L-806BD | Vietnamese Yorkshire pig breed
SRT | L-806BE | Yujiang pig breed
SRT | L-806BF | Wanzhe Spotted pig breed
SRT | L-806C1 | Wanzhe Spotted, Chunan Spotted pig breed
SRT | L-806C2 | Wanzhe Spotted, Wannan Spotted pig breed
SRT | L-806C3 | Shengxian Spotted pig breed
SRT | L-806C4 | Qingping pig breed
SRT | L-806C5 | Xiangxi Black pig breed
SRT | L-806C6 | Bamaxiang pig breed
SRT | L-806C7 | Taihu pig breed
SRT | L-806C8 | Erhulian pig breed
SRT | L-806C9 | Jiaxing Black pig breed
SRT | L-806CA | Mi pig breed
SRT | L-806CB | Shahutou pig breed
SRT | L-806CC | Jiaoxi pig breed
SRT | L-806CD | Shanghai White pig breed
SRT | L-806CE | Hubei White pig breed
SRT | L-806CF | Xinjin pig breed
SRT | L-806D1 | Xinjin, Jilin Black pig breed
SRT | L-806D2 | Xinjin, Ning-an pig breed
SRT | L-806D3 | Í pig breed
SRT | L-806D4 | DBI pig breed
SRT | L-806D5 | Xinjin, Xinjin pig breed
SRT | L-806D6 | Meixin pig breed
SRT | L-806D7 | North East China Spotted pig breed
SRT | L-806D8 | Fannong Spotted pig breed
SRT | L-806D9 | Laoshan pig breed
SRT | L-806DA | Nanjing Black pig breed
SRT | L-806DB | Shanxi Black pig breed
SRT | L-806DC | Ganzhou White pig breed
SRT | L-806DD | Guangxi White pig breed
SRT | L-806DE | Hanzhong White pig breed
SRT | L-806DF | Lutai White pig breed
SRT | L-806E1 | Yili White pig breed
SRT | L-806E2 | Xinjiang White pig breed
SRT | L-806E3 | BSI pig breed
SRT | L-806E4 | Mong Cai pig breed
SRT | L-806E5 | Lang Hong pig breed
SRT | L-806E6 | Muong Khuong pig breed
SRT | L-806E7 | Meo pig breed
SRT | L-806E8 | Tong Con pig breed
SRT | L-806E9 | Ha Bac pig breed
SRT | L-806EA | Thai Binh pig breed
SRT | L-806EB | Co pig breed
SRT | L-806EC | Swiss Improved Landrace pig breed
SRT | L-806ED | German Landrace B pig breed
SRT | L-806EE | Edelschwein pig breed
SRT | L-806EF | Swabian-Hall pig breed
SRT | L-806F1 | Bentheim Black Pied pig breed
SRT | L-806F2 | Baldinger Spotted pig breed
SRT | L-806F3 | German Red Pied pig breed
SRT | L-806F4 | German Cornwall pig breed
SRT | L-806F5 | Göttingen Miniature pig breed
SRT | L-806F6 | Munich Miniature pig breed
SRT | L-806F8 | Leicoma pig breed
SRT | L-806F9 | Schwerfurt Meat pig breed
SRT | L-806FA | Hungarian White pig breed
SRT | L-806FB | Hungahyb pig breed
SRT | L-806FC | Bulgarian Native pig breed
SRT | L-806FD | East Balkan pig breed
SRT | L-806FE | Kula pig breed
SRT | L-806FF | Nghia Binh pig breed
SRT | L-807E2 | Bichon Teneriffe dog breed
SRT | L-807E3 | Bizanian Hound dog breed
SRT | L-807E4 | Bloodhound, St. Hubert dog breed
SRT | L-807E5 | Bloodhound, Southern Hound dog breed
SRT | L-807E6 | Bordeaux Dog breed
SRT | L-807E7 | Brandlbracke dog breed
SRT | L-807E8 | Braque d'Ariége dog breed
SRT | L-807E9 | Portuguese Guard Dog breed
SRT | L-807EA | Great Münsterländer dog breed
SRT | L-807EB | Beagle, Smooth dog breed
SRT | L-807EC | Beagle, Rough dog breed
SRT | L-807ED | Belgian Griffon, Rough dog breed
SRT | L-807EE | Belgian Griffon, Smooth dog breed
SRT | L-807EF | Braque Belge dog breed
SRT | L-807F1 | Belgian Street Dog breed
SRT | L-807F2 | Bernese Hound dog breed
SRT | L-808A1 | Eurasier dog breed
SRT | L-808A2 | English Bulldog breed
SRT | L-808A3 | Dogue de Bourdeaux dog breed
SRT | L-808A4 | Kai Ken dog breed
SRT | L-808A5 | Kui Mlk dog breed
SRT | L-808A6 | Argentine Dogo dog breed
SRT | L-808A7 | Alentejo herder dog breed
SRT | L-808A8 | Saint Bernard, Long-haired dog breed
SRT | L-808A9 | Saint Bernard, Short-haired dog breed
SRT | L-808AA | West Siberian Laika dog breed
SRT | L-808AB | Basset Fauve de Bretagne dog breed
SRT | L-808AC | Japanese Retriever dog breed
SRT | L-808AD | Kai Dog breed
SRT | L-808AE | American Blue Gascon Hound dog breed
SRT | L-808AF | Beagle Harrier dog breed
SRT | L-808B1 | Kangal Dog breed
SRT | L-808B2 | Leopard Cur dog breed
SRT | L-808B3 | Patterdale Terrier dog breed
SRT | L-808B4 | Petit Brabaçon dog breed
SRT | L-808B5 | Aidi dog breed
SRT | L-808B6 | American Indian Dog breed
SRT | L-808B7 | Austrian Pinscher dog breed
SRT | L-808B8 | American Eskimo, standard dog breed
SRT | L-808B9 | American Eskimo, Miniature dog breed
SRT | L-808BA | American Eskimo, Toy dog breed
SRT | L-808BB | Basset Griffon Vendéen dog breed
SRT | L-808BC | Batard dog breed
SRT | L-808BD | Basset Bleu de Gascogne dog breed
SRT | L-808BE | Braque Dupuy dog breed
SRT | L-808BF | Bruno de Jura dog breed
SRT | L-808C1 | Cão da Serra de Aires dog breed
SRT | L-808C2 | Cão de Castro Laboreiro dog breed
SRT | L-808C3 | Cão de Fila Miguel dog breed
SRT | L-808C4 | Catalan Sheepdog breed
SRT | L-808C5 | Caucasian Shepherd Dog breed
SRT | L-808C6 | Cirneco dell'Etna dog breed
SRT | L-808C7 | English Toy Terrier dog breed
SRT | L-808C8 | German Spitz dog breed
SRT | L-808C9 | Dingo dog breed
SRT | L-808CA | Fauve de Bretagne dog breed
SRT | L-808CB | Hellenic Hound dog breed
SRT | L-808CC | Holland Shepherd dog breed
SRT | L-808CD | Japanese Spitz dog breed
SRT | L-808CE | Jämthund dog breed
SRT | L-808CF | Jindo dog breed
SRT | L-808D1 | Karelo-Finnish Laika dog breed
SRT | L-808D2 | King Shepherd dog breed
SRT | L-808D3 | Kishu dog breed
SRT | L-808D4 | Kirhiz dog breed
SRT | L-808D5 | Magyar Agår dog breed
SRT | L-808D6 | Middle Asian Ovtcharka dog breed
SRT | L-808D7 | Mi-Ki dog breed
SRT | L-808D8 | Miniature Australian Shepherd dog breed
SRT | L-808D9 | Min-pei dog breed
SRT | L-808DA | Mountain Cur dog breed
SRT | L-808DB | Moscow Longhaired Toy Terrier dog breed
SRT | L-808DC | Perdigueiro Portuguese dog breed
SRT | L-808DD | Podengo Canario dog breed
SRT | L-808DE | Podengo Pequeno dog breed
SRT | L-808DF | Pressa Mallorquin dog breed
SRT | L-808E1 | Pyrenean Mastiff dog breed
SRT | L-808E2 | Rastreador Brasileiro dog breed
SRT | L-808E3 | Sabuesos Españoles dog breed
SRT | L-808E4 | Schiller Hound dog breed
SRT | L-808E5 | South Russian Steppe Hound dog breed
SRT | L-808E6 | Styrian Mountain dog breed
SRT | L-808E7 | Berger du Languedoc dog breed
SRT | L-808E8 | Teddy Roosevelt Terrier dog breed
SRT | L-808E9 | Transylvanian Hound dog breed
SRT | L-808EA | Trigg Hound dog breed
SRT | L-808EB | Tyrolean Hound dog breed
SRT | L-808EC | White Shepherd dog breed
SRT | L-808ED | Wirehair Styrian mountain dog breed
SRT | L-808EE | Yugoslavian Hound dog breed
SRT | L-808EF | Old Farm Collie dog breed
SRT | L-808F1 | Old German Shepherd dog breed
SRT | L-808F2 | New Zealand Heading Dog breed
SRT | L-808F3 | German Koolie dog breed
SRT | L-808F4 | Smithfield dog breed
SRT | L-808F5 | Spanish Greyhound dog breed
SRT | L-808F6 | Armant dog breed
SRT | L-808F8 | Australian Greyhound dog breed
SRT | L-808F9 | Australian Terrier, rough-coated dog breed
SRT | L-808FA | Australian Terrier, silky dog breed
SRT | L-808FB | Austrian Hound dog breed
SRT | L-808FC | Austrian Smooth-Haired Bracke dog breed
SRT | L-808FD | Balkan Hound dog breed
SRT | L-808FE | Banjara greyhound dog breed
SRT | L-808FF | Beagle, Standard dog breed
SRT | L-80916 | Estrela Mountain Dog breed
SRT | L-80917 | Epagneul Picard dog breed
SRT | L-80918 | Epagneul Bleu de Picardie dog breed
SRT | L-80919 | Estonian Hound dog breed
SRT | L-80920 | Epagneul Pont-Audemer dog breed
SRT | L-80921 | Eurasian dog breed
SRT | L-80922 | Fell Terrier dog breed
SRT | L-80923 | Fila Brasileiro dog breed
SRT | L-80924 | Finnish Hound dog breed
SRT | L-80925 | Finnish Lapphund dog breed
SRT | L-80926 | Entlebucher dog breed
SRT | L-80927 | French Guard Dog breed
SRT | L-80928 | French Spaniel dog breed
SRT | L-80929 | Coton de Tuléar dog breed
SRT | L-80930 | Hamiltonstövare dog breed
SRT | L-80931 | Danish Broholmer dog breed
SRT | L-80932 | English Shepherd dog breed
SRT | L-80933 | Drentse Patrijshond dog breed
SRT | L-80934 | Dunker dog breed
SRT | L-80935 | Dutch Kooiker Dog breed
SRT | L-80936 | Dutch Shepherd dog breed
SRT | L-80937 | East Siberian Laika dog breed
SRT | L-80938 | Deutsche bracke dog breed
SRT | L-80939 | Hanoverian Hound dog breed
SRT | L-80940 | Hovawart dog breed
SRT | L-80941 | Icelandic Sheepdog breed
SRT | L-80942 | Inca Hairless Dog breed
SRT | L-80943 | Irish Red and White Setter dog breed
SRT | L-80944 | Jagdterrier dog breed
SRT | L-80945 | German Spaniel dog breed
SRT | L-80946 | Grand Anglo-Français dog breed
SRT | L-80947 | Grand Bassett Griffon Vendeen dog breed
SRT | L-80948 | Grand Bleu de Gascogne dog breed
SRT | L-80949 | Grand Gascon-Saintongeois dog breed
SRT | L-80950 | German Pinscher dog breed
SRT | L-80951 | Greater Swiss Mountain Dog breed
SRT | L-80952 | Greenland Dog breed
SRT | L-80953 | Griffon Fauve de Bretegne dog breed
SRT | L-80954 | Griffon Nivernais dog breed
SRT | L-80955 | Grand Griffon Vendeen dog breed
SRT | L-80956 | Ainu dog breed
SRT | L-80957 | Basset Artésian Normand dog breed
SRT | L-80958 | Bavarian Mountain Hound dog breed
SRT | L-80959 | Beauceron dog breed
SRT | L-80960 | Azawakh dog breed
SRT | L-80961 | Australian Shepherd dog breed
SRT | L-80962 | Belgian Wolfhound dog breed
SRT | L-80963 | Bergamasco dog breed
SRT | L-80964 | Berger de Picard dog breed
SRT | L-80965 | Berger de Pyrenees dog breed
SRT | L-80966 | Billy dog breed
SRT | L-80967 | Belgian Griffon dog breed
SRT | L-80968 | American Hairless Terrier dog breed
SRT | L-80969 | Beagle, Elizabethan dog breed
SRT | L-80970 | Japanese Pointer dog breed
SRT | L-80971 | Akbash dog breed
SRT | L-80972 | Alapaha blueblood bullDog breed
SRT | L-80973 | Barbet dog breed
SRT | L-80974 | American Bulldog breed
SRT | L-80975 | Black Russian Terrier dog breed
SRT | L-80976 | Anglo-Francais de moyen venerie dog breed
SRT | L-80977 | Anglo-Francais de petit venerie dog breed
SRT | L-80978 | Appenzeller dog breed
SRT | L-80979 | Ariégeois dog breed
SRT | L-80980 | Alano Español dog breed
SRT | L-80981 | Australian Kelpie dog breed
SRT | L-80982 | Alpine dachsbracke dog breed
SRT | L-80983 | Chien Français Blanc et Noir dog breed
SRT | L-80984 | Carolina Dog breed
SRT | L-80985 | Catahoula Leopard dog breed
SRT | L-80986 | Caucasian Mountain Dog breed
SRT | L-80987 | Cesky Fousek dog breed
SRT | L-80988 | Cesky Terrier dog breed
SRT | L-80989 | Chart Polski dog breed
SRT | L-80990 | Black Forest Hound dog breed
SRT | L-80991 | Chien d'Artois dog breed
SRT | L-80992 | Canaan dog breed
SRT | L-80993 | Chien Français Tricolore dog breed
SRT | L-80994 | Chinese Crested dog breed
SRT | L-80995 | Chinese Foo Dog breed
SRT | L-80996 | Chinese Imperial ch'in dog breed
SRT | L-80997 | Chinook dog breed
SRT | L-80998 | Chien Français Blanc et Orange dog breed
SRT | L-80999 | Braque Français de Grand Taille dog breed
SRT | L-809A1 | Bolognese dog breed
SRT | L-809A2 | Border Collie dog breed
SRT | L-809A3 | Bracco Italiano dog breed
SRT | L-809A4 | Cane Corso dog breed
SRT | L-809A5 | Braque du Bourbonnais dog breed
SRT | L-809A6 | Braque Francais de Petite Taille dog breed
SRT | L-809A7 | Braque Saint-Germain dog breed
SRT | L-809A8 | Briquet Basset Griffon Vendeen dog breed
SRT | L-809A9 | Black Mouth Cur dog breed
SRT | L-809AA | Braque d'Auvergne dog breed
SRT | L-809AB | Schapendoes dog breed
SRT | L-809AC | Sarplaninac dog breed
SRT | L-809AD | Russo-Laika dog breed
SRT | L-809AE | Bosnian Hound dog breed
SRT | L-809AF | Rat Terrier dog breed
SRT | L-809B1 | Pumi dog breed
SRT | L-809B2 | Presa Canario dog breed
SRT | L-809B3 | Portuguese Pointer dog breed
SRT | L-809B4 | Porcelaine dog breed
SRT | L-809B5 | Shropshire Terrier dog breed
SRT | L-809B6 | Boykin Spaniel dog breed
SRT | L-809B7 | Southern Blackmouth Cur dog breed
SRT | L-809B8 | South Russian Ovcharka dog breed
SRT | L-809B9 | Small Spanish Hound dog breed
SRT | L-809BA | Small Münsterländer dog breed
SRT | L-809BB | Slovak Cuvak dog breed
SRT | L-809BC | Shiloh Shepherd dog breed
SRT | L-809BD | Shiba Inu dog breed
SRT | L-809BE | Welsh Sheepdog breed
SRT | L-809BF | Shar-pei dog breed
SRT | L-809C1 | Sloughi dog breed
SRT | L-809C2 | Owczarek Podhalandski dog breed
SRT | L-809C3 | Norbottenspets dog breed
SRT | L-809C4 | Norwegian Dunkerhound dog breed
SRT | L-809C5 | Old Danish Bird Dog breed
SRT | L-809C6 | Old Format Dachsund dog breed
SRT | L-809C7 | Old Format Manchester Terrier dog breed
SRT | L-809C8 | Old Format Min/Toy Poodle dog breed
SRT | L-809C9 | Old Format Welsh Corgi dog breed
SRT | L-809CA | Neopolitan Mastiff dog breed
SRT | L-809CB | Perdiguero de Burgos dog breed
SRT | L-809CC | Perdiguero Navarro dog breed
SRT | L-809CD | Peruvian Inca Orchid dog breed
SRT | L-809CE | Petit Bleu de Gascogne dog breed
SRT | L-809CF | Petit Gascon-Saintongeois dog breed
SRT | L-809D1 | Petit Griffon Bleu de Gascogne dog breed
SRT | L-809D2 | Olde English Bulldogge dog breed
SRT | L-809D3 | Löwchen dog breed
SRT | L-809D4 | Polski Owczarek Nizinny dog breed
SRT | L-809D5 | Polish Hound dog breed
SRT | L-809D6 | Poitevin dog breed
SRT | L-809D7 | Spanish Pointer dog breed
SRT | L-809D8 | Kyi-Leo dog breed
SRT | L-809D9 | Large Spanish Hound dog breed
SRT | L-809DA | Lundehund dog breed
SRT | L-809DB | Lurcher Hound dog breed
SRT | L-809DC | Maremma Sheepdogs dog breed
SRT | L-809DD | McNab dog breed
SRT | L-809DE | Miniature Bull Terrier dog breed
SRT | L-809E1 | Mudi dog breed
SRT | L-809E2 | Munster Lander Pointer dog breed
SRT | L-809E3 | Loenberger dog breed
SRT | L-809E4 | Chi Terrier dog breed
SRT | L-809E5 | Krasky Ovcar dog breed
SRT | L-809E6 | Kromfohrländer dog breed
SRT | L-809E7 | Havanese dog breed
SRT | L-809E8 | American lamalese dog breed
SRT | L-809E9 | Perro de Pressa Canario dog breed
SRT | L-809EA | Norwegian Lundehund dog breed
SRT | L-809EB | North American Shepherd dog breed
SRT | L-809EC | Kyi Apso dog breed
SRT | L-809ED | Swedish Lapphund dog breed
SRT | L-809EE | Treeing Tennessee Brindle dog breed
SRT | L-809EF | Telomain dog breed
SRT | L-809F1 | Swedish Vallhund dog breed
SRT | L-809F2 | Stumpy Tail Cattle Dog breed
SRT | L-809F3 | Stabyhoun dog breed
SRT | L-809F4 | Spinone Italiano dog breed
SRT | L-809F5 | Spanish Mastiff dog breed
SRT | L-809F6 | Berger Shetland dog breed
SRT | L-809F7 | Thai Ridgeback dog breed
SRT | L-809F8 | Swiss Mountain Dog breed
SRT | L-809F9 | Tibetan Mastiff dog breed
SRT | L-809FA | Glen of Imaal Terrier dog breed
SRT | L-809FB | Tosa Inu dog breed
SRT | L-809FC | Toy Havanese Terrier dog breed
SRT | L-809FD | Treeing Cur dog breed
SRT | L-809FE | Treeing Feist dog breed
SRT | L-809FF | Greater Swiss Mountain Hound dog breed
SRT | L-80A70 | Harlequin cat breed
SRT | L-80A71 | Manxamese cat breed
SRT | L-80A73 | Maltese cat breed
SRT | L-80A75 | Ragdoll cat breed
SRT | L-80A76 | Turkish van cat breed
SRT | L-80A77 | British Blue cat breed
SRT | L-80A78 | American Bobtail Shorthair cat breed
SRT | L-80A79 | American Bobtail Longhair cat breed
SRT | L-80A80 | American Curl cat breed
SRT | L-80A81 | Australian Mist cat breed
SRT | L-80A83 | Bengal cat breed
SRT | L-80A84 | Brazilian Shorthair cat breed
SRT | L-80A85 | California Spangled cat breed
SRT | L-80A86 | Chantilly/Tiffany cat breed
SRT | L-80A87 | Shorthair cat breed
SRT | L-80A88 | German Rex cat breed
SRT | L-80A89 | LaPerm Shorthair cat breed
SRT | L-80A90 | LaPerm Longhair cat breed
SRT | L-80A91 | Munchkin Shorthair cat breed
SRT | L-80A92 | Munchkin Longhair cat breed
SRT | L-80A93 | Nebelung cat breed
SRT | L-80A94 | Norwegian Forest cat breed
SRT | L-80A95 | Oriental Longhair cat breed
SRT | L-80A96 | Pixiebob cat breed
SRT | L-80A97 | Ragamuffin cat breed
SRT | L-80A99 | Selkirk Rex cat breed
SRT | L-80AA1 | Siberian cat breed
SRT | L-80AA2 | Snowshoe cat breed
SRT | L-80AA3 | Sokoke cat breed
SRT | L-80AA4 | Sphynx cat breed
SRT | L-80B01 | Bergamasca sheep breed
SRT | L-80B02 | Portland sheep breed
SRT | L-80B04 | Weisse Hornlose Heidschnucke sheep breed
SRT | L-80B05 | Drents Heideschaap sheep breed
SRT | L-80B06 | Kameroen sheep breed
SRT | L-80B07 | Mergelland sheep breed
SRT | L-80B08 | Ouessant sheep breed
SRT | L-80B09 | Canadian Arcott sheep breed
SRT | L-80B10 | Noordhollander sheep breed
SRT | L-80B17 | Rijnlam-A sheep breed
SRT | L-80B18 | Schoonebeker sheep breed
SRT | L-80B19 | Wallis Blacknosed Sheep breed
SRT | L-80B22 | Newfoundland sheep breed
SRT | L-80B23 | Wallis Country Sheep breed
SRT | L-80B24 | Rideau Arcott sheep breed
SRT | L-80B25 | Tukidale sheep breed
SRT | L-80B26 | Polwarth sheep breed
SRT | L-80B27 | Ryeland sheep breed
SRT | L-80B2A | Thalli sheep breed
SRT | L-80B2B | Tong sheep breed
SRT | L-80B2C | Touabire sheep breed
SRT | L-80B2D | Tunis sheep breed
SRT | L-80B2E | Tyrol Mountain sheep breed
SRT | L-80B2F | Uda sheep breed
SRT | L-80B33 | German Mutton Merino sheep breed
SRT | L-80B34 | Medium-Wool Merino sheep breed
SRT | L-80B35 | Fonthill Merino sheep breed
SRT | L-80B36 | South African Mutton Merino sheep breed
SRT | L-80B37 | Strong Wool Merino sheep breed
SRT | L-80B38 | Poll Merino sheep breed
SRT | L-80B39 | Fine Merino sheep breed
SRT | L-80B3A | South African Merino sheep breed
SRT | L-80B40 | Superfine Merino sheep breed
SRT | L-80B47 | Baden Wurttemburg horse breed
SRT | L-80B48 | British Warmblood horse breed
SRT | L-80B49 | Israeli horse breed
SRT | L-80B4A | French Ardennais horse breed
SRT | L-80B4B | Booroola Merino sheep breed
SRT | L-80B50 | Cukurova horse breed
SRT | L-80B51 | Czech Coldblood horse breed
SRT | L-80B52 | Czechoslovakian Small Riding Horse horse breed
SRT | L-80B53 | Jianchang horse breed
SRT | L-80B54 | Jielin horse breed
SRT | L-80B55 | Wielkopolski horse breed
SRT | L-80B56 | Eleia horse breed
SRT | L-80B57 | English Cob horse breed
SRT | L-80B58 | Welsh Pony horse breed
SRT | L-80B59 | Welsh Pony of Cob Type horse breed
SRT | L-80B5A | English Hunter horse breed
SRT | L-80B5B | Eriskay Pony horse breed
SRT | L-80B5C | Hackney Pony horse breed
SRT | L-80B5D | Estonian Draft horse breed
SRT | L-80B5E | Heihe horse breed
SRT | L-80B5F | Heilongkaing horse breed
SRT | L-80B65 | Danish Sport Pony horse breed
SRT | L-80B66 | Kabarda horse breed
SRT | L-80B67 | Kalmyk horse breed
SRT | L-80B68 | Mangalarga Marchador horse breed
SRT | L-80B69 | Don horse breed
SRT | L-80B6A | Manipuri horse breed
SRT | L-80B6B | Swiss Warmblood horse breed
SRT | L-80B6C | Tavda horse breed
SRT | L-80B6D | East Bulgarian horse breed
SRT | L-80B6E | East Friesian (Old Type) horse breed
SRT | L-80B6F | East Friesian Warmblood (Modern Type) horse breed
SRT | L-80B70 | Kakhetian pig breed
SRT | L-80B71 | West French White pig breed
SRT | L-80B80 | Miniature Hereford cattle breed
SRT | L-80B81 | Jem-Jem Zebu cattle breed
SRT | L-80B82 | Minusin horse breed
SRT | L-80B83 | Morochuco horse breed
SRT | L-80B84 | French Trotter horse breed
SRT | L-80B85 | Furioso horse breed
SRT | L-80B86 | Murghese horse breed
SRT | L-80B87 | Mytilene horse breed
SRT | L-80B88 | Namib Desert Horse horse breed
SRT | L-80B89 | Danish Oldenborg horse breed
SRT | L-80B8A | Volynsk cattle breed
SRT | L-80B8B | Senepol cattle breed
SRT | L-80B8C | Shilluk cattle breed
SRT | L-80B8D | Sar Planina sheep breed
SRT | L-80B8E | Santa Inês sheep breed
SRT | L-80B8F | Sahel-type sheep breed
SRT | L-80B90 | Rygja sheep breed
SRT | L-80B91 | Rya sheep breed
SRT | L-80B92 | Moghani sheep breed
SRT | L-80B93 | Rouge de l'Quest sheep breed
SRT | L-80B94 | Soay sheep breed
SRT | L-80B95 | South Suffolk sheep breed
SRT | L-80B96 | South Wales Mountain sheep breed
SRT | L-80B97 | Spælsau sheep breed
SRT | L-80B98 | Spiegel sheep breed
SRT | L-80B99 | St. Croix sheep breed
SRT | L-80B9A | Steigar sheep breed
SRT | L-80B9B | Steinschaf sheep breed
SRT | L-80B9C | Welsh Mountain sheep breed
SRT | L-80B9D | Swedish Fur Sheep breed
SRT | L-80B9E | Teeswater sheep breed
SRT | L-80B9F | Texel sheep breed
SRT | L-80BA1 | Pelibüey sheep breed
SRT | L-80BA2 | Morada Nova sheep breed
SRT | L-80BA3 | Balkhi sheep breed
SRT | L-80BA4 | Bavarian Forest sheep breed
SRT | L-80BA5 | Barbados Blackbelly sheep breed
SRT | L-80BA6 | Romney sheep breed
SRT | L-80BA7 | Awassi sheep breed
SRT | L-80BA8 | Arapawa Island sheep breed
SRT | L-80BA9 | Arabi sheep breed
SRT | L-80BB1 | Apennine sheep breed
SRT | L-80BB2 | American Tunis sheep breed
SRT | L-80BB3 | Balwen Welsh Mountain sheep breed
SRT | L-80BB4 | Priangan sheep breed
SRT | L-80BB5 | Rabo Largo sheep breed
SRT | L-80BE6 | Muban pig breed
SRT | L-80BE7 | Iban pig breed
SRT | L-80BE8 | Altay sheep breed
SRT | L-80BE9 | Faeroes sheep breed
SRT | L-80BF6 | Pitt Island sheep breed
SRT | L-80BF8 | Pinzirita sheep breed
SRT | L-80BF9 | Sardinian sheep breed
SRT | L-80C01 | East Friesian sheep breed
SRT | L-80C02 | Ujumqin sheep breed
SRT | L-80C22 | DLS sheep breed
SRT | L-80C23 | Walachenschaf sheep breed
SRT | L-80C24 | Outaouais Arcott sheep breed
SRT | L-80C25 | Ossimi sheep breed
SRT | L-80C29 | Bentheimer Landschaf sheep breed
SRT | L-80C30 | Barbado sheep breed
SRT | L-80C31 | Baluchi sheep breed
SRT | L-86B36 | Blanc de Bouscat rabbit breed
SRT | L-8A111 | American Indian Horse horse breed
SRT | L-8A112 | American Mustang horse breed
SRT | L-8A113 | American Quarter Horse horse breed
SRT | L-8A115 | American Shetland pony horse breed
SRT | L-8A116 | Anadolu horse breed
SRT | L-8A117 | Andean horse breed
SRT | L-8A118 | Anglo-Kabarda horse breed
SRT | L-8A125 | Narym horse breed
SRT | L-8A126 | National Spotted Saddle Horse horse breed
SRT | L-8A127 | Nigerian horse breed
SRT | L-8A128 | North Swedish Trotter horse breed
SRT | L-8A129 | Oriental Horse horse breed
SRT | L-8A12A | Rhineland Heavy Draft horse breed
SRT | L-8A12B | Romanian Saddle Horse horse breed
SRT | L-8A12C | Rottal horse breed
SRT | L-8A12D | Royal Canadian Mounted Police Horse horse breed
SRT | L-8A12E | Russian Saddle Horse horse breed
SRT | L-8A12F | Sable Island Horse horse breed
SRT | L-8A130 | Panje horse breed
SRT | L-8A131 | Patibarcina horse breed
SRT | L-8A132 | Pechora horse breed
SRT | L-8A133 | Peneia horse breed
SRT | L-8A134 | Periangan horse breed
SRT | L-8A135 | Persian Arab horse breed
SRT | L-8A136 | Petiso Argentino horse breed
SRT | L-8A137 | Polish Draft horse breed
SRT | L-8A138 | Priob horse breed
SRT | L-8A139 | Rahvan horse breed
SRT | L-8A13A | Salerno horse breed
SRT | L-8A13B | Sandalwood horse breed
SRT | L-8A13C | Sandan horse breed
SRT | L-8A13D | Pindos horse breed
SRT | L-8A13E | Piquira Pony horse breed
SRT | L-8A13F | Pleven horse breed
SRT | L-8A14A | Garrano tarpan horse X domestic horse breed
SRT | L-8A14B | Konink tarpan horse X domestic horse breed
SRT | L-8A14C | Asturian tarpan horse X domestic horse breed
SRT | L-8A14D | Pottok tarpan horse X domestic horse breed
SRT | L-8A150 | Russian Trotter horse breed
SRT | L-8A151 | West African Barb horse breed
SRT | L-8A152 | Fell Pony horse breed
SRT | L-8A153 | National Show Horse horse breed
SRT | L-8A154 | Zhemaichu horse breed
SRT | L-8A155 | Yonaguni horse breed
SRT | L-8A156 | Yakut horse breed
SRT | L-8A157 | Tawleed horse breed
SRT | L-8A158 | Western Sudan Pony horse breed
SRT | L-8A159 | Welera Pony horse breed
SRT | L-8A15A | Vyatka horse breed
SRT | L-8A15B | Vladimir Heavy Draft horse breed
SRT | L-8A15C | Vlaamperd horse breed
SRT | L-8A15D | Ukrainian Saddle Horse horse breed
SRT | L-8A15E | Tori horse breed
SRT | L-8A15F | Tokara horse breed
SRT | L-8A160 | New Kirgiz horse breed
SRT | L-8A161 | Oldenburg horse breed
SRT | L-8A162 | Misaki horse breed
SRT | L-8A163 | Miyako horse breed
SRT | L-8A164 | Mongolian horse breed
SRT | L-8A165 | Waler horse breed
SRT | L-8A166 | Dutch Draft horse breed
SRT | L-8A167 | Egyptian horse breed
SRT | L-8A168 | Estonian Native horse breed
SRT | L-8A169 | Exmoor Pony horse breed
SRT | L-8A16A | Faeroes Island Horse horse breed
SRT | L-8A16B | Falabella horse breed
SRT | L-8A16C | Dutch Warmblood horse breed
SRT | L-8A16D | Dongola horse breed
SRT | L-8A16E | Døle horse breed
SRT | L-8A16F | Djerma horse breed
SRT | L-8A170 | Deliboz horse breed
SRT | L-8A171 | Dartmoor Pony horse breed
SRT | L-8A172 | Crioulo horse breed
SRT | L-8A173 | Finnhorse horse breed
SRT | L-8A174 | Sanfratello horse breed
SRT | L-8A175 | Morab horse breed
SRT | L-8A176 | Moyle horse breed
SRT | L-8A177 | Mustang horse breed
SRT | L-8A178 | M'Bayar horse breed
SRT | L-8A179 | Lusitano horse breed
SRT | L-8A17A | Newfoundland Pony horse breed
SRT | L-8A17B | Noma horse breed
SRT | L-8A17C | Nooitgedacht Pony horse breed
SRT | L-8A17D | Nordland horse breed
SRT | L-8A17E | Noric horse breed
SRT | L-8A17F | North Swedish Horse horse breed
SRT | L-8A180 | Northeastern horse breed
SRT | L-8A181 | Kisber Felver horse breed
SRT | L-8A182 | Anglo-Arab horse breed
SRT | L-8A183 | Nonius horse breed
SRT | L-8A184 | Nooitgedacht horse breed
SRT | L-8A185 | Iomud horse breed
SRT | L-8A186 | Jutland horse breed
SRT | L-8A187 | Karabair horse breed
SRT | L-8A188 | Karabakh horse breed
SRT | L-8A189 | Kazakh horse breed
SRT | L-8A18A | Mangalarga horse breed
SRT | L-8A18B | Kirdi Pony horse breed
SRT | L-8A18C | Kiso horse breed
SRT | L-8A18D | Kladruby horse breed
SRT | L-8A18E | Knabstrup horse breed
SRT | L-8A18F | Kushum horse breed
SRT | L-8A190 | Kustanai horse breed
SRT | L-8A191 | Latvian horse breed
SRT | L-8A192 | Lithuanian Heavy Draft horse breed
SRT | L-8A193 | Lokai horse breed
SRT | L-8A194 | Kiger Mustang horse breed
SRT | L-8A195 | Pony of the Americas horse breed
SRT | L-8A196 | Pintabian horse breed
SRT | L-8A197 | Pantaneiro horse breed
SRT | L-8A198 | Orlov Trotter horse breed
SRT | L-8A199 | Northern Ardennais horse breed
SRT | L-8A19A | Abtenauer horse breed
SRT | L-8A19B | Adaev horse breed
SRT | L-8A19C | Albanian horse breed
SRT | L-8A19E | Alter Real horse breed
SRT | L-8A19F | American Bashkir Curly horse breed
SRT | L-8A1A1 | Poitou Mule Producer horse breed
SRT | L-8A1A2 | Polesian horse breed
SRT | L-8A1A3 | Sardinian Anglo-Arab horse breed
SRT | L-8A1A4 | Sardinian Pony horse breed
SRT | L-8A1A5 | Sarvar horse breed
SRT | L-8A1A6 | Schleswig horse breed
SRT | L-8A1A7 | Schwarzwalder Fuchse horse breed
SRT | L-8A1A8 | Senne horse breed
SRT | L-8A1A9 | Shan horse breed
SRT | L-8A1AA | Silesian horse breed
SRT | L-8A1AB | Sini horse breed
SRT | L-8A1AC | Skyros horse breed
SRT | L-8A1AD | Slovak Warmblood horse breed
SRT | L-8A1AE | Sokolka horse breed
SRT | L-8A1AF | South African Miniature horse breed
SRT | L-8A1B1 | South German Coldblood horse breed
SRT | L-8A1B2 | Southwest Spanish Mustang horse breed
SRT | L-8A1B4 | Spanish-American Horse horse breed
SRT | L-8A1B5 | Spanish Anglo-Arab horse breed
SRT | L-8A1B6 | Spanish Colonial Horse horse breed
SRT | L-8A1B7 | Spiti horse breed
SRT | L-8A1B8 | Sulawesi horse breed
SRT | L-8A1B9 | Criollo horse breed
SRT | L-8A1BA | Hequ horse breed
SRT | L-8A1BB | Connemara Pony horse breed
SRT | L-8A1BC | Colorado Ranger horse breed
SRT | L-8A1BD | Dales Pony horse breed
SRT | L-8A1BE | Gotland horse breed
SRT | L-8A1BF | Chincoteague Pony horse breed
SRT | L-8A1C1 | Hokkaido horse breed
SRT | L-8A1C2 | Highland Pony horse breed
SRT | L-8A1C3 | Groningen horse breed
SRT | L-8A1C4 | Cuban Pinto horse breed
SRT | L-8A1C5 | Fleuve horse breed
SRT | L-8A1C6 | Golden American Saddlebred horse breed
SRT | L-8A1C7 | Gidran horse breed
SRT | L-8A1C8 | Gelderland horse breed
SRT | L-8A1C9 | Galician Pony horse breed
SRT | L-8A1CA | Friesian horse breed
SRT | L-8A1CB | Frederiksborg horse breed
SRT | L-8A1CC | Fouta horse breed
SRT | L-8A1CD | Florida Cracker horse breed
SRT | L-8A1CE | Guangxi horse breed
SRT | L-8A1CF | Ardennes horse breed
SRT | L-8A1D1 | American Walking Pony horse breed
SRT | L-8A1D2 | Azteca horse breed
SRT | L-8A1D3 | American Cream Draft horse breed
SRT | L-8A1D4 | Altai horse breed
SRT | L-8A1D5 | Akhal-Teke horse breed
SRT | L-8A1D6 | Abyssinian horse breed
SRT | L-8A1D7 | Bhirum Pony horse breed
SRT | L-8A1D8 | Cheju horse breed
SRT | L-8A1D9 | Cayuse horse breed
SRT | L-8A1DA | Caspian horse breed
SRT | L-8A1DB | Carthusian horse breed
SRT | L-8A1DC | Campolina horse breed
SRT | L-8A1DD | Byelorussian Harness horse breed
SRT | L-8A1DE | Budyonny horse breed
SRT | L-8A1DF | Australian Brumby horse breed
SRT | L-8A1E1 | Australian Stock Horse horse breed
SRT | L-8A1E2 | Basuto Pony horse breed
SRT | L-8A1E3 | Bashkir Curly horse breed
SRT | L-8A1E4 | Bashkir horse breed
SRT | L-8A1E5 | Barb horse breed
SRT | L-8A1E6 | Ban-ei horse breed
SRT | L-8A1E7 | Carpathian Pony horse breed
SRT | L-8A1E8 | Baluchi horse breed
SRT | L-8A1E9 | Balearic horse breed
SRT | L-8A1EA | Chilean Corralero horse breed
SRT | L-8A1EB | Breton horse breed
SRT | L-8A1EC | Taishuh horse breed
SRT | L-8A1ED | Swedish Warmblood horse breed
SRT | L-8A1EE | Sudan Country-Bred horse breed
SRT | L-8A1EF | Spanish-Norman horse breed
SRT | L-8A1F1 | Spanish Barb horse breed
SRT | L-8A1F2 | Soviet Heavy Draft horse breed
SRT | L-8A1F3 | Sorraia horse breed
SRT | L-8A1F4 | Somali Pony horse breed
SRT | L-8A1F5 | Tersk horse breed
SRT | L-8A1F6 | Shagya horse breed
SRT | L-8A1F7 | Selle Francais horse breed
SRT | L-8A1F8 | Sanhe horse breed
SRT | L-8A1FA | Russian Heavy Draft horse breed
SRT | L-8A1FB | Rocky Mountain Horse horse breed
SRT | L-8A1FC | Racking Horse horse breed
SRT | L-8A1FD | Quarter Pony horse breed
SRT | L-8A1FE | Quarab horse breed
SRT | L-8A1FF | Single-Footing Horse horse breed
SRT | L-8B105 | Tuy Hoa Hairless pig breed
SRT | L-8B106 | Hainan pig breed
SRT | L-8B107 | Sino-Vietnamese pig breed
SRT | L-8B108 | Bo Xu pig breed
SRT | L-8B109 | Thuoc Nhieu pig breed
SRT | L-8B111 | Burmese pig breed
SRT | L-8B112 | Chin pig breed
SRT | L-8B113 | Siamese pig breed
SRT | L-8B114 | Hailum pig breed
SRT | L-8B115 | Kwai pig breed
SRT | L-8B116 | Raad pig breed
SRT | L-8B117 | Akha pig breed
SRT | L-8B118 | South China pig breed
SRT | L-8B119 | South China Black pig breed
SRT | L-8B121 | Balinese pig breed
SRT | L-8B122 | Diani pig breed
SRT | L-8B123 | Kaman pig breed
SRT | L-8B124 | Ashanti Dwarf pig breed
SRT | L-8B125 | Koronadal pig breed
SRT | L-8B126 | Ohmini pig breed
SRT | L-8B127 | Clawn pig breed
SRT | L-8B128 | Inobuta (inter-species hybrid) pig breed
SRT | L-8B129 | Kangaroo Island pig breed
SRT | L-8B130 | Captain Cooker pig breed
SRT | L-8B131 | West African pig breed
SRT | L-8B132 | Nigerian pig breed
SRT | L-8B133 | Bakosi pig breed
SRT | L-8B134 | Windsnyer pig breed
SRT | L-8B135 | Kolbroek pig breed
SRT | L-8B136 | South African Landrace pig breed
SRT | L-8B137 | Bulgarian White pig breed
SRT | L-8B139 | Bulgarian Landrace pig breed
SRT | L-8B140 | Danube White pig breed
SRT | L-8B141 | Dermantsi Pied pig breed
SRT | L-8B142 | Romanian Native, Stocli pig breed
SRT | L-8B143 | Romanian Native, Baltaret pig breed
SRT | L-8B144 | Banat White pig breed
SRT | L-8B145 | Bazna pig breed
SRT | L-8B146 | Dobrogea Black pig breed
SRT | L-8B147 | Strei pig breed
SRT | L-8B148 | Romanian Large White pig breed
SRT | L-8B149 | Romanian Meat Pig pig breed
SRT | L-8B150 | Gurktal pig breed
SRT | L-8B151 | Black Slavonian pig breed
SRT | L-8B152 | Resava pig breed
SRT | L-8B153 | Morava pig breed
SRT | L-8B155 | Dzumalia pig breed
SRT | L-8B156 | Macedonian pig breed
SRT | L-8B157 | Albanian Native pig breed
SRT | L-8B158 | Shkodra pig breed
SRT | L-8B159 | Slovenian White pig breed
SRT | L-8B160 | Subotica White pig breed
SRT | L-8B161 | Prestice pig breed
SRT | L-8B162 | Slovakian Black Pied pig breed
SRT | L-8B163 | Czech Improved White pig breed
SRT | L-8B164 | Moravian Large Yorkshire pig breed
SRT | L-8B165 | Slovakian White pig breed
SRT | L-8B166 | Slovhyb-1 pig breed
SRT | L-8B167 | Nitra Hybrid pig breed
SRT | L-8B168 | Synthetic SL98 pig breed
SRT | L-8B169 | SL96 pig breed
SRT | L-8B170 | Czech Meat pig breed
SRT | L-8B171 | Czech Miniature pig breed
SRT | L-8B172 | Small Polish Prick-Eared pig breed
SRT | L-8B173 | Polesian pig breed
SRT | L-8B174 | Nadbuzanska pig breed
SRT | L-8B175 | Sarny pig breed
SRT | L-8B176 | Krolevets pig breed
SRT | L-8B177 | Polish Marsh pig breed
SRT | L-8B178 | Large Polish Long-Eared pig breed
SRT | L-8B958 | Herens cattle breed
SRT | L-8B959 | Hinterwald cattle breed
SRT | L-8B95A | Hungarian Gray cattle breed
SRT | L-8B95B | Icelandic cattle breed
SRT | L-8B95C | Illawarra cattle breed
SRT | L-8B95D | Irish Moiled cattle breed
SRT | L-8B95E | Israeli Holstein cattle breed
SRT | L-8B95F | Istoben cattle breed
SRT | L-8B961 | Jaulan cattle breed
SRT | L-8B962 | Kazakh cattle breed
SRT | L-8B963 | Kerry cattle breed
SRT | L-8B964 | Kholmogory cattle breed
SRT | L-8B966 | Latvian Brown cattle breed
SRT | L-8B967 | Lincoln Red Shorthorn cattle breed
SRT | L-8B968 | Lithuanian Red cattle breed
SRT | L-8B969 | Mashona cattle breed
SRT | L-8B96A | Milking Devon cattle breed
SRT | L-8B96B | Mirandesa cattle breed
SRT | L-8B96C | Mixed dairy cattle breed
SRT | L-8B96D | Mongolian cattle breed
SRT | L-8B96E | Morucha cattle breed
SRT | L-8B96F | Kurdi cattle breed
SRT | L-8B971 | N'dama cattle breed
SRT | L-8B972 | Norwegian Red cattle breed
SRT | L-8B973 | Parthenais cattle breed
SRT | L-8B974 | Polish Red cattle breed
SRT | L-8B975 | Rätien Gray cattle breed
SRT | L-8B976 | Red and White cattle breed
SRT | L-8B977 | Red Angus cattle breed
SRT | L-8B978 | Red Polled Østland cattle breed
SRT | L-8B979 | Red Steppe cattle breed
SRT | L-8B97A | Reggiana cattle breed
SRT | L-8B97B | Retinta cattle breed
SRT | L-8B97C | Romosinuano cattle breed
SRT | L-8B97D | Russian Black Pied cattle breed
SRT | L-8B97E | RX3 cattle breed
SRT | L-8B97F | Salorn cattle breed
SRT | L-8B983 | Murboden cattle breed
SRT | L-8B984 | San Martinero cattle breed
SRT | L-8B985 | Sarabi cattle breed
SRT | L-8B987 | Sharabi cattle breed
SRT | L-8B988 | Shetland cattle breed
SRT | L-8B989 | Simbrah cattle breed
SRT | L-8B98A | South Devon cattle breed
SRT | L-8B98B | Suffolk cattle breed
SRT | L-8B98C | Sussex cattle breed
SRT | L-8B98D | Swedish Red Polled cattle breed
SRT | L-8B98E | Telemark cattle breed
SRT | L-8B98F | Texas Longhorn cattle breed
SRT | L-8B990 | Texon cattle breed
SRT | L-8B991 | Vestland Fjord cattle breed
SRT | L-8B992 | Vestland Red Polled cattle breed
SRT | L-8B993 | Wagyu cattle breed
SRT | L-8B994 | White Cáceres cattle breed
SRT | L-8B995 | Xinjiang Brown cattle breed
SRT | L-8B996 | Yanbian cattle breed
SRT | L-8B998 | Zaobei cattle breed
SRT | L-8B999 | Zavot cattle breed
SRT | L-8B99A | Znamensk cattle breed
SRT | L-8B99B | Alistana-Sanabresa cattle breed
SRT | L-8B99C | Andalusian Blond cattle breed
SRT | L-8B99D | Aosta Black Pied cattle breed
SRT | L-8B99E | Aosta Chestnut cattle breed
SRT | L-8B99F | Aosta Red Pied cattle breed
SRT | L-8B9A0 | Aracena cattle breed
SRT | L-8B9A1 | Argentine Friesian cattle breed
SRT | L-8B9A2 | Armorican cattle breed
SRT | L-8B9A3 | Arouquesa cattle breed
SRT | L-8B9A4 | Aure et Saint-Girons cattle breed
SRT | L-8B9A5 | Australian White cattle breed
SRT | L-8B9A6 | Austrian Simmental cattle breed
SRT | L-8B9A7 | Austrian Yellow cattle breed
SRT | L-8B9A8 | Avetonou cattle breed
SRT | L-8B9A9 | Avilena cattle breed
SRT | L-8B9AA | Avilena-Black Iberian cattle breed
SRT | L-8B9AB | Bakosi cattle breed
SRT | L-8B9AC | Bakwiri cattle breed
SRT | L-8B9AD | Baltic Black Pied cattle breed
SRT | L-8B9AE | Baoule cattle breed
SRT | L-8B9AF | Barrosa cattle breed
SRT | L-8B9B0 | Barroso cattle breed
SRT | L-8B9B1 | Bearnais cattle breed
SRT | L-8B9B2 | Beef shorthorn cattle breed
SRT | L-8B9B3 | Beef synthetic cattle breed
SRT | L-8B9B4 | Beijing Black Pied cattle breed
SRT | L-8B9B5 | Beiroa cattle breed
SRT | L-8B9B6 | Belgian Black Pied Holstein cattle breed
SRT | L-8B9B7 | Belgian Red Pied cattle breed
SRT | L-8B9B8 | Belgian White and Red cattle breed
SRT | L-8B9B9 | Belted Welsh cattle breed
SRT | L-8B9BA | Bestuzhev cattle breed
SRT | L-8B9BB | Betizuak cattle breed
SRT | L-8B9BC | Black Baldy cattle breed
SRT | L-8B9BD | Black Forrest cattle breed
SRT | L-8B9BE | Black Iberian cattle breed
SRT | L-8B9BF | Northern Blue cattle breed
SRT | L-8B9C0 | Bragado do Sorraia cattle breed
SRT | L-8B9C1 | Braganca cattle breed
SRT | L-8B9C2 | Brandrood ljsselvee cattle breed
SRT | L-8B9C3 | Brazilian Polled cattle breed
SRT | L-8B9C4 | Breton Black Pied cattle breed
SRT | L-8B9C5 | Brown Atlas cattle breed
SRT | L-8B9C6 | Bulgarian Brown cattle breed
SRT | L-8B9C7 | Bulgarian Red cattle breed
SRT | L-8B9C8 | Burlina cattle breed
SRT | L-8B9C9 | Burwash cattle breed
SRT | L-8B9CA | Byelorussian Red cattle breed
SRT | L-8B9CB | Byelorussian Synthetic cattle breed
SRT | L-8B9CC | Cabannina cattle breed
SRT | L-8B9CD | Caldeano cattle breed
SRT | L-8B9CE | Caldelana cattle breed
SRT | L-8B9CF | Calvana cattle breed
SRT | L-8B9D0 | Camargue cattle breed
SRT | L-8B9D1 | Cambodian cattle breed
SRT | L-8B9D2 | Caracu cattle breed
SRT | L-8B9D3 | Carpathian Brown cattle breed
SRT | L-8B9D4 | Casanareno cattle breed
SRT | L-8B9D5 | Central Russian Black Pied cattle breed
SRT | L-8B9D6 | Chaouia cattle breed
SRT | L-8B9D7 | Charollandais cattle breed
SRT | L-8B9D8 | Char-swiss cattle breed
SRT | L-8B9D9 | Korean Black cattle breed
SRT | L-8B9DA | Chesi cattle breed
SRT | L-8B9DB | Cheurfa cattle breed
SRT | L-8B9DC | Chiford cattle breed
SRT | L-8B9DD | Chimaine cattle breed
SRT | L-8B9DE | Chinampo cattle breed
SRT | L-8B9DF | Cildir cattle breed
SRT | L-8B9E0 | COOPELSO 93 cattle breed
SRT | L-8B9E1 | Thrace cattle breed
SRT | L-8B9E2 | Corsican cattle breed
SRT | L-8B9E3 | Cretan Lowland cattle breed
SRT | L-8B9E4 | Cretan Mountain cattle breed
SRT | L-8B9E5 | Croatian Red cattle breed
SRT | L-8B9E6 | Cukurova cattle breed
SRT | L-8B9E7 | Curraleiro cattle breed
SRT | L-8B9E8 | Cyprus cattle breed
SRT | L-8B9E9 | Czech Pied cattle breed
SRT | L-8B9EA | Dagestan Mountain cattle breed
SRT | L-8B9EB | Dairy Shorthorn cattle breed
SRT | L-8B9EC | Dairy Synthetic cattle breed
SRT | L-8B9ED | Danish Red Pied cattle breed
SRT | L-8B9EE | Dengchuan cattle breed
SRT | L-8B9EF | Dexter-Kerry cattle breed
SRT | L-8B9F0 | Doran cattle breed
SRT | L-8B9F1 | Dorna cattle breed
SRT | L-8B9F2 | Dortyol cattle breed
SRT | L-8B9F3 | East Anatolian Red cattle breed
SRT | L-8B9F4 | East Finnish cattle breed
SRT | L-8B9F5 | East Macedonian cattle breed
SRT | L-8B9F6 | Epirus cattle breed
SRT | L-8B9F7 | Estonian Black Pied cattle breed
SRT | L-8B9FA | Ferrandais cattle breed
SRT | L-8B9FB | Finnish Ayrshire cattle breed
SRT | L-8B9FC | Flemish cattle breed
SRT | L-8B9FD | Red Flemish cattle breed
SRT | L-8B9FE | Fort Cross cattle breed
SRT | L-8B9FF | Frati cattle breed
SRT | L-8BA00 | Estonian Native cattle breed
SRT | L-8BA01 | Faeroes cattle breed
SRT | L-8BA02 | French Brown cattle breed
SRT | L-8BA03 | Frijolillo cattle breed
SRT | L-8BA04 | FRS cattle breed
SRT | L-8BA05 | Gacko cattle breed
SRT | L-8BA06 | Gado da Terra cattle breed
SRT | L-8BA07 | Georgian Mountain cattle breed
SRT | L-8BA08 | German Black Pied cattle breed
SRT | L-8BA09 | German Black Pied Dairy cattle breed
SRT | L-8BA0A | Pechora cattle breed
SRT | L-8BA0B | Pee Wee cattle breed
SRT | L-8BA0C | Peloponnesus cattle breed
SRT | L-8BA0D | Pester cattle breed
SRT | L-8BA0E | Pie Rouge de l'Est cattle breed
SRT | L-8BA0F | Pisana cattle breed
SRT | L-8BA10 | German Brown cattle breed
SRT | L-8BA11 | German Shorthorn cattle breed
SRT | L-8BA12 | Ghana Shorthorn cattle breed
SRT | L-8BA13 | Glan-Donnersberg cattle breed
SRT | L-8BA14 | Gole cattle breed
SRT | L-8BA15 | Golpayegani cattle breed
SRT | L-8BA16 | Gorbatov Red cattle breed
SRT | L-8BA17 | Goryn cattle breed
SRT | L-8BA19 | Greater Caucasus cattle breed
SRT | L-8BA1A | Polish Black and White Lowland cattle breed
SRT | L-8BA1B | Polish Simmental cattle breed
SRT | L-8BA1C | Polled Jersey cattle breed
SRT | L-8BA1D | Polled Lincoln Red cattle breed
SRT | L-8BA1E | Polled Shorthorn (US) cattle breed
SRT | L-8BA1F | Polled Simmental cattle breed
SRT | L-8BA20 | Greek Shorthorn cattle breed
SRT | L-8BA21 | Greek Steppe cattle breed
SRT | L-8BA22 | Gray Alpine cattle breed
SRT | L-8BA23 | Guadiana Spotted cattle breed
SRT | L-8BA24 | Guelma cattle breed
SRT | L-8BA25 | Harz Red cattle breed
SRT | L-8BA26 | Hawaiian wild cattle breed
SRT | L-8BA27 | Hereland cattle breed
SRT | L-8BA28 | Holgus cattle breed
SRT | L-8BA29 | Hrbinecky cattle breed
SRT | L-8BA2A | Polled Sussex cattle breed
SRT | L-8BA2B | Polled Welsh Black cattle breed
SRT | L-8BA2C | Pontremolese cattle breed
SRT | L-8BA2D | Preta cattle breed
SRT | L-8BA2E | Puerto Rican Criollo cattle breed
SRT | L-8BA2F | Pyrenean cattle breed
SRT | L-8BA30 | Huertana cattle breed
SRT | L-8BA31 | Hungarian Pied cattle breed
SRT | L-8BA32 | Hungarofries cattle breed
SRT | L-8BA33 | Improved Rodopi cattle breed
SRT | L-8BA34 | INRA 95 cattle breed
SRT | L-8BA35 | Italian Brown cattle breed
SRT | L-8BA36 | Italian Red Pied cattle breed
SRT | L-8BA37 | Japanese Black cattle breed
SRT | L-8BA38 | Japanese Brown cattle breed
SRT | L-8BA39 | Japanese Poll cattle breed
SRT | L-8BA3A | Qinchuan cattle breed
SRT | L-8BA3B | Ramo Grande cattle breed
SRT | L-8BA3C | Randall Lineback cattle breed
SRT | L-8BA3D | Red Galloway cattle breed
SRT | L-8BA3E | Regus cattle breed
SRT | L-8BA3F | Rendena cattle breed
SRT | L-8BA40 | Japanese Shorthorn cattle breed
SRT | L-8BA41 | Jarmelista cattle breed
SRT | L-8BA42 | Kabyle cattle breed
SRT | L-8BA43 | Kapsiki cattle breed
SRT | L-8BA44 | Katerini cattle breed
SRT | L-8BA45 | Kenran cattle breed
SRT | L-8BA46 | Khevsurian cattle breed
SRT | L-8BA47 | Kilis cattle breed
SRT | L-8BA48 | Kochi cattle breed
SRT | L-8BA49 | Korean Native cattle breed
SRT | L-8BA4A | Rhaetian Gray cattle breed
SRT | L-8BA4B | Rio Limon Dairy Criollo cattle breed
SRT | L-8BA4C | Rodopi cattle breed
SRT | L-8BA4D | Romanian Red cattle breed
SRT | L-8BA4E | Romanian Brown cattle breed
SRT | L-8BA4F | Russian Brown cattle breed
SRT | L-8BA50 | Kostroma cattle breed
SRT | L-8BA51 | Kravarsky cattle breed
SRT | L-8BA52 | Kuchinoshima cattle breed
SRT | L-8BA53 | Murray Gray cattle breed
SRT | L-8BA54 | Australian Shorthorn cattle breed
SRT | L-8BA55 | Kumamoto cattle breed
SRT | L-8BA56 | Lagune cattle breed
SRT | L-8BA57 | Lakenvelder cattle breed
SRT | L-8BA58 | Latvian Blue Roan cattle breed
SRT | L-8BA59 | La Velasquez cattle breed
SRT | L-8BA5A | Sardinian cattle breed
SRT | L-8BA5B | Sardinian brown cattle breed
SRT | L-8BA5C | Savinja Gray cattle breed
SRT | L-8BA5D | Sayaguesa cattle breed
SRT | L-8BA5E | Seferihisar cattle breed
SRT | L-8BA5F | Shkodra Red cattle breed
SRT | L-8BA60 | Lebanese cattle breed
SRT | L-8BA61 | Lebedin cattle breed
SRT | L-8BA62 | Lesser Caucasus cattle breed
SRT | L-8BA63 | Liberian Dwarf cattle breed
SRT | L-8BA64 | Libyan cattle breed
SRT | L-8BA65 | Lim cattle breed
SRT | L-8BA66 | Limiana cattle breed
SRT | L-8BA67 | Limpurger cattle breed
SRT | L-8BA68 | Lobi cattle breed
SRT | L-8BA69 | Lourdais cattle breed
SRT | L-8BA6A | Slovakian Pied cattle breed
SRT | L-8BA6B | Slovakian Pinzgau cattle breed
SRT | L-8BA6C | Slovenian Brown cattle breed
SRT | L-8BA6D | Somba cattle breed
SRT | L-8BA6E | South African Brown Swiss cattle breed
SRT | L-8BA6F | South Anatolian Red cattle breed
SRT | L-8BA70 | Lucerna cattle breed
SRT | L-8BA71 | Luxi cattle breed
SRT | L-8BA72 | Macedonian Busa cattle breed
SRT | L-8BA73 | Makaweli cattle breed
SRT | L-8BA74 | Marinhoa cattle breed
SRT | L-8BA75 | Maronesa cattle breed
SRT | L-8BA76 | Mazury cattle breed
SRT | L-8BA77 | Messaoria cattle breed
SRT | L-8BA78 | Metohija Red cattle breed
SRT | L-8BA79 | Mingrelian Red cattle breed
SRT | L-8BA7A | Southern Ukranian cattle breed
SRT | L-8BA7B | Spanish Brown Alpine cattle breed
SRT | L-8BA7C | Suksun cattle breed
SRT | L-8BA7D | Swiss Black Pied cattle breed
SRT | L-8BA7E | Sychevka cattle breed
SRT | L-8BA7F | Sykia cattle breed
SRT | L-8BA80 | Minhota cattle breed
SRT | L-8BA81 | Minorcan cattle breed
SRT | L-8BA82 | Mishima cattle breed
SRT | L-8BA83 | Modenese cattle breed
SRT | L-8BA84 | Monchina cattle breed
SRT | L-8BA85 | Montafon cattle breed
SRT | L-8BA86 | Montbeliard cattle breed
SRT | L-8BA87 | Morenas del Noroeste cattle breed
SRT | L-8BA88 | Murcian cattle breed
SRT | L-8BA89 | Murnau-Werdenfels cattle breed
SRT | L-8BA8A | Tagil cattle breed
SRT | L-8BA8B | Tajma cattle breed
SRT | L-8BA8C | Tambov Red cattle breed
SRT | L-8BA8D | Tarina cattle breed
SRT | L-8BA8E | Thessaly cattle breed
SRT | L-8BA8F | Tinima cattle breed
SRT | L-8BA90 | Nantais cattle breed
SRT | L-8BA91 | Nejdi cattle breed
SRT | L-8BA92 | N'Gabou cattle breed
SRT | L-8BA93 | North Finncattle cattle breed
SRT | L-8BA94 | Oropa cattle breed
SRT | L-8BA95 | Oulmes Blond cattle breed
SRT | L-8BA96 | Pajuna cattle breed
SRT | L-8BA97 | Palmera cattle breed
SRT | L-8BA98 | Pankota Red cattle breed
SRT | L-8BA99 | Paphos cattle breed
SRT | L-8BA9A | Tinos cattle breed
SRT | L-8BA9B | Transylvanian Pinzgua cattle breed
SRT | L-8BA9C | Tropical Dairy Cattle cattle breed
SRT | L-8BA9D | Tropicana cattle breed
SRT | L-8BA9E | Tudanca cattle breed
SRT | L-8BA9F | Turino cattle breed
SRT | L-8BAA0 | Turkish Brown cattle breed
SRT | L-8BAA1 | Tux-Zillertal cattle breed
SRT | L-8BAA2 | Tyrol Gray cattle breed
SRT | L-8BAA3 | Abondance cattle breed
SRT | L-8BAA4 | Ala-Tau cattle breed
SRT | L-8BAA5 | Albanian Illyrian cattle breed
SRT | L-8BAA6 | Albanian Dwarf cattle breed
SRT | L-8BAA7 | Ukrainian Whiteheaded cattle breed
SRT | L-8BAA8 | Ural Black Pied cattle breed
SRT | L-8BAA9 | Valdres cattle breed
SRT | L-8BAAA | Vaynol cattle breed
SRT | L-8BAAB | Verinesa cattle breed
SRT | L-8BAAC | Vianesa cattle breed
SRT | L-8BAAD | Villard-de-Lans cattle breed
SRT | L-8BAAE | Vogelsberg cattle breed
SRT | L-8BAAF | Pie Rouge des Plaines cattle breed
SRT | L-8BAB0 | Vorderwald cattle breed
SRT | L-8BAB1 | West African Dwarf Shorthorn cattle breed
SRT | L-8BAB2 | West Finnish cattle breed
SRT | L-8BAB3 | West Macedonian cattle breed
SRT | L-8BAB4 | Whitebred Shorthorn cattle breed
SRT | L-8BAB5 | White Galloway cattle breed
SRT | L-8BAB6 | White Welsh cattle breed
SRT | L-8BAB7 | Witrik cattle breed
SRT | L-8BAB8 | Yacumento cattle breed
SRT | L-8BAB9 | Yaroslavl cattle breed
SRT | L-8BABA | Yurino cattle breed
SRT | L-8BABB | Aleppo cattle breed
SRT | L-8BABC | Schwyz cattle breed
SRT | L-8BABD | Busa cattle breed
SRT | L-8BABE | Chiangus cattle breed
SRT | L-8BABF | Hallingdal cattle breed
SRT | L-8BAC0 | Danish Jersey cattle breed
SRT | L-8BAC1 | Enderby Island cattle breed
SRT | L-8BAC2 | German Angus cattle breed
SRT | L-8BAC3 | Israeli Red cattle breed
SRT | L-8BAC4 | Lineback cattle breed
SRT | L-8BAC5 | Mertolenga cattle breed
SRT | L-8BAC6 | Red Friesian cattle breed
SRT | L-8BAC7 | Senegus cattle breed
SRT | L-8BAC8 | Southern Crioulo cattle breed
SRT | L-8BAC9 | Vosges cattle breed
SRT | L-8BACA | Montanara cattle breed
SRT | L-8BACB | Almanzorena cattle breed
SRT | L-8BACC | Lorquina cattle breed
SRT | L-8BACD | Calasparrena cattle breed
SRT | L-8BACE | Amrit Mahal zebu cattle breed
SRT | L-8BACF | Bachaur cattle breed
SRT | L-8BAD0 | Barka zebu cattle breed
SRT | L-8BAD1 | Bengali cattle breed
SRT | L-8BAD2 | Bhagnari cattle breed
SRT | L-8BAD3 | Boran cattle breed
SRT | L-8BAD4 | Channi cattle breed
SRT | L-8BAD5 | Cholistani cattle breed
SRT | L-8BAD6 | Dajal cattle breed
SRT | L-8BAD7 | Dangi cattle breed
SRT | L-8BAD8 | Deoni cattle breed
SRT | L-8BAD9 | Dhanni cattle breed
SRT | L-8BADA | Gaolao cattle breed
SRT | L-8BADB | Hallikar cattle breed
SRT | L-8BADC | Hariana cattle breed
SRT | L-8BADD | Indo-Brazilian cattle breed
SRT | L-8BADE | Kangayam cattle breed
SRT | L-8BADF | Kankrej cattle breed
SRT | L-8BAE0 | Kenkatha cattle breed
SRT | L-8BAE1 | Kherigarh cattle breed
SRT | L-8BAE2 | Khillari cattle breed
SRT | L-8BAE3 | Krishna Valley cattle breed
SRT | L-8BAE4 | Lohani cattle breed
SRT | L-8BAE5 | Malvi cattle breed
SRT | L-8BAE6 | Mewati cattle breed
SRT | L-8BAE7 | Nagori cattle breed
SRT | L-8BAE9 | Nelore cattle breed
SRT | L-8BAEA | Nimari cattle breed
SRT | L-8BAEB | Ponwar cattle breed
SRT | L-8BAEC | Rath cattle breed
SRT | L-8BAED | Rathi cattle breed
SRT | L-8BAEE | Red Sindhi cattle breed
SRT | L-8BAEF | Rojhan cattle breed
SRT | L-8BAF0 | Sahiwal cattle breed
SRT | L-8BAF1 | Siri zebu cattle breed
SRT | L-8BAF2 | Tharparkar cattle breed
SRT | L-8BAF3 | Zanzibar Zebu cattle breed
SRT | L-8BAF4 | Arsi cattle breed
SRT | L-8BAF5 | Atpadi Mahal cattle breed
SRT | L-8BAF6 | Azaouak cattle breed
SRT | L-8BAF7 | Azerbaijan Zebu cattle breed
SRT | L-8BAF8 | Baggara cattle breed
SRT | L-8BAF9 | Bambawa cattle breed
SRT | L-8BAFA | Bami cattle breed
SRT | L-8BAFB | Banyo cattle breed
SRT | L-8BAFC | Bargur cattle breed
SRT | L-8BAFD | Bari cattle breed
SRT | L-8BAFE | Bimal cattle breed
SRT | L-8BAFF | Borneo Zebu cattle breed
SRT | L-8BB00 | Butana cattle breed
SRT | L-8BB01 | Chittagong Red cattle breed
SRT | L-8BB02 | Cutchi cattle breed
SRT | L-8BB03 | Dairy Zebu of Uberaba cattle breed
SRT | L-8BB04 | Dashtiari cattle breed
SRT | L-8BB05 | Diali cattle breed
SRT | L-8BB06 | Didinga cattle breed
SRT | L-8BB07 | Dongola cattle breed
SRT | L-8BB09 | Fellata cattle breed
SRT | L-8BB0A | Turkmen zebu cattle breed
SRT | L-8BB0B | Abyssinian Highland Zebu cattle breed
SRT | L-8BB0C | Abyssinian Shorthorned Zebu cattle breed
SRT | L-8BB0E | Aceh cattle breed
SRT | L-8BB0F | Achham cattle breed
SRT | L-8BB10 | Garre cattle breed
SRT | L-8BB11 | Gasara cattle breed
SRT | L-8BB12 | Gobra cattle breed
SRT | L-8BB13 | Goomsur cattle breed
SRT | L-8BB14 | Gujamavu cattle breed
SRT | L-8BB15 | Leiqiong cattle breed
SRT | L-8BB16 | Hissar cattle breed
SRT | L-8BB17 | Ingessana cattle breed
SRT | L-8BB18 | Jamaica Brahman cattle breed
SRT | L-8BB19 | Jellicut cattle breed
SRT | L-8BB1A | Adamawa cattle breed
SRT | L-8BB1B | Aden Zebu cattle breed
SRT | L-8BB1C | Afghan cattle breed
SRT | L-8BB1D | Alambadi cattle breed
SRT | L-8BB1E | Umblachery cattle breed
SRT | L-8BB1F | Venezuelan Zebu cattle breed
SRT | L-8BB20 | Pantaneiro cattle breed
SRT | L-8BB21 | Jenubi cattle breed
SRT | L-8BB22 | Jiddu cattle breed
SRT | L-8BB23 | Jijjiga Zebu cattle breed
SRT | L-8BB24 | Kabota cattle breed
SRT | L-8BB25 | Kachcha Siri cattle breed
SRT | L-8BB26 | Kalakheri cattle breed
SRT | L-8BB27 | Kamdhino cattle breed
SRT | L-8BB28 | Kandahari cattle breed
SRT | L-8BB29 | Kaningan cattle breed
SRT | L-8BB2A | Wakwa cattle breed
SRT | L-8BB2B | White Fulani cattle breed
SRT | L-8BB2C | Yemeni Zebu cattle breed
SRT | L-8BB2D | Iranian Zebu cattle breed
SRT | L-8BB2E | Khorsan cattle breed
SRT | L-8BB2F | Polled Gir cattle breed
SRT | L-8BB30 | Kappiliyan cattle breed
SRT | L-8BB31 | Karamajong cattle breed
SRT | L-8BB32 | Kenana cattle breed
SRT | L-8BB33 | Kenya Boran cattle breed
SRT | L-8BB34 | Kenya Zebu cattle breed
SRT | L-8BB35 | Khamala cattle breed
SRT | L-8BB36 | Khurasani zebu cattle breed
SRT | L-8BB37 | Kilara cattle breed
SRT | L-8BB38 | Kinniya cattle breed
SRT | L-8BB39 | Konari cattle breed
SRT | L-8BB3A | Guzerat cattle breed
SRT | L-8BB3B | Tadzhik zebu cattle breed
SRT | L-8BB3C | Deogir cattle breed
SRT | L-8BB3D | Gayal cattle breed
SRT | L-8BB3E | American bison X cattle breed
SRT | L-8BB3F | Australian Braford X zebu cattle breed
SRT | L-8BB40 | Krishnagari cattle breed
SRT | L-8BB41 | Kumauni cattle breed
SRT | L-8BB42 | Ladakhi cattle breed
SRT | L-8BB43 | Latuka cattle breed
SRT | L-8BB44 | Lugware cattle breed
SRT | L-8BB45 | Madagascar Zebu cattle breed
SRT | L-8BB46 | Madaripur cattle breed
SRT | L-8BB47 | Magal cattle breed
SRT | L-8BB48 | Malawi Zebu cattle breed
SRT | L-8BB49 | Malnad Gidda cattle breed
SRT | L-8BB4A | Australian Friesian Sahiwal X zebu cattle breed
SRT | L-8BB4B | Braford X zebu cattle breed
SRT | L-8BB4C | Brahmousin X zebu cattle breed
SRT | L-8BB4D | Canchim X zebu cattle breed
SRT | L-8BB4E | Charbray X zebu cattle breed
SRT | L-8BB4F | Droughtmaster X zebu cattle breed
SRT | L-8BB50 | Mampati cattle breed
SRT | L-8BB51 | Manapari cattle breed
SRT | L-8BB52 | Maure cattle breed
SRT | L-8BB53 | Mazandarani cattle breed
SRT | L-8BB54 | Merauke cattle breed
SRT | L-8BB55 | Mere cattle breed
SRT | L-8BB56 | Mhaswad cattle breed
SRT | L-8BB57 | Miniature Zebu cattle breed
SRT | L-8BB58 | Mongalla cattle breed
SRT | L-8BB59 | Morang cattle breed
SRT | L-8BB5A | Gelbray X zebu cattle breed
SRT | L-8BB5B | Jamaica Black X zebu cattle breed
SRT | L-8BB5C | Jamaica Hope X zebu cattle breed
SRT | L-8BB5D | Jamaica Red X zebu cattle breed
SRT | L-8BB5E | Karan Fries X zebu cattle breed
SRT | L-8BB5F | Karan Swiss X zebu cattle breed
SRT | L-8BB60 | Mozambique Angoni cattle breed
SRT | L-8BB61 | Mpwapwa cattle breed
SRT | L-8BB62 | Murle cattle breed
SRT | L-8BB63 | Nakali cattle breed
SRT | L-8BB64 | Nepalese Hill Zebu cattle breed
SRT | L-8BB65 | N'Gaoundere cattle breed
SRT | L-8BB66 | Nkedi cattle breed
SRT | L-8BB67 | North Bangladesh Gray cattle breed
SRT | L-8BB68 | North Somali Zebu cattle breed
SRT | L-8BB69 | Polled Guzerat cattle breed
SRT | L-8BB6A | Mandalong X zebu cattle breed
SRT | L-8BB6B | Australian Milking Zebu X zebu cattle breed
SRT | L-8BB6C | Red Brangus X zebu cattle breed
SRT | L-8BB6D | Santa Cruz X zebu cattle breed
SRT | L-8BB6E | Siboney X zebu cattle breed
SRT | L-8BB6F | Bambara X zebu cattle breed
SRT | L-8BB70 | Polled Nelore cattle breed
SRT | L-8BB71 | Prewakwa cattle breed
SRT | L-8BB72 | Pul-M'bor cattle breed
SRT | L-8BB73 | Punganur cattle breed
SRT | L-8BB74 | Ramgarhi cattle breed
SRT | L-8BB75 | Red Bororo cattle breed
SRT | L-8BB76 | Red Desert cattle breed
SRT | L-8BB77 | Red Kandhari cattle breed
SRT | L-8BB78 | Shakhansurri cattle breed
SRT | L-8BB79 | Sheko cattle breed
SRT | L-8BB7A | Bambey X zebu cattle breed
SRT | L-8BB7B | Batanes Black X zebu cattle breed
SRT | L-8BB7C | Borgou X zebu cattle breed
SRT | L-8BB7D | Brahorn X zebu cattle breed
SRT | L-8BB7E | Bralers X zebu cattle breed
SRT | L-8BB7F | Bra-Maine X zebu cattle breed
SRT | L-8BB80 | Shendi cattle breed
SRT | L-8BB81 | Shuwa cattle breed
SRT | L-8BB82 | Sinhala cattle breed
SRT | L-8BB83 | Sistani cattle breed
SRT | L-8BB84 | Small East African Zebu cattle breed
SRT | L-8BB85 | Sokoto Gudali cattle breed
SRT | L-8BB86 | Somali cattle breed
SRT | L-8BB87 | Sonkheri cattle breed
SRT | L-8BB88 | Son Valley cattle breed
SRT | L-8BB89 | South China Zebu cattle breed
SRT | L-8BB8A | Bra-Swiss X zebu cattle breed
SRT | L-8BB8B | Bravon X zebu cattle breed
SRT | L-8BB8C | Brazilian Dairy Hybrid X zebu cattle breed
SRT | L-8BB8D | Burmese X zebu cattle breed
SRT | L-8BB8E | Bushuev X zebu cattle breed
SRT | L-8BB8F | Caiua X zebu cattle breed
SRT | L-8BB90 | South Malawi Zebu cattle breed
SRT | L-8BB91 | Sudanese Fulani cattle breed
SRT | L-8BB92 | Tabapua cattle breed
SRT | L-8BB93 | Tamankaduwa cattle breed
SRT | L-8BB94 | Tanzanian Zebu cattle breed
SRT | L-8BB95 | Tarai cattle breed
SRT | L-8BB96 | Thillari cattle breed
SRT | L-8BB97 | Toposa cattle breed
SRT | L-8BB98 | Toronke cattle breed
SRT | L-8BB99 | Toupouri cattle breed
SRT | L-8BB9A | Carazebu X zebu cattle breed
SRT | L-8BB9B | Central Asian Zebu X zebu cattle breed
SRT | L-8BB9C | Charford X zebu cattle breed
SRT | L-8BB9D | Cuban Criollo X zebu cattle breed
SRT | L-8BB9E | Cuban Zebu X zebu cattle breed
SRT | L-8BB9F | Dishty X zebu cattle breed
SRT | L-8BC00 | Djakore X zebu cattle breed
SRT | L-8BC01 | Gambian N'Dama X zebu cattle breed
SRT | L-8BC03 | Ghana Sanga X zebu cattle breed
SRT | L-8BC04 | Girolando X zebu cattle breed
SRT | L-8BC05 | Guzerando X zebu cattle breed
SRT | L-8BC06 | Hatton X zebu cattle breed
SRT | L-8BC07 | Ibage X zebu cattle breed
SRT | L-8BC08 | Iraqi X zebu cattle breed
SRT | L-8BC09 | Jerdi X zebu cattle breed
SRT | L-8BC10 | Jersind X zebu cattle breed
SRT | L-8BC11 | Jotko X zebu cattle breed
SRT | L-8BC12 | Kanem X zebu cattle breed
SRT | L-8BC13 | Keteku X zebu cattle breed
SRT | L-8BC14 | Lavinia X zebu cattle breed
SRT | L-8BC15 | Local Indian Dairy X zebu cattle breed
SRT | L-8BC16 | Mantiqueira X zebu cattle breed
SRT | L-8BC17 | Ndagu X zebu cattle breed
SRT | L-8BC18 | Normanzu X zebu cattle breed
SRT | L-8BC19 | Nuba Mountain X zebu cattle breed
SRT | L-8BC20 | Pabna X zebu cattle breed
SRT | L-8BC21 | Mixed Perijanero X zebu cattle breed
SRT | L-8BC22 | Pitangueiras X zebu cattle breed
SRT | L-8BC23 | Quasah X zebu cattle breed
SRT | L-8BC24 | Rana X zebu cattle breed
SRT | L-8BC25 | Ranger X zebu cattle breed
SRT | L-8BC26 | Renitelo X zebu cattle breed
SRT | L-8BC27 | Riopardenze X zebu cattle breed
SRT | L-8BC28 | Rustaqi X zebu cattle breed
SRT | L-8BC29 | Sabre X zebu cattle breed
SRT | L-8BC30 | Sahford X zebu cattle breed
SRT | L-8BC31 | Schwyz-Zeboid X zebu cattle breed
SRT | L-8BC32 | Suia X zebu cattle breed
SRT | L-8BC33 | Suisbu X zebu cattle breed
SRT | L-8BC34 | Sunandini X zebu cattle breed
SRT | L-8BC35 | Taino X zebu cattle breed
SRT | L-8BC36 | Thibar X zebu cattle breed
SRT | L-8BC37 | Toubou X zebu cattle breed
SRT | L-8BC38 | Tropical X zebu cattle breed
SRT | L-8BC39 | TSSH-1 X zebu cattle breed
SRT | L-8BC40 | Victoria X zebu cattle breed
SRT | L-8BC41 | Wokalup X zebu cattle breed
SRT | L-8BC42 | Madura wild javan X zebu cattle breed
SRT | L-80A40 | Rex cat breed
SRT | L-80770 | Dachshund superbreed of dog
SRT | L-80320 | Dorset sheep superbreed
SRT | L-80A42 | Devon rex cat breed
SRT | L-80A41 | Cornish rex cat breed
SRT | L-80A45 | Oregon rex cat breed
SRT | L-80A05 | Abyssinian cat
SRT | L-80A06 | American shorthair cat
SRT | L-80A07 | American wirehaired cat
SRT | L-80A08 | Balinese cat
SRT | L-80A09 | Birman cat
SRT | L-80A10 | Bombay cat
SRT | L-80A11 | British shorthaired cat
SRT | L-80A12 | Burmese cat
SRT | L-D9814 | Cestrum parqui
SRT | L-80A13 | Chartreux cat
SRT | L-80A51 | Colourpoint shorthaired cat
SRT | L-80A19 | Domestic leopard cat
SRT | L-80A20 | Domestic longhaired cat
SRT | L-8880B | Domestic medium-haired cat
SRT | L-80A52 | Domestic shorthaired cat
SRT | L-80A14 | Egyptian mau cat
SRT | L-80A53 | Exotic shorthaired cat
SRT | L-80A15 | Havana brown cat
SRT | L-80A16 | Japanese bobtail cat
SRT | L-80A17 | Javanese cat
SRT | L-80A18 | Korat cat
SRT | L-80A31 | Longhaired manx
SRT | L-80A21 | Maine coon cat
SRT | L-80A30 | Manx
SRT | L-80A32 | Ocicat
SRT | L-80A54 | Oriental shorthaired cat
SRT | L-80A33 | Persian cat
SRT | L-80A43 | Russian blue cat
SRT | L-80A44 | Scottish fold cat
SRT | L-80A50 | Shorthaired cat
SRT | L-80A55 | Siamese cat
SRT | L-80A56 | Singapura cat
SRT | L-80A57 | Somali cat
SRT | L-80A58 | Tonkinese cat
SRT | L-80A59 | Turkish angora cat
SRT | L-80705 | Affenpinscher
SRT | L-80706 | Afghan hound
SRT | L-80707 | Airedale terrier
SRT | L-80708 | Akita dog
SRT | L-80709 | Alaskan malamute
SRT | L-807A4 | American foxhound
SRT | L-80711 | Australian cattle dog
SRT | L-80710 | Australian terrier
SRT | L-80712 | Basenji
SRT | L-80713 | Basset hound
SRT | L-80714 | Beagle
SRT | L-80715 | Bedlington terrier
SRT | L-80716 | Belgian groenendael dog
SRT | L-80717 | Belgian laeken dog
SRT | L-80718 | Belgian malinois dog
SRT | L-80719 | Belgian sheepdog
SRT | L-80720 | Belgian tervuren dog
SRT | L-80721 | Bernese mountain dog
SRT | L-80722 | Bichons frise dog
SRT | L-80723 | Bloodhound
SRT | L-80724 | Border terrier
SRT | L-80725 | Borzoi dog
SRT | L-80726 | Boston terrier
SRT | L-80727 | Bouvier des Flandres
SRT | L-80728 | Boxer dog
SRT | L-80729 | Briard dog
SRT | L-80730 | Bull terrier
SRT | L-80735 | Bulldog
SRT | L-80736 | Bullmastiff
SRT | L-80737 | Cairn terrier
SRT | L-80738 | Cavalier King Charles spaniel
SRT | L-80744 | Chow Chow
SRT | L-80750 | Collie
SRT | L-80760 | Coonhound
SRT | L-80777 | Dalmatian dog
SRT | L-80778 | Dandie dinmont terrier
SRT | L-80780 | Doberman pinscher
SRT | L-80781 | Drever dog
SRT | L-807A5 | English foxhound
SRT | L-80782 | English toy spaniel
SRT | L-80790 | Eskimo dog
SRT | L-80793 | Finnish spitz dog
SRT | L-807B0 | Foxhound
SRT | L-807B1 | French bulldog
SRT | L-807B2 | German shepherd dog
SRT | L-807B4 | Great Pyrenee dog
SRT | L-807B3 | Great dane dog
SRT | L-807B5 | Greyhound
SRT | L-807C0 | Griffon dog
SRT | L-807C3 | Harrier dog
SRT | L-80702 | Hound
SRT | L-807C4 | Ibizan hound
SRT | L-807C6 | Irish terrier
SRT | L-807C5 | Irish wolfhound
SRT | L-807C7 | Italian greyhound
SRT | L-807C8 | Jack Russel terrier
SRT | L-807C9 | Japanese chin dog
SRT | L-807D0 | Japanese spaniel
SRT | L-807D1 | Karelian bear dog
SRT | L-807D2 | Keeshond
SRT | L-807D3 | Kerry blue terrier
SRT | L-807D4 | Komondor dog
SRT | L-807D5 | Kuvasz dog
SRT | L-807D6 | Lakeland terrier
SRT | L-807D7 | Lhasa apso
SRT | L-807D8 | Maltese dog
SRT | L-80803 | Mastiff dog
SRT | L-80804 | Mexican hairless dog
SRT | L-80805 | Miniature pinscher dog
SRT | L-80806 | Newfoundland dog
SRT | L-80807 | Norfolk terrier
SRT | L-80808 | Norwegian elkhound
SRT | L-80809 | Norwich terrier
SRT | L-80810 | Old English sheepdog
SRT | L-80811 | Otter hound
SRT | L-80812 | Papillon dog
SRT | L-80813 | Pekingese dog
SRT | L-80814 | Petit basset griffon vendeen dog
SRT | L-80815 | Pharaoh hound
SRT | L-80816 | Plott hound
SRT | L-80820 | Pointer
SRT | L-80824 | Pomeranian dog
SRT | L-80834 | Portuguese water dog
SRT | L-80835 | Pudelpointer
SRT | L-80836 | Pug dog
SRT | L-80837 | Puli dog
SRT | L-80840 | Retriever
SRT | L-80847 | Rhodesian ridgeback dog
SRT | L-80848 | Rottweiler dog
SRT | L-80849 | Saluki dog
SRT | L-80850 | Samoyed dog
SRT | L-80851 | Schipperke dog
SRT | L-80779 | Scottish deerhound
SRT | L-80864 | Scottish terrier
SRT | L-80865 | Sealyham terrier
SRT | L-80870 | Setter
SRT | L-80874 | Shetland sheepdog
SRT | L-80875 | Shih tzu dog
SRT | L-80876 | Siberian huskie
SRT | L-80877 | Silky terrier
SRT | L-80878 | Skye terrier
SRT | L-80879 | Soft-coated wheaten terrier
SRT | L-80880 | Spaniel
SRT | L-80895 | St. Bernard dog
SRT | L-80801 | Standard Manchester terrier
SRT | L-80896 | Tahltan bear dog
SRT | L-80703 | Terrier
SRT | L-80897 | Tibetan spaniel
SRT | L-80898 | Tibetan terrier
SRT | L-80802 | Toy Manchester terrier
SRT | L-80903 | Weimaraner
SRT | L-80904 | Welsh terrier
SRT | L-80913 | West Highland white terrier
SRT | L-80914 | Whippet dog
SRT | L-807C2 | Wirehaired pointing griffon dog
SRT | L-88120 | Wolf
SRT | L-80915 | Yorkshire terrier
SRT | L-80105 | Aberdeen Angus cow breed
SRT | L-80106 | Ayrshire cow breed
SRT | L-80108 | Black Angus cow breed
SRT | L-80112 | Blonde d'Aquitaine cow breed
SRT | L-80113 | Brahma cow breed
SRT | L-80115 | Brown Swiss cow breed
SRT | L-80116 | Canadian cow breed
SRT | L-80118 | Chianina cow breed
SRT | L-80119 | Criollo cow breed
SRT | L-80120 | Dexter cow breed
SRT | L-80130 | Galloway cow breed
SRT | L-80131 | Belted Galloway cow breed
SRT | L-80132 | Gelbveih cow breed
SRT | L-80133 | German Fleck-Vieh cow breed
SRT | L-80134 | Gir cow breed
SRT | L-80135 | Guernsey cow breed
SRT | L-80136 | Gujarati cow breed
SRT | L-80137 | Hays converter cow breed
SRT | L-80141 | Horned Hereford cow breed
SRT | L-80142 | Polled Hereford cow breed
SRT | L-80143 | Holstein-Friesian cow breed
SRT | L-80144 | Jersey cow breed
SRT | L-80145 | Limousin cow breed
SRT | L-80146 | Lincoln red cow breed
SRT | L-80147 | Longhorn cow breed
SRT | L-80148 | Luing cow breed
SRT | L-80149 | Maine Anjou cow breed
SRT | L-80150 | Marchigiana cow breed
SRT | L-80151 | Meusse-Rhine-Ijssel cow breed
SRT | L-80153 | Nellore cow breed
SRT | L-80154 | Normandie cow breed
SRT | L-80156 | Pinzgauer cow breed
SRT | L-80157 | Red Poll cow breed
SRT | L-80158 | Salers cow breed
SRT | L-80160 | Scottish Highland cow breed
SRT | L-80170 | Shorthorn cow breed
SRT | L-80171 | Milking Shorthorn cow breed
SRT | L-80172 | Simmental cow breed
SRT | L-80173 | Tarentaise cow breed
SRT | L-80174 | Black Welsh cow breed
SRT | L-80175 | Brown Welsh cow breed
SRT | L-80176 | White Park cow breed
SRT | L-801E8 | Bison bison X Simmental hybrid
SRT | L-80205 | Alpine goat breed
SRT | L-80206 | French alpine goat breed
SRT | L-80207 | Rock alpine goat breed
SRT | L-80208 | Angora goat breed
SRT | L-80209 | Camarron goat breed
SRT | L-80210 | Chamoisee goat breed
SRT | L-80211 | La Mancha goat breed
SRT | L-80212 | Anglo nubian goat breed
SRT | L-80213 | Pygmy goat breed
SRT | L-80214 | Saanen goat breed
SRT | L-80215 | Swiss alpine goat breed
SRT | L-80216 | Toggenburg goat breed
SRT | L-80218 | Australian goat breed
SRT | L-80219 | Arapawa Island goat breed
SRT | L-8021A | Maltese goat breed
SRT | L-8021B | Provençale goat breed
SRT | L-8021C | Negra Serrana goat breed
SRT | L-8021D | Orobica goat breed
SRT | L-8021E | Roya-Vesubie goat breed
SRT | L-8021F | Retinta Extremena goat breed
SRT | L-80220 | Appenzell goat breed
SRT | L-80221 | American Cashmere goat breed
SRT | L-80222 | Altai Mountain goat breed
SRT | L-80223 | Pyrenean goat breed
SRT | L-80224 | Bagot goat breed
SRT | L-80225 | Russian White goat breed
SRT | L-80226 | Moxotó goat breed
SRT | L-80227 | Myotonic goat breed
SRT | L-80228 | Nachi goat breed
SRT | L-80229 | Nigerian Dwarf goat breed
SRT | L-8022A | Sarda goat breed
SRT | L-8022B | Serpentina goat breed
SRT | L-8022C | Serrana goat breed
SRT | L-8022D | Verata goat breed
SRT | L-8022E | Verzasca goat breed
SRT | L-80230 | Norwegian goat breed
SRT | L-80231 | Oberhasli goat breed
SRT | L-80232 | Peacock goat breed
SRT | L-80233 | Philippine goat breed
SRT | L-80234 | Loashan goat breed
SRT | L-80235 | San Clemente goat breed
SRT | L-80236 | Somali goat breed
SRT | L-80237 | Spanish goat breed
SRT | L-80238 | Rove goat breed
SRT | L-80239 | SRD goat breed
SRT | L-80240 | Swedish Landrace goat breed
SRT | L-80241 | Thuringian goat breed
SRT | L-80242 | Uzbek Black goat breed
SRT | L-80243 | Zhongwei goat breed
SRT | L-80244 | Barbari goat breed
SRT | L-80245 | Poitou goat breed
SRT | L-80246 | Repartida goat breed
SRT | L-80247 | Booted goat breed
SRT | L-80248 | Corsican goat breed
SRT | L-80249 | Chapar goat breed
SRT | L-80250 | Canindé goat breed
SRT | L-80251 | Canary Island goat breed
SRT | L-80252 | Daera Din Panah goat breed
SRT | L-80253 | British Alpine goat breed
SRT | L-80254 | Bhuj goat breed
SRT | L-80255 | Boer goat breed
SRT | L-80256 | Benadir goat breed
SRT | L-80257 | Creole Antilles goat breed
SRT | L-80258 | Beetal goat breed
SRT | L-80259 | Golden Guernsey goat breed
SRT | L-80260 | Danish Landrace goat breed
SRT | L-80261 | Kaghani goat breed
SRT | L-80263 | Irish goat breed
SRT | L-80265 | Grisons Striped goat breed
SRT | L-80266 | Jining Gray goat breed
SRT | L-80267 | Finnish Landrace goat breed
SRT | L-80268 | Erzgebirg goat breed
SRT | L-80269 | Kamori goat breed
SRT | L-80270 | Don goat breed
SRT | L-80271 | Kiko goat breed
SRT | L-80272 | Kinder goat breed
SRT | L-80273 | Pygora goat breed
SRT | L-80274 | Wooden Leg goat breed
SRT | L-80275 | Alpine Chamoisee goat breed
SRT | L-80276 | Massif Central goat breed
SRT | L-80277 | Malagueña goat breed
SRT | L-80278 | Algarvia goat breed
SRT | L-80279 | British Saanen goat breed
SRT | L-80280 | British Toggenburg goat breed
SRT | L-80281 | Bündner goat breed
SRT | L-80282 | Blanca Andaluza goat breed
SRT | L-80283 | Blanca Celtiberica goat breed
SRT | L-80284 | Bravia goat breed
SRT | L-80285 | Black Grisonne goat breed
SRT | L-80286 | Chamois of the Alps goat breed
SRT | L-80287 | Charnequeria goat breed
SRT | L-80288 | Carpathe goat breed
SRT | L-80289 | Col Noir du Valais goat breed
SRT | L-80290 | Damani goat breed
SRT | L-80291 | Des Fosses (Communes de l'Ouest) goat breed
SRT | L-80292 | English goat breed
SRT | L-80293 | English Guernsey goat breed
SRT | L-80294 | German colored goat breed
SRT | L-80295 | Guadarrama goat breed
SRT | L-80296 | Garganica goat breed
SRT | L-80297 | Girgentana goat breed
SRT | L-80298 | Jonica goat breed
SRT | L-80299 | Murciana-Granadina goat breed
SRT | L-80306 | Barbados sheep breed
SRT | L-80307 | Black faced Highland sheep breed
SRT | L-80308 | Cheviot sheep breed
SRT | L-80309 | Clun Forest sheep breed
SRT | L-80310 | Corriedale sheep breed
SRT | L-80311 | Cotswold sheep breed
SRT | L-80312 | Debouillet sheep breed
SRT | L-80321 | Horn dorset sheep breed
SRT | L-80322 | Finnish landrace sheep breed
SRT | L-80323 | Karakul sheep breed
SRT | L-80324 | Kerry Hill sheep breed
SRT | L-80325 | Leicester sheep breed
SRT | L-80326 | Lincoln sheep breed
SRT | L-80327 | Hampshire Down sheep breed
SRT | L-80331 | American merino sheep breed
SRT | L-80332 | Delaine merino sheep breed
SRT | L-80333 | Montdale sheep breed
SRT | L-80334 | Mouflon sheep breed
SRT | L-80335 | Navajo sheep breed
SRT | L-80336 | No-tail sheep breed
SRT | L-80337 | North County cheviot sheep breed
SRT | L-80338 | Oxford Down sheep breed
SRT | L-80339 | Panama sheep breed
SRT | L-80340 | Perendale sheep breed
SRT | L-80341 | Rambouillet sheep breed
SRT | L-80342 | Romanov sheep breed
SRT | L-80343 | Romedale sheep breed
SRT | L-80344 | Romnelet sheep breed
SRT | L-80345 | Romney marsh sheep breed
SRT | L-80346 | Shropshire sheep breed
SRT | L-80347 | Southdown sheep breed
SRT | L-80348 | Suffolk sheep breed
SRT | L-80349 | Targhee sheep breed
SRT | L-80350 | Wiltshire horn sheep breed
SRT | L-80405 | American Albino horse breed
SRT | L-80406 | American Buckskin horse breed
SRT | L-80407 | American cream horse breed
SRT | L-80408 | American miniature horse breed
SRT | L-80409 | American paint horse breed
SRT | L-80410 | American saddlebred horse breed
SRT | L-80411 | American trotter horse breed
SRT | L-80412 | American tunis horse breed
SRT | L-80413 | Andalusian horse breed
SRT | L-80414 | Appaloosa horse breed
SRT | L-80415 | Arabian horse breed
SRT | L-80416 | Belgian horse breed
SRT | L-80417 | Canadian horse breed
SRT | L-80418 | Cleveland bay horse breed
SRT | L-80419 | Clydesdale horse breed
SRT | L-80421 | Fjord horse breed
SRT | L-80422 | Galiceno horse breed
SRT | L-80423 | Hackney horse breed
SRT | L-80424 | Haflinger horse breed
SRT | L-80425 | Hanoverian horse breed
SRT | L-80426 | Holsteiner horse breed
SRT | L-80427 | Hunter horse breed
SRT | L-80428 | Icelandic horse breed
SRT | L-80429 | Lipizzaner horse breed
SRT | L-80430 | Missouri fox trotting horse breed
SRT | L-80431 | Morgan horse breed
SRT | L-80433 | New Forest pony horse breed
SRT | L-80435 | Norman coach horse breed
SRT | L-80436 | Palomino horse breed
SRT | L-80437 | Paso Fino horse breed
SRT | L-80438 | Percheron horse breed
SRT | L-80439 | Peruvian Paso horse breed
SRT | L-80440 | Pinto horse breed
SRT | L-80450 | Pony horse breed
SRT | L-80451 | American pony horse breed
SRT | L-80452 | Shetland pony horse breed
SRT | L-80453 | Ariégeois pony horse breed
SRT | L-80454 | Quarter horse breed
SRT | L-80455 | Shire horse breed
SRT | L-80456 | Spanish mustang horse breed
SRT | L-80457 | Standardbred horse breed
SRT | L-80458 | Suffolk horse breed
SRT | L-80459 | Tennessee walking horse breed
SRT | L-80461 | Trakehner horse breed
SRT | L-80462 | Viking horse breed
SRT | L-80463 | Welsh walking horse breed
SRT | L-80464 | Westphalian horse breed
SRT | L-80465 | Yorkshire coach horse breed
SRT | L-80495 | Draft pony superbreed horse breed
SRT | L-804A0 | American draft pony horse breed
SRT | L-804B0 | Pindos pony horse breed
SRT | L-804C0 | Skyros pony horse breed
SRT | L-80505 | Beltsville pig breed
SRT | L-80506 | Beltsville pig #1 pig breed
SRT | L-80507 | Beltsville pig #2 pig breed
SRT | L-80510 | Berkshire pig breed
SRT | L-80511 | Kentucky red berkshire pig breed
SRT | L-80520 | Boar power pig breed
SRT | L-80521 | Boar power pig 27 pig breed
SRT | L-80522 | Boar power pig 48 pig breed
SRT | L-80523 | Boar power pig 59 pig breed
SRT | L-80524 | Boar power pig 72 pig breed
SRT | L-80525 | Boar power pig 84 pig breed
SRT | L-80526 | Boar power pig 141 pig breed
SRT | L-80527 | Boar power pig 161 pig breed
SRT | L-80528 | Boar power pig 282 pig breed
SRT | L-80529 | Boar power pig 292 pig breed
SRT | L-80530 | Boar power pig 414 pig breed
SRT | L-80531 | Boar power pig 454 pig breed
SRT | L-80532 | Boar power pig 474 pig breed
SRT | L-80533 | Boar power pig 545 pig breed
SRT | L-80534 | Boar power pig 565 pig breed
SRT | L-80535 | Boar power pig 616 pig breed
SRT | L-80536 | Boar power pig 656 pig breed
SRT | L-80537 | Boar power pig 747 pig breed
SRT | L-80538 | Boar power pig 828 pig breed
SRT | L-80539 | Boar power pig 929 pig breed
SRT | L-80540 | British lop pig breed
SRT | L-80541 | British saddleback pig breed
SRT | L-80550 | CPF pig breed
SRT | L-80551 | CPF pig #1 pig breed
SRT | L-80552 | CPF pig #2 pig breed
SRT | L-80553 | Chester white pig breed
SRT | L-80554 | Connor prairie pig breed
SRT | L-80560 | DK pig breed
SRT | L-80561 | DK pig 30 pig breed
SRT | L-80562 | DK pig 31 pig breed
SRT | L-80563 | DK pig 33 pig breed
SRT | L-80564 | DK pig 51 pig breed
SRT | L-80565 | DK pig 61 pig breed
SRT | L-80566 | DK pig 63 pig breed
SRT | L-80567 | DK pig 77 pig breed
SRT | L-80568 | Duroc pig breed
SRT | L-80570 | FHC pig breed
SRT | L-80571 | FHC elite pig 1 pig breed
SRT | L-80572 | FHC elite pig 2 pig breed
SRT | L-80573 | FHC elite pig 3 pig breed
SRT | L-80574 | FHC elite pig 4 pig breed
SRT | L-80575 | FHC elite pig 5 pig breed
SRT | L-80576 | FHC elite pig 6 pig breed
SRT | L-80577 | FHC elite pig 7 pig breed
SRT | L-80578 | FHC elite pig 8 pig breed
SRT | L-80579 | FHC elite pig 9 pig breed
SRT | L-8057A | Gloucester old spot pig breed
SRT | L-80580 | Hampshire pig breed
SRT | L-80581 | Hereford pig breed
SRT | L-80582 | Hormel miniature pig breed
SRT | L-80590 | Kleen leen pig breed
SRT | L-80591 | Kleen leen black pig breed
SRT | L-80592 | Kleen leen red pig breed
SRT | L-80593 | Kleen leen white pig breed
SRT | L-80594 | Lacombe pig breed
SRT | L-80600 | Landrace pig breed
SRT | L-80601 | Belgium landrace pig breed
SRT | L-80602 | British landrace pig breed
SRT | L-80603 | Danish landrace pig breed
SRT | L-80604 | Dutch landrace pig breed
SRT | L-80605 | French landrace pig breed
SRT | L-80606 | German landrace pig breed
SRT | L-80607 | Italian landrace pig breed
SRT | L-80608 | Norwegian landrace pig breed
SRT | L-80609 | Swedish landrace pig breed
SRT | L-80610 | Large black pig breed
SRT | L-80611 | Large white pig breed
SRT | L-80612 | Lucie pig breed
SRT | L-80620 | Maryland pig breed
SRT | L-80621 | Maryland pig #1 pig breed
SRT | L-80622 | Middle white pig breed
SRT | L-80630 | Minnesota pig breed
SRT | L-80631 | Minnesota pig #1 pig breed
SRT | L-80632 | Minnesota pig #2 pig breed
SRT | L-80633 | Minnesota pig #3 pig breed
SRT | L-80640 | Montana pig breed
SRT | L-80641 | Montana pig #1 pig breed
SRT | L-80642 | OIC pig breed
SRT | L-80643 | Oxford sandy block pig breed
SRT | L-80644 | Palouse pig breed
SRT | L-80650 | Pic pig breed
SRT | L-80651 | Pic Cambourgh pig breed
SRT | L-80652 | Pic line pig 24 pig breed
SRT | L-80653 | Pic line pig 26 pig breed
SRT | L-80654 | Pietrain pig breed
SRT | L-80655 | Poland China pig breed
SRT | L-80656 | Red wattle pig breed
SRT | L-80657 | San Pierre pig breed
SRT | L-80658 | Spotted pig breed
SRT | L-80659 | Tamworth pig breed
SRT | L-80660 | Welsh pig breed
SRT | L-80661 | Wessex saddleback pig breed
SRT | L-80662 | Yorkshire pig breed
SRT | L-80663 | Yuca pig breed
SRT | L-80731 | American pit bull terrier dog breed
SRT | L-80732 | Colored bull terrier dog breed
SRT | L-80733 | Staffordshire bull terrier dog breed
SRT | L-80734 | White bull terrier dog breed
SRT | L-80740 | Chihuahua superbreed dog breed
SRT | L-80741 | Long coat chihuahua dog breed
SRT | L-80742 | Short coat chihuahua dog breed
SRT | L-80743 | Long and short coat chihuahua dog breed
SRT | L-80751 | Bearded collie dog breed
SRT | L-80752 | Rough collie dog breed
SRT | L-80753 | Rough and smooth dog breed
SRT | L-80754 | Smooth collie dog breed
SRT | L-80761 | American coonhound dog breed
SRT | L-80762 | Black and tan coonhound dog breed
SRT | L-80763 | Blue tick coonhound dog breed
SRT | L-80764 | English coonhound dog breed
SRT | L-80765 | Redbone coonhound dog breed
SRT | L-80766 | Treeing walker coonhound dog breed
SRT | L-80771 | Longhaired miniature dachshund dog breed
SRT | L-80772 | Smooth miniature dachshund dog breed
SRT | L-80773 | Wirehaired miniature dachshund dog breed
SRT | L-80774 | Longhaired standard dachshund dog breed
SRT | L-80775 | Smooth standard dachshund dog breed
SRT | L-80776 | Wirehaired standard dachshund dog breed
SRT | L-8077A | Dachshund, Miniature dog breed
SRT | L-8077B | Standard dachshund dog breed
SRT | L-80791 | American eskimo dog breed
SRT | L-80792 | Canadian eskimo dog breed
SRT | L-807A0 | Fox terrier superbreed dog breed
SRT | L-807A1 | Smooth fox terrier dog breed
SRT | L-807A2 | Wire fox terrier dog breed
SRT | L-807A3 | Toy fox terrier dog breed
SRT | L-80800 | Manchester terrier superbreed dog breed
SRT | L-80821 | German longhaired pointer dog breed
SRT | L-80822 | German shorthaired pointer dog breed
SRT | L-80823 | German wirehaired pointer dog breed
SRT | L-80830 | Poodle superbreed dog breed
SRT | L-80831 | Toy poodle dog breed
SRT | L-80832 | Miniature poodle dog breed
SRT | L-80833 | Standard poodle dog breed
SRT | L-80841 | Chesapeake Bay retriever dog breed
SRT | L-80842 | Curly-coated retriever dog breed
SRT | L-80843 | Flat-coated retriever dog breed
SRT | L-80844 | Golden retriever dog breed
SRT | L-80845 | Labrador retriever dog breed
SRT | L-80846 | Nova Scotia duck tolling retriever dog breed
SRT | L-80860 | Schnauzer superbreed dog breed
SRT | L-80861 | Miniature schnauzer dog breed
SRT | L-80862 | Giant schnauzer dog breed
SRT | L-80863 | Standard schnauzer dog breed
SRT | L-80871 | English setter dog breed
SRT | L-80872 | Gordon setter dog breed
SRT | L-80873 | Irish setter dog breed
SRT | L-80881 | American water spaniel dog breed
SRT | L-80882 | Brittany spaniel dog breed
SRT | L-80883 | Clumber spaniel dog breed
SRT | L-80884 | American cocker spaniel dog breed
SRT | L-80885 | Black cocker spaniel dog breed
SRT | L-80886 | A.S.C.O.B. cocker spaniel dog breed
SRT | L-80887 | Parti-color cocker spaniel dog breed
SRT | L-80888 | English Springer spaniel dog breed
SRT | L-80889 | Field spaniel dog breed
SRT | L-80891 | Irish water spaniel dog breed
SRT | L-80892 | Sussex spaniel dog breed
SRT | L-80893 | Welsh Springer spaniel dog breed
SRT | L-80894 | English cocker spaniel dog breed
SRT | L-80900 | Vizsla superbreed dog breed
SRT | L-80901 | Smooth haired vizsla dog breed
SRT | L-80902 | Wirehaired vizsla dog breed
SRT | L-80910 | Welsh corgi superbreed dog breed
SRT | L-80911 | Cardigan Welsh corgi dog breed
SRT | L-80912 | Pembroke Welsh corgi dog breed
SRT | L-88106 | Alaskan Klee Kai dog breed
SRT | L-88107 | Anatolian shepherd dog breed
SRT | L-88108 | Boerboel dog breed
SRT | L-8810A | Victorian Bulldogge dog breed
SRT | L-8880C | American bobtail cat breed
SRT | L-8880D | Pixie-bob cat breed
SRT | L-8A105 | Warmblood horse breed
SRT | L-8A106 | Brabant horse breed
SRT | L-8A10B | Equus caballus gmelini horse breed
SRT | L-8A10C | Gypsy Vanner horse breed
SRT | L-8A10D | Murgese horse breed
SRT | L-8A114 | Saddlebred horse superbreed horse breed
SRT | L-8B102 | Ukrainian steppe white pig breed
SRT | L-8B943 | Bos taurus indicus cow breed
SRT | L-8B946 | Bos taurus taurus subspecies domestic European cow breed
SRT | L-8B948 | Masai cow breed
SRT | L-8B949 | Bos taurus X Bison bison hybrid cow breed
SRT | L-8C339 | Galway sheep breed

## CID 7481 Breed Registry

Type:

  Extensible

Version:

  20060822

Table CID 7481. Breed Registry

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109200 | America Kennel Club
DCM | 109201 | America's Pet Registry Inc.
DCM | 109202 | American Canine Association
DCM | 109203 | American Purebred Registry
DCM | 109204 | American Rare Breed Association
DCM | 109205 | Animal Registry Unlimited
DCM | 109206 | Animal Research Foundation
DCM | 109207 | Canadian Border Collie Association
DCM | 109208 | Canadian Kennel Club
DCM | 109209 | Canadian Livestock Records Association
DCM | 109210 | Canine Federation of Canada
DCM | 109211 | Continental Kennel Club
DCM | 109212 | Dog Registry of America
DCM | 109213 | Federation of International Canines
DCM | 109214 | International Progressive Dog Breeders' Alliance
DCM | 109215 | National Kennel Club
DCM | 109216 | North American Purebred Dog Registry
DCM | 109217 | United All Breed Registry
DCM | 109218 | United Kennel Club
DCM | 109219 | Universal Kennel Club International
DCM | 109220 | Working Canine Association of Canada
DCM | 109221 | World Kennel Club
DCM | 109222 | World Wide Kennel Club

### Note

The contents of this table were derived from the information available at http://www.canadasguidetodogs.com/breederinfo/breedregistries.htm.

## CID 7482 DX Anatomy Imaged for Animals

Type:

  Extensible

Version:

  20090717

Table CID 7482. DX Anatomy Imaged for Animals

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 7483 “Common Anatomic Regions for Animals”

## CID 7483 Common Anatomic Regions for Animals

Type:

  Extensible

Version:

  20130617

Table CID 7483. Common Anatomic Regions for Animals

Coding Scheme Designator | Code Value | Code Meaning | Notes
--- | --- | --- | ---
SRT | T-D4000 | Abdomen |
SRT | T-D8030 | All legs |
SRT | T-1531B | Atlantal-axial joint |
SRT | T-15311 | Atlanto-occipital joint |
SRT | T-74000 | Bladder |
SRT | T-12771 | Calcaneal tubercle | See Note 1.
SRT | T-D0788 | Carpus | See Note 2.
SRT | T-11501 | Cervical spine |
SRT | T-D00F7 | Cervico-thoracic spine |
SRT | T-D3000 | Chest |
SRT | R-FAB55 | Chest and Abdomen |
SRT | T-11B02 | Coccygeal vertrebrae | See Note 3.
SRT | T-59300 | Colon |
SRT | T-D0310 | Digit |
SRT | T-110A2 | Distal phalanx |
SRT | T-15430 | Elbow joint |
SRT | T-D0010 | Entire body |
SRT | T-56000 | Esophagus |
SRT | T-12710 | Femur |
SRT | T-D8640 | Fetlock of forelimb |
SRT | T-D9540 | Fetlock of hindlimb |
SRT | T-D04F2 | Forefoot |
SRT | T-22200 | Frontal sinus |
SRT | T-D9713 | Hindfoot |
SRT | T-15710 | Hip joint |
SRT | T-12410 | Humerus |
SRT | T-11503 | Lumbar spine |
SRT | T-D00F9 | Lumbo-sacral spine |
SRT | T-11180 | Mandible |
SRT | T-54170 | Mandibular dental arch |
SRT | T-540EE | Mandibular incisor teeth |
SRT | T-54160 | Maxillary dental arch |
SRT | T-540ED | Maxillary incisor teeth |
SRT | T-1254D | Metacarpus |
SRT | T-12847 | Metatarsus |
SRT | T-22000 | Nasal sinus |
SRT | T-12450 | Navicular of forefoot | See Note 4.
SRT | T-127EC | Navicular of hindfoot | See Note 4.
SRT | T-D14AD | Orbital region |
SRT | T-D8650 | Pastern of forefoot |
SRT | T-D9550 | Pastern of hindfoot |
SRT | T-12730 | Patella |
SRT | T-D6000 | Pelvis |
SRT | T-12403 | Radius and ulna |
SRT | T-11AD0 | Sacrum |
SRT | T-D2220 | Shoulder |
SRT | T-11100 | Skull |
SRT | T-15728 | Stiffle |
SRT | T-11096 | Tarsus | See Note 5.
SRT | T-11502 | Thoracic spine |
SRT | T-D00F8 | Thoraco-lumbar spine |
SRT | T-12701 | Tibia and fibula |
SRT | T-50110 | Upper gastro-intestinal tract |
SRT | T-75000 | Urethra |
SRT | T-70000 | Urinary tract |
SRT | T-D8040 | Wing |

### Note

1. T-12771 is used in preference to (T-12770, SRT, "Calcaneus").

2. T-D0788 is used in preference to carpal joint.

3. T-11B02 is used in preference to (T-11BF0, SRT, "coccyx") as used for humans, since the animal possess a tail.

4. T-127EC assumes correspondence between equine hindfoot and human navicular, and T-12450 the equine forefoot navicular and human scaphoid (distal sesamoid).

5. T-11096 is used for the hock joint.

6. In a prior version of this table, the code T-D8300 was used for T-15430. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 7484 DX View for Animals

Type:

  Extensible

Version:

  20090717

Table CID 7484. DX View for Animals

Coding Scheme Designator | Code Value | Code Meaning | Equivalent per Smallwood et al (see Note 1) | View Position (0018,5101) (see Note 2)
--- | --- | --- | --- | ---
SRT | R-40AC9 | Caudodistal-cranioproximal oblique | CdDi-CrPrO | CDDI_CRPRO
DCM | 123019 | Caudal 10 degree distal-cranioproximal oblique | Cd10Di-CrPrO | CD10DI_CRPRO
SRT | R-10244 | Caudocranial | CdCr | CDCR
SRT | R-40AAC | Dorso-ventral | DV | DV
SRT | R-40AE8 | Dorsolateral-palmaromedial oblique | DL-PaMO | DL_PAMO
SRT | R-40AFC | Dorsal 35 degree lateral-palmaromedial oblique | D35L-PaMO | D35L_PAMO
SRT | R-40AC2 | Dorsal 45 degree lateral-palmaromedial oblique | D40L-PaMO | D40L_PAMO
SRT | R-40AE1 | Dorsal 60 degree lateral-palmaromedial oblique | D60L-PaMO | D60L_PAMO
SRT | R-40ACF | Dorsolateral-plantaromedial oblique | DL-PlMO | DL_PLMO
SRT | R-40ACB | Dorsal 35 degree lateral-plantaromedial oblique | D35L-PlMO | D35L_PLMO
SRT | R-40AB6 | Dorsal 40 degree lateral-plantaromedial oblique | D40L-PlMO | D40L_PLMO
SRT | R-40AE4 | Dorsal 45 degree lateral-plantaromedial oblique | D45L-PlMO | D45L_PLMO
SRT | R-40AC6 | Dorsal 60 degree lateral-plantaromedial oblique | D60L-PlMO | D60L_PLMO
SRT | R-40AF2 | Dorsomedial-palmarolateral | DM-PaLO | DM_PALO
SRT | R-40AB5 | Dorsal 35 degree medial-palmarolateral oblique | D35M-PaLO | D35M_PALO
SRT | R-40AD2 | Dorsal 40 degree medial-palmarolateral oblique | D40M-PaLO | D40M_PALO
SRT | R-40AD4 | Dorsal 45 degree medial-palmarolateral | D45M-PaLO | D45M_PALO
SRT | R-40AC7 | Dorsal 60 degree medial-palmarolateral oblique | D60M-PaLO | D60M_PALO
SRT | R-40AD0 | Dorsomedial-plantarolateral oblique | DM-PlLO | DM_PLLO
SRT | R-40ACD | Dorsal 35 degree medial-plantarolateral oblique | D35M-PlLO | D35M_PLLO
SRT | R-40AD3 | Dorsal 40 degree medial-plantarolateral oblique | D40M-PlLO | D40M_PLLO
SRT | R-40AC5 | Dorsal 45 degree medial-plantarolateral oblique | D45M-PlLO | D45M_PLLO
SRT | R-40AE3 | Dorsal 60 degree medial-plantarolateral oblique | D60M-PlLO | D60M_PLLO
SRT | R-40AA9 | Dorsopalmar | DPa | DPA
SRT | R-102C4 | Dorsoplantar | DPl | DPL
SRT | R-40AFA | Dorsoproximal-palmarodistal oblique | DPr-PaDiO | DPR_PADIO
SRT | R-40ACE | Dorsal 65 degree proximal-palmarodistal oblique | D65Pr-PaDiO | D65PR_PADIO
SRT | R-40ABD | Dorsoproximal-plantarodistal oblique | DPr-PlDiO | DPR_PLDIO
SRT | R-40AD5 | Dorsal 65 degree proximal-plantarodistal oblique | D65Pr-PlDiO | D65PR_PLDIO
SRT | R-40AEA | Dorsorostral-ventrocaudal oblique | DR-VcdO | DR_VCDO
SRT | R-40AFB | Dorsal 20 degree rostral-ventrocaudal oblique | D20R-VcdO | D20R_VCDO
SRT | R-40ADB | Laterodorsoproximal-mediopalmarodistal oblique | LDPr-MpaDiO | LDPR_MPADIO
SRT | R-40AB4 | Lateral 45 deg dorsal 50 deg proximal-mediopalmarodistal oblique | L45D50Pr-MpaDiO | L45D50PR_MPADIO
SRT | R-40ADC | Laterodorsoproximal-medioplantarodistal oblique | LDPr-MplDiO | LDPR_MPLDIO
SRT | R-40AEC | Lateral 45 deg dorsal 50 deg proximal-medioplantarodistal obliq | L45D50Pr-MplDiO | L45D50PR_MPLDIO
SRT | R-10228 | Lateromedial | LM | LM
SRT | R-40AE0 | Left caudal-right rostral oblique | LeCd-RtRO | LECD_RTRO
SRT | R-40AC1 | Left 30 degree caudal-right rostral oblique | Le30Cd-RtRO | LE30CD_RTRO
SRT | R-40AE5 | Left dorsal-right ventral oblique | LeD-RtVO | LED_RTVO
SRT | R-40AFE | Left 20 degree dorsal-right ventral oblique | Le20D-RtVO | LE20D_RTVO
SRT | R-40AC3 | Left 45 degree dorsal-right ventral oblique | Le45D-RtVO | LE45D_RTVO
SRT | R-40AE6 | Left rostral-right caudal oblique | LeR-RtCdO | LER_RTCDO
SRT | R-40ADD | Left 20 degree rostral-right caudal oblique | Le20R-RtCdO | LE20R_RTCDO
SRT | R-40AF5 | Left ventral-right dorsal oblique | LeV-RtDO | LEV_RTDO
SRT | R-40ADE | Left 20 degree ventral-right dorsal oblique | Le20V-RtDO | LE20V_RTDO
SRT | R-40AC4 | Left 45 degree ventral-right dorsal oblique | Le45V-RtDO | LE45V_RTDO
SRT | R-10232 | Left-right lateral | LeRtL | LERTL
SRT | R-10224 | Mediolateral | ML | ML
SRT | R-40AF8 | Palmaromedial-dorsolateral | PaM-DLO | PAM_DLO
SRT | R-40AF6 | Palmar 45 degree medial-dorsolateral | Pa45M-DLO | PA45M_DLO
SRT | R-40AEE | Palmaroproximal-dorsodistal oblique | PaPr-DdiO | PAPR_DDIO
SRT | R-40ABC | Palmar 75 degree proximal-dorsodistal oblique | Pa75Pr-DdiO | PA75PR_DDIO
SRT | R-40AE9 | Plantarolateral-dorsomedial oblique | PlL-DMO | PLL_DMO
SRT | R-40AEF | Plantar 60 degree lateral-dorsomedial oblique | Pl60L-DMO | PL60L_DMO
SRT | R-40AD6 | Plantaroproximal-dorsodistal oblique | PlPr-DdiO | PLPR_DDIO
SRT | R-40AC8 | Plantar 75 degree proximal-dorsodistal oblique | Pl75Pr-DdiO | PL75PR_DDIO
SRT | R-40AD7 | Proximo-distal | PrDi | PRDI
SRT | R-40ADA | Right caudal-left rostral oblique | RtCd-LeRO | RTCD_LERO
SRT | R-40ACA | Right 30 degree caudal-left rostral oblique | Rt30Cd-LeRO | RT30CD_LERO
SRT | R-40ACC | Right dorsal-left ventral oblique | RtD-LeVO | RTD_LEVO
SRT | R-40AD8 | Right 20 degree dorsal-left ventral oblique | Rt20D-LeVO | RT20D_LEVO
SRT | R-40AEB | Right 45 degree dorsal-left ventral oblique | Rt45D-LeVO | RT45D_LEVO
SRT | R-40AFD | Right rostral-left caudal oblique | RtR-LeCdO | RTR_LECDO
SRT | R-40AF9 | Right 20 degree rostral-left caudal oblique | Rt20R-LeCdO | RT20R_LECDO
SRT | R-40AC0 | Right ventral-left dorsal oblique | RtV-LeDO | RTV_LEDO
SRT | R-40AD1 | Right 20 degree ventral-left dorsal oblique | Rt20V-LeDO | RT20V_LEDO
SRT | R-40AD9 | Right 45 degree ventral-left dorsal oblique | Rt45V-LeDO | RT45V_LEDO
SRT | R-10236 | Right-left lateral | RtLeL | RTLEL
SRT | R-40AF0 | Rostrocaudal | RCd | RCD
SRT | R-40ADF | Rostrodorsal-caudoventral oblique | RD-CdVO | RD_CDVO
SRT | R-40AF3 | Rostral 20 degree dorsal-caudoventral oblique | R20D-CdVO | R20D_CDVO
SRT | R-40AB7 | Rostroventral-caudodorsal | RV-CdDO | RV_CDDO
SRT | R-40AB9 | Rostral 30 degree ventral-caudodorsal | R30V-CdDO | R30V_CDDO
SRT | R-40ABB | Ventral left-dorsal right oblique | VLe-DrtO | VLE_DRTO
SRT | R-40ABA | Ventral 30 degree left-dorsal right oblique | V30Le-DrtO | V30LE_DRTO
SRT | R-40AF4 | Ventral right-dorsal left oblique | VRt-DleO | VRT_DLEO
SRT | R-40AB8 | Ventral 30 degree right-dorsal left oblique | V30Rt-DleO | V30RT_DLEO
SRT | R-40AB0 | Ventro-dorsal | VD | VD
SRT | R-40AF7 | Ventrorostral-dorsocaudal oblique | VR-DCdO | VR_DCDO
SRT | R-40AF1 | Ventral 20 degree rostral-dorsocaudal oblique | V20R-DCdO | V20R_DCDO

### Note

1. The Smallwood et al equivalent may also be used as the Code Meaning (0008,0104) (i.e., as a synonym) in place of the full text described here, and as the value for View Name (0008,2127), if sent.

2. The Defined Terms for View Position are derived from the Smallwood et al equivalent by capitalizing and replacing hyphens with underscores.

## CID 7486 Mixed Breeds

Type:

  Extensible

Version:

  20080324

Table CID 7486. Mixed Breeds

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | L-80A74 | Mixed breed cat
SRT | L-80217 | Mixed breed goat
SRT | L-809DF | Mixed breed dog
SRT | L-8A10F | Mixed breed horse
SRT | L-8C33A | Mixed breed sheep
SRT | L-93791 | Mixed breed chicken
SRT | L-8B947 | Mixed breed cattle
SRT | L-8B103 | Mixed breed pig

## CID 8101 Container Types

Type:

  Extensible

Version:

  20080626

Table CID 8101. Container Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-0101E | Tissue cassette
SRT | A-01022 | Tissue microarray cassette
SRT | A-01024 | Specimen vial
SRT | A-0101B | Microscope slide
SRT | A-01023 | Specimen container
SRT | A-01021 | Electron microscopy grid
SRT | A-01025 | Specimen well

## CID 8102 Container Component Types

Type:

  Extensible

Version:

  20080626

Table CID 8102. Container Component Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 8101 “Container Types”
SRT | A-0101D | Microscope slide cover slip
SRT | F-62219 | Microscope slide mounting media
SRT | A-0101F | Specimen container lid

## CID 8103 Anatomic Pathology Specimen Types

Type:

  Extensible

Version:

  20080626

Table CID 8103. Anatomic Pathology Specimen Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-D0010 | Entire body
SRT | G-80A5 | Body substance sample
SRT | G-80A6 | Body fluid sample
SRT | G-8300 | Tissue specimen
SRT | G-843A | Gross specimen
SRT | G-8439 | Tissue section
SRT | G-843B | Core sample of tissue block
SRT | G-843C | Tissue spot
SRT | G-81EA | Slide
SRT | G-803C | Smear sample
SRT | T-1A404 | Touch preparation cytologic material
SRT | T-1A403 | Liquid based cytologic material
SRT | G-8003 | Aspirate
SRT | G-81A0 | Genetic sample
Include CID 8104 “Breast Tissue Specimen Types”

## CID 8104 Breast Tissue Specimen Types

Type:

  Extensible

Version:

  20080626

Table CID 8104. Breast Tissue Specimen Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-8346 | breast duct sample
SRT | G-8339 | frozen section breast sample
SRT | G-833D | lumpectomy breast sample
SRT | G-8430 | specimen from breast obtained by excision
SRT | G-8311 | specimen from breast obtained by total mastectomy
SRT | G-833F | segmentectomy breast sample
SRT | G-832D | breast tru-cut biopsy sample
SRT | G-8318 | specimen from breast obtained by core needle biopsy
SRT | G-8319 | specimen from breast, stereotactically guided core needle biopsy
SRT | G-831B | specimen from breast by incisional biopsy of breast mass
SRT | R-003AC | specimen from breast obtained by image guided core biopsy

## CID 8109 Specimen Collection Procedure

Type:

  Extensible

Version:

  20080626

Table CID 8109. Specimen Collection Procedure

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P1-03130 | Aspiration
SRT | P1-03100 | Biopsy
SRT | P1-03000 | Excision
SRT | P1-03000 | Resection
SRT | P1-0D300 | Harvesting of tissue
SRT | P1-03021 | Removal of device
SRT | P1-38200 | Venipuncture
SRT | P0-00593 | Taking of swab
SRT | P3-02000 | Specimen collection
SRT | P1-03154 | Scraping

## CID 8110 Specimen Sampling Procedure

Type:

  Extensible

Version:

  20080626

Table CID 8110. Specimen Sampling Procedure

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P1-01003 | Dissection
DCM | 111726 | Dissection with entire specimen submission
DCM | 111727 | Dissection with representative sections submission
SRT | P3-40011 | Core sampling
SRT | P3-4000D | Block sectioning
SRT | P3-40004 | Laser microdissection
SRT | P3-4000E | Block surface recut
SRT | P3-4000F | Block step sectioning
SRT | P3-4500A | Touch preparation (procedure)
SRT | P3-00048 | Smear procedure

## CID 8111 Specimen Preparation Procedure

Type:

  Extensible

Version:

  20080626

Table CID 8111. Specimen Preparation Procedure

Coding Scheme Designator | Code Value | Code Meaning | HL7 v3 ActClass equivalent
--- | --- | --- | ---
SRT | P3-02000 | Specimen collection | SPECCOLLECT
SRT | P3-05013 | Specimen receiving | CONTREG
SRT | P3-4000A | Sampling of tissue specimen | PROC
SRT | P3-00003 | Staining | SPCTRT
SRT | P3-05000 | Specimen processing | SPCTRT
DCM | 111729 | Specimen storage | STORE

## CID 8112 Specimen Stains

Type:

  Extensible

Version:

  20080626

Table CID 8112. Specimen Stains

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-22860 | acid fast stain
SRT | C-2280A | acid phosphatase stain
SRT | C-2280B | Albert's stain
SRT | C-22963 | alcian blue 8GX stain
SRT | C-2288B | alcian blue stain
SRT | C-22932 | alcohol soluble nigrosine stain
SRT | C-2286D | aldehyde fuchsin stain
SRT | C-22961 | alizarin blue S stain
SRT | C-22959 | alizarin cyanine green stain
SRT | C-22953 | alizarin red S stain
SRT | C-22813 | alizarin yellow GG stain
SRT | C-22814 | alizarin yellow R stain
SRT | C-2285B | alkaline phosphatase stain
SRT | C-2287E | aniline blue stain
SRT | C-2280C | auramine stain
SRT | C-22873 | azo black stain
SRT | C-22929 | azocarmine G (GX) stain
SRT | C-22842 | azophloxin stain
SRT | C-22831 | azorubin S stain
SRT | C-22945 | azure A stain
SRT | C-22946 | azure B stain
SRT | C-22944 | azure C stain
SRT | C-2286E | bauer's chromic acid leucofuchsin stain
SRT | C-22872 | benzo fast scarlet stain
SRT | C-2280D | beta-glucuronidase stain
SRT | C-22866 | biebrich scarlet stain
SRT | C-22848 | bismark brown R stain
SRT | C-22848 | bismark brown Y stain
SRT | C-22921 | blue shade eosin stain
SRT | C-22965 | brazilin stain
SRT | C-22934 | brilliant cresyl blue stain
SRT | C-22869 | brilliant crocein stain
SRT | C-22865 | brilliant orange stain
SRT | C-22857 | brilliant yellow stain
SRT | C-2283C | butyrate esterase stain
SRT | C-2286B | carbol fuchsin stain
SRT | C-22971 | carmine stain
SRT | C-22972 | carminic acid stain
SRT | C-22822 | carmoisine A stain
SRT | C-22936 | celestine blue B stain
SRT | C-2280E | chloroacetate esterase stain
SRT | C-2287B | chromic acid stain
SRT | C-22838 | chromotrope 2R stain
SRT | C-22806 | chrysoidine R stain
SRT | C-22805 | chrysoidine Y stain
SRT | C-22973 | cochineal stain
SRT | C-22837 | colloidal iron stain
SRT | C-22851 | Congo red stain
SRT | C-22847 | cresyl echt violet stain
SRT | C-22840 | cresyl violet stain
SRT | C-22833 | crystal ponceau stain
SRT | C-2283D | crystal violet stain
SRT | C-22966 | curcumin stain
SRT | C-22826 | diamond black stain
SRT | C-22871 | durazol red stain
SRT | C-22852 | erie garnet stain
SRT | C-22839 | eriochrome blue black SE stain
SRT | C-22924 | erythrosin B stain
SRT | C-22923 | erythrosin Y stain
SRT | C-22854 | Evans blue stain
SRT | C-22883 | fast blue B salt stain
SRT | C-22881 | fast blue BB salt stain
SRT | C-22878 | fast blue RR salt stain
SRT | C-22882 | fast garnet GBC salt stain
SRT | C-22886 | fast green FCF stain
SRT | C-22876 | fast red B salt stain
SRT | C-22877 | fast red ITR stain
SRT | C-22875 | fast red TR salt stain
SRT | C-22867 | fast sulfon black F stain
SRT | C-22879 | fast violet B salt stain
SRT | C-22859 | fat red 7B stain
SRT | C-2280F | Feulgen reaction stain
SRT | C-22810 | field's stain
SRT | C-22816 | Flagellar stain
SRT | C-22A00 | fluorescent stain
SRT | C-2286C | fouchet stain
SRT | C-22902 | fuchsin acid stain
SRT | C-22889 | fuchsin basic stain
SRT | C-22935 | gallocyanine stain
SRT | F-61968 | giemsa stain
SRT | C-22830 | gram stain
SRT | C-2286F | hansel stain
SRT | C-22967 | hematein stain
SRT | C-22968 | hematoxylin stain
SRT | C-22817 | immunofluorescent stain
SRT | C-2285C | India ink stain
SRT | C-22962 | indigo carmine stain
SRT | C-22927 | indophenol from naphthol stain
SRT | C-22974 | insoluble berlin blue stain
SRT | C-22804 | janus green B stain
SRT | C-22818 | Jenner-Giemsa stain
SRT | C-22899 | kenacid blue R stain
SRT | C-22942 | lacmoid stain
SRT | C-22819 | Leishman stain
SRT | C-22887 | light green SF stain
SRT | C-22841 | lissamine fast red B stain
SRT | C-22843 | lissamine fast yellow stain
SRT | C-22914 | lissamine green B stain
SRT | C-22917 | lissamine rhodamine stain
SRT | C-2283F | luxol fast blue stain
SRT | C-22890 | malachite green stain
SRT | C-2283A | Mallory bleach stain
SRT | C-22802 | martius yellow stain
SRT | C-2281A | may-Grunwald giemsa stain
SRT | C-22937 | meldola blue stain
SRT | C-22811 | metanil yellow stain
SRT | C-22907 | methyl blue stain
SRT | C-2281B | methyl green pyronin stain
SRT | C-22809 | methyl orange stain
SRT | C-22808 | methyl red stain
SRT | C-2287C | methyl violet stain
SRT | C-22947 | methylene blue stain
SRT | C-2284A | methylene violet stain
SRT | C-22952 | methylene violet stain (Bernthsen)
SRT | C-2287F | modified trichrome stain
SRT | C-2284B | mucicarmine stain
SRT | C-2281C | myeloperoxidase stain
SRT | C-22846 | naphthalene black 12B stain
SRT | C-22801 | naphthol green B stain
SRT | C-22803 | naphthol yellow S stain
SRT | C-2285D | naphthol-AS-D-chloracetate esterase stain
SRT | C-22928 | neutral red stain
SRT | C-2281D | neutrophil alkaline phosphatase stain
SRT | C-22891 | new fuchsin stain
SRT | C-2284C | night blue stain
SRT | C-22941 | nile blue stain
SRT | C-22823 | nitrazine yellow stain
SRT | C-2281E | nonspecific esterase stain
SRT | C-22955 | nuclear fast red stain
SRT | C-22863 | oil red O stain
SRT | C-22832 | orange G stain
SRT | C-22824 | orange II stain
SRT | C-2284D | orcein stain
SRT | C-22901 | page blue 83 stain
SRT | C-22989 | page blue G-90 stain
SRT | C-22885 | patent blue V sodium salt stain
SRT | C-2281F | periodic acid Schiff stain
SRT | R-F748A | permethrin stain
SRT | C-2285E | peroxidase stain
SRT | C-22922 | phloxin B stain
SRT | C-2284E | phosphotungstic acid-hematoxylin stain
SRT | C-22829 | ponceau 3R stain
SRT | C-22868 | ponceau S stain
SRT | C-22828 | ponceau xylidine stain
SRT | C-22855 | pontamine sky blue 5BX stain
SRT | C-22856 | pontamine sky blue 6BX stain
SRT | C-22870 | potassium hydroxide stain
SRT | C-22956 | procion brilliant blue MRS stain
SRT | C-2288A | protargol S stain
SRT | C-22820 | Prussian blue stain
SRT | C-2284F | quinacrine fluorescent stain
SRT | C-2286A | rhodamine stain
SRT | C-2282A | Romanowsky stain
SRT | C-22925 | rose bengal stain
SRT | C-22908 | rosolic acid sodium salt stain
SRT | C-22964 | saffron stain
SRT | F-61DA5 | safranin stain
SRT | C-22931 | safranine O stain
SRT | C-2287A | silver nitrate stain
SRT | C-22836 | silver stain
SRT | C-22874 | sirius red F3B stain
SRT | C-22912 | solochrome azurine (BS) stain
SRT | C-22821 | solochrome black 6B stain
SRT | C-22909 | solochrome cyanine R stain
SRT | C-22825 | solochrome dark blue stain
SRT | C-22975 | soluble berlin blue stain
SRT | C-22906 | spirit soluble aniline blue stain
SRT | C-22920 | spirit soluble eosin stain
SRT | C-2282B | spore stain
SRT | C-2282D | Sudan stain
SRT | C-22827 | sunset yellow FCF stain
SRT | C-2285A | tartrate resistant acid phosphatase
SRT | C-22844 | tartrazine stain
SRT | C-2285F | terminal deoxynucleotidyl transferase stain
SRT | C-2288D | thioflavine S stain
SRT | C-22926 | thioflavine T stain
SRT | C-22850 | thionin stain
SRT | C-22943 | thionine stain
SRT | C-22845 | titan yellow stain
SRT | C-2287D | trichrome stain
SRT | C-22815 | tropaeolin O stain
SRT | C-22812 | tropaeolin OO stain
SRT | C-22853 | trypan blue stain
SRT | C-2283E | Van Gieson stain
SRT | C-22880 | verhoeff's hematoxylin stain
SRT | C-22858 | vital new red stain
SRT | C-22904 | water soluble aniline blue stain
SRT | C-22954 | water soluble anthracene brown stain
SRT | C-22933 | water soluble nigrosine stain
SRT | C-22957 | waxoline blue stain
SRT | F-61E5A | wayson stain
SRT | F-619B7 | wright stain
SRT | C-22888 | xylene cyanol FF stain
SRT | C-2282C | Ziehl-Neelsen stain
SRT | C-22A08 | acridine orange stain
SRT | C-22A07 | acriflavine stain
SRT | C-22A03 | atebrin FS stain
SRT | C-22A02 | auramine G stain
SRT | C-22A01 | auramine O stain
SRT | C-22A11 | coriphosphine stain
SRT | C-22A05 | fluorescein stain
SRT | C-22AA1 | fluorexon stain
SRT | C-22A04 | rhodamine B stain
SRT | C-22A06 | Fluorescein sodium stain
SRT | C-22864 | Sudan black B stain
SRT | C-2282E | Sudan black stain
SRT | C-22958 | Sudan blue stain
SRT | C-22807 | Sudan II stain
SRT | C-22861 | Sudan III stain
SRT | C-22862 | Sudan IV stain
SRT | C-22903 | alkali blue 5B (4B) stain
SRT | C-22905 | alkali blue 6B stain
SRT | C-22911 | chrome azurol S stain
SRT | C-22918 | dibromofluorescein stain
SRT | C-22897 | ethyl violet stain
SRT | C-22896 | methyl green stain
SRT | C-22892 | methyl violet 2B stain
SRT | C-22894 | methyl violet 6B stain
SRT | C-22916 | pyronine B stain
SRT | C-22915 | pyronine G stain
SRT | C-22951 | toluidine blue stain
SRT | C-22895 | victoria blue 4R stain
SRT | C-22913 | victoria blue B stain
SRT | C-22919 | water soluble eosin stain

## CID 8113 Specimen Preparation Steps

Type:

  Extensible

Version:

  20080626

Table CID 8113. Specimen Preparation Steps

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P3-40005 | Specimen microwave heating
SRT | P3-40009 | Specimen steam heating
SRT | P3-40006 | Protease digestion of tissue specimen
SRT | P3-4000B | Specimen dehydration
SRT | P3-05050 | Specimen freezing
SRT | P3-40003 | Specimen clearing

## CID 8114 Specimen Fixatives

Type:

  Extensible

Version:

  20080626

Table CID 8114. Specimen Fixatives

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-2141C | Neutral Buffered Formalin
SRT | F-62235 | Bouin's fluid
SRT | C-2141B | Formalin aqueous solution of formaldehyde
SRT | F-62231 | Carnoy's fluid
SRT | F-62238 | Formol sublimate
SRT | F-62233 | Helly's fluid
SRT | F-6220F | Michel's medium
SRT | F-62234 | Zenker's fluid
SRT | C-21403 | Paraformaldehyde
SRT | C-21624 | Acetic acid
SRT | C-20830 | Chloroform
SRT | C-12916 | Chromium trioxide
SRT | C-21047 | Ethanol
SRT | C-21402 | Formaldehyde
SRT | C-13321 | Mercuric chloride
SRT | C-2102B | Methanol
SRT | C-15211 | Osmium tetroxide
SRT | C-21919 | Picric acid
SRT | C-13518 | Potassium dichromate

## CID 8115 Specimen Embedding Media

Type:

  Extensible

Version:

  20080626

Table CID 8115. Specimen Embedding Media

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-616D8 | Paraffin wax
SRT | F-62232 | Tissue freezing medium
SRT | C-2A000 | Plastic
SRT | C-84085 | Agar
SRT | C-2A400 | Epoxy resin
SRT | C-100EA | Acrylic resin

## CID 8120 WSI Referenced Image Purposes of Reference

Type:

  Extensible

Version:

  20100824

Table CID 8120. WSI Referenced Image Purposes of Reference

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121311 | Localizer
DCM | 121350 | Same acquisition at lower resolution
DCM | 121351 | Same acquisition at higher resolution
DCM | 121352 | Same acquisition at different focal depth
DCM | 121353 | Same acquisition at different spectral band
DCM | 121354 | Imaged container label

## CID 8121 Microscopy Lens Type

Type:

  Extensible

Version:

  20100824

Table CID 8121. Microscopy Lens Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-0011A | High power non-immersion lens
SRT | A-0011B | Oil immersion lens
SRT | A-00118 | Slide overview lens

## CID 8122 Microscopy Illuminator and Sensor Color

Type:

  Extensible

Version:

  20100824

Table CID 8122. Microscopy Illuminator and Sensor Color

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-102C0 | Full Spectrum
SRT | R-102BE | Infrared
SRT | G-A11A | Red
SRT | G-A11E | Green
SRT | G-A12F | Blue
SRT | R-102BF | Ultraviolet

## CID 8123 Microscopy Illumination Method

Type:

  Extensible

Version:

  20100824

Table CID 8123. Microscopy Illumination Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111741 | Transmission illumination
DCM | 111742 | Reflection illumination
DCM | 111743 | Epifluorescence illumination
DCM | 111744 | Brightfield illumination
DCM | 111745 | Darkfield illumination
DCM | 111746 | Oblique illumination
DCM | 111747 | Phase contrast illumination
DCM | 111748 | Differential interference contrast
DCM | 111749 | Total internal reflection fluorescence

## CID 8124 Microscopy Filter

Type:

  Extensible

Version:

  20100824

Table CID 8124. Microscopy Filter

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-010E2 | Green optical filter
SRT | A-010DF | Red optical filter
SRT | A-010DA | Blue optical filter
SRT | A-010DC | Infrared optical filter
SRT | A-010E1 | Polarizing optical filter
SRT | A-010DE | Violet optical filter
SRT | A-010DD | Ultraviolet optical filter
SRT | A-0010F | Dichroic beamsplitter
SRT | A-00121 | Hoffman modulator
SRT | A-0011D | Darkfield stop
SRT | A-0011C | Rheinberg filter
SRT | A-0011E | Phase contrast plate
SRT | A-00120 | Condenser annulus
SRT | A-0011F | Nomarski prism
SRT | A-00123 | de Sénarmont compensator
DCM | 111609 | No filter

## CID 8125 Microscopy Illuminator Type

Type:

  Extensible

Version:

  20100824

Table CID 8125. Microscopy Illuminator Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-00125 | Tungsten halogen lamp
SRT | A-00127 | Mercury arc lamp
SRT | A-00124 | Xenon arc lamp
SRT | A-00126 | Light emitting diode
SRT | A-23000 | Laser

## CID 8130 Staining Protocols

This Context Group is intended for use in the Scheduled Protocol Code Sequence (0040,0008) and the Performed Protocol Code Sequence (0040,0260) attributes for an automated slide stainer. When so used, an Item with value (P3-00003, SRT, "Staining") will also include a Protocol Context Sequence(0040,0440) using TID 8003 “Specimen Staining” Specimen Staining to identify the specific stain substance.

Type:

  Extensible

Version:

  20111028

Table CID 8130. Staining Protocols

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
SRT | P3-00003 | Staining
SRT | P3-50495 | Hematoxylin and eosin stain method

## CID 8131 Pathology Imaging Protocols

This Context Group is intended for use in the Scheduled Protocol Code Sequence (0040,0008) and the Performed Protocol Code Sequence (0040,0260).When so used, an Item with value (112703, DCM, "Whole Slide Imaging") may also include a Protocol Context Sequence(0040,0440) using TID 8010Slide Imaging Parameters.

Type:

  Extensible

Version:

  20120605

Table CID 8131. Pathology Imaging Protocols

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
DCM | 112700 | Peri-operative Photographic Imaging
DCM | 112701 | Gross Specimen Imaging
DCM | 112702 | Slide Microscopy
DCM | 112703 | Whole Slide Imaging
DCM | 112704 | WSI 20X RGB
DCM | 112705 | WSI 40X RGB

## CID 8132 Magnification Selection

Type:

  Extensible

Version:

  20120605

Table CID 8132. Magnification Selection

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
DCM | 112715 | 5X
DCM | 112716 | 10X
DCM | 112717 | 20X
DCM | 112718 | 40X

## CID 8133 Tissue Selection

Type:

  Extensible

Version:

  20120605

Table CID 8133. Tissue Selection

Coding Scheme | Code Value | Code Meaning
--- | --- | ---
DCM | 112719 | Nominal empty tile suppression
DCM | 112720 | High threshold empty tile suppression
DCM | 112721 | No empty tile suppression

## CID 8201 Surface Scan Acquisition Types

Type:

  Extensible

Version:

  20121129

Table CID 8201. Surface Scan Acquisition Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 114201 | Time of flight
DCM | 114202 | Interferometry
DCM | 114203 | Laser scanning
DCM | 114204 | Pattern projection
DCM | 114205 | Shape from shading
DCM | 114206 | Shape from motion
DCM | 114207 | Confocal imaging
DCM | 114208 | Point Cloud Algorithmic

## CID 8202 Surface Scan Mode Types

Type:

  Extensible

Version:

  20121129

Table CID 8202. Surface Scan Mode Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 114209 | Turntable Scan Method
DCM | 114210 | High resolution
DCM | 114211 | Fast mode
DCM | 114216 | Checkerboard

## CID 8203 Surface Scan Registration Method Types

Type:

  Extensible

Version:

  20121129

Table CID 8203. Surface Scan Registration Method Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 114213 | Iterative Closest Point
DCM | 125022 | Fiducial Alignment
DCM | 114215 | Freehand

## CID 8300 Visual Evaluation Methods

Type:

  Extensible

Version:

  20140331

Table CID 8300. Visual Evaluation Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109701 | Overall image quality evaluation
DCM | 109702 | Grayscale resolution evaluation
DCM | 109703 | Luminance response evaluation
DCM | 109704 | Luminance uniformity evaluation
DCM | 109705 | Chromaticity evaluation
DCM | 109706 | Pixel faults evaluation
DCM | 109707 | Veiling glare evaluation
DCM | 109708 | Geometrical image evaluation
DCM | 109709 | Angular viewing evaluation
DCM | 109710 | Clinical evaluation

## CID 8301 Test Pattern Codes

Test patterns for display calibration jobs.

Type:

  Extensible

Version:

  20140331

Table CID 8301. Test Pattern Codes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109801 | TG18-QC Pattern
DCM | 109802 | TG18-BR Pattern
DCM | 109803 | TG18-PQC Pattern
DCM | 109804 | TG18-CT Pattern
DCM | 109805 | TG18-LN8-01 Pattern
DCM | 109806 | TG18-LN8-02 Pattern
DCM | 109807 | TG18-LN8-03 Pattern
DCM | 109808 | TG18-LN8-04 Pattern
DCM | 109809 | TG18-LN8-05 Pattern
DCM | 109810 | TG18-LN8-06 Pattern
DCM | 109811 | TG18-LN8-07 Pattern
DCM | 109812 | TG18-LN8-08 Pattern
DCM | 109813 | TG18-LN8-09 Pattern
DCM | 109814 | TG18-LN8-10 Pattern
DCM | 109815 | TG18-LN8-11 Pattern
DCM | 109816 | TG18-LN8-12 Pattern
DCM | 109817 | TG18-LN8-13 Pattern
DCM | 109818 | TG18-LN8-14 Pattern
DCM | 109819 | TG18-LN8-15 Pattern
DCM | 109820 | TG18-LN8-16 Pattern
DCM | 109821 | TG18-LN8-17 Pattern
DCM | 109822 | TG18-LN8-18 Pattern
DCM | 109823 | TG18-LN12-01 Pattern
DCM | 109824 | TG18-LN12-02 Pattern
DCM | 109825 | TG18-LN12-03 Pattern
DCM | 109826 | TG18-LN12-04 Pattern
DCM | 109827 | TG18-LN12-05 Pattern
DCM | 109828 | TG18-LN12-06 Pattern
DCM | 109829 | TG18-LN12-07 Pattern
DCM | 109830 | TG18-LN12-08 Pattern
DCM | 109831 | TG18-LN12-09 Pattern
DCM | 109832 | TG18-LN12-10 Pattern
DCM | 109833 | TG18-LN12-11 Pattern
DCM | 109834 | TG18-LN12-12 Pattern
DCM | 109835 | TG18-LN12-13 Pattern
DCM | 109836 | TG18-LN12-14 Pattern
DCM | 109837 | TG18-LN12-15 Pattern
DCM | 109838 | TG18-LN12-16 Pattern
DCM | 109839 | TG18-LN12-17 Pattern
DCM | 109840 | TG18-LN12-18 Pattern
DCM | 109841 | TG18-UN10 Pattern
DCM | 109842 | TG18-UN80 Pattern
DCM | 109843 | TG18-UNL10 Pattern
DCM | 109844 | TG18-UNL80 Pattern
DCM | 109845 | TG18-AD Pattern
DCM | 109846 | TG18-MP Pattern
DCM | 109847 | TG18-RH10 Pattern
DCM | 109848 | TG18-RH50 Pattern
DCM | 109849 | TG18-RH89 Pattern
DCM | 109850 | TG18-RV10 Pattern
DCM | 109851 | TG18-RV50 Pattern
DCM | 109852 | TG18-RV89 Pattern
DCM | 109853 | TG18-PX Pattern
DCM | 109854 | TG18-CX Pattern
DCM | 109855 | TG18-LPH10 Pattern
DCM | 109856 | TG18-LPH50 Pattern
DCM | 109857 | TG18-LPH89 Pattern
DCM | 109858 | TG18-LPV10 Pattern
DCM | 109859 | TG18-LPV50 Pattern
DCM | 109860 | TG18-LPV89 Pattern
DCM | 109861 | TG18-AFC Pattern
DCM | 109862 | TG18-NS10 Pattern
DCM | 109863 | TG18-NS50 Pattern
DCM | 109864 | TG18-NS89 Pattern
DCM | 109865 | TG18-GV Pattern
DCM | 109866 | TG18-GVN Pattern
DCM | 109867 | TG18-GQ Pattern
DCM | 109868 | TG18-GQN Pattern
DCM | 109869 | TG18-GQB Pattern
DCM | 109870 | TG18-GA03 Pattern
DCM | 109871 | TG18-GA05 Pattern
DCM | 109872 | TG18-GA08 Pattern
DCM | 109873 | TG18-GA10 Pattern
DCM | 109874 | TG18-GA15 Pattern
DCM | 109875 | TG18-GA20 Pattern
DCM | 109876 | TG18-GA25 Pattern
DCM | 109877 | TG18-GA30 Pattern
DCM | 109878 | TG18-CH Image
DCM | 109879 | TG18-KN Image
DCM | 109880 | TG18-MM1 Image
DCM | 109881 | TG18-MM2 Image
DCM | 109901 | OIQ Pattern
DCM | 109902 | ANG Pattern
DCM | 109903 | GD Pattern
DCM | 109904 | BN01 Pattern
DCM | 109905 | BN02 Pattern
DCM | 109906 | BN03 Pattern
DCM | 109907 | BN04 Pattern
DCM | 109908 | BN05 Pattern
DCM | 109909 | BN06 Pattern
DCM | 109910 | BN07 Pattern
DCM | 109911 | BN08 Pattern
DCM | 109912 | BN09 Pattern
DCM | 109913 | BN10 Pattern
DCM | 109914 | BN11 Pattern
DCM | 109915 | BN12 Pattern
DCM | 109916 | BN13 Pattern
DCM | 109917 | BN14 Pattern
DCM | 109918 | BN15 Pattern
DCM | 109919 | BN16 Pattern
DCM | 109920 | BN17 Pattern
DCM | 109921 | BN18 Pattern
DCM | 109931 | DIN Geometry Pattern
DCM | 109932 | DIN Grayscale Pattern
DCM | 109933 | DIN Resolution Pattern
DCM | 109941 | White Pattern
DCM | 109943 | SMPTE Pattern

## CID 8302 Measurement Pattern Codes

Test pattern images that define measurement points for display calibration jobs.

Type:

  Extensible

Version:

  20140331

Table CID 8302. Measurement Pattern Codes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109843 | TG18-UNL10 Pattern
DCM | 109844 | TG18-UNL80 Pattern

## CID 8303 Display Device Type

The type of image display device.

Type:

  Non-Extensible

Version:

  20140331

Table CID 8303. Display Device Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 109991 | CRT Display
DCM | 109992 | Liquid Crystal Display
DCM | 109993 | Plasma Display
DCM | 109994 | OLED
DCM | 109995 | DLP Rear Projection System
DCM | 109996 | DLP Front Projection System
DCM | 109997 | CRT Rear Projection System
DCM | 109998 | CRT Front Projection System
DCM | 109999 | Other Projection System

## CID 9000 Physical Quantity Descriptors

Type:

  Extensible

Version:

  20141110

Table CID 9000. Physical Unit Descriptors

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-C1C6 | Quantity
DCM | 121401 | Derivation
SRT | G-C036 | Measurement Method

### Note

The concept (G-C1C6, SRT, "Quantity"), lacking a formal definition in SNOMED, is assumed in this usage to be synonymous with the concept defined for "quantity" in Joint Committee for Guides in Metrology (JCGM), International Vocabulary of Metrology, Basic and General Concepts and Associated Terms (http://www.bipm.org/utils/common/documents/jcgm/JCGM_200_2012.pdf); the definition is "property of a phenomenon, body, or substance, where the property has a magnitude that can be expressed as a number and a reference". That document further distinguishes a "physical quantity", "chemical quantity", and "biological quantity", though no such distinction is implied here, and "quantity" is assumed to be all inclusive.

## CID 9231 Workitem Definition

Type:

  Extensible

Version:

  20020904

Table CID 9231. Workitem Definition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110001 | Image Processing
DCM | 110002 | Quality Control
DCM | 110003 | Computer Aided Diagnosis
DCM | 110004 | Computer Aided Detection
DCM | 110005 | Interpretation
DCM | 110006 | Transcription
DCM | 110007 | Report Verification
DCM | 110008 | Print
DCM | 110009 | No subsequent Workitems
DCM | 110013 | Media Import

## CID 9232 Non-DICOM Output Types (Retired)

See PS3.16-2011.

## CID 9241 Radiotherapy General Workitem Definition

Type:

  Extensible

Version:

  20130408

Table CID 9241. Radiotherapy General Workitem Definition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121701 | RT Patient Setup
DCM | 121722 | RT Patient Position Adjustment
DCM | 121723 | RT Patient Position In-treatment-session Review
DCM | 121724 | RT Treatment Simulation with Internal Verification
DCM | 121725 | RT Treatment Simulation with External Verification
DCM | 121726 | RT Treatment with Internal Verification
DCM | 121727 | RT Treatment with External Verification
DCM | 121728 | RT Treatment QA with Internal Verification
DCM | 121729 | RT Treatment QA with External Verification
DCM | 121730 | RT Machine QA
DCM | 121731 | RT Treatment QA by RT Plan Dose Check
DCM | 121732 | RT Treatment QA by RT Plan Difference Check
DCM | 121733 | RT Treatment QA by RT Ion Plan Dose Check
DCM | 121734 | RT Treatment QA by RT Ion Plan Difference Check

## CID 9242 Radiotherapy Acquisition Workitem Definition

Type:

  Extensible

Version:

  20110406

Table CID 9242. Radiotherapy Acquisition Workitem Definition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121702 | RT Patient Position Acquisition, single plane MV
DCM | 121703 | RT Patient Position Acquisition, dual plane MV
DCM | 121704 | RT Patient Position Acquisition, single plane kV
DCM | 121705 | RT Patient Position Acquisition, dual plane kV
DCM | 121706 | RT Patient Position Acquisition, dual plane kV/MV
DCM | 121707 | RT Patient Position Acquisition, CT kV
DCM | 121708 | RT Patient Position Acquisition, CT MV
DCM | 121709 | RT Patient Position Acquisition, Optical
DCM | 121710 | RT Patient Position Acquisition, Ultrasound
DCM | 121711 | RT Patient Position Acquisition, Spatial Fiducials

## CID 9243 Radiotherapy Registration Workitem Definition

Type:

  Extensible

Version:

  20110406

Table CID 9243. Radiotherapy Registration Workitem Definition

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121712 | RT Patient Position Registration, single plane
DCM | 121713 | RT Patient Position Registration, dual plane
DCM | 121714 | RT Patient Position Registration, 3D CT general
DCM | 121715 | RT Patient Position Registration, 3D CT marker-based
DCM | 121716 | RT Patient Position Registration, 3D CT volume-based
DCM | 121717 | RT Patient Position Registration, 3D on 2D reference
DCM | 121718 | RT Patient Position Registration, 2D on 3D reference
DCM | 121719 | RT Patient Position Registration, Optical
DCM | 121720 | RT Patient Position Registration, Ultrasound
DCM | 121721 | RT Patient Position Registration, Spatial Fiducials

## CID 9250 Scheduled Processing Parameter Concept Codes for RT Treatment

Type:

  Extensible

Version:

  20140402

Table CID 9250. Scheduled Processing Parameter Concept Codes for RT Treatment

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121712 | Treatment Delivery Type

## CID 9300 Procedure Discontinuation Reasons

Type:

  Extensible

Version:

  20110128

Table CID 9300. Procedure Discontinuation Reasons

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110526 | Resource pre-empted
DCM | 110527 | Resource inadequate
DCM | 110528 | Discontinued Procedure Step rescheduled
DCM | 110529 | Discontinued Procedure Step rescheduling recommended
Include CID 9301 “Modality PPS Discontinuation Reasons”
Include CID 9302 “Media Import PPS Discontinuation Reasons”

## CID 9301 Modality PPS Discontinuation Reasons

Type:

  Extensible

Version:

  20140419

Table CID 9301. Modality PPS Discontinuation Reasons

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110500 | Doctor canceled procedure
DCM | 110501 | Equipment failure
DCM | 110502 | Incorrect procedure ordered
DCM | 110503 | Patient allergic to media/contrast
DCM | 110504 | Patient died
DCM | 110505 | Patient refused to continue procedure
DCM | 110506 | Patient taken for treatment or surgery
DCM | 110507 | Patient did not arrive
DCM | 110508 | Patient pregnant
DCM | 110509 | Change of procedure for correct charging
DCM | 110510 | Duplicate order
DCM | 110511 | Nursing unit cancel
DCM | 110512 | Incorrect side ordered
DCM | 110513 | Discontinued for unspecified reason
DCM | 110514 | Incorrect worklist entry selected
DCM | 110515 | Patient condition prevented continuing
DCM | 110516 | Equipment change
SRT | D0-B0330 | Injection Site Extravasation
SRT | DF-10780 | Radiopharmaceutical Adverse Reaction

## CID 9302 Media Import PPS Discontinuation Reasons

Type:

  Extensible

Version:

  20090616

Table CID 9302. Media Import PPS Discontinuation Reasons

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110521 | Objects incorrectly formatted
DCM | 110522 | Object Types not supported
DCM | 110523 | Object Set incomplete
DCM | 110524 | Media Failure
DCM | 110501 | Equipment failure
DCM | 110510 | Duplicate order
DCM | 110513 | Discontinued for unspecified reason
DCM | 110514 | Incorrect worklist entry selected

## CID 10000 Scope of Accumulation

Type:

  Extensible

Version:

  20131010

Table CID 10000. Scope of Accumulation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113014 | Study
DCM | 113015 | Series
DCM | 113016 | Performed Procedure Step
DCM | 113970 | Procedure Step To This Point
DCM | 113852 | Irradiation Event

## CID 10001 UID Types

Type:

  Extensible

Version:

  20081028

Table CID 10001. UID Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 110180 | Study Instance UID
DCM | 112002 | Series Instance UID
DCM | 121126 | Performed Procedure Step SOP Instance UID
DCM | 113853 | Irradiation Event UID

## CID 10002 Irradiation Event Types

Type:

  Extensible

Version:

  20051101

Table CID 10002. Irradiation Event Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P5-06000 | Fluoroscopy
DCM | 113611 | Stationary Acquisition
DCM | 113612 | Stepping Acquisition
DCM | 113613 | Rotational Acquisition

## CID 10003 Equipment Plane Identification

Type:

  Extensible

Version:

  20081028

Table CID 10003. Equipment Plane Identification

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113620 | Plane A
DCM | 113621 | Plane B
DCM | 113622 | Single Plane
DCM | 113890 | All Planes

## CID 10004 Fluoro Modes

Type:

  Extensible

Version:

  20051101

Table CID 10004. Fluoro Modes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113630 | Continuous
DCM | 113631 | Pulsed

## CID 10006 X-Ray Filter Materials

Type:

  Extensible

Version:

  20081028

Table CID 10006. X-Ray Filter Materials

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-150F9 | Molybdenum or Molybdenum compound
SRT | C-120F9 | Aluminum or Aluminum compound
SRT | C-127F9 | Copper or Copper compound
SRT | C-167F9 | Rhodium or Rhodium compound
SRT | C-1190E | Niobium or Niobium compound
SRT | C-1190F | Europium or Europium compound
SRT | C-132F9 | Lead or Lead compound
SRT | C-156F9 | Tantalum or Tantalum compound
SRT | C-137F9 | Silver or Silver compound

## CID 10007 X-Ray Filter Types

Type:

  Extensible

Version:

  20131010

Table CID 10007. X-Ray Filter Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113650 | Strip filter
DCM | 113651 | Wedge filter
DCM | 113652 | Butterfly filter
DCM | 113653 | Flat filter
DCM | 111609 | No Filter

## CID 10008 Dose Related Distance Measurements

Type:

  Extensible

Version:

  20131010

Table CID 10008. Dose Related Distance Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113748 | Distance Source to Isocenter
DCM | 113737 | Distance Source to Reference Point
DCM | 113750 | Distance Source to Detector
DCM | 113751 | Table Longitudinal Position
DCM | 113752 | Table Lateral Position
DCM | 113753 | Table Height Position
DCM | 113792 | Distance Source to Table Plane
DCM | 113759 | Table Longitudinal End Position
DCM | 113760 | Table Lateral End Position
DCM | 113761 | Table Height End Position

## CID 10009 Measured/calculated

Type:

  Extensible

Version:

  20051101

Table CID 10009. Measured/Calculated

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-41D41 | Measured
SRT | R-41D2D | Calculated
SRT | R-10260 | Estimated

## CID 10010 Dose Measurement Devices

Type:

  Extensible

Version:

  20051101

Table CID 10010. Dose Measurement Devices

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | A-2C090 | Dosimeter

## CID 10011 Effective Dose Evaluation Method

Type:

  Extensible

Version:

  20071031

Table CID 10011. Effective Dose Evaluation Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113800 | DLP to E conversion via MC computation
DCM | 113801 | CTDIfreeair to E conversion via MC computation
DCM | 113802 | DLP to E conversion via measurement
DCM | 113803 | CTDIfreeair to E conversion via measurement

## CID 10013 CT Acquisition Type

Type:

  Extensible

Version:

  20071031

Table CID 10013. CT Acquisition Type

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113804 | Sequenced Acquisition
SRT | P5-08001 | Spiral Acquisition
DCM | 113805 | Constant Angle Acquisition
DCM | 113806 | Stationary Acquisition
DCM | 113807 | Free Acquisition

## CID 10014 Contrast Imaging Technique

Type:

  Extensible

Version:

  20071031

Table CID 10014. Contrast Imaging Technique

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P5-00100 | Diagnostic radiography with contrast media
SRT | P5-0808E | CT without contrast

## CID 10015 CT Dose Reference Authorities

Type:

  Extensible

Version:

  20081027

Table CID 10015. CT Dose Reference Authorities

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113808 | ICRP Pub 60
DCM | 113841 | ICRP Pub 103

## CID 10016 Anode Target Material

Type:

  Extensible

Version:

  20070827

Table CID 10016. Anode Target Material

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | C-150F9 | Molybdenum or Molybdenum compound
SRT | C-167F9 | Rhodium or Rhodium compound
SRT | C-164F9 | Tungsten or Tungsten compound

## CID 10017 X-Ray Grid

Type:

  Extensible

Version:

  20070827

Table CID 10017. X-Ray Grid

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 111641 | Fixed grid
DCM | 111642 | Focused grid
DCM | 111643 | Reciprocating grid
DCM | 111644 | Parallel grid
DCM | 111645 | Crossed grid
DCM | 111646 | No grid

## CID 10020 Source of Projection X-Ray Dose Information

Type:

  Extensible

Version:

  20120406

Table CID 10020. Source of Projection X-Ray Dose Information

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113856 | Automated Data Collection
DCM | 113857 | Manual Entry
DCM | 113858 | MPPS Content
SRT | A-2C090 | Dosimeter
DCM | 113866 | Copied From Image Attributes
DCM | 113867 | Computed From Image Attributes
DCM | 113868 | Derived From Human-Readable Reports
DCM | 113940 | System Calculated

## CID 10021 Source of CT Dose Information

Type:

  Extensible

Version:

  20081028

Table CID 10021. Source of CT Dose Information

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113856 | Automated Data Collection
DCM | 113857 | Manual Entry
DCM | 113866 | Copied From Image Attributes
DCM | 113867 | Computed From Image Attributes
DCM | 113868 | Derived From Human-Readable Reports

## CID 10022 Label Types

Type:

  Extensible

Version:

  20110816

Table CID 10022. Label Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113607 | Series Number
DCM | 113608 | Acquisition Number
DCM | 113609 | Instance Number

## CID 10023 Size Specific Dose Estimation Method for CT

Type:

  Extensible

Version:

  20120171

Table CID 10023. Size Specific Dose Estimation Method for CT

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113934 | AAPM 204 Lateral Dimension
DCM | 113935 | AAPM 204 AP Dimension
DCM | 113936 | AAPM 204 Sum of Lateral and AP Dimension
DCM | 113937 | AAPM 204 Effective Diameter Estimated From Patient Age

## CID 10025 Radiation Dose Reference Points

Type:

  Extensible

Version:

  20120406

Table CID 10025. Radiation Dose Reference Points

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113860 | 15cm from Isocenter toward Source
DCM | 113861 | 30cm in Front of Image Input Surface
DCM | 113862 | 1cm above Tabletop
DCM | 113863 | 30cm above Tabletop
DCM | 113864 | 15cm from Table Centerline
DCM | 113865 | Entrance exposure to a 4.2 cm breast thickness
DCM | 113941 | In Detector Plane

## CID 10030 Detector Types

Type:

  Extensible

Version:

  20120406

Table CID 10030. Detector Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113948 | Direct Detector
DCM | 113949 | Indirect Detector
DCM | 113950 | Storage Detector
DCM | 113951 | Film

## CID 10031 CR/DR Mechanical Configuration

Type:

  Extensible

Version:

  20120406

Table CID 10031. CR/DR Mechanical Configuration

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113953 | Unmounted Detector
DCM | 113952 | Table Mount
DCM | 113954 | Upright Stand Mount
DCM | 113955 | C-Arm Mount

## CID 10032 Projection X-Ray Acquisition Device Types

Type:

  Extensible

Version:

  20120406

Table CID 10032. Projection X-Ray Acquisition Device Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113957 | Fluoroscopy-Guided Projection Radiography System
DCM | 113958 | Integrated Projection Radiography System
DCM | 113959 | Cassette-based Projection Radiography System

## CID 10033 CT Reconstruction Algorithm

Type:

  Extensible

Version:

  20130207

Table CID 10033. CT Reconstruction Algorithm

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113962 | Filtered Back Projection
DCM | 113963 | Iterative Reconstruction

### Note

The values in this Context Group correspond to the Defined Terms for Reconstruction Algorithm (0018,9315) used in the CT Reconstruction Functional Group Macro in PS3.3.

## CID 10040 Radiopharmaceutical Organ Dose Reference Authority

Type:

  Extensible

Version:

  20140419

Table CID 10040. Radiopharmaceutical Organ Dose Reference Authority

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113520 | MIRD Pamphlet 1
DCM | 113521 | ICRP Publication 53
DCM | 113526 | MIRDOSE
DCM | 113527 | OLINDA-EXM
DCM | 113528 | Package Insert
DCM | 113529 | Institutionally Approved Estimates
DCM | 113530 | Investigational New Drug
DCM | 113522 | ICRP Publication 80
DCM | 113523 | ICRP Publication 106

## CID 10041 Source of Radioisotope Activity Information

Type:

  Extensible

Version:

  20140419

Table CID 10041. Source of Radioisotope Activity Information

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113541 | Dose Calibrator
DCM | 113542 | Infusion System
DCM | 113543 | Radioisotope Generator

## CID 10043 Intravenous Extravasation Symptoms

Type:

  Extensible

Version:

  20140419

Table CID 10043. Intravenous Extravasation Symptoms

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D0-B0324 | Injection site abscess
SRT | D0-B0380 | Injection site anesthesia
SRT | D0-B03A4 | Injection site atrophy
SRT | D0-B0394 | Injection site bruising
SRT | D0-B0342 | Injection site burning
SRT | D0-B0364 | Injection site cyst
SRT | D0-B0354 | Injection site dermatitis
SRT | D0-B0300 | Injection site disorder
SRT | D0-B0352 | Injection site edema
SRT | D0-B03A2 | Injection site fibrosis
SRT | M-44150 | Injection site granuloma
SRT | D0-B0334 | Injection site hemorrhage
SRT | D0-B0311 | Injection site hypersensitivity
SRT | D0-B03A0 | Injection site induration
SRT | D0-B0320 | Injection site infection
SRT | D0-B0350 | Injection site inflammation
SRT | D0-B0312 | Injection site irritation
SRT | D0-B0339 | Injection site malabsorption
SRT | D0-B0360 | Injection site mass
SRT | D0-B0370 | Injection site necrosis
SRT | D0-B0346 | Injection site nerve damage
SRT | D0-B0340 | Injection site pain
SRT | D0-B0382 | Injection site paresthesia
SRT | D0-B0314 | Injection site pigmentation change
SRT | D0-B0310 | Injection site reaction
SRT | M-78066 | Injection site scar
SRT | D0-B0326 | Injection site sterile abscess
SRT | D0-B0338 | Injection site thrombosis
SRT | D0-B0390 | Injection site ulcer
SRT | D0-B0356 | Injection site urticaria
DCM | 113568 | Extravasation visible in image

## CID 10044 Radiosensitive Organs

Type:

  Extensible

Version:

  20140419

Table CID 10044. Radiosensitive Organs

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-B3000 | Adrenal gland
SRT | T-74000 | Bladder
SRT | T-A0100 | Brain
SRT | T-04000 | Breast
SRT | T-C1000 | Bone Marrow
SRT | T-D0859 | Bone Surface
SRT | T-59300 | Colon
SRT | T-56000 | Esophagus
SRT | T-AA700 | Eye lenses
SRT | T-63000 | Gall bladder
SRT | T-32000 | Heart
SRT | T-71000 | Kidney
SRT | T-62002 | Liver
SRT | T-28000 | Lung
SRT | T-C4000 | Lymph node
SRT | T-13001 | Muscle
SRT | T-51300 | Oral mucosa
SRT | T-87000 | Ovary
SRT | T-65000 | Pancreas
SRT | T-9200B | Prostate
SRT | T-61007 | Salivary Glands
SRT | T-00009 | Skin
SRT | T-58000 | Small intestine
SRT | T-C3000 | Spleen
SRT | T-57000 | Stomach
SRT | T-94000 | Testis
SRT | T-C8000 | Thymus
SRT | T-B6000 | Thyroid
SRT | T-83000 | Uterus

## CID 10045 Radiopharmaceutical Patient State

Type:

  Extensible

Version:

  20140419

Table CID 10045. Radiopharmaceutical Patient State

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3102 “Rest-Stress”
SRT | F-70102 | Abnormal Renal Function
DCM | 113560 | Acute unilateral renal blockage
DCM | 113561 | Low Thyroid Uptake
DCM | 113562 | High Thyroid Uptake
DCM | 113563 | Severely Jaundiced

## CID 10046 GFR Measurements

Type:

  Extensible

Version:

  20140419

Table CID 10046. GFR Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 33914‐3 | Glomerular Filtration Rate (MDRD)
LN | 48642-3 | Glomerular Filtration Rate non-black (MDRD)
LN | 48643-1 | Glomerular Filtration Rate black (MDRD)
LN | 50044-7 | Glomerular Filtration Rate female (MDRD)
LN | 50210-4 | Glomerular Filtration Rate Cystatin-based formula
LN | 50384-7 | Glomerular Filtration Rate Creatinine-based formula (Schwartz)
LN | 35591-7 | Cockroft-Gault Formula estimation of GFR
LN | 62238-1 | CKD-EPI Formula estimation of GFR

## CID 10047 GFR Measurement Methods

Type:

  Extensible

Version:

  20140419

Table CID 10047. GFR Measurement Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 113570 | Cockroft-Gault Formula estimation of GFR
DCM | 113571 | CKD-EPI Formula estimation of GFR
DCM | 113572 | Glomerular Filtration Rate (MDRD)
DCM | 113573 | Glomerular Filtration Rate non-black (MDRD)
DCM | 113574 | Glomerular Filtration Rate black (MDRD)
DCM | 113575 | Glomerular Filtration Rate female (MDRD)
DCM | 113576 | Glomerular Filtration Rate Cystatin-based formula
DCM | 113577 | Glomerular Filtration Rate Creatinine-based formula (Schwartz)

## CID 12001 Ultrasound Protocol Types

Type:

  Extensible

Version:

  20081027

Table CID 12001. Ultrasound Protocol Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P5-B3000 | Echocardiography
SRT | P5-B3002 | Transesophageal echocardiography
SRT | P5-B3012 | Transthoracic echocardiography
SRT | P0-05F95 | Epicardial echocardiography
SRT | P5-B3005 | Intravascular echocardiography
SRT | P5-B3006 | Intracardiac echocardiography
SRT | P5-B3050 | Exercise stress echocardiography
SRT | P5-B300F | Pediatric echocardiography
SRT | P5-B300C | Intraoperative echocardiography
SRT | P5-B3090 | Contrast echocardiography
SRT | P5-B8215 | Fetal echocardiography
Include CID 3261 “Stress Protocols”

### Note

In a prior version of this context group, Transthoracic echocardiography was assigned the code P5-B3003 and Epicardial echocardiography was assigned the code P5-B3004; these codes conflict with other SNOMED code assignments. In addition, the prior version used many codes that are not actually in SNOMED. Receiving applications should be aware of this change, and the possibility of misinterpretation of SOP Instances that may include the deprecated codes; see Annex J.

## CID 12002 Ultrasound Protocol Stage Types

Type:

  Extensible

Version:

  20081027

Table CID 12002. Ultrasound Protocol Stage Types

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 3207 “Stress Test Procedure Phases”
Include CID 12102 “Temporal Periods Relating to Procedure or Therapy”
SRT | P2-35000 | Cardiac pacing
SRT | P2-71306 | Hand grip
SRT | R-40928 | Valsalva maneuver

### Note

A prior version of this context group used many codes that are not actually in SNOMED. Although there is minimal possibility of misinterpretation with SOP Instances that may include the deprecated use, receiving applications should be aware of this change; see Annex J.

## CID 12003 OB-GYN Dates

Type:

  Extensible

Version:

  20030130

Table CID 12003. OB-GYN Dates

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11778-8 | EDD
LN | 11779-6 | EDD from LMP
LN | 11781-2 | EDD from average ultrasound age
LN | 11780-4 | EDD from ovulation date
LN | 11955-2 | LMP
LN | 33066-2 | Estimated LMP by EDD
LN | 11976-8 | Ovulation date
LN | 33067-0 | Conception Date

## CID 12004 Fetal Biometry Ratios

Type:

  Extensible

Version:

  20030130

Table CID 12004. Fetal Biometry Ratios

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11947-9 | HC/AC
LN | 11871-1 | FL/AC
LN | 11872-9 | FL/BPD
LN | 11823-2 | Cephalic Index
LN | 11873-7 | FL/HC

## CID 12005 Fetal Biometry Measurements

Type:

  Extensible

Version:

  20030130

Table CID 12005. Fetal Biometry Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11979-2 | Abdominal Circumference
LN | 11818-2 | Anterior-Posterior Abdominal Diameter
LN | 11819-0 | Anterior-Posterior Trunk Diameter
LN | 11820-8 | Biparietal Diameter
LN | 11824-0 | BPD area corrected
LN | 11860-4 | Cisterna Magna
LN | 11963-6 | Femur Length
LN | 11965-1 | Foot length
LN | 11984-2 | Head Circumference
LN | 11851-3 | Occipital-Frontal Diameter
LN | 11988-3 | Thoracic Circumference
LN | 33068-8 | Thoracic Area
LN | 11862-0 | Tranverse Abdominal Diameter
LN | 11863-8 | Trans Cerebellar Diameter
LN | 11864-6 | Transverse Thoracic Diameter
LN | 11853-9 | Left Kidney thickness
LN | 11834-9 | Left Kidney length
LN | 11825-7 | Left Kidney width
LN | 11855-4 | Right Kidney thickness
LN | 11836-4 | Right Kidney length
LN | 11827-3 | Right Kidney width
LN | 33191-8 | APAD * TAD

## CID 12006 Fetal Long Bones Biometry Measurements

Type:

  Extensible

Version:

  20030130

Table CID 12006. Fetal Long Bones Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11966-9 | Humerus length
LN | 11967-7 | Radius length
LN | 11969-3 | Ulna length
LN | 11968-5 | Tibia length
LN | 11964-4 | Fibula length
LN | 11962-8 | Clavicle length
LN | 11963-6 | Femur Length

## CID 12007 Fetal Cranium

Type:

  Extensible

Version:

  20030130

Table CID 12007. Fetal Cranium

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 12171-5 | Lateral Ventricle width
LN | 11860-4 | Cisterna Magna length
LN | 12146-7 | Nuchal Fold thickness
LN | 33070-4 | Inner Orbital Diameter
LN | 11629-3 | Outer Orbital Diameter
LN | 11863-8 | Trans Cerebellar Diameter
LN | 33069-6 | Nuchal Translucency
LN | 33197-5 | Anterior Horn Lateral ventricular width
LN | 33196-7 | Posterior Horn Lateral ventricular width
LN | 12170-7 | Width of Hemisphere

## CID 12008 OB-GYN Amniotic Sac

Type:

  Extensible

Version:

  20030130

Table CID 12008. OB-GYN Amniotic Sac

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11624-4 | First Quadrant Diameter
LN | 11626-9 | Second Quadrant Diameter
LN | 11625-1 | Third Quadrant Diameter
LN | 11623-6 | Fourth Quadrant Diameter
SRT | M-02550 | Diameter

## CID 12009 Early Gestation Biometry Measurements

Type:

  Extensible

Version:

  20030130

Table CID 12009. Early Gestation Biometry Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11957-8 | Crown Rump Length
LN | 11850-5 | Gestational Sac Diameter
LN | 33071-2 | Spine Length
LN | 11816-6 | Yolk Sac length
LN | 33069-6 | Nuchal Translucency

## CID 12011 Ultrasound Pelvis and Uterus

Type:

  Extensible

Version:

  20030130

Table CID 12011. Ultrasound Pelvis and Uterus

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11961-0 | Cervix Length
LN | 12145-9 | Endometrium Thickness

## CID 12012 OB Equations and Tables

Type:

  Extensible

Version:

  20030130

Table CID 12012. OB Equations and Tables

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12013 “Gestational Age Equations and Tables”
Include CID 12014 “OB Fetal Body Weight Equations and Tables”
Include CID 12015 “Fetal Growth Equations and Tables”
Include CID 12016 “Estimated Fetal Weight Percentile Equations and Tables”

## CID 12013 Gestational Age Equations and Tables

These terms define a functional relationship of the gestational age from a biometric measurement.

Type:

  Extensible

Version:

  20061024

Table CID 12013. Gestational Age Equations and Tables

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11885-1 | Gestational Age by LMP
LN | 11884-4 | Average Ultrasound Age
LN | 33072-0 | AC, ASUM 2000
LN | 11889-3 | AC, Campbell 1975
LN | 11892-7 | AC, Hadlock 1984
LN | 33073-8 | AC, Hansmann1985
LN | 33537-2 | AC, Jeanty 1982
LN | 11893-5 | AC, Jeanty 1984
LN | 33074-6 | AC, Lessoway 1998
LN | 33075-3 | AC, Mertz 1988
LN | 33076-1 | AC, Shinozuka 1996
LN | 33077-9 | A-P Abdominal Diameter, Lessoway 1998
LN | 33078-7 | AxT, Shinozuka 1996
LN | 33079-5 | BPD, ASUM 1989
LN | 11900-8 | BPD, Doubilet 1993
LN | 11902-4 | BPD, Hadlock 1984
LN | 11903-2 | BPD, Hansmann 1985
LN | 33538-0 | BPD, Hansmann 1986
LN | 33539-8 | BPD, Jeanty 1982
LN | 11905-7 | BPD, Jeanty 1984
LN | 11906-5 | BPD, Kurtz 1980
LN | 33080-3 | BPD, Lessoway 1998
LN | 33081-1 | BPD, Mertz 1988
LN | 33082-9 | BPD, Osaka 1989
LN | 33083-7 | BPD, Rempen 1991
LN | 11907-3 | BPD, Sabbagha 1978
LN | 33084-5 | BPD, Shinozuka 1996
LN | 33085-2 | BPD, Tokyo 1986
LN | 11901-6 | BPDa, Hadlock 1982
LN | 33086-0 | BPD-oi, Chitty 1997
LN | 33087-8 | BPD-oo, Chitty 1997
LN | 33088-6 | Clavicle length, Yarkoni 1985
LN | 33089-4 | CRL, ASUM 1991
LN | 33090-2 | CRL, ASUM 2000
LN | 33091-0 | CRL, Daya 1993
LN | 11910-7 | CRL, Hadlock 1992
LN | 11911-5 | CRL, Hansmann 1985
LN | 33540-6 | CRL, Hansmann 1986
LN | 33092-8 | CRL, Jeanty 1982
LN | 11917-2 | CRL, Jeanty 1984
LN | 11913-1 | CRL, Nelson 1981
LN | 33093-6 | CRL, Osaka 1989
LN | 33094-4 | CRL, Rempen 1991
LN | 11914-9 | CRL, Robinson 1975
LN | 33095-1 | CRL, Shinozuka 1996
LN | 33096-9 | CRL, Tokyo 1986
LN | 33097-7 | Fibula, Jeanty 1983
LN | 11918-0 | Fibula, Merz 1987
LN | 33098-5 | FL, Chitty 1997
LN | 11920-6 | FL, Hadlock 1984
LN | 11921-4 | FL, Hansmann 1985
LN | 33541-4 | FL, Hansmann 1986
LN | 11922-2 | FL, Hohler 1982
LN | 33099-3 | FL, Jeanty 1982
LN | 11923-0 | FL, Jeanty 1984
LN | 33100-9 | FL, Lessoway 1998
LN | 11924-8 | FL, Merz 1987
LN | 33542-2 | FL, Merz 1988
LN | 33101-7 | FL, Osaka 1989
LN | 33102-5 | FL, Shinozuka 1996
LN | 33103-3 | FL, Tokyo 1986
LN | 11926-3 | Foot Length, Mercer 1987
LN | 33104-1 | GS, Daya 1991
LN | 33105-8 | GS, Hansmann 1979
LN | 33106-6 | GS, Hansmann 1982
LN | 11928-9 | GS, Hellman 1969
LN | 33107-4 | GS, Nyberg 1992
LN | 11929-7 | GS, Rempen 1991
LN | 33108-2 | GS, Tokyo 1986
LN | 33109-0 | HC, ASUM 2000
LN | 33110-8 | HC measured, Chitty 1997
LN | 33111-6 | HC derived, Chitty 1997
LN | 11932-1 | HC, Hadlock 1984
LN | 33112-4 | HC, Hansmann 1985
LN | 33543-0 | HC, Hansmann 1986
LN | 33113-2 | HC, Jeanty 1982
LN | 11934-7 | HC, Jeanty 1984
LN | 33114-0 | HC, Lessoway 1998
LN | 33115-7 | HC Merz, 1988
LN | 33116-5 | Humerus Length, ASUM 2000
LN | 11936-2 | Humerus, Jeanty 1984
LN | 11937-0 | Humerus, Merz 1987
LN | 33117-3 | Humerus Length, Osaka 1989
LN | 33118-1 | Length of Vertebra, Tokyo 1986
LN | 33119-9 | OFD, ASUM 2000
LN | 33544-8 | OFD, Hansmann 1985
LN | 33120-7 | OFD, Hansmann 1986
LN | 33121-5 | OFD, Lessoway 1998
LN | 33122-3 | IOD, Mayden 1982
LN | 33123-1 | IOD, Trout 1994
LN | 33545-5 | BD, Jeanty 1982
LN | 33124-9 | OOD, Mayden, 1982
LN | 33125-6 | OOD, Trout 1994
LN | 33126-4 | Radius, Jeanty 1983
LN | 11939-6 | Radius, Merz 1987
LN | 33127-2 | Spine Length, Tokyo, 1989
LN | 11941-2 | Tibia, Jeanty 1984
LN | 33128-0 | TAD, Eriksen 1985
LN | 33129-8 | TAD Hansmann, 1979
LN | 33130-6 | TAD, Tokyo 1986
LN | 33131-4 | ThC, Chitkara 1987
LN | 33132-2 | TCD, Chitty 1994
LN | 33133-0 | TCD, Goldstein 1987
LN | 33134-8 | TCD, Hill 1990
LN | 33135-5 | TCD, Nimrod 1986
LN | 33136-3 | Transverse Thoracic Diameter, Hansmann 1985
LN | 33137-1 | Transverse Thoracic Diameter, Lessoway 1998
LN | 33138-9 | Fetal Trunk Cross-Sectional Area, Osaka 1989
LN | 11944-6 | Ulna, Jeanty 1984
LN | 11945-3 | Ulna, Merz 1987

## CID 12014 OB Fetal Body Weight Equations and Tables

These terms define a functional relationship to estimated fetal body mass from a biometric measurement.

Type:

  Extensible

Version:

  20030130

Table CID 12014. OB Fetal Body Weight Equations and Tables

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11756-4 | EFW by AC, Campbell 1975
LN | 11738-2 | EFW by AC, BPD, Hadlock 1984
LN | 11734-1 | EFW by AC, BPD, FL, Hadlock 1984
LN | 11735-8 | EFW by AC, BPD, FL, Hadlock 1985
LN | 11732-5 | EFW by AC, BPD, FL, HC, Hadlock 1985
LN | 11750-7 | EFW by AC, FL, Hadlock 1984
LN | 11751-5 | EFW by AC, FL, Hadlock 1985
LN | 11746-5 | EFW by AC, FL, HC, Hadlock 1985
LN | 11754-9 | EFW by AC, HC Hadlock 1984
LN | 33139-7 | EFW by BPD, TTD, Hansmann 1986
LN | 11739-0 | EFW by AC and BPD, Shepard 1982
LN | 33140-5 | EFW by BPD, FTA, FL, Osaka 1990
LN | 33141-3 | EFW1 by Shinozuka 1996
LN | 33142-1 | EFW2 by Shinozuka 1996
LN | 33143-9 | EFW3 by Shinozuka 1996
LN | 33144-7 | EFW by BPD, APAD, TAD, FL, Tokyo 1987

## CID 12015 Fetal Growth Equations and Tables

These terms specify biometric growth parameter of a population distribution as a function of gestational age. The term may also specify the population's distribution, and so enable calculating a percentile rank or Z-score relative to the distribution.

Type:

  Extensible

Version:

  20030130

Table CID 12015. Fetal Growth Equations and Tables

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 33145-4 | AC by GA, ASUM 2000
LN | 33146-2 | AC by GA, Hadlock 1984
LN | 33147-0 | AC (measured) by GA, Chitty 1994
LN | 33546-3 | AC (derived) by GA, Chitty 1994
LN | 33148-8 | AC by GA, Merz 1988
LN | 33149-6 | AC by GA, Shinozuka 1996
LN | 33150-4 | AxT by GA, Shinozuka 1996
LN | 33151-2 | BPD by GA, ASUM 2000
LN | 33198-3 | BPD by GA, Hadlock 1984
LN | 33556-2 | BPD outer-inner by GA, Chitty 1994
LN | 33152-0 | BPD outer-outer by GA, Chitty 1994
LN | 33153-8 | BPD by GA, Jeanty 1982
LN | 33154-6 | BPD by GA, Merz 1988
LN | 33155-3 | BPD by GA, Rempen 1991
LN | 33156-1 | BPD by GA, Shinozuka 1996
LN | 33157-9 | Cephalic Index by GA, Chitty 1994
LN | 33158-7 | Cephalic Index by GA, Hadlock 1981
LN | 33159-5 | CRL by GA ASUM 2000
LN | 33160-3 | CRL by GA, Rempen1991
LN | 33161-1 | CRL by GA, Shinozuka 1996
LN | 33162-9 | EFW by GA, Hadlock 1991
LN | 33163-7 | EFW by GA, Hansmann 1986
LN | 33164-5 | Fibula by GA, Jeanty 1983
LN | 33165-2 | FL by GA, ASUM 2000
LN | 33166-0 | FL by GA, Hadlock 1984
LN | 33167-8 | FL by GA, Chitty 1994
LN | 33168-6 | FL by GA, Jeanty 1982
LN | 33169-4 | FL by GA, Merz 1988
LN | 33170-2 | FL by GA, Shinozuka 1996
LN | 33171-0 | GS by GA, Rempen 1991
LN | 33172-8 | HC by GA, ASUM 2000
LN | 33173-6 | HC by GA, Hadlock 1984
LN | 33174-4 | HC derived by GA, Chitty 1994
LN | 33175-1 | HC by GA, Jeanty 1982
LN | 33176-9 | HC by GA, Merz 1988
LN | 33177-7 | Humerus Length by GA, ASUM 2000
LN | 33178-5 | OFD by GA, ASUM 2000
LN | 33179-3 | OFD by GA, Chitty 1994
LN | 33180-1 | Radius by GA, Jeanty 1983
LN | 33181-9 | TCD by GA Goldstein 1987
LN | 33182-7 | HC/AC by GA, Campbell 1977

## CID 12016 Estimated Fetal Weight Percentile Equations and Tables

These terms specify the population distribution for use in Z-score and percentile rank.

Type:

  Extensible

Version:

  20030130

Table CID 12016. Estimated Fetal Weight Percentile Equations and Tables

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 33183-5 | FWP by GA, Hadlock 1991
LN | 33184-3 | FWP by GA, Williams, 1982
LN | 33185-0 | FWP by GA, Alexander, 1996
LN | 33186-8 | Male Singleton BWP by GA, Arbuckle 1993
LN | 33187-6 | Female Singleton BWP by GA, Arbuckle 1993
LN | 33199-1 | Male Twins BWP by GA, Arbuckle 1993
LN | 33188-4 | Female Twins BWP by GA, Arbuckle 1993
LN | 33189-2 | FWP by GA, Brenner 1976
LN | 33190-0 | FWP by MA, Hadlock 1985

## CID 12017 Growth Distribution Rank

Type:

  Extensible

Version:

  20030130

Table CID 12017. Growth Distribution Rank

Code Scheme | Code Value | Code Meaning
--- | --- | ---
DCM | 125012 | Growth Percentile Rank
DCM | 125013 | Growth Z-score

## CID 12018 OB-GYN Summary

Type:

  Extensible

Version:

  20030130

Table CID 12018. OB-GYN Summary

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11878-6 | Number of Fetuses by US
LN | 11886-9 | Gestational Age by ovulation date

## CID 12019 OB-GYN Fetus Summary

Type:

  Extensible

Version:

  20030130

Table CID 12019. OB-GYN Fetus Summary

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 18185-9 | Gestational Age
LN | 11888-5 | Composite Ultrasound Age
LN | 11885-1 | Gestational Age by LMP
LN | 11727-5 | Estimated Weight
LN | 11767-1 | EFW percentile rank
LN | 11948-7 | Fetal Heart Rate

## CID 12020 Fetal Biometry Anatomic Sites

Type:

  Extensible

Version:

  20141110

Table CID 12020. Fetal Biometry Anatomic Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-D4000 | Abdomen
SRT | T-A6000 | Cerebellum
SRT | T-A1520 | Cisterna Magna
SRT | T-12710 | Femur
SRT | T-D9700 | Foot
SRT | T-71000 | Kidney
SRT | T-11100 | Skull
SRT | T-D3000 | Thorax
SRT | T-D2000 | Trunk

## CID 12021 Fetal Long Bone Anatomic Sites

Type:

  Extensible

Version:

  20141110

Table CID 12021. Fetal Long Bone Anatomic Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-12310 | Clavicle
SRT | T-12710 | Femur
SRT | T-12750 | Fibula
SRT | T-12750 | Radius
SRT | T-12740 | Tibia
SRT | T-12430 | Ulna

## CID 12022 Fetal Cranium Bone Anatomic Sites

Type:

  Extensible

Version:

  20141110

Table CID 12022. Fetal Cranium Anatomic Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-A1700 | Anterior Horn Lateral Ventricle
SRT | T-A6000 | Cerebellum
SRT | T-A010F | Cerebral hemisphere
SRT | T-A1520 | Cisterna magna
SRT | T-A1650 | Lateral Ventricle
SRT | T-D06B6 | Nuchal region of scalp
SRT | T-D14AE | Orbit
SRT | T-A1710 | Posterior Horn Lateral Ventricle

## CID 12023 Pelvis and Uterus Anatomic Sites

Type:

  Extensible

Version:

  20141110

Table CID 12023. Pelvis and Uterus Anatomic Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-83200 | Cervix
SRT | T-83400 | Endometrium
SRT | T-83000 | Uterus

## CID 12030 Ultrasound Contrast/Bolus Agents

Type:

  Extensible

Version:

  20090409

Table CID 12030. Ultrasound Contrast/Bolus Agents

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125901 | CARDIOsphere
NDC | 11994-*011-04 | Definity
DCM | 125902 | Echovist
DCM | 125903 | Imagify
DCM | 125904 | Levovist
NDC | 00019-2707-03 | Optison
DCM | 125905 | Sonazoid
DCM | 125906 | SonoVue
DCM | 125907 | Targestar-B
DCM | 125908 | Targestar-P

### Note

1. See Controlled Terminology descriptions in Annex D for manufacturer references.

2. The generic formulation is not used for Code Meaning (0008,0104) because for ultrasonic contrast agents the physical properties of the agent are more significant than chemical formula in determining its acoustic properties.

## CID 12031 Protocol Interval Events

Type:

  Extensible

Version:

  20090409

Table CID 12031. Protocol Interval Events

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125233 | Start of drug dose administration
DCM | 125234 | Start of contrast agent administration
DCM | 125235 | Destruction of microbubbles
DCM | 125236 | Onset of exercise
DCM | 125237 | Cessation of exercise
DCM | 125238 | Onset of stimulation
DCM | 125239 | Cessation of stimulation

## CID 12032 Transducer Scan Pattern

Type:

  Extensible

Version:

  20090409

Table CID 12032. Transducer Scan Pattern

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125240 | Line scan pattern
DCM | 125241 | Plane scan pattern
DCM | 125242 | Volume scan pattern

## CID 12033 Ultrasound Transducer Geometry

Type:

  Extensible

Version:

  20090409

Table CID 12033. Ultrasound Transducer Geometry

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125251 | Non-imaging Doppler ultrasound transducer geometry
DCM | 125252 | Linear ultrasound transducer geometry
DCM | 125253 | Curved linear ultrasound transducer geometry
DCM | 125254 | Sector ultrasound transducer geometry
DCM | 125255 | Radial ultrasound transducer geometry
DCM | 125256 | Ring ultrasound transducer geometry

## CID 12034 Ultrasound Transducer Beam Steering

Type:

  Extensible

Version:

  20090409

Table CID 12034. Ultrasound Transducer Beam Steering

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125257 | Fixed beam direction
DCM | 125258 | Mechanical beam steering
DCM | 125259 | Phased beam steering

## CID 12035 Ultrasound Transducer Application

Type:

  Extensible

Version:

  20090409

Table CID 12035. Ultrasound Transducer Application

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125261 | External Transducer
DCM | 125262 | Transesophageal Transducer
DCM | 125263 | Endovaginal Transducer
DCM | 125264 | Endorectal Transducer
DCM | 125265 | Intravascular Transducer

## CID 12101 Vascular Summary

Type:

  Extensible

Version:

  20030327

Table CID 12101. Vascular Summary

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 121106 | Comment

## CID 12102 Temporal Periods Relating to Procedure or Therapy

Type:

  Extensible

Version:

  20050110

Table CID 12102. Temporal Periods Relating to Procedure or Therapy

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-422A4 | After Procedure
SRT | R-40FBA | During Procedure
SRT | R-40FB9 | Before Procedure

## CID 12103 Vascular Ultrasound Anatomic Location

Type:

  Extensible

Version:

  20030327

Table CID 12103. Vascular Ultrasound Anatomic Location

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12104 “Extracranial Arteries”
Include CID 12105 “Intracranial Cerebral Vessels”
Include CID 12106 “Intracranial Cerebral Vessels (unilateral)”
Include CID 12107 “Upper Extremity Arteries”
Include CID 12108 “Upper Extremity Veins”
Include CID 12109 “Lower Extremity Arteries”
Include CID 12110 “Lower Extremity Veins”
Include CID 12111 “Abdominal Arteries (lateral)”
Include CID 12112 “Abdominal Arteries (unilateral)”
Include CID 12113 “Abdominal Veins (lateral)”
Include CID 12114 “Abdominal Veins (unilateral)”
Include CID 12115 “Renal Vessels”

## CID 12104 Extracranial Arteries

This context group specifies the anatomic location for vascular observations

Type:

  Extensible

Version:

  20030327

Table CID 12104. Extracranial Arteries

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-45160 | Carotid Bifurcation
SRT | T-45170 | Carotid Bulb
SRT | T-45100 | Common Carotid Artery
SRT | T-45200 | External Carotid Artery
SRT | T-45300 | Internal Carotid Artery
SRT | T-46100 | Subclavian Artery
SRT | T-45700 | Vertebral Artery

## CID 12105 Intracranial Cerebral Vessels

Type:

  Extensible

Version:

  20050110

Table CID 12105. Intracranial Cerebral Vessels

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-45540 | Anterior Cerebral Artery
SRT | T-45530 | Anterior Communicating Artery
SRT | G-0368 | Anterior-Middle Cerebral Artery Bifurcation
SRT | G-0369 | Anterior-Posterior Cerebral Artery Bifurcation
SRT | T-45308 | Carotid Siphon
SRT | T-45430 | Central Retinal Artery
SRT | T-48286 | Central Retinal Vein
SRT | T-45300 | Internal Carotid Artery
SRT | R-102BB | Internal Carotid Artery C5 segment
SRT | R-102BC | Internal Carotid Artery C6 segment
SRT | R-102BD | Terminal internal carotid artery
SRT | T-45600 | Middle Cerebral Artery
SRT | R-1024F | Middle Cerebral Artery M1 Segment
SRT | R-10251 | Middle Cerebral Artery M2 Segment
SRT | T-45400 | Ophthalmic Artery
SRT | T-45900 | Posterior Cerebral Artery
SRT | R-10253 | Posterior Cerebral Artery P1 Segment
SRT | R-10255 | Posterior Cerebral Artery P2 Segment
SRT | T-45320 | Posterior Communicating Artery

## CID 12106 Intracranial Cerebral Vessels (unilateral)

Type:

  Extensible

Version:

  20030327

Table CID 12106. Intracranial Cerebral Vessels (Unilateral)

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-45800 | Basilar Artery

## CID 12107 Upper Extremity Arteries

Type:

  Extensible

Version:

  20050110

Table CID 12107. Upper Extremity Arteries

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-47100 | Axillary Artery
SRT | T-47160 | Brachial Artery
SRT | T-47340 | Deep Palmar Arch of Radial Artery
SRT | T-46010 | Innominate Artery
SRT | T-47300 | Radial Artery
T-12420 | Radius | RADIUS
T-12403 | Radius and ulna | RADIUSULNA
SRT | T-46100 | Subclavian Artery
SRT | T-47240 | Superficial Palmar Arch
SRT | T-47200 | Ulnar Artery
SRT | T-47260 | Digital artery of hand

## CID 12108 Upper Extremity Veins

Type:

  Extensible

Version:

  20030327

Table CID 12108. Upper Extremity Veins

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-49110 | Axillary vein
SRT | T-48052 | Basilic vein
SRT | T-49350 | Brachial vein
SRT | T-49240 | Cephalic vein
SRT | T-48620 | Innominate vein
SRT | T-48170 | Internal Jugular vein
SRT | T-49250 | Median Cubital vein
SRT | T-49340 | Radial vein
SRT | T-48330 | Subclavian vein
SRT | T-49330 | Ulnar vein
SRT | T-48610 | Superior Vena Cava
SRT | T-49218 | Deep Palmar Venous Arch
SRT | T-49217 | Superficial Palmar Venous Arch

## CID 12109 Lower Extremity Arteries

Type:

  Extensible

Version:

  20050110

Table CID 12109. Lower Extremity Arteries

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-46710 | Common Iliac Artery
SRT | R-10258 | Common Iliac Artery Bifurcation
SRT | T-47700 | Anterior Tibial Artery
SRT | T-47400 | Common Femoral Artery
SRT | T-47741 | Dorsalis Pedis Artery
SRT | T-46910 | External Iliac Artery
SRT | T-46740 | Internal Iliac Artery
SRT | T-47630 | Peroneal Artery
SRT | T-47690 | Plantar Arterial Arch
SRT | T-47500 | Popliteal Artery
SRT | T-47600 | Posterior Tibial Artery
SRT | T-47440 | Profunda Femoris Artery
SRT | T-47403 | Superficial Femoral Artery

## CID 12110 Lower Extremity Veins

Type:

  Extensible

Version:

  20101103

Table CID 12110. Lower Extremity Veins

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-49630 | Anterior Tibial Vein
SRT | T-49423 | Lateral calf perforator
SRT | G-035B | Common Femoral Vein
SRT | T-48920 | Common Iliac Vein
SRT | T-48930 | External Iliac Vein
SRT | T-4942D | Gastrocnemius vein
SRT | G-036F | Giacomini vein
SRT | T-49530 | Great Saphenous Vein
SRT | R-10259 | Great Saphenous Vein of Thigh
SRT | R-1025A | Great Saphenous Vein of Calf
SRT | T-49550 | Lesser Saphenous Vein
SRT | T-49640 | Peroneal Vein
SRT | T-49650 | Popliteal Vein
SRT | G-036E | Posterior arch vein
SRT | T-49620 | Posterior Tibial Vein
SRT | T-49660 | Profunda Femoris Vein
SRT | T-D930A | Saphenofemoral Junction
SRT | G-036B | Soleal vein
SRT | G-035A | Superficial Femoral Vein
SRT | T-4942C | Thigh perforator
SRT | T-48940 | Internal iliac vein
SRT | T-4941A | Saphenopopliteal junction
SRT | T-4942A | Hunterian perforating vein
SRT | T-49426 | Cockett's perforating vein
SRT | T-49424 | Boyd's perforating vein

## CID 12111 Abdominal Arteries (lateral)

Type:

  Extensible

Version:

  20050110

Table CID 12111. Abdominal Arteries (Lateral)

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-46640 | Accessory Renal Artery
SRT | T-46410 | Gastric Artery
SRT | T-46980 | Ovarian Artery
SRT | T-46970 | Testicular Artery
SRT | T-F1810 | Umbilical Artery
SRT | T-46820 | Uterine Artery

## CID 12112 Abdominal Arteries (unilateral)

Type:

  Extensible

Version:

  20050110

Table CID 12112. Abdominal Arteries (Unilateral)

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-42000 | Aorta
SRT | T-42520 | Infra-renal Aorta
SRT | T-42510 | Supra-renal Aorta
SRT | T-46400 | Celiac Axis
SRT | T-46421 | Common Hepatic Artery
SRT | T-46710 | Common Iliac Artery
SRT | T-46440 | Gastroduodenal Artery
SRT | T-46520 | Inferior Mesenteric Artery
SRT | T-46960 | Lumbar Artery
SRT | T-46422 | Proper Hepatic Artery
SRT | T-46423 | Right Branch of Hepatic Artery
SRT | T-46427 | Left Branch of Hepatic Artery
SRT | T-46460 | Splenic Artery
SRT | T-46510 | Superior Mesenteric Artery

## CID 12113 Abdominal Veins (lateral)

Type:

  Extensible

Version:

  20030327

Table CID 12113. Abdominal Veins (Lateral)

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-48920 | Common iliac vein
SRT | T-48820 | Gastric vein
SRT | G-0370 | Ileal vein
SRT | T-48780 | Ovarian vein
SRT | T-48770 | Testicular Vein
SRT | G-035E | First Lumbar Artery
SRT | G-035F | Second Lumbar Artery
SRT | G-0360 | Third Lumbar Artery
SRT | G-0361 | Fourth Lumbar Artery
SRT | G-0362 | Fifth Lumbar Artery
SRT | G-0363 | Sixth Lumbar Artery

## CID 12114 Abdominal Veins (unilateral)

Type:

  Extensible

Version:

  20030327

Table CID 12114. Abdominal Veins (Unilateral)

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-48720 | Hepatic Vein
SRT | G-036D | Inferior Right Hepatic Vein
SRT | T-48727 | Left Hepatic Vein
SRT | T-48726 | Middle Hepatic Vein
SRT | T-48725 | Right Hepatic Vein
SRT | T-48810 | Portal Vein
SRT | T-4881F | Left Main Branch of Portal Vein
SRT | T-4882A | Right Main Branch of Portal Vein
SRT | T-48910 | Inferior Mesenteric Vein
SRT | T-48710 | Inferior Vena Cava
SRT | T-48890 | Splenic Vein
SRT | T-48840 | Superior Mesenteric Vein
SRT | G-036C | Transjugular Intrahepatic Portosystemic Shunt
SRT | T-48817 | Umbilical Vein

## CID 12115 Renal Vessels

Type:

  Extensible

Version:

  20030327

Table CID 12115. Renal Vessels

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-46600 | Renal Artery
SRT | G-035C | Hilar Artery
SRT | T-46659 | Segmental Artery
SRT | T-4667C | Lobar Artery
SRT | T-4668A | Arcuate Artery of the Kidney
SRT | T-4667D | Interlobar Artery of Kidney
SRT | T-46640 | Accessory Renal Artery
SRT | T-46668 | Perforating Artery of Kidney
SRT | T-48740 | Renal Vein

## CID 12116 Vessel Segment Modifiers

This context group is the set of modifiers that specify the position along a vessel segment.

Type:

  Extensible

Version:

  20050110

Table CID 12116. Vessel Segment Modifiers

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A119 | Distal
SRT | G-A188 | Mid-longitudinal
SRT | G-036A | Origin of vessel
SRT | G-A118 | Proximal
SRT | R-1025B | Dilated portion of segment

## CID 12117 Vessel Branch Modifiers

This context group is the set of modifiers to specify a particular vessel segment or branch.

Type:

  Extensible

Version:

  20030327

Table CID 12117. Vessel Branch Modifiers

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-035D | Collateral branch of vessel
SRT | G-A115 | Inferior
SRT | G-A104 | Lateral
SRT | G-A101 | Left
SRT | G-A332 | Main
SRT | G-A109 | Medial
SRT | G-A100 | Right
SRT | G-A116 | Superior

## CID 12118 Measurement Orientation

Type:

  Extensible

Version:

  20110125

Table CID 12118. Measurement Orientation

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122675 | Anterior-Posterior
SRT | G-A117 | Transverse
SRT | G-A143 | Longitudinal

## CID 12119 Vascular Ultrasound Property

Type:

  Extensible

Version:

  20030327

Table CID 12119. Vascular Ultrasound Property

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12120 “Blood Velocity Measurements by Ultrasound”
Include CID 12121 “Vascular Indices and Ratios”
Include CID 12122 “Other Vascular Properties”

## CID 12120 Blood Velocity Measurements by Ultrasound

Type:

  Extensible

Version:

  20030327

Table CID 12120. Blood Velocity Measurements by Ultrasound

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 11653-3 | End Diastolic Velocity
LN | 11665-7 | Minimum Diastolic Velocity
LN | 11726-7 | Peak Systolic Velocity
LN | 20352-1 | Time averaged mean velocity
LN | 11692-1 | Time averaged peak velocity

## CID 12121 Vascular Indices and Ratios

Type:

  Extensible

Version:

  20050110

Table CID 12121. Vascular Indices and Ratios

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 20167-3 | Acceleration Index
SRT | R-101BA | Lumen Area Stenosis
SRT | R-101BB | Lumen Diameter Stenosis
LN | 12008-9 | Pulsatility Index
LN | 12023-8 | Resistivity Index
LN | 12144-2 | Systolic to Diastolic Velocity Ratio
LN | 33867-3 | Velocity ratio

### Note

This Context Group formerly included SNOMED codes G-0371 and G-0372, which have been replaced by R-101BA and R-101BB, respectively. See Annex J.

## CID 12122 Other Vascular Properties

Type:

  Extensible

Version:

  20050110

Table CID 12122. Other Vascular Properties

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 20168-1 | Acceleration Time
LN | 20217-6 | Deceleration Time
SRT | G-0364 | Vessel lumen diameter
SRT | R-1025C | Vessel Intimal Diameter
SRT | R-1025D | Vessel Intimal Cross-Sectional Area
SRT | G-0365 | Vessel outside diameter
SRT | G-0366 | Vessel lumen cross-sectional area
LN | 33878-0 | Volume flow
SRT | R-1025E | Vessel depth from surface
LN | 20247-3 | Peak Gradient
LN | 20256-4 | Mean Gradient
SRT | R-1025F | Length of Segment

## CID 12123 Carotid Ratios

Type:

  Extensible

Version:

  20030327

Table CID 12123. Carotid Ratios

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 33868-1 | ICA/CCA velocity ratio

## CID 12124 Renal Ratios

Type:

  Extensible

Version:

  20030327

Table CID 12124. Renal Ratios

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 33869-9 | Renal Artery/Aorta velocity ratio

## CID 12140 Pelvic Vasculature Anatomical Location

Type:

  Extensible

Version:

  20040322

Table CID 12140. Pelvic Vasculature Anatomical Location

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-F1810 | Umbilical Artery
SRT | T-F1820 | Umbilical Vein
SRT | T-46980 | Ovarian Artery
SRT | T-48780 | Ovarian Vein
SRT | T-46820 | Uterine Artery
SRT | T-49010 | Uterine Vein
SRT | T-F1412 | Vitelline Artery of Placenta
SRT | T-F1413 | Vitelline Vein of Placenta
SRT | T-46710 | Common Iliac Artery

## CID 12141 Fetal Vasculature Anatomical Location

Type:

  Extensible

Version:

  20040322

Table CID 12141. Fetal Vasculature Anatomical Location

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-42000 | Aorta
SRT | T-D0765 | Descending Aorta
SRT | T-45600 | Middle Cerebral Artery
SRT | T-48581 | Pulmonary Vein
SRT | T-44000 | Pulmonary Artery

## CID 12200 Echocardiography Left Ventricle

Type:

  Extensible

Version:

  20030918

Table CID 12200. Echocardiography Left Ventricle

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12201 “Left Ventricle Linear”
Include CID 12240 “Left Ventricle Area”
Include CID 12202 “Left Ventricle Volume”
Include CID 12222 “Orifice Flow Properties”
Include CID 12203 “Left Ventricle Other”
Include CID 12239 “Cardiac Output Properties”

## CID 12201 Left Ventricle Linear

Type:

  Extensible

Version:

  20030918

Table CID 12201. Left Ventricle Linear

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 29436-3 | Left Ventricle Internal End Diastolic Dimension
LN | 29438-9 | Left Ventricle Internal Systolic Dimension
LN | 18051-3 | Left Ventricular Fractional Shortening
LN | 18154-5 | Interventricular Septum Diastolic Thickness
LN | 18155-2 | Interventricular Septum to Posterior Wall Thickness Ratio
LN | 18054-7 | Interventricular Septum % Thickening
LN | 18158-6 | Interventricular Septum Systolic Thickness
LN | 18053-9 | Left Ventricle Posterior Wall % Thickening
LN | 18077-8 | Left Ventricle diastolic major axis
LN | 18076-0 | Left Ventricle systolic major axis
LN | 18156-0 | Left Ventricle Posterior Wall Systolic Thickness
LN | 18152-9 | Left Ventricle Posterior Wall Diastolic Thickness
SRT | G-0377 | Left Ventricle Semi-major Axis Diastolic Dimension
SRT | G-0378 | Left Ventricle Truncated Semi-major Axis Diastolic Dimension

## CID 12202 Left Ventricle Volume

Type:

  Extensible

Version:

  20030918

Table CID 12202. Left Ventricle Volume

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 18026-5 | Left Ventricular End Diastolic Volume
LN | 18148-7 | Left Ventricular End Systolic Volume
LN | 18043-0 | Left Ventricular Ejection Fraction by US

## CID 12203 Left Ventricle Other

Type:

  Extensible

Version:

  20030918

Table CID 12203. Left Ventricle Other

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 18087-7 | Left Ventricle Mass
LN | 18071-1 | Left Ventricular Isovolumic Relaxation Time
SRT | G-037E | Left Ventricular Isovolumic Contraction Time
SRT | G-037A | Left Ventricular Peak Early Diastolic Tissue Velocity
SRT | G-037B | Ratio of MV Peak Velocity to LV Peak Tissue Velocity E-Wave
SRT | G-037C | LV Peak Diastolic Tissue Velocity During Atrial Systole
SRT | G-037D | Left Ventricular Peak Systolic Tissue Velocity
SRT | G-037F | Left Ventricular Index of Myocardial Performance

## CID 12204 Echocardiography Right Ventricle

Type:

  Extensible

Version:

  20080623

Table CID 12204. Echocardiography Right Ventricle

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
SRT | F-04FD8 | RV Stroke Volume
SRT | F-04FA5 | RV Cardiac Output
SRT | F-04F84 | RV Cardiac Index
SRT | F-04FE5 | RV Stroke Index
LN | 20304-2 | Right Ventricular Internal Diastolic Dimension
LN | 20305-9 | Right Ventricular Internal Systolic Dimension
SRT | G-0381 | Right Ventricular Index of Myocardial Performance
SRT | G-0380 | Right Ventricular Peak Systolic Pressure
LN | 18153-7 | Right Ventricular Anterior Wall Diastolic Thickness
LN | 18157-8 | Right Ventricular Anterior Wall Systolic Thickness

## CID 12205 Echocardiography Left Atrium

Type:

  Extensible

Version:

  20030918

Table CID 12205. Echocardiography Left Atrium

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
LN | 29469-4 | Left Atrium Antero-posterior Systolic Dimension
LN | 17985-3 | Left Atrium to Aortic Root Ratio
LN | 29486-8 | Left Atrial Appendage Peak Velocity
LN | 17977-0 | Left Atrium Area A4C view
SRT | G-0383 | Left Atrium Systolic Volume

## CID 12206 Echocardiography Right Atrium

Type:

  Extensible

Version:

  20030918

Table CID 12206. Echocardiography Right Atrium

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
LN | 18070-3 | Right Atrium Systolic Pressure
LN | 17988-7 | Right Atrium Area A4C view

## CID 12207 Echocardiography Mitral Valve

Type:

  Extensible

Version:

  20030918

Table CID 12207. Echocardiography Mitral Valve

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12239 “Cardiac Output Properties”
LN | 17978-8 | Mitral Valve A-Wave Peak Velocity
LN | 18037-2 | Mitral Valve E-Wave Peak Velocity
LN | 18038-0 | Mitral Valve E to A Ratio
SRT | G-0386 | Mitral Valve AT/DT Ratio
SRT | G-0384 | Mitral Valve E-Wave Deceleration Time
LN | 18040-6 | Mitral Valve E-F Slope by M-Mode
LN | 18036-4 | Mitral Valve EPSS, E wave
SRT | G-0385 | Mitral Valve A-Wave Duration
LN | 18057-0 | Mitral Valve Diastolic Peak Instantaneous Gradient
SRT | G-0387 | Mitral Valve Closure to Opening Time
LN | 18035-6 | Mitral Regurgitation dP/dt derived from Mitral Reg. velocity

### Note

This Context Group includes measurements of the left ventricle only. For right ventricle measurements, see CID 12204 “Echocardiography Right Ventricle”.

## CID 12208 Echocardiography Tricuspid Valve

Type:

  Extensible

Version:

  20030918

Table CID 12208. Echocardiography Tricuspid Valve

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
LN | 18031-5 | Tricuspid Valve E Wave Peak Velocity
LN | 18030-7 | Tricuspid Valve A Wave Peak Velocity
LN | 18039-8 | Tricuspid Valve E to A Ratio
LN | 20296-0 | Time from Q wave to Tricuspid Valve Opens
SRT | G-0389 | Tricuspid Valve Closure to Opening Time
LN | 18034-9 | Tricuspid Regurgitation dP/dt

## CID 12209 Echocardiography Pulmonic Valve

Type:

  Extensible

Version:

  20030918

Table CID 12209. Echocardiography Pulmonic Valve

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
LN | 18096-8 | Pulmonic Valve Area by continuity
LN | 18042-2 | Pulmonic Valve Ejection Time
SRT | G-0388 | Ratio of Pulmonic Valve Acceleration Time to Ejection Time
LN | 20295-2 | Time from Q wave to Pulmonic Valve Closes

## CID 12210 Echocardiography Pulmonary Artery

Type:

  Extensible

Version:

  20030918

Table CID 12210. Echocardiography Pulmonary Artery

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
LN | 18020-8 | Main Pulmonary Artery Diameter
LN | 18021-6 | Right Pulmonary Artery Diameter
LN | 18019-0 | Left Pulmonary Artery Diameter
SRT | G-038A | Main Pulmonary Artery Peak Velocity

## CID 12211 Echocardiography Aortic Valve

Type:

  Extensible

Version:

  20030918

Table CID 12211. Echocardiography Aortic Valve

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
LN | 17996-0 | Aortic Valve Cusp Separation
LN | 18041-4 | Aortic Valve Ejection Time
SRT | G-0382 | Ratio of Aortic Valve Acceleration Time to Ejection Time

## CID 12212 Echocardiography Aorta

Type:

  Extensible

Version:

  20030918

Table CID 12212. Echocardiography Aorta

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
LN | 18015-8 | Aortic Root Diameter
LN | 18011-7 | Aortic Arch Diameter
LN | 18012-5 | Ascending Aortic Diameter
LN | 18014-1 | Aortic Isthmus Diameter
LN | 18013-3 | Descending Aortic Diameter
LN | 17995-2 | Thoracic Aorta Coarctation Systolic Peak Instantaneous Gradient
LN | 29460-3 | Thoracic Aorta Coarctation Systolic Peak Velocity

## CID 12214 Echocardiography Pulmonary Veins

Type:

  Extensible

Version:

  20030918

Table CID 12214. Echocardiography Pulmonary Veins

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
LN | 29450-4 | Pulmonary Vein Systolic Peak Velocity
LN | 29451-2 | Pulmonary Vein Diastolic Peak Velocity
LN | 29452-0 | Pulmonary Vein Systolic to Diastolic Ratio
LN | 29453-8 | Pulmonary Vein Atrial Contraction Reversal Peak Velocity
SRT | G-038B | Pulmonary Vein A-Wave Duration
SRT | G-038D | Pulmonary Vein D-Wave Velocity Time Integral
SRT | G-038C | Pulmonary Vein S-Wave Velocity Time Integral

## CID 12215 Echocardiography Vena Cavae

Type:

  Extensible

Version:

  20030918

Table CID 12215. Echocardiography Vena Cavae

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
LN | 18006-7 | Inferior Vena Cava Diameter
LN | 18050-5 | Inferior Vena Cava % Collapse

## CID 12216 Echocardiography Hepatic Veins

Type:

  Extensible

Version:

  20030918

Table CID 12216. Echocardiography Hepatic Veins

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
LN | 29471-0 | Hepatic Vein Systolic Peak Velocity
LN | 29472-8 | Hepatic Vein Diastolic Peak Velocity
LN | 29473-6 | Hepatic Vein Systolic to Diastolic Ratio
LN | 29474-4 | Hepatic Vein Atrial Contraction Reversal Peak Velocity

## CID 12217 Echocardiography Cardiac Shunt

Type:

  Extensible

Version:

  20030918

Table CID 12217. Echocardiography Cardiac Shunt

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
LN | 29462-9 | Pulmonary-to-Systemic Shunt Flow Ratio

## CID 12218 Echocardiography Congenital

Type:

  Extensible

Version:

  20100317

Table CID 12218. Echocardiography Congenital

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12250 “Cardiac Ultrasound Common Linear Measurements”

## CID 12219 Pulmonary Vein Modifiers

Type:

  Extensible

Version:

  20030918

Table CID 12219. Pulmonary Vein Modifiers

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-404A0 | Right Upper Segment
SRT | R-4049E | Right Lower Segment
SRT | R-40491 | Left Upper Segment
SRT | R-4214B | Left Lower Segment

## CID 12220 Echocardiography Common Measurements

Type:

  Extensible

Version:

  20030918

Table CID 12220. Echocardiography Common Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 8867-4 | Heart rate

## CID 12221 Flow Direction

Type:

  Extensible

Version:

  20100317

Table CID 12221. Flow Direction

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-42047 | Antegrade Flow
SRT | R-42E61 | Retrograde Flow
SRT | G-0367 | Regurgitant Flow
SRT | F-32330 | Left to right cardiovascular shunt
SRT | F-32340 | Right to left cardiovascular shunt

### Note

In a prior version of this Context Group, the code R-42E61 was specified for Regurgitant Flow. This has been corrected to be Retrograde Flow. Some applications might continue to send codeR-42E61 instead ofG-0367 for Regurgitant Flow.

## CID 12222 Orifice Flow Properties

Type:

  Extensible

Version:

  20100317

Table CID 12222. Orifice Flow Properties

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 33878-0 | Volume Flow
LN | 34141-2 | Peak Instantaneous Flow Rate
SRT | G-038E | Cardiovascular Orifice Area
SRT | G-038F | Cardiovascular Orifice Diameter
SRT | G-0390 | Regurgitant Fraction
LN | 11653-3 | End Diastolic Velocity
LN | 11726-7 | Peak Systolic Velocity
LN | 20352-1 | Time Averaged Mean Velocity
LN | 11692-1 | Time Averaged Peak Velocity
LN | 20247-3 | Peak Gradient
LN | 20256-4 | Mean Gradient
LN | 20354-7 | Velocity Time Integral
LN | 20280-4 | Pressure Half-Time
LN | 20168-1 | Acceleration Time
LN | 20217-6 | Deceleration Time
LN | 20216-8 | Deceleration Slope
LN | 12144-2 | Systolic to Diastolic Velocity Ratio
LN | 59102-4 | Flow Radius
LN | 59130-5 | Alias velocity
LN | 20167-3 | Acceleration Slope
LN | 59127-1 | D-E Slope
LN | 59128-9 | E-F Slope
LN | 59103-2 | A-C Interval
LN | 59104-0 | Peak E wave/Peak A wave by US
LN | 59106-5 | Stenosis Peak Gradient
LN | 59107-3 | Stenosis Peak Velocity
LN | 59079-4 | Peak Reversal Velocity during Atrial Contraction
LN | 59080-2 | E-Wave Peak Velocity
LN | 59081-0 | A-Wave Peak Velocity
LN | 59111-5 | E Velocity to Annulus E Velocity Ratio
LN | 59115-6 | Velocity of Flow Propagation

### Note

1. In a prior version of this Context Group, the code 11726-7 was specified for Peak Velocity. This has been corrected to be Peak Systolic Velocity. Some applications might continue to send code 11726-7 instead of 20351-3 for Peak Velocity.

2. In a prior version of this Context Group, the code 20352-1 was specified for Mean Velocity. This has been corrected to be Time Averaged Mean Velocity. Some applications might continue to send code 20352-1 instead of 11692-1 for Time Averaged Peak Velocity.

## CID 12223 Echocardiography Stroke Volume Origin

Type:

  Extensible

Version:

  20030918

Table CID 12223. Echocardiography Stroke Volume Origin

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-32600 | Left Ventricle
SRT | T-32650 | Left Ventricle Outflow Tract
SRT | T-32550 | Right Ventricle Outflow Tract
SRT | T-35300 | Mitral Valve
SRT | T-42000 | Aorta

## CID 12224 Ultrasound Image Modes

Type:

  Extensible

Version:

  20100317

Table CID 12224. Ultrasound Image Modes

Coding Scheme Designator | Code Value | Code Meaning | SNOMED-CT Concept ID Equivalent
--- | --- | --- | ---
SRT | G-03A2 | 2D mode | 399064001
SRT | R-409E2 | Doppler Color Flow | 261197005
SRT | G-0394 | M mode | 399155008
SRT | R-409E4 | Doppler Pulsed | 261199008
SRT | R-409E3 | Doppler Continuous Wave | 261198000
SRT | P0-02241 | Power Doppler | 425704008
SRT | P0-02242 | 3D mode | 426865009
SRT | P5-B0128 | Tissue Doppler Imaging | 439858009

## CID 12226 Echocardiography Image View

Type:

  Extensible

Version:

  20100317

Table CID 12226. Echocardiography Image View

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A19B | Apical two chamber
SRT | G-A19C | Apical four chamber
SRT | G-0395 | Apical long axis
SRT | G-0396 | Parasternal long axis
SRT | G-0577 | Parasternal long axis view of the RV inflow tract
SRT | G-0578 | Parasternal long axis view of the RV outflow tract
SRT | G-0397 | Parasternal short axis
SRT | G-0398 | Parasternal short axis at the aortic valve level
SRT | G-0399 | Parasternal short axis at the level of the mitral chords
SRT | G-039A | Parasternal short axis at the Mitral Valve level
SRT | G-039B | Parasternal short axis at the Papillary Muscle level
SRT | G-039C | Right Ventricular Inflow Tract View
SRT | G-039D | Right Ventricular Outflow Tract View
SRT | G-039E | Subcostal long axis
SRT | G-039F | Subcostal short axis
SRT | G-03A0 | Suprasternal long axis
SRT | G-03A1 | Suprasternal short axis
SRT | R-40B0E | Transesophageal short axis view
SRT | R-40AFF | Subcostal view of cardiac outlets directed anteriorly
SRT | G-0579 | Subcostal short axis view at papillary muscle level
SRT | G-057B | Subcostal short axis view at mitral valve level
SRT | G-057E | Subcostal short axis view at aortic valve level
SRT | G-057C | Subcostal short axis view at venous inflow level
SRT | R-40B0A | Subcostal oblique coronal view
SRT | R-40B00 | Suprasternal coronal view
SRT | R-40B01 | Suprasternal sagittal view
SRT | G-057D | Suprasternal long axis view of aortic arch

## CID 12227 Echocardiography Measurement Method

Type:

  Extensible

Version:

  20030918

Table CID 12227. Echocardiography Measurement Method

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12228 “Echocardiography Volume Methods”
Include CID 12229 “Echocardiography Area Methods”
Include CID 12230 “Gradient Methods”
Include CID 12231 “Volume Flow Methods”
Include CID 12232 “Myocardium Mass Methods”

## CID 12228 Echocardiography Volume Methods

Type:

  Extensible

Version:

  20100317

Table CID 12228. Echocardiography Volume Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125204 | Area-Length Biplane
DCM | 125205 | Area-Length Single Plane
DCM | 125211 | Biplane Ellipse
DCM | 125226 | Single Plane Ellipse
DCM | 125206 | Cube Method
DCM | 125207 | Method of Disks, Biplane
DCM | 125208 | Method of Disks, Single Plane
DCM | 125209 | Teichholz
DCM | 125227 | Modified Simpson
DCM | 125228 | Bullet Method

## CID 12229 Echocardiography Area Methods

Type:

  Extensible

Version:

  20030918

Table CID 12229. Echocardiography Area Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125210 | Area by Pressure Half-Time
DCM | 125212 | Continuity Equation
DCM | 125213 | Continuity Equation by Mean Velocity
DCM | 125214 | Continuity Equation by Peak Velocity
DCM | 125215 | Continuity Equation by Velocity Time Integral
DCM | 125216 | Proximal Isovelocity Surface Area
DCM | 125220 | Planimetry

## CID 12230 Gradient Methods

Type:

  Extensible

Version:

  20030918

Table CID 12230. Gradient Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125217 | Full Bernoulli
DCM | 125218 | Simplified Bernoulli

## CID 12231 Volume Flow Methods

Type:

  Extensible

Version:

  20030918

Table CID 12231. Volume Flow Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125219 | Doppler Volume Flow
DCM | 125216 | Proximal Isovelocity Surface Area

## CID 12232 Myocardium Mass Methods

Type:

  Extensible

Version:

  20100317

Table CID 12232. Myocardium Mass Methods

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125221 | Left Ventricle Mass by M-mode
DCM | 125222 | Left Ventricle Mass by Truncated Ellipse
DCM | 125270 | Left Ventricle Mass by Area Length
DCM | 125271 | Left Ventricle Mass by M-mode - adjusted by Height
DCM | 125272 | Left Ventricle Mass by Truncated Ellipse - adjusted by Height
DCM | 125273 | Left Ventricle Mass by Area Length - adjusted by Height

## CID 12233 Cardiac Phase

Type:

  Extensible

Version:

  20100317

Table CID 12233. Cardiac Phase

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-32020 | Systole
SRT | F-32010 | Diastole
SRT | F-32011 | End Diastole
SRT | R-FAB5B | End Systole
SRT | R-40B1B | Early Diastole
SRT | F-32021 | Peak Systolic
SRT | F-32030 | Atrial Systole
SRT | F-32040 | Ventricular Systole
SRT | R-40B12 | Ventricular Isovolumic Contraction
SRT | R-40B11 | Ventricular Ejection
SRT | R-40B10 | Ventricular Isovolumic Relaxation
SRT | R-40B1C | Diastolic Rapid Inflow
SRT | R-40B21 | Diastasis

## CID 12234 Respiration State

Type:

  Extensible

Version:

  20030918

Table CID 12234. Respiration State

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-20010 | Inspiration
SRT | F-20020 | Expiration

## CID 12235 Mitral Valve Anatomic Sites

Type:

  Extensible

Version:

  20040614

Table CID 12235. Mitral Valve Anatomic Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-0391 | Medial Mitral Annulus
SRT | G-0392 | Lateral Mitral Annulus
SRT | T-35313 | Mitral Annulus

## CID 12236 Echo Anatomic Sites

Type:

  Extensible

Version:

  20110818

Table CID 12236. Echo Anatomic Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12235 “Mitral Valve Anatomic Sites”
Include CID 12223 “Echocardiography Stroke Volume Origin”
Include CID 12241 “Tricuspid Valve Finding Sites”
Include CID 12242 “Aortic Valve Finding Sites”
Include CID 12243 “Left Ventricle Finding Sites”
Include CID 12244 “Congenital Finding Sites”
SRT | D4-32030 | Thoracic Aortic Coarctation
SRT | D3-90008 | Pericardial effusion

## CID 12237 Echocardiography Anatomic Site Modifiers

Type: Extensible Version: 20030918

Type:

  Extensible

Version:

  20030918

Table CID 12237. Echocardiography Anatomic Site Modifiers

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12219 “Pulmonary Vein Modifiers”

## CID 12238 Wall Motion Scoring Schemes

Type:

  Extensible

Version:

  20050321

Table CID 12238. Wall Motion Scoring Schemes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125223 | 4 Point Segment Finding Scale
DCM | 125224 | 5 Point Segment Finding Scale
DCM | 125225 | 5 Point Segment Finding Scale With Graded Hypokinesis

## CID 12239 Cardiac Output Properties

Type:

  Extensible

Version:

  20030918

Table CID 12239. Cardiac Output Properties

Coding Scheme Designator | Code Value | Code Meaning | Equivalent LOINC Code Value
--- | --- | --- | ---
SRT | F-32120 | Stroke Volume | 20562-5
SRT | F-32100 | Cardiac Output | 8741-1
SRT | F-32110 | Cardiac Index
SRT | F-00078 | Stroke Index

## CID 12240 Left Ventricle Area

Type:

  Extensible

Version:

  20030918

Table CID 12240. Left Ventricle Area

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-0374 | Left Ventricular Systolic Area
SRT | G-0375 | Left Ventricular Diastolic Area
SRT | G-0376 | Left Ventricular Fractional Area Change
SRT | G-0379 | Left Ventricle Epicardial Diastolic Area, psax pap view

## CID 12241 Tricuspid Valve Finding Sites

Type:

  Extensible

Version:

  20040614

Table CID 12241. Tricuspid Valve Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-35111 | Tricuspid Annulus

## CID 12242 Aortic Valve Finding Sites

Type:

  Extensible

Version:

  20040614

Table CID 12242. Aortic Valve Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-35410 | Aortic Valve Ring

## CID 12243 Left Ventricle Finding Sites

Type:

  Extensible

Version:

  20040614

Table CID 12243. Left Ventricle Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-32650 | Left Ventricle Outflow Tract

## CID 12244 Congenital Finding Sites

Type:

  Extensible

Version:

  20040614

Table CID 12244. Congenital Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D4-31150 | Ventricular Septal Defect
SRT | D4-31220 | Atrial Septal Defect

## CID 12245 Cardiac Ultrasound Report Titles

Type:

  Extensible

Version:

  20100317

Table CID 12245. Cardiac Ultrasound Report Titles

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 125195 | Pediatric Cardiac Ultrasound Report
DCM | 125196 | Fetal Cardiac Ultrasound Report
DCM | 125197 | Adult Congenital Cardiac Ultrasound Report

## CID 12246 Cardiac Ultrasound Indication for Study

Type:

  Extensible

Version:

  20100317

Table CID 12246. Cardiac Ultrasound Indication for Study

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-0A44A | Fever
SRT | F-24210 | Hemoptysis
SRT | R-00302 | Murmur
SRT | D4-31000 | Congenital heart disease
SRT | F-37000 | Chest Pain
SRT | D3-13040 | Coronary Artery Disease
SRT | F-03C97 | Heart disease risk factors
SRT | F-201B3 | Dyspnea
SRT | F-38002 | Abnormal ECG
SRT | D3-30000 | Arrhythmia
SRT | D3-13012 | Angina pectoris
SRT | D3-02000 | Hypertension
SRT | F-37150 | Palpitations
SRT | D3-31290 | Supraventricular tachycardia
SRT | D3-00006 | Syncope
SRT | D3-33120 | Left bundle branch block
SRT | D3-10800 | Valvular heart disease
SRT | P0-05DA0 | Imaging guidance

## CID 12247 Pediatric, Fetal and Congenital Cardiac Surgical Interventions

Type:

  Extensible

Version:

  20100317

Table CID 12247. Pediatric, Fetal and Congenital Cardiac Surgical Interventions

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | P1-31919 | Arterial switch operation
SRT | P1-31018 | Implantation of baffle, atrial or interatrial
SRT | P1-31872 | Atrial septal defect repair
SRT | P1-31846 | Percutaneous prosthetic closure of atrial septal defect
SRT | P1-31037 | Repair of defect of the atrioventricular septum
SRT | P1-36957 | Blalock-Taussig shunt, pulmonary-subclavian artery anastomosis
SRT | P1-36956 | Central aortopulmonary shunt operation
SRT | P1-34001 | Repair of coarctation of aorta
SRT | P5-39106 | Coarctation of the Aorta Balloon Angioplasty
SRT | P0-06135 | Coarctation of the Aorta Angioplasty with Implant of Stent
SRT | P1-31088 | Damus-Stansel-Kaye operation
SRT | P1-31926 | Creation of conduit of right atrium and pulmonary artery
SRT | P1-36993 | Lateral-Caval Fontan procedure
SRT | P1-3696A | Hemi-Fontan operation
SRT | P1-36997 | Left Glenn shunt procedure
SRT | P1-36994 | Left-sided bidirectional Glenn shunt procedure
SRT | P1-31917 | Mustard operation
SRT | P1-31089 | Norwood type operation
SRT | P0-057E8 | Closure of ductus arteriosus with clip
SRT | P0-00E0B | Patent ductus arteriosus coil or device closure
SRT | P1-38803 | Partial anomalous pulmonary venous connection operation
SRT | P1-31920 | Rastelli operation
SRT | P1-36995 | Right Glenn shunt procedure
SRT | P1-36996 | Right-sided bidirectional Glenn shunt procedure
SRT | P0-00C6B | Construction of LV to aorta tunnel w RV to PA valved conduit
SRT | P1-30A31 | Radical aortopulmonary reconstruct w RV to PA valveless conduit
SRT | P1-3180D | Sano procedure
SRT | P1-31003 | Atrial inversion operation using atrial wall
SRT | P0-0530F | Repair of total anomalous pulmonary venous connection
SRT | P1-32504 | Implantation of heart valve prosthesis or synthetic device
SRT | P1-32502 | Implantation of heart valve with tissue graft
SRT | P1-31876 | Correction of ventricular septal defect
SRT | P1-31850 | Ventricular septal defect device closure

## CID 12248 Cardiac Ultrasound Summary Codes

Type:

  Extensible

Version:

  20100317

Table CID 12248. Cardiac Ultrasound Summary Codes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D4-31810 | Congenital stenosis of aortic valve
SRT | D4-31220 | Atrial Septal Defect
SRT | D3-29022 | Aortic regurgitation
SRT | D3-29021 | Aortic stenosis
SRT | D3-10008 | Cardiomegaly
SRT | D4-32014 | Coarctation of the Aorta
SRT | D4-31303 | Common atrioventricular canal
SRT | D4-31010 | Complete transposition of great vessels
SRT | M-04100 | Cyanosis
SRT | D4-31B16 | Dextrocardia
SRT | D3-83001 | Interrupted Aortic Arch
SRT | D4-31B24 | Mesocardia
SRT | D3-81660 | Acute febrile mucocutaneous lymph node syndrome
SRT | D3-29013 | Mitral valve prolapse
SRT | D3-29012 | Mitral regurgitation
SRT | D3-29011 | Mitral stenosis
SRT | D4-33622 | Partial anomalous pulmonary venous connection
SRT | D4-31310 | Atrial septal defect with endocardial cushion defect, partial type
SRT | D3-29051 | Pulmonic valve stenosis
SRT | D3-17100 | Rheumatic Fever
SRT | D4-31110 | Tetralogy of Fallot
SRT | D4-31159 | Ventricular Septal Defect (VSD)
SRT | D4-31040 | Corrected transposition of great vessels
SRT | D3-29082 | Pulmonary atresia with intact ventricular septum
SRT | D4-31611 | Pulmonary atresia with ventricular septal defect
SRT | D4-31A00 | Hypoplastic left heart syndrome
SRT | D4-31125 | Functional Single Ventricle

## CID 12249 Cardiac Ultrasound Fetal Summary Codes

Type:

  Extensible

Version:

  20100317

Table CID 12249. Cardiac Ultrasound Fetal Summary Codes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12248 “Cardiac Ultrasound Summary Codes”
SRT | F-0518A | Edema of fetal scalp
SRT | F-8612F | Edema of fetal chest wall

## CID 12250 Cardiac Ultrasound Common Linear Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12250. Cardiac Ultrasound Common Linear Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A22A | Length
SRT | M-02550 | Diameter
SRT | G-A193 | Major Axis
SRT | G-A194 | Minor Axis
SRT | M-02560 | Circumference
SRT | G-A196 | Radius
LN | 59089-3 | ROI Thickness by US
LN | 59090-1 | ROI Internal Dimension by US

## CID 12251 Cardiac Ultrasound Linear Valve Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12251. Cardiac Ultrasound Linear Valve Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12250 “Cardiac Ultrasound Common Linear Measurements”
LN | 59091-9 | D-E Excursion
LN | 59109-9 | Leaflet Separation
LN | 59110-7 | Leaflet Thickness
LN | 59122-2 | C-E Distance

## CID 12252 Cardiac Ultrasound Cardiac Function

Type:

  Extensible

Version:

  20100317

Table CID 12252. Cardiac Ultrasound Cardiac Function

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | F-32070 | Cardiac ejection fraction
LN | 59117-2 | Mean Velocity of Circumferential Fiber Shortening (Mean VcFv)
LN | 59118-0 | HR-Corrected Mean Velocity of Circumferential Fiber Shortening
LN | 59092-7 | % Thickening
LN | 59132-1 | Fractional Shortening

## CID 12253 Cardiac Ultrasound Area Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12253. Cardiac Ultrasound Area Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
LN | 20226-7 | Flow Area
SRT | G-A166 | Area
LN | 59123-0 | Jet Area

## CID 12254 Cardiac Ultrasound Hemodynamic Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12254. Cardiac Ultrasound Hemodynamic Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122182 | R-R interval
DCM | 109072 | Tau
DCM | 109071 | Indicator mean transit time
LN | 59082-8 | Closure to Opening Time
LN | 59083-6 | Isovolumic Relaxation Time
LN | 59084-4 | Isovolumic Contraction Time
LN | 20222-6 | Ejection Time
LN | 59085-1 | Pre-Ejection Period
LN | 59119-8 | Filling Time
SRT | F-31000 | Blood Pressure
LN | 59086-9 | Heart Rate-Corrected Ejection Time
LN | 59087-7 | Heart Rate-Corrected Pre-Ejection Period
LN | 59105-7 | A-Wave Duration
LN | 59088-5 | Pre-Ejection Period/Ejection Time Ratio
LN | 59108-1 | Envelope Duration
LN | 59121-4 | Time to Peak by US
LN | 59120-6 | dP/dt by US

## CID 12255 Cardiac Ultrasound Myocardium Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12255. Cardiac Ultrasound Myocardium Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
DCM | 122447 | Wall Mass
LN | 59099-2 | Myocardial Performance Index (Tei)
LN | 59094-3 | Endocardial Area
LN | 59093-5 | Epicardial Area

## CID 12257 Cardiac Ultrasound Left Ventricle

Type:

  Extensible

Version:

  20100317

Table CID 12257. Cardiac Ultrasound Left Ventricle

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12239 “Cardiac Output Properties”
Include CID 12250 “Cardiac Ultrasound Common Linear Measurements”
Include CID 12252 “Cardiac Ultrasound Cardiac Function”
Include CID 12253 “Cardiac Ultrasound Area Measurements”
Include CID 12254 “Cardiac Ultrasound Hemodynamic Measurements”
Include CID 12255 “Cardiac Ultrasound Myocardium Measurements”
SRT | G-D705 | Volume
LN | 18155-2 | Interventricular Septum to Posterior Wall Thickness Ratio
SRT | G-037B | Ratio of MV Peak Velocity to LV Peak Tissue Velocity E-Wave
LN | 59097-6 | Left Ventricle Meridional Wall Stress
LN | 59095-0 | Time to Left Ventricle S Tissue Velocity
LN | 59096-8 | Time to Left Ventricle E Tissue Velocity
LN | 59124-8 | Tissue Velocity Time Integral (VTI) for the area under Left Ventricle E wave
LN | 59125-5 | Tissue Velocity Time Integral (VTI) for the area under Left Ventricle A wave
LN | 59129-7 | Left Ventricle E to A Tissue Velocity Ratio
LN | 59133-9 | Peak Tissue Velocity

## CID 12258 Cardiac Ultrasound Right Ventricle

Type:

  Extensible

Version:

  20100317

Table CID 12258. Cardiac Ultrasound Right Ventricle

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12250 “Cardiac Ultrasound Common Linear Measurements”
Include CID 12253 “Cardiac Ultrasound Area Measurements”
Include CID 12254 “Cardiac Ultrasound Hemodynamic Measurements”
Include CID 12255 “Cardiac Ultrasound Myocardium Measurements”
SRT | G-D705 | Volume

## CID 12259 Cardiac Ultrasound Ventricles Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12259. Cardiac Ultrasound Ventricles Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12257 “Cardiac Ultrasound Left Ventricle”
Include CID 12258 “Cardiac Ultrasound Right Ventricle”

## CID 12260 Cardiac Ultrasound Pulmonary Artery

Type:

  Extensible

Version:

  20100317

Table CID 12260. Cardiac Ultrasound Pulmonary Artery

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12250 “Cardiac Ultrasound Common Linear Measurements”
Include CID 12253 “Cardiac Ultrasound Area Measurements”
Include CID 12254 “Cardiac Ultrasound Hemodynamic Measurements”

## CID 12261 Cardiac Ultrasound Pulmonary Vein

Type:

  Extensible

Version:

  20100317

Table CID 12261. Cardiac Ultrasound Pulmonary Vein

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12254 “Cardiac Ultrasound Hemodynamic Measurements”
Include CID 3612 “Blood Velocity Measurements”
LN | 59112-3 | Pulmonary Vein A Duration Mitral Valve A Duration Ratio
LN | 59113-1 | Pulmonary Vein A VTI to Mitral Valve VTI Ratio
LN | 59114-9 | Pulm Vein A duration to MV A duration difference

## CID 12262 Cardiac Ultrasound Pulmonary Valve

Type:

  Extensible

Version:

  20100317

Table CID 12262. Cardiac Ultrasound Pulmonary Valve

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12239 “Cardiac Output Properties”
Include CID 12251 “Cardiac Ultrasound Linear Valve Measurements”
Include CID 12252 “Cardiac Ultrasound Cardiac Function”
Include CID 12253 “Cardiac Ultrasound Area Measurements”
Include CID 12254 “Cardiac Ultrasound Hemodynamic Measurements”
LN | 59101-6 | Pulmonary Artery Pressure using Accel Time
LN | 20295-2 | Time from Q wave to Pulmonic Valve Closes
LN | 59100-8 | A-Wave Amplitude
LN | 59126-3 | B-C Slope

## CID 12263 Cardiac Ultrasound Venous Return Pulmonary Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12263. Cardiac Ultrasound Venous Return Pulmonary Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12261 “Cardiac Ultrasound Pulmonary Vein”
Include CID 12262 “Cardiac Ultrasound Pulmonary Valve”

## CID 12264 Cardiac Ultrasound Venous Return Systemic Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12264. Cardiac Ultrasound Venous Return Systemic Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12239 “Cardiac Output Properties”
Include CID 12250 “Cardiac Ultrasound Common Linear Measurements”
Include CID 12252 “Cardiac Ultrasound Cardiac Function”
Include CID 12253 “Cardiac Ultrasound Area Measurements”
Include CID 12254 “Cardiac Ultrasound Hemodynamic Measurements”
Include CID 3612 “Blood Velocity Measurements”

## CID 12265 Cardiac Ultrasound Atria and Atrial Septum Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12265. Cardiac Ultrasound Atria and Atrial Septum Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12250 “Cardiac Ultrasound Common Linear Measurements”
Include CID 12253 “Cardiac Ultrasound Area Measurements”
Include CID 12255 “Cardiac Ultrasound Myocardium Measurements”
SRT | G-D705 | Volume
LN | 17985-3 | Left Atrium to Aortic Root Ratio
LN | 59131-3 | Left Atrium Volume to Right Atrium Volume Ratio

## CID 12266 Cardiac Ultrasound Mitral Valve

Type:

  Extensible

Version:

  20100317

Table CID 12266. Cardiac Ultrasound Mitral Valve

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12239 “Cardiac Output Properties”
Include CID 12251 “Cardiac Ultrasound Linear Valve Measurements”
Include CID 12252 “Cardiac Ultrasound Cardiac Function”
Include CID 12253 “Cardiac Ultrasound Area Measurements”
Include CID 12254 “Cardiac Ultrasound Hemodynamic Measurements”
Include CID 3612 “Blood Velocity Measurements”
LN | 18035-6 | Mitral Regurgitation dP/dt derived from Mitral Reg. velocity
LN | 59098-4 | Mitral Valve E-septal Separation
LN | 18036-4 | Mitral Valve EPSS, E wave

## CID 12267 Cardiac Ultrasound Tricuspid Valve

Type:

  Extensible

Version:

  20100317

Table CID 12267. Cardiac Ultrasound Tricuspid Valve

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12239 “Cardiac Output Properties”
Include CID 12251 “Cardiac Ultrasound Linear Valve Measurements”
Include CID 12252 “Cardiac Ultrasound Cardiac Function”
Include CID 12253 “Cardiac Ultrasound Area Measurements”
Include CID 12254 “Cardiac Ultrasound Hemodynamic Measurements”
Include CID 3612 “Blood Velocity Measurements”
SRT | R-003A9 | Tricuspid Diastolic Filling Period (DFPt)
LN | 20296-0 | Time from Q wave to Tricuspid Valve Opens

## CID 12268 Cardiac Ultrasound Atrioventricular Valves Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12268. Cardiac Ultrasound Atrioventricular Valves Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12266 “Cardiac Ultrasound Mitral Valve”
Include CID 12267 “Cardiac Ultrasound Tricuspid Valve”

## CID 12269 Cardiac Ultrasound Interventricular Septum Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12269. Cardiac Ultrasound Interventricular Septum Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12250 “Cardiac Ultrasound Common Linear Measurements”
Include CID 12253 “Cardiac Ultrasound Area Measurements”

## CID 12270 Cardiac Ultrasound Aortic Valve

Type:

  Extensible

Version:

  20100317

Table CID 12270. Cardiac Ultrasound Aortic Valve

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12239 “Cardiac Output Properties”
Include CID 12250 “Cardiac Ultrasound Common Linear Measurements”
Include CID 12252 “Cardiac Ultrasound Cardiac Function”
Include CID 12253 “Cardiac Ultrasound Area Measurements”
Include CID 12254 “Cardiac Ultrasound Hemodynamic Measurements”
LN | 17996-0 | Aortic Valve Cusp Separation

## CID 12271 Cardiac Ultrasound Outflow Tracts Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12271. Cardiac Ultrasound Outflow Tracts Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12257 “Cardiac Ultrasound Left Ventricle”
Include CID 12258 “Cardiac Ultrasound Right Ventricle”
Include CID 12262 “Cardiac Ultrasound Pulmonary Valve”
Include CID 12270 “Cardiac Ultrasound Aortic Valve”

## CID 12272 Cardiac Ultrasound Semilunar Valves, Annulate and Sinuses Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12272. Cardiac Ultrasound Semilunar Valves, Annulate and Sinuses Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12262 “Cardiac Ultrasound Pulmonary Valve”
Include CID 12270 “Cardiac Ultrasound Aortic Valve”

## CID 12273 Cardiac Ultrasound Aortic Sinotubular Junction

Type:

  Extensible

Version:

  20100317

Table CID 12273. Cardiac Ultrasound Aortic Sinotubular Junction

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12250 “Cardiac Ultrasound Common Linear Measurements”
LN | 59116-4 | Aortic Sinotubular Junction to Aortic Root Ratio

## CID 12274 Cardiac Ultrasound Aorta Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12274. Cardiac Ultrasound Aorta Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12250 “Cardiac Ultrasound Common Linear Measurements”
Include CID 12270 “Cardiac Ultrasound Aortic Valve”
Include CID 12273 “Cardiac Ultrasound Aortic Sinotubular Junction”

## CID 12275 Cardiac Ultrasound Coronary Arteries Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12275. Cardiac Ultrasound Coronary Arteries Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 12250 “Cardiac Ultrasound Common Linear Measurements”

## CID 12276 Cardiac Ultrasound Aorto Pulmonary Connections Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12276. Cardiac Ultrasound Aorto Pulmonary Connections Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12220 “Echocardiography Common Measurements”
Include CID 12222 “Orifice Flow Properties”
Include CID 3612 “Blood Velocity Measurements”
LN | 29462-9 | Pulmonary-to-Systemic Shunt Flow Ratio

## CID 12277 Cardiac Ultrasound Pericardium and Pleura Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12277. Cardiac Ultrasound Pericardium and Pleura Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12250 “Cardiac Ultrasound Common Linear Measurements”

## CID 12279 Cardiac Ultrasound Fetal General Measurements

Type:

  Extensible

Version:

  20100317

Table CID 12279. Cardiac Ultrasound Fetal General Measurements

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
Include CID 12004 “Fetal Biometry Ratios”
LN | 11988-3 | Thoracic Circumference
LN | 33068-8 | Thoracic Area
LN | 59073-7 | Cardiac Circumference, transverse by US
LN | 59074-5 | Cardiothoracic Circumference Ratio
LN | 59075-2 | Cardiac Cross-sectional Area, transverse by US
LN | 59076-0 | Cardiothoracic Area Ratio
LN | 11820-8 | Biparietal Diameter
LN | 33069-6 | Nuchal Translucency
LN | 11963-6 | Femur Length
LN | 11979-2 | Abdominal Circumference
LN | 11818-2 | Anterior-Posterior Abdominal Diameter
LN | 11819-0 | Anterior-Posterior Trunk Diameter
LN | 11824-0 | BPD area corrected
LN | 11860-4 | Cisterna Magna
LN | 11984-2 | Head Circumference
LN | 11851-3 | Occipital-Frontal Diameter
LN | 11862-0 | Tranverse Abdominal Diameter
LN | 11863-8 | Trans Cerebellar Diameter
LN | 11864-6 | Transverse Thoracic Diameter
LN | 59077-8 | Foramen Ovale Diameter/Aortic Root Diameter
LN | 59078-6 | Left Ventricle/Right Ventricle Diameter Ratio
SRT | F-00AA0 | Number of umbilical arteries

## CID 12280 Cardiac Ultrasound Target Sites

Type:

  Extensible

Version:

  20100317

Table CID 12280. Cardiac Ultrasound Target Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | R-4210B | Anterior Wall
SRT | R-42175 | Posterior Wall
SRT | T-3500E | Cardiac valve annulus
SRT | T-42340 | Preductal region of aortic arch
SRT | T-42350 | Postductal region of aortic arch
SRT | R-421AA | Vena Contracta
SRT | T-42304 | Transverse Aortic Arch
SRT | M-36700 | Effusion
SRT | T-35002 | Cardiac Valve Leaflet
SRT | T-44200 | Right Pulmonary Artery
SRT | T-44400 | Left Pulmonary Artery
SRT | T-32550 | Right Ventricle Outflow Tract
SRT | T-32650 | Left Ventricle Outflow Tract
SRT | T-48510 | Right Superior Pulmonary Vein
SRT | T-48520 | Right Inferior Pulmonary Vein
SRT | T-48530 | Left Superior Pulmonary Vein
SRT | T-48540 | Left Inferior Pulmonary Vein
SRT | T-48505 | Pulmonary Vein Right Middle Segment
SRT | T-F6859 | Pulmonary Vein Common Left Segment
SRT | T-F6858 | Pulmonary Vein Common Right Segment
SRT | M-2460C | Pulmonary Vein confluence to Atrium Connection

## CID 12281 Cardiac Ultrasound Target Site Modifiers

Type:

  Extensible

Version:

  20100317

Table CID 12281. Cardiac Ultrasound Target Site Modifiers

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | G-A104 | Lateral
SRT | G-A109 | Medial
SRT | G-A105 | Anterior
SRT | G-A106 | Posterior
SRT | G-A115 | Inferior
SRT | G-A119 | Distal
SRT | G-A118 | Proximal
SRT | G-A109 | Middle

## CID 12282 Cardiac Ultrasound Venous Return Systemic Finding Sites

Type:

  Extensible

Version:

  20100317

Table CID 12282. Cardiac Ultrasound Venous Return Systemic Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-48610 | Superior Vena Cava
SRT | M-2460D | Right Superior Vena Cava
SRT | T-48611 | Left Superior Vena Cava
SRT | T-48710 | Inferior Vena Cava
SRT | T-48720 | Hepatic Vein
SRT | T-D087E | Hemi-Fontan Pathway
SRT | T-D0884 | Glenn Pathway
SRT | T-D087C | Fontan Pathway
SRT | T-D087D | Fontan Inferior Vena Cava Pathway
SRT | T-D0882 | Fontan Fenestration
SRT | T-D0880 | Fontan Pulmonary Artery Connection
SRT | DD-66228 | Fontan Baffle Leak
SRT | T-D0887 | Mustard or Senning Superior Vena Cava Pathway
SRT | T-D0888 | Mustard or Senning Inferior Vena Cava Pathway
SRT | T-D0885 | Mustard or Senning Common Systemic Venous Pathway

## CID 12283 Cardiac Ultrasound Venous Return Pulmonary Finding Sites

Type:

  Extensible

Version:

  20100317

Table CID 12283. Cardiac Ultrasound Venous Return Pulmonary Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-48581 | Pulmonary Vein
SRT | T-4858F | Pulmonary Vein Great Vessel
SRT | M-20103 | Cor Triatriatum Orifice
SRT | T-D087B | Pulmonary Vein to Atrium Connection
SRT | T-D0886 | Mustard or Senning Pulmonary Venous Pathway

## CID 12284 Cardiac Ultrasound Atria and Atrial Septum Finding Sites

Type:

  Extensible

Version:

  20100317

Table CID 12284. Cardiac Ultrasound Atria and Atrial Septum Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-32200 | Right Atrium
SRT | T-32300 | Left Atrium
SRT | D4-31320 | Common Atrium
SRT | T-32310 | Left Auricular Appendage
SRT | T-32210 | Right Auricular Appendage
SRT | T-32150 | Interatrial Septum Structure
SRT | D4-31220 | Atrial Septal Defect
SRT | T-32156 | Limbus of Fossa Ovalis
SRT | D4-31012 | Patent Foramen Ovale
SRT | T-D0882 | Fontan Fenestration
SRT | DD-66228 | Fontan Baffle Leak

## CID 12285 Cardiac Ultrasound Atrioventricular Valves Finding Sites

Type:

  Extensible

Version:

  20100317

Table CID 12285. Cardiac Ultrasound Atrioventricular Valves Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-35300 | Mitral Valve
SRT | T-35100 | Tricuspid Valve
SRT | T-35008 | Common (non-mitral, non-tricuspid) Atrioventricular Valve Structure

## CID 12286 Cardiac Ultrasound Interventricular Septum Finding Sites

Type:

  Extensible

Version:

  20100317

Table CID 12286. Cardiac Ultrasound Interventricular Septum Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-32410 | Interventricular Septum
SRT | D4-31150 | Ventricular Septal Defect
SRT | M-20102 | Bulboventricular Foramen

## CID 12287 Cardiac Ultrasound Ventricles Finding Sites

Type:

  Extensible

Version:

  20100317

Table CID 12287. Cardiac Ultrasound Ventricles Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-32600 | Left Ventricle
SRT | T-32500 | Right Ventricle
SRT | D4-31120 | Common Ventricle

## CID 12288 Cardiac Ultrasound Outflow Tracts Finding Sites

Type:

  Extensible

Version:

  20100317

Table CID 12288. Cardiac Ultrasound Outflow Tracts Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-32600 | Left Ventricle
SRT | T-32500 | Right Ventricle
SRT | T-D0879 | Rastelli Interventricular Tunnel
SRT | T-D087F | Right Ventricle to Pulmonary Artery Conduit Anastomosis
SRT | T-D0881 | Left Ventricle to Pulmonary Artery Conduit Anastomosis
SRT | T-35400 | Aortic Valve
SRT | T-35200 | Pulmonic Valve

## CID 12289 Cardiac Ultrasound Semilunar Valves, Annulate and Sinuses Finding Sites

Type:

  Extensible

Version:

  20100317

Table CID 12289. Cardiac Ultrasound Semilunar Valves, Annulus and Sinuses Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-35400 | Aortic Valve
SRT | T-42110 | Root of Aorta
SRT | T-35200 | Pulmonic Valve
SRT | T-35014 | Truncal Valve Structure
SRT | T-D087A | Neoaortic Valve
SRT | T-D0883 | Neoaortic Root

## CID 12290 Cardiac Ultrasound Pulmonary Arteries Finding Sites

Type:

  Extensible

Version:

  20100317

Table CID 12290. Cardiac Ultrasound Pulmonary Arteries Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-44100 | Pulmonary Trunk
SRT | T-44000 | Pulmonary Artery
SRT | T-D0877 | Aorta to Pulmonary Artery Connection

## CID 12291 Cardiac Ultrasound Aorta Finding Sites

Type:

  Extensible

Version:

  20100317

Table CID 12291. Cardiac Ultrasound Aorta and Coronary Arteries Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-42110 | Root of Aorta
SRT | T-42200 | Structure Sinus of Valsalva
SRT | T-42220 | Left Sinus of Valsalva
SRT | T-42210 | Right Sinus of Valsalva
SRT | T-42231 | Non-coronary Sinus
SRT | T-42102 | Aortic Sinotubular Junction
SRT | T-42100 | Ascending Aorta
SRT | T-42300 | Aortic Arch
SRT | T-42310 | Aortic Isthmus
SRT | D4-32014 | Coarctation of Aorta
SRT | T-42070 | Thoracic Aorta
SRT | T-42500 | Abdominal Aorta
SRT | T-42510 | Supra Renal Aorta
SRT | T-42520 | Infra-Renal Aorta
SRT | T-46010 | Innominate Artery
SRT | T-45110 | Right Common Carotid Artery
SRT | T-46110 | Right Subclavian Artery
SRT | T-45120 | Left Common Carotid Artery
SRT | T-46120 | Left Subclavian Artery

## CID 12292 Cardiac Ultrasound Coronary Arteries Finding Sites

Type:

  Extensible

Version:

  20100317

Table CID 12292. Cardiac Ultrasound Aorta and Coronary Arteries Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-43107 | Left Main Coronary Artery
SRT | T-43120 | Circumflex Coronary Artery
SRT | T-43121 | Proximal Circumflex Coronary Artery
SRT | T-43127 | Mid Circumflex Coronary Artery
SRT | T-43122 | Distal Circumflex Coronary Artery
SRT | T-43110 | Anterior Descending Branch of Left Coronary Artery
SRT | T-43002 | Septal Artery
SRT | T-43203 | Right Coronary Artery
SRT | T-D0878 | Posterior Descending Coronary Artery

## CID 12293 Cardiac Ultrasound Aortopulmonary Connections Finding Sites

Type:

  Extensible

Version:

  20100317

Table CID 12293. Cardiac Ultrasound Aorto Pulmonary Connections Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | D4-32012 | Patent Ductus Arteriosus
SRT | T-D0877 | Aorta to Pulmonary Artery Connection
SRT | D1-50666 | Arteriovenous Fistula

## CID 12294 Cardiac Ultrasound Pericardium and Pleura Finding Sites

Type:

  Extensible

Version:

  20100317

Table CID 12294. Cardiac Ultrasound Pericardium and Pleura Finding Sites

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
SRT | T-39000 | Pericardium
SRT | T-29000 | Pleura
SRT | T-29200 | Left Pleura
SRT | T-29100 | Right Pleura

# C Acquisition and Protocol Context Templates (Normative)

This Annex specifies the content of Templates for Acquisition and Protocol Context required by DICOM IODs.

## Templates for Acquisition and Protocol Context

### TID 3401 ECG Acquisition Context

Type:

  Extensible

Order:

  Non-Significant

Table TID 3401. ECG Acquisition Context

| VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | ---
1 | CODE | DT (10:11345, MDC, "Lead System") | 1 | U |  | BCID 3263 “Electrode Placement Values”
2 | CODE | DT (109054, DCM, "Patient State") | 1 | U |  | BCID 3262 “ECG Patient State Values”
3 | NUMERIC | DT (109055, DCM, "Protocol Stage") | 1 | U |  | UNITS = EV ({stage}, UCUM, "stage")
4 | CODE | DT (109056, DCM, "Stress Protocol") | 1 | U |  | BCID 3261 “Stress Protocols”
5 | NUMERIC | DCID 3690 “ECG Control Variables Numeric” | 1-n | U |  |
6 | TEXT | DCID 3691 “ECG Control Variables Text” | 1-n | U |  |

### TID 3403 Catheterization Acquisition Context

Type:

  Extensible

Order:

  Non-Significant

Table TID 3403. Catheterization Acquisition Context

| VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | ---
1 | CODE | EV (G-72BB, SRT, "Catheterization Procedure Phase") | 1 | U |  | BCID 3250 “Catheterization Procedure Phase”
2 | CODE | EV (109058, DCM, "Contrast Phase") | 1 | U |  | BCID 3600 “Relative Times”
3 | CODE | EV (109059, DCM, "Physiological challenges") | 1 | U |  | BCID 3271 “Hemodynamic Physiological Challenges”
4 | NUMERIC | EV (109060, DCM, "Procedure Step Number") | 1 | U |  | UNITS = EV ({step}, UCUM, "step")
5 | TEXT | EV (121124, DCM, "Procedure Action ID") | 1 | U |  |

### Note

See TID 3100 “Procedure Action” in Annex A for description of Procedure Action ID used in Row 5.

### TID 3450 Cardiac Electrophysiology Acquisition Context

Type:

  Extensible

Order:

  Non-Significant

Table TID 3450. Cardiac Electrophysiology Acquisition Context

| VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | ---
1 | CODE | EV (109061, DCM, "EP Procedure Phase") | 1 | U |  | BCID 3254 “Electrophysiology Procedure Phase”
2 | NUM | EV (109060, DCM, "Procedure Step Number") | 1 | U |  | UNITS = EV ({step}, UCUM, "step")
3 | TEXT | EV (109063, DCM, "Pulse train definition") | 1 | U |  |

### TID 3460 Projection Radiography Acquisition Context

Type:

  Extensible

Order:

  Non-Significant

Table TID 3460. Projection Radiography Acquisition Context

| VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | ---
1 | CODE | DT (F-047E7, SRT, "Functional observable") | 1-n | U |  | BCID 91 “Functional Condition Present During Acquisition”
2 | CODE | DT (F-043E6, SRT, "Respiration Observable") | 1 | U |  | BCID 3823 “Respiratory Status”
3 | CODE | DT (F-13006, SRT, "Joint position") | 1 | U |  | BCID 92 “Joint Position During Acquisition”
4 | CODE | DT (109132, DCM, "Joint positioning method") | 1 | U |  | BCID 93 “Joint Positioning Method”
5 | CODE | DT (109133, DCM, "Physical force") | 1-n | U |  | BCID 94 “Physical Force Applied During Acquisition”

### TID 3470 NM/PET Acquisition Context

Type:

  Extensible

Order:

  Non-Significant

Table TID 3470. NM/PET Acquisition Context

| VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | ---
1 | CODE | DT (109054, DCM, "Patient State") | 1 | M |  | DCID 3101 “Cardiac Procedural State Values”
2 | INCLUDE | BTID 3471 “PET Covariates Acquisition Context” | 1 | U |  |

### TID 3471 PET Covariates Acquisition Context

Type:

  Extensible

Order:

  Non-Significant

Table TID 3471. PET Covariates Acquisition Context

| VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | ---
1 | NUMERIC | (14749-6, LN, "Glucose") | 1 | U |  | UNITS = EV (mmol/l, UCUM, "mmol/l")
2 | DATE | (109081, DCM, "Glucose Measurement Date") | 1 | MC | IFF Row 1 is present |
3 | TIME | (109082, DCM, "Glucose Measurement Time") | 1 | MC | IFF Row 1 is present |

### TID 8001 Specimen Preparation

This template describes a single specimen preparation step.

Type:

  Extensible

Order:

  Significant

Table TID 8001. Specimen Preparation

| VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | ---
1 | TEXT | EV (121041, DCM, "Specimen Identifier") | 1 | M |  |
2 | TEXT | EV (111724, DCM, "Issuer of Specimen Identifier") | 1 | U |  |
3 | CODE | EV (111701, DCM, "Processing type") | 1 | M |  | DCID 8111 “Specimen Preparation Procedure”
4 | DATETIME | DT (111702, DCM, "DateTime of processing") | 1 | U |  |
5 | TEXT | DT (111703, DCM, "Processing step description") | 1 | U |  |
6 | CODE | DT (111703, DCM, "Processing step description") | 1 | U |  | DCID 8113 “Specimen Preparation Steps”
7 | CODE | DT (P3-02000, SRT, "Specimen Collection") | 1 | MC | IFF Row 3 Processing Type value is (P3-02000, SRT, "Specimen Collection") | BCID 8109 “Specimen Collection Procedure”
8 | INCLUDE | DTID 8002 “Specimen Sampling” | 1 | MC | IFF Row 3 Processing Type value is (P3-4000A, SRT, "Specimen Sampling") |
9 | INCLUDE | DTID 8003 “Specimen Staining” | 1 | MC | IFF Row 3 Processing type value is (P3-00003, SRT, "Staining") |
10 | CODE | DT (F-6221B, SRT, "Tissue Fixative") | 1 | U |  | BCID 8114 “Specimen Fixatives”
11 | CODE | DT (F-6221A, SRT, "Embedding medium") | 1 | U |  | BCID 8115 “Specimen Embedding Media”

Content Item Descriptions

Row 1 | For sampling steps (which create a child specimen from a parent), the ID is that of the child specimen. For other preparation steps, the ID of a specimen does not change during the processing.
Rows 5, 6 | The issuer shall be formatted in accordance with the HL7v2 Hierarchic Designator Data Type. That format is [ Namespace ID]^[ Universal ID^ Universal ID Type], where Namespace IDidentifies an entity within the local namespace or domain, Universal IDis a universal or unique identifier for an entity, and Universal ID Typespecifies the standard format of the Universal ID (see HL7 v2 Section 2.A.33).

### TID 8002 Specimen Sampling

Type:

  Extensible

Order:

  Significant

Table TID 8002. Specimen Sampling

| VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | ---
1 | CODE | DT (111704, DCM, "Sampling Method") | 1 | M |  | BCID 8110 “Specimen Sampling Procedure”
2 | TEXT | DT (111705, DCM, "Parent Specimen Identifier") | 1 | M |  |
3 | TEXT | DT (111706, DCM, "Issuer of Parent Specimen Identifier") | 1 | U |  |
4 | CODE | DT (111707, DCM, "Parent specimen type") | 1 | M |  | BCID 8103 “Anatomic Pathology Specimen Types”
5 | TEXT | DT (111708, DCM, "Position Frame of Reference") | 1 | U |  |
6 | TEXT | DT (111709, DCM, "Location of sampling site") | 1 | U |  |
7 | NUMERIC | DT (111710, DCM, "Location of sampling site X offset") | 1 | U |  |
8 | NUMERIC | DT (111711, DCM, "Location of sampling site Y offset") | 1 | U |  |
9 | NUMERIC | DT (111712, DCM, "Location of sampling site Z offset") | 1 | U |  |
10 | IMAGE | DT (111709, DCM, "Location of sampling site") | 1 | U |  |

Content Item Descriptions

Rows 5, 6 | The Issuer of Specimen Identifier shall be formatted in accordance with the HL7 v2 Hierarchic Designator data type (see HL7 v2.6 Section 2.A.33), i.e., [ Namespace ID]^[ Universal ID^Universal ID Type]
Row 5 | Description of coordinate system and origin reference point on parent specimen or parent specimen container used for localizing the sampling site
Rows 7-9 | The X, Y and Z locations are used as needed to describe the sampling site; not all may be needed. E.g., resection from 10 cm along the colon may be described as only a Y dimension location.
Row 10 | Reference to image of parent specimen localizing the sampling site; may include referenced Presentation State object

### TID 8003 Specimen Staining

Type:

  Extensible

Order:

  Significant

Table TID 8003. Specimen Staining

| VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | ---
1 | CODE | DT (G-C350, SRT, "Using substance") | 1-n | MC | IF Row 2 not present | DCID 8112 “Specimen Stains”
2 | TEXT | DT (G-C350, SRT, "Stain") | 1 | MC | IF Row 1 not present |

### TID 8004 Specimen Localization

Type:

  Extensible

Order:

  Significant

Table TID 8004. Specimen Localization

| VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | ---
1 | TEXT | DT (111708, DCM, "Position Frame of Reference") | 1 | U |  |
2 | TEXT | DT (111718, DCM, "Location of Specimen") | 1 | U |  |
3 | NUMERIC | DT (111719, DCM, "Location of Specimen X offset") | 1 | U |  |
4 | NUMERIC | DT (111720, DCM, "Location of Specimen Y offset") | 1 | U |  |
5 | NUMERIC | DT (111721, DCM, "Location of Specimen Z offset") | 1 | U |  |
6 | IMAGE | DT (111718, DCM, "Location of Specimen") | 1 | U |  |
7 | COMPOSITE | DT (111718, DCM, "Location of Specimen") | 1 | U |  | Presentation State SOP Instance reference
8 | TEXT | DT (111723, DCM, "Visual Marking of Specimen") | 1 | U |  |

Content Item Descriptions

Row 1 | Description of coordinate system and origin reference point used for localizing the Specimen. The value "CURRENT IMAGE " identifies the frame of reference as the pixel space of the Image SOP Instance in which this Content Item occurs.
Row 2 | Description of specimen location, either in absolute terms or relative to the Position Frame Reference of Row 1
Rows 3-5 | Location of specimen (nominal center) relative to the Position Frame Reference of Row 1. The Content Items include the units of measurement (e.g., mm). If Row 1 value is "CURRENT IMAGE ", measurement shall be from the top left hand corner of the Pixel Data of the SOP Instance, using units of ({pixel}, UCUM, "Pixels").
Row 6 | Reference to image of container localizing the specimen; may include referenced Presentation State object
Row 7 | Reference to Presentation State object for this SOP Instance, with annotations localizing the specimen
Row 8 | Description of visual distinguishing identifiers, e.g., ink, or a particular shape of the specimen

### TID 8010 Slide Imaging Parameters

This template describes protocol parameters for a Slide Imaging Procedure Step. As an extensible template, additional items may be included using other concept names from standard or private coding schemes.

Type:

  Extensible

Order:

  Significant

Table TID 8010. Slide Imaging Parameters

| VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | ---
1 | CODE | EV (112706, DCM, "Illumination Method") | 1-n | U |  | DCID 8123 “Microscopy Illumination Method”
2 | NUMERIC | EV (112707, DCM, "Number of focal planes") | 1 | UC | XOR Row 3 | UNITS = EV ({planes}, UCUM, "planes")
3 | CODE | EV (112707, DCM, "Number of focal planes") | 1 | UC | XOR Row 2 | DT (112714, DCM, "Multiple planes")
4 | NUMERIC | EV (112708, DCM, "Focal plane Z offset") | 1-n | U |  | UNITS = EV (um, UCUM, "um")
5 | CODE | EV (112709, DCM, "Magnification selection") | 1 | U |  | DCID 8132 “Magnification Selection”
6 | NUMERIC | EV (112710, DCM, "Illumination wavelength") | 1-n | U |  | UNITS = EV (nm, UCUM, "nm")
7 | CODE | EV (112711, DCM, "Illumination spectral band") | 1-n | U |  | DCID 8122 “Microscopy Illuminator and Sensor Color”
8 | CODE | EV (112712, DCM, "Optical filter type") | 1-n | U |  | DCID 8124 “Microscopy Filter”
9 | CODE | EV (112713, DCM, "Tissue selection method") | 1 | U |  | DCID 8133 “Tissue Selection”

### TID 15100 Contrast Agent/Pre-Medication Protocol Context

This Template specifies medications to be administered prior to a diagnostic imaging protocol, imaging contrast agents to be used in the protocol, and/or bolus agents to be used in the protocol. Each medication or agent may be modified by a specified route of administration. The top level Content Items of this template may appear in any order in the Protocol Context Sequence, hence the order in this template is not significant. There may be significance in the order in which the Content Items are included in the Protocol Context Sequence, e.g., the requested order in which pre-medications are to be administered.

Type:

  Extensible

Order:

  Non-Significant

Table TID 15100. Contrast Agent/Pre-Medication Protocol Context

| NL | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | ---
1 |  | CODE | EV (123011, DCM, "Contrast/Bolus Agent") | 1-n | U |  | BCID 12 “Radiographic Contrast Agent”
2 | > | CODE | EV (G-C340, SRT, "Route of Administration") | 1 | U |  | BCID 11 “Route of Administration”
3 |  | CODE | EV (123012, DCM, "Pre-Medication") | 1-n | U |  |
4 | > | CODE | EV (G-C340, SRT, "Route of Administration") | 1 | U |  | BCID 11 “Route of Administration”

### TID 15101 NM/PET Protocol Context

Type:

  Extensible

Order:

  Significant

Table TID 15101. NM/PET Protocol Context

| NL | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | ---
1 |  | CODE | EV (F-61FDB, SRT, "Radiopharmaceutical agent") | 1 | M |  | BCID 25 “Radiopharmaceuticals” BCID 4021 “PET Radiopharmaceutical”
2 | > | CODE | EV (C-10072, SRT, "Radionuclide) | 1 | U |  | BCID 18 “Isotopes in Radiopharmaceuticals” BCID 4020 “PET Radionuclide”
3 | > | UIDREF | EV (113503, DCM, "Radiopharmaceutical Administration Event UID") | 1 | U |  |
4 | > | DATETIME | EV (123003, DCM, "Radiopharmaceutical Start Time") | 1 | U |  |
5 | > | DATETIME | EV (123004, DCM, "Radiopharmaceutical Stop Time") | 1 | U |  |
6 | > | NUMERIC | EV (123005, DCM, "Radiopharmaceutical Volume") | 1 | U |  | UNITS = DT (cm3, UCUM, "cm3")
7 | > | NUMERIC | EV (123006, DCM, "Radionuclide Total Dose") | 1 | U |  | UNITS = DT (Bq, UCUM, "Bq")
8 | > | NUMERIC | EV (123007, DCM, "Radiopharmaceutical Specific Activity") | 1 | U |  | UNITS = DT (Bq/mol, UCUM, "Bq/mol")
9 | > | CODE | EV (G-C340, SRT, "Route of Administration") | 1 | U |  | BCID 11 “Route of Administration”
10 | > | NUMERIC | EV (123009, DCM, "Radionuclide Syringe Counts") | 1 | U |  | UNITS = DT ({counts}/s, UCUM "counts/s")
11 | > | NUMERIC | EV (123010, DCM, "Radionuclide Residual Syringe Counts") | 1 | U |  | UNITS = DT ({counts}/s, UCUM "counts/s")
12 |  | NUMERIC | EV (14749-6, LN, "Glucose") | 1 | U |  | UNITS = EV (mmol/l, UCUM, "mmol/l")
13 | > | DATE | EV (109081, DCM, "Glucose Measurement Date") | 1 | M |  |
14 | > | TIME | EV (109082, DCM, "Glucose Measurement Time") | 1 | M |  |

### TID 15200 JJ1017 Protocol Context

This Template defines protocol context concepts to support the requirements of Japanese Guideline JJ1017. This is expected to be used with Scheduled or Performed Protocol Codes from Coding Scheme JJ1017-16M defined in Guideline JJ1017.

Type:

  Extensible

Order:

  Significant

Table TID 15200. JJ1017 Protocol Context

| NL | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | ---
1 |  | CODE | EV (123016, DCM, "Imaging Conditions") | 1 | M |  | Baseline terms from Coding Scheme JJ1017-16S of JJ1017 version 3.0

# D DICOM Controlled Terminology Definitions (Normative)

This Annex specifies the meanings of codes defined in DICOM, either explicitly or by reference to another part of DICOM or an external reference document or standard.

DICOM Code Definitions (Coding Scheme Designator "DCM" Coding Scheme Version "01")

Table D-1. DICOM Controlled Terminology Definitions

Code Value | Code Meaning | Definition | Notes
--- | --- | --- | ---
ARCHIVE | Archive | Archive device. |
AR | Autorefraction | Autorefraction device. |
AS | Angioscopy | Angioscopy device | Retired
AU | Audio | Audio object. |
BDUS | Ultrasound Bone Densitometry | Ultrasound Bone Densitometry (modality). |
BI | Biomagnetic imaging | Biomagnetic imaging device. |
BMD | Bone Mineral Densitometry | Bone Mineral Densitometry by X-Ray (modality), including dual-energy X-Ray absorptiometry (DXA) and morphometric X-Ray absorptiometry (MXA). |
CAD | Computer Assisted Detection/Diagnosis | Computer Assisted Detection/Diagnosis device. |
CAPTURE | Image Capture | Image Capture Device, includes video capture. |
CD | Color flow Doppler | Color flow Doppler | Retired Replaced by (US, DCM, "Ultrasound")
CF | Cinefluorography | Cinefluorography | Retired Replaced by (RF, DCM, "Radiofluoroscopy")
COMP | Computation Server | Computation Server; includes radiotherapy planning. |
CP | Culposcopy | Culposcopy | Retired
CR | Computed Radiography | Computed Radiography device. |
CS | Cystoscopy | Cystoscopy | Retired
CT | Computed Tomography | Computed Tomography device. |
DD | Duplex Doppler | Duplex Doppler | Retired Replaced by (US, DCM, "Ultrasound")
DF | Digital fluoroscopy | Digital fluoroscopy | Retired Replaced by (RF, DCM, "Radiofluoroscopy")
DG | Diaphanography | Diaphanography device. |
DM | Digital microscopy | Digital microscopy | Retired
DOCD | Document Digitizer Equipment | Equipment that digitized hardcopy documents and imported them. |
DS | Digital Subtraction Angiography | Digital Subtraction Angiography | Retired Replaced by (XA, DCM, "X-Ray Angiography")
DSS | Department System Scheduler | Department System Scheduler, workflow manager; includes RIS. |
DX | Digital Radiography | Digital Radiography device. |
EC | Echocardiography | Echocardiography | Retired Replaced by (US, DCM, "Ultrasound")
ECG | Electrocardiography | Electrocardiography device. |
EPS | Cardiac Electrophysiology | Cardiac Electrophysiology device. |
ES | Endoscopy | Endoscopy device. |
F | Female | Female sex. |
FA | Fluorescein angiography | Fluorescein angiography | Retired Replaced by (OP, DCM, "Ophthalmic photography")
FC | Female changed to Male | Female sex changed to Male sex. |
FILMD | Film Digitizer | Film Digitizer. |
FP | Female Pseudohermaphrodite | Female Pseudohermaphrodite. |
FS | Fundoscopy | Fundoscopy | Retired
GM | General Microscopy | General Microscopy device. |
H | Hermaphrodite | Hermaphrodite. |
HC | Hard Copy | Hard Copy. |
HD | Hemodynamic Waveform | Hemodynamic Waveform acquisition device. |
IO | Intra-oral Radiography | Intra-oral Radiography device. |
IVUS | Intravascular Ultrasound | Intravascular Ultrasound device. |
KER | Keratometry | Keratometry device. |
KO | Key Object Selection | Key Object Selection object. |
LEN | Lensometry | Lensometry device. |
LOG | Procedure Logging | Procedure Logging device; includes cath lab logging. |
LP | Laparoscopy | Laparoscopy | Retired
LS | Laser surface scan | Laser surface scan device. |
M | Male | Male sex. |
MA | Magnetic resonance angiography | Magnetic resonance angiography | Retired Replaced by (MR, DCM, "Magnetic resonance")
MC | Male changed to Female | Male sex changed to Female sex. |
MCD | Media Creation Device | A device that creates DICOM PS3.10 interchange media. E.g., a CD creator that is managed by the Media Creation Management Service Class. |
MEDIM | Portable Media Importer Equipment | Equipment that retrieved and imported objects from interchange Media. |
MG | Mammography | Mammography device. |
MP | Male Pseudohermaphrodite | Male Pseudohermaphrodite. |
MR | Magnetic Resonance | Magnetic Resonance device. |
MS | Magnetic resonance spectroscopy | Magnetic resonance spectroscopy | Retired Replaced by (MR, DCM, "Magnetic resonance")
NEARLINE | Nearline | Instances need to be retrieved from relatively slow media such as optical disk or tape. |
NM | Nuclear Medicine | Nuclear Medicine device. |
OAM | Ophthalmic Axial Measurements | Measurements of the axial length of the eye, which are done by various devices. |
OCT | Optical Coherence Tomography | Modality device that uses an interferometric, non-invasive optical tomographic technique to image 2D slices and 3D volumes of tissue using visible and near visible frequencies. |
OFFLINE | Offline | Instances need to be retrieved by manual intervention. |
ONLINE | Online | Instances are immediately available. |
OP | Ophthalmic photography | Ophthalmic photography modality. |
OPM | Ophthalmic Mapping | Modality device that measures corneal topography, corneal or retinal thickness, and other similar parameters that are typically displayed as maps. |
OPR | Ophthalmic Refraction | Modality device that measures the refractive characteristics of the eye. |
OPT | Ophthalmic Tomography | Tomography of the eye acquired by a modality that is based on light and optical principles. Tomography based on other principles, such as ultrasound, is excluded. |
OPV | Ophthalmic Visual Field | Modality device that measures visual fields and perform visual perimetry. |
OT | Other Modality | Other Modality device. |
PR | Presentation State | Presentation State object. |
PRINT | Hard Copy Print Server | Hard Copy Print Server; includes printers with embedded DICOM print server. |
PT | Positron emission tomography | Positron emission tomography (PET) device. |
PX | Panoramic X-Ray | Panoramic X-Ray device. |
REG | Registration | Registration. |
RF | Radiofluoroscopy | Radiofluoroscopy device. |
RG | Radiographic imaging | Radiographic imaging (conventional film/screen). |
RT | Radiation Therapy Device | Radiation Therapy Device; includes linear accelerator, proton therapy. |
RTDOSE | Radiotherapy Dose | Radiotherapy Dose. |
RTIMAGE | Radiotherapy Image | Radiotherapy Imaging device; includes portal imaging. |
RTPLAN | Radiotherapy Plan | Radiotherapy Plan. |
RTRECORD | Radiotherapy Treatment Record | Radiotherapy Treatment Record. |
RTSTRUCT | Radiotherapy Structure Set | Radiotherapy Structure Set. |
SEG | Segmentation | Segmentation. |
SM | Slide Microscopy | Slide Microscopy. |
SMR | Stereometric Relationship | Stereometric image pairing modality. |
SR | Structured Report Document | Structured Report Document. |
SRF | Subjective Refraction | Subjective Refraction device. |
ST | Single-photon emission computed tomography | Single-photon emission computed tomography (SPECT) device | Retired Replaced by (NM, DCM, "Nuclear Medicine")
TG | Thermography | Thermography device. |
U | Unknown Sex | Unknown Sex. |
UNAVAILABLE | Unavailable | Instances cannot be retrieved. |
US | Ultrasound | Ultrasound device. |
VA | Visual Acuity | Visual Acuity device. |
VF | Videofluorography | Videofluorography | Retired Replaced by (RF, DCM, "Radiofluoroscopy")
VIDD | Video Tape Digitizer Equipment | Equipment that digitizes video tape and imports it. |
WSD | Workstation | Workstation. |
XA | X-Ray Angiography | X-Ray Angiography device. |
XC | External-camera Photography | External-camera Photography device. |
109001 | Digital timecode (NOS) | A signal transmitted for the purpose of interchange of the current time, not specific to any source or methodology. |
109002 | ECG-based gating signal, processed | A signal that is generated for each detection of a heart beat. |
109003 | IRIG-B timecode | A signal transmitted by the Inter-Range Instrumentation Group for the purpose of synchronizing time clocks. |
109004 | X-Ray Fluoroscopy On Signal | A signal that indicates that X-Ray source has been activated for fluoroscopy use. |
109005 | X-Ray On Trigger | A signal that indicated that the X-Ray source has been activated for image recording. |
109006 | Differential signal | An electrical signal derived from two electrodes. |
109007 | His bundle electrogram | An electrophysiological recording from the HIS nerve bundle. |
109008 | Monopole signal | An electrical signal from one electrode relative to an indifferent potential. |
109009 | Pacing (electrical) stimulus, voltage | The voltage stimulus during cardiac pacing. |
109010 | Radio frequency ablation, power | The power injected during RF ablation procedure. |
109011 | Voltage measurement by basket catheter | Electrophysiological signals acquired using a multi-splined catheter each equipped with multiple electrodes. |
109012 | Voltage measurement by mapping catheter | Electrophysiological signals acquired using a steerable catheter. |
109013 | Voltage measurement, NOS | A voltage measurement not otherwise specified. |
109014 | 35% of thermal CO | A signal point that is 35% of the peak thermal cardiac output signal. |
109015 | 70% of thermal CO | A signal point that is 70% of the peak thermal cardiac output signal. |
109016 | A wave peak pressure | The peak pressure of each heart beat in the atrium caused by the atrial contraction. |
109017 | A wave pressure, average | The average of several A wave pressure measurements. |
109018 | Beat detected (accepted) | An identified cardiac beat used in the determination of a measurement. |
109019 | Beat detected (rejected) | An identified cardiac beat not used in the determination of a measurement. |
109020 | Diastolic pressure, average | The average of several diastolic pressure measurements | Retired. Replaced by (F-00E22, SRT, "Average diastolic blood pressure")
109021 | Diastolic pressure nadir | The lowest pressure value excluding any undershoot artifact. | Retired. Replaced by (F-00E1F, SRT, "Minimum diastolic blood pressure")
109022 | End diastole | The moment at the end of the diastolic phase of the cardiac cycle. | Retired. Replaced by (F-32011, SRT, "End diastole")
109023 | End of expiration | The moment at the end of respiratory expiration. |
109024 | End of inspiration | The moment at the end of respiratory inspiration. |
109025 | Max dp/dt | The maximum positive rate of change of pressure. |
109026 | Max neg dp/dt | The maximum negative rate of change of pressure. |
109027 | Mean blood pressure | The average blood pressure value, generally over 2 or more seconds | Retired. Replaced by (F-31150, SRT, "Mean blood pressure")
109028 | Peak of thermal cardiac output bolus | The peak change in blood temperature during a thermal cardiac output measurement. |
109029 | Start of expiration | The moment respiratory expiration begins. |
109030 | Start of inspiration | The moment of respiratory inspiration begins. |
109031 | Start of thermal cardiac output bolus | The first discernible blood temperature change following the injectate during a thermal cardiac output measurement. |
109032 | Systolic pressure, average | The average of several systolic blood pressure measurements. | Retired. Replaced by (F-00E14, SRT, "Average systolic blood pressure")
109033 | Systolic peak pressure | The highest systolic blood pressure value excluding any overshoot artifact | Retired. Replaced by (F-00E11, SRT, "Maximum systolic blood pressure")
109034 | V wave peak pressure | The peak pressure of each heart beat in the atrium caused by the filling of the atrium. |
109035 | V wave pressure, average | The average of several V wave pressure measurements. |
109036 | Valve close | The moment at which a heart valve closes. |
109037 | Valve open | The moment at which a heart valve opens. |
109038 | Ablation off | The moment when RF ablation current is turned off. |
109039 | Ablation on | The moment when RF ablation current is turned on. |
109040 | HIS bundle wave | The moment in the cardiac cycle when the HIS bundle nerves depolarize. |
109041 | P wave | The surface electrocardiogram of the atrial contraction. |
109042 | Q wave | The first negative deflection of the electrocardiogram cause by ventricular depolarization. |
109043 | R wave | The first positive deflection the electrocardiogram cause by ventricular depolarization. |
109044 | S wave | The first negative deflection after the R wave. |
109045 | Start of atrial contraction | The beginning of the atrial contraction. |
109046 | Start of atrial contraction (subsequent) | The beginning of the second atrial contraction of two consecutive beats. |
109047 | Stimulation at rate 1 interval | The stimulation interval during cardiac stimulation first used in a pacing train. |
109048 | Stimulation at rate 2 interval | The stimulation interval different from the first stimulation interval used in a pacing train. |
109049 | Stimulation at rate 3 interval | A stimulation interval different from and subsequent to the second interval in a pacing train. |
109050 | Stimulation at rate 4 interval | Describes a stimulation interval different from and subsequent to the third interval in a pacing train. |
109051 | T wave | The electrocardiogram deflection caused by ventricular repolarization. |
109052 | V wave | The peak pressure of each heart beat monitored in the atrium caused by the filling of the atrium. |
109053 | V wave of next beat | The second V wave measurement of two consecutive beats. |
109054 | Patient State | A description of the physiological condition of the patient. |
109055 | Protocol Stage | The exercise level during a progressive cardiac stress test. |
109056 | Stress Protocol | A series of physiological challenges designed to progressively increase the work of the heart. |
109057 | Catheterization Procedure Phase | A subpart of a cardiac catheterization procedure | Retired. Replaced by (G-72BB, SRT, "Catheterization Procedure Phase")
109058 | Contrast Phase | The subpart of a cardiac catheterization procedure in which a radio-opaque contrast medium is injected into the patient. |
109059 | Physiological challenges | Physical changes administered to a patient in order to elicit an physiological response. |
109060 | Procedure Step Number | Enumeration of a subpart of a catheterization procedure. |
109061 | EP Procedure Phase | A subpart of en electrophysiological procedure. |
109063 | Pulse train definition | A means of defining a series of cardiac stimulation pulses. |
109070 | End of systole |  | Retired. Replaced by (R-FAB5B, SRT, "End systole")
109071 | Indicator mean transit time | Time for a median particle to travel from point of injection to point of detection. |
109072 | Tau | The time constant of isovolumic pressure fall. |
109073 | V max myocardial | Maximum velocity of myocardial contractility. |
109080 | Real time acquisition | Total time for the acquisition is shorter than cardiac cycle, no gating is applied; see Cardiac Synchronization Technique (0018,9037). |
109081 | Prospective gating | Certain thresholds have been set for a gating window that defines the acceptance of measurement data during the acquisition; see Cardiac Synchronization Technique (0018,9037). |
109082 | Retrospective gating | Certain thresholds have been set for a gating window that defines the acceptance of measurement data after the acquisition; see Cardiac Synchronization Technique (0018,9037). |
109083 | Paced | There is a constant RR interval, which makes thresholding not required; see Cardiac Synchronization Technique (0018,9037). E.g., Pacemaker. |
109091 | Cardiac Stress State | Imaging after injection of tracer during increased cardiac workload or increased myocardial blood flow, achieved by either exercise or pharmacologic means. | Retired. Replaced by (F-05019, SRT, "Cardiac stress state").
109092 | Reinjection State | Imaging after injection of additional tracer under resting conditions. |
109093 | Redistribution State | Imaging after allowing a moderate amount of time for tracer to move from its initial sites of uptake. Example: For Thallium imaging this would correspond to imaging 2-6 hours after injection. |
109094 | Delayed Redistribution State | Imaging after allowing an extended amount of time for tracer to move from its initial sites of uptake. Example: For Thallium imaging this would correspond to imaging more than 6 hours after injection. |
109095 | Peak stress state | Peak Cardiac stress state | Retired. Replaced by (F-05028, SRT, "Peak stress state")
109096 | Recovery state | Recovery from cardiac stress | Retired. Replaced by (F-05018, SRT, "Cardiac stress Recovery state")
109101 | Acquisition Equipment | Equipment that originally acquired the data stored within composite instances. E.g., a CT, MR or Ultrasound modality. |
109102 | Processing Equipment | Equipment that has processed composite instances to create new composite instances. E.g., a 3D Workstation. |
109103 | Modifying Equipment | Equipment that has modified existing composite instances (without creating new composite instances). E.g., a QA Station or Archive. |
109104 | De-identifying Equipment | Equipment that has modified an existing composite instance to remove patient identifying information. |
109105 | Frame Extracting Equipment | Equipment that has processed composite instances to create new composite instances by extracting selected frames from the original instance. |
109106 | Enhanced Multi-frame Conversion Equipment | Equipment that has processed composite instances to create new composite instances by converting classic single frame images to enhanced multi-frame image, or vice versa and updating other instances to maintain referential integrity. |
109110 | Voice | The sound of a human's speech, recorded during a procedure. | May include the patient's voice, or the voice of staff present in the room, or an operator's voice (whether for the purpose of recording a narrative accompanying a procedure or not).
109111 | Operator's narrative | The voice of a device operator, recorded during a procedure. |
109112 | Ambient room environment | The ambient sound recorded during a procedure, which may or may not include voice and other types of sound. |
109113 | Doppler audio | The Doppler waveform recorded as an audible signal. |
109114 | Phonocardiogram | The sound of the human heart beating. | Such as might be recorded from an electronic stethoscope.
109115 | Physiological audio signal | Any sound made by the human body. | May include the sound of the heart, but also sound from other organs, such as bowel sounds or bruits from vessels, or sounds of respiration. Not intended to include voice.
109116 | Arterial Pulse Waveform | A digitized signal from the patient arterial system collected through pulse oximetry or other means. |
109117 | Respiration Waveform | A digitized signal from the patient respiratory system representing respiration. |
109120 | On admission to unit | The occasion on which a procedure was performed on admission to a specialist unit. E.g., intensive care. |
109121 | On discharge | The occasion on which a procedure was performed on discharge from hospital as an in-patient. |
109122 | On discharge from unit | The occasion on which a procedure was performed on discharge from a specialist unit. E.g., intensive care. |
109123 | Pre-intervention | The occasion on which a procedure was performed immediately prior to non-surgical intervention. E.g, percutaneous angioplasty, biopsy. |
109124 | Post-intervention | The occasion on which a procedure was performed immediately after to non-surgical intervention. E.g, percutaneous angioplasty, biopsy. |
109125 | At last appointment | The occasion on which a procedure was performed at the most recent outpatient visit. |
109132 | Joint position method | The active or passive joint positioning during acquisition. |
109133 | Physical force | A physical force applied during acquisition. |
109134 | Prior to voiding | Prior to voiding. |
109135 | Post voiding | Post voiding. |
109136 | Neutral musculoskeletal position | Neutral musculoskeletal position. |
109200 | America Kennel Club | America Kennel Club. |
109201 | America's Pet Registry Inc. | America's Pet Registry Inc. |
109202 | American Canine Association | American Canine Association. |
109203 | American Purebred Registry | American Purebred Registry. |
109204 | American Rare Breed Association | American Rare Breed Association. |
109205 | Animal Registry Unlimited | Animal Registry Unlimited. |
109206 | Animal Research Foundation | Animal Research Foundation. |
109207 | Canadian Border Collie Association | Canadian Border Collie Association. |
109208 | Canadian Kennel Club | Canadian Kennel Club. |
109209 | Canadian Livestock Records Association | Canadian Livestock Records Association. |
109210 | Canine Federation of Canada | Canine Federation of Canada. |
109211 | Continental Kennel Club | Continental Kennel Club. |
109212 | Dog Registry of America | Dog Registry of America. |
109213 | Federation of International Canines | Federation of International Canines. |
109214 | International Progressive Dog Breeders' Alliance | International Progressive Dog Breeders' Alliance. |
109215 | National Kennel Club | National Kennel Club. |
109216 | North American Purebred Dog Registry | North American Purebred Dog Registry. |
109217 | United All Breed Registry | United All Breed Registry. |
109218 | United Kennel Club | United Kennel Club. |
109219 | Universal Kennel Club International | Universal Kennel Club International. |
109220 | Working Canine Association of Canada | Working Canine Association of Canada. |
109221 | World Kennel Club | World Kennel Club. |
109222 | World Wide Kennel Club | World Wide Kennel Club. |
109701 | Overall image quality evaluation | Evaluation of overall image quality as described in section 7.3.2 of [IEC 62563-1]. |
109702 | Grayscale resolution evaluation | Visual verification of sufficient grayscale resolution based on 8 and 10-bit markers as described in section 7.3.3 of [IEC 62563-1]. |
109703 | Luminance response evaluation | Visual evaluation of luminance response using the TG18-CT test pattern as described in section 7.3.4 of [IEC 62563-1]. |
109704 | Luminance uniformity evaluation | Visual detection of luminance non-uniformities as described in section 7.3.5 of [IEC 62563-1]. |
109705 | Chromaticity evaluation | Visual verification of color uniformity as described in section 7.3.6 of [IEC 62563-1]. |
109706 | Pixel faults evaluation | Visual detection of defective pixels on dark (TG18-UN80) and bright (TG18-UN10) images as described in section 7.3.7 of [IEC 62563-1]. |
109707 | Veiling glare evaluation | Visual evaluation of veiling glare by looking at low contrast objects on 2 test patterns as described in section 7.3.8 of [IEC 62563-1]. |
109708 | Geometrical image evaluation | Visual evaluation of geometry, phase/clock correction and clipping as described in section 7.3.9 of [IEC 62563-1]. |
109709 | Angular viewing evaluation | Visual evaluation of viewing angle as described in section 7.3.10 of [IEC 62563-1]. |
109710 | Clinical evaluation | Visual evaluation of the appearance of clinical images as described in section 7.3.11 of [IEC 62563-1]. |
109801 | TG18-QC Pattern | AAPM TG18-QC Pattern used for evaluation of resolution, luminance, distortion, artifacts. See [AAPM OR 03]. |
109802 | TG18-BR Pattern | AAPM TG18-BR Pattern used for the evaluation of the display of low-contrast, fine-detail image structures See [AAPM OR 03]. |
109803 | TG18-PQC Pattern | AAPM TG18-PQC Pattern used for evaluation of resolution, luminance, contrast transfer for prints. See [AAPM OR 03]. |
109804 | TG18-CT Pattern | AAPM TG18-CT Pattern used for evaluation of luminance response. See [AAPM OR 03]. |
109805 | TG18-LN8-01 Pattern | The 1st image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109806 | TG18-LN8-02 Pattern | The 2nd image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109807 | TG18-LN8-03 Pattern | The 3rd image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109808 | TG18-LN8-04 Pattern | The 4th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109809 | TG18-LN8-05 Pattern | The 5th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109810 | TG18-LN8-06 Pattern | The 6th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109811 | TG18-LN8-07 Pattern | The 7th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109812 | TG18-LN8-08 Pattern | The 8th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109813 | TG18-LN8-09 Pattern | The 9th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109814 | TG18-LN8-10 Pattern | The 10th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration series. See [AAPM OR 03]. |
109815 | TG18-LN8-11 Pattern | The 11th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109816 | TG18-LN8-12 Pattern | The 12th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109817 | TG18-LN8-13 Pattern | The 13th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109818 | TG18-LN8-14 Pattern | The 14th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109819 | TG18-LN8-15 Pattern | The 15th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109820 | TG18-LN8-16 Pattern | The 16th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109821 | TG18-LN8-17 Pattern | The 17th image in the AAPM TG18-LN8 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109822 | TG18-LN8-18 Pattern | The 18th image in the AAPM TG18-LN8- set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109823 | TG18-LN12-01 Pattern | The 1st image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109824 | TG18-LN12-02 Pattern | The 2 nd image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109825 | TG18-LN12-03 Pattern | The 3rd image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109826 | TG18-LN12-04 Pattern | The 4th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109827 | TG18-LN12-05 Pattern | The 5th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109828 | TG18-LN12-06 Pattern | The 6th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109829 | TG18-LN12-07 Pattern | The 7th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109830 | TG18-LN12-08 Pattern | The 8th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109831 | TG18-LN12-09 Pattern | The 9th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109832 | TG18-LN12-10 Pattern | The 10th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109833 | TG18-LN12-11 Pattern | The 11th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109834 | TG18-LN12-12 Pattern | The 12th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109835 | TG18-LN12-13 Pattern | The 13th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109836 | TG18-LN12-14 Pattern | The 14th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109837 | TG18-LN12-15 Pattern | The 15th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109838 | TG18-LN12-16 Pattern | The 16th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109839 | TG18-LN12-17 Pattern | The 17th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109840 | TG18-LN12-18 Pattern | The 18th image in the AAPM TG18-LN12 set used for DICOM grayscale calibration. See [AAPM OR 03]. |
109841 | TG18-UN10 Pattern | The AAPM TG18-UN10 Pattern used for evaluation of luminance and color uniformity, and angular response. See [AAPM OR 03]. |
109842 | TG18-UN80 Pattern | The AAPM TG18-UN80 Pattern used for evaluation of luminance and color uniformity, and angular response. See [AAPM OR 03]. |
109843 | TG18-UNL10 Pattern | The AAPM TG18-UNL10 Pattern is the AAPM TG-18 UN10 Pattern with added defining lines. See [AAPM OR 03]. |
109844 | TG18-UNL80 Pattern | The AAPM TG18-UNL80 Pattern is the AAPM TG-18 UN80 Pattern with added defining lines. See [AAPM OR 03]. |
109845 | TG18-AD Pattern | The AAPM TG18-AD Pattern used for visual evaluation of the reflection of ambient light from the display. See [AAPM OR 03]. |
109846 | TG18-MP Pattern | The AAPM TG18-MP Pattern used for evaluation of Luminance response (bit-depth resolution). See [AAPM OR 03]. |
109847 | TG18-RH10 Pattern | The AAPM TG18-RH10 Pattern used for LSF-line spectra function-(1k and 2k) evaluation by 5 horizontal lines at 10% luminance level. See [AAPM OR 03]. |
109848 | TG18-RH50 Pattern | The AAPM TG18-RH50 Pattern used for LSF-line spectra function-(1k and 2k) evaluation by 5 horizontal lines at 50% luminance level. See [AAPM OR 03]. |
109849 | TG18-RH89 Pattern | The AAPM TG18-RH89 Pattern used for LSF-line spectra function-(1k and 2k) evaluation by 5 horizontal lines at 89% luminance level. See [AAPM OR 03]. |
109850 | TG18-RV10 Pattern | The AAPM TG18-RV10 Pattern used for LSF-line spectra function-(1k and 2k) evaluation by 5 vertical lines at 10% luminance level. See [AAPM OR 03]. |
109851 | TG18-RV50 Pattern | The AAPM TG18-RV50 Pattern used for LSF-line spectra function-(1k and 2k) evaluation by 5 vertical lines at 50% luminance level. See [AAPM OR 03]. |
109852 | TG18-RV89 Pattern | The AAPM TG18-RV89 Pattern used for LSF-line spectra function-(1k and 2k) evaluation by 5 vertical lines at 89% luminance level. See [AAPM OR 03]. |
109853 | TG18-PX Pattern | The AAPM TG18-PX Pattern used for the assessment of display resolution. See [AAPM OR 03]. |
109854 | TG18-CX Pattern | The AAPM TG18-CX Pattern used to assess display resolution and resolution uniformity. See [AAPM OR 03]. |
109855 | TG18-LPH10 Pattern | The AAPM TG18-LPH10 Pattern used to assess display resolution. This pattern has horizontal bars consisting of alternating single-pixel-wide lines across the faceplate of display. The lines have a 12% positive contrast against 10% background level of the maximum pixel value. See [AAPM OR 03]. |
109856 | TG18-LPH50 Pattern | The AAPM TG18-LPH50 Pattern used to assess display resolution. This pattern has horizontal bars consisting of alternating single-pixel-wide lines across the faceplate of display. The lines have a 50% positive contrast against 10% background level of the maximum pixel value. See [AAPM OR 03]. |
109857 | TG18-LPH89 Pattern | The AAPM TG18-LPH89 Pattern used to assess display resolution. This pattern has horizontal bars consisting of alternating single-pixel-wide lines across the faceplate of display. The lines have a 12% positive contrast against 89% background level of the maximum pixel value. See [AAPM OR 03]. |
109858 | TG18-LPV10 Pattern | The AAPM TG18-LPV10 Pattern used to assess display resolution. This pattern has vertical bars consisting of alternating single-pixel-wide lines across the faceplate of display. The lines have a 12% positive contrast against 10% background level of the maximum pixel value. See [AAPM OR 03]. |
109859 | TG18-LPV50 Pattern | The AAPM TG18-LPV50 Pattern used to assess display resolution. This pattern has vertical bars consisting of alternating single-pixel-wide lines across the faceplate of display. The lines have a 12% positive contrast against 50% background level of the maximum pixel value. See [AAPM OR 03]. |
109860 | TG18-LPV89 Pattern | The AAPM TG18-LPV89 Pattern used to assess display resolution. This pattern has vertical bars consisting of alternating single-pixel-wide lines across the faceplate of display. The lines have a 12% positive contrast against 89% background level of the maximum pixel value. See [AAPM OR 03]. |
109861 | TG18-AFC Pattern | The AAPM TG18-AFC Pattern used to assess display noise. See [AAPM OR 03] |
109862 | TG18-NS10 Pattern | The AAPM TG18-NS10 Pattern is AAPM TG18-RV10/RH10 with only difference being the absence of the single line at the center of the measurement area. See [AAPM OR 03]. |
109863 | TG18-NS50 Pattern | The AAPM TG18-NS50 Pattern is AAPM TG18-RV50/RH50 with only difference being the absence of the single line at the center of the measurement area. See [AAPM OR 03]. |
109864 | TG18-NS89 Pattern | The AAPM TG18-NS89 Pattern is AAPM TG18-RV89/RH89 with only difference being the absence of the single line at the center of the measurement area. See [AAPM OR 03]. |
109865 | TG18-GV Pattern | The TG18-GV Pattern used to assess display veiling. See [AAPM OR 03]. |
109866 | TG18-GVN Pattern | The TG18-GVN Pattern used to assess display veiling. This pattern is identical to AAPM TG18-GV Pattern except that the large-diameter white circle is replaced with a black circle, creating a completely black pattern except for the presence of low-contrast targets. See [AAPM OR 03]. |
109867 | TG18-GQ Pattern | The TG18-GQ Pattern used for quantitative assessment of veiling glare. This pattern is identical to TG18-GV except that is lacks the central low-contrast objects. See [AAPM OR 03]. |
109868 | TG18-GQN Pattern | TG18-GQN Pattern used for the quantitative assessment of veiling glare. This pattern is identical to AAPM TG18-GQ Pattern except that the large-diameter white circle is replaced with a black circle, creating a completely black pattern except for the presence of low-contrast targets. See [AAPM OR 03]. |
109869 | TG18-GQB Pattern | The TG18-GQB Pattern used for the quantitative assessment of veiling glare. This pattern is identical to AAPM TG18-GQ Pattern except eliminating the central black circle. See [AAPM OR 03]. |
109870 | TG18-GA03 Pattern | The TG18-GA03 Pattern used for quantitative assessment of veiling glare. This pattern is identical to TG18-GQ except that the radius of the central black circle is varied as r = 3. See [AAPM OR 03]. |
109871 | TG18-GA05 Pattern | The TG18-GA05 Pattern This pattern is identical to TG18-GQ except that the radius of the central black circle is varied as r = 5. See [AAPM OR 03]. |
109872 | TG18-GA08 Pattern | The TG18-GA08 Pattern used for quantitative assessment of veiling glare. This pattern is identical to TG18-GQ except that the radius of the central black circle is varied as r = 8. See [AAPM OR 03]. |
109873 | TG18-GA10 Pattern | The TG18-GA10 Pattern used for quantitative assessment of veiling glare. This pattern is identical to TG18-GQ except that the radius of the central black circle is varied as r = 10. See [AAPM OR 03]. |
109874 | TG18-GA15 Pattern | The TG18-GA15 Pattern used for quantitative assessment of veiling glare. This pattern is identical to TG18-GQ except that the radius of the central black circle is varied as r = 15. |
109875 | TG18-GA20 Pattern | The TG18-GA20 Pattern used for quantitative assessment of veiling glare. This pattern is identical to TG18-GQ except that the radius of the central black circle is varied as r = 20. See [AAPM OR 03]. |
109876 | TG18-GA25 Pattern | The TG18-GA25 Pattern used for quantitative assessment of veiling glare. This pattern is identical to TG18-GQ except that the radius of the central black circle is varied as r = 25. See [AAPM OR 03]. |
109877 | TG18-GA30 Pattern | The TG18-GA30 Pattern used for quantitative assessment of veiling glare. This pattern is identical to TG18-GQ except that the radius of the central black circle is varied as r = 30. See [AAPM OR 03]. |
109878 | TG18-CH Image | The AAPM TG18-CH Image is a reference anatomical PA chest image. See [AAPM OR 03]. |
109879 | TG18-KN Image | The AAPM TG18-KN Image is a reference anatomical knee image. See [AAPM OR 03]. |
109880 | TG18-MM1 Image | The AAPM TG18-MM1 Image is a reference anatomical mammogram image. See [AAPM OR 03]. |
109881 | TG18-MM2 Image | The AAPM TG18-MM2 Image is a reference anatomical mammogram image. See [AAPM OR 03]. |
109901 | OIQ Pattern | The IEC OIQ Pattern is used as an alternative to the TG18-QC Pattern. See [IEC 62563-1]. |
109902 | ANG Pattern | The IEC ANG Pattern used for angular viewing evaluation. See [IEC 62563-1]. |
109903 | GD Pattern | The IEC GD Pattern used for geometrical image evaluation. See [IEC 62563-1]. |
109904 | BN01 Pattern | The IEC BN01 Pattern is used as analternative to the TG18-LN-01 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109905 | BN02 Pattern | The IEC BN02 Pattern is used as analternative to the TG18-LN-02 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109906 | BN03 Pattern | The IEC BN03 Pattern is used as analternative to the TG18-LN-03 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109907 | BN04 Pattern | The IEC BN04 Pattern is used as analternative to the TG18-LN-04 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109908 | BN05 Pattern | The IEC BN05 Pattern is used as analternative to the TG18-LN-05 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109909 | BN06 Pattern | The IEC BN06 Pattern is used as analternative to the TG18-LN-06 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109910 | BN07 Pattern | The IEC BN07 Pattern is used as analternative to the TG18-LN-07 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109911 | BN08 Pattern | The IEC BN08 Pattern is used as analternative to the TG18-LN-08 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109912 | BN09 Pattern | The IEC BN09 Pattern is used as analternative to the TG18-LN-09 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109913 | BN10 Pattern | The IEC BN10 Pattern is used as analternative to the TG18-LN-10 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109914 | BN11 Pattern | The IEC BN11 Pattern is used as analternative to the TG18-LN-11 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109915 | BN12 Pattern | The IEC BN12 Pattern is used as analternative to the TG18-LN-12 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109916 | BN13 Pattern | The IEC BN13 Pattern is used as analternative to the TG18-LN-13 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109917 | BN14 Pattern | The IEC BN14 Pattern is used as analternative to the TG18-LN-14 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109918 | BN15 Pattern | The IEC BN15 Pattern is used as analternative to the TG18-LN-15 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109919 | BN16 Pattern | The IEC BN16 Pattern is used as analternative to the TG18-LN-16 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109920 | BN17 Pattern | The IEC BN17 Pattern is used as analternative to the TG18-LN-17 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109921 | BN18 Pattern | The IEC BN18 Pattern is used as analternative to the TG18-LN-18 Pattern, to avoid the use of a cone or baffle with LCDs. See [IEC 62563-1]. |
109931 | DIN Grayscale Pattern | Test image "Bild 2" for the gray-scale reproduction of imaging devices. See [DIN 6868-57]. |
109932 | DIN Geometry Pattern | Test image "Bild 3" for the geometrical imaging properties of imaging devices. See [DIN 6868-57]. |
109933 | DIN Resolution Pattern | Test image "Bild 5" for displaying the spatial and contrast resolution as well as the line structure of imaging devices. See [DIN 6868-57]. |
109941 | White Pattern | An alternative to AAPM TG18-UN80, specified at 100% of maximum pixel value. |
109943 | SMPTE Pattern | A standard display test pattern. See [SMPTE RP133]. A pattern is available at http://www.dclunie.com/images/smpte.512.512.8.gif. |
109991 | CRT Display | A Display Device that displays images on a Cathode Ray Tube. |
109992 | Liquid Crystal Display | A Display Device that displays images on a Liquid Crystal Display. |
109993 | Plasma Display | A Display Device that displays images on a Plasma Display. |
109994 | OLED | A Display Device that displays images on an Organic Light Emitting Diode based display. |
109995 | DLP Rear Projection System | A Display Device that projects images on a surface from behind using a Digital Light Processing Projector. |
109996 | DLP Front Projection System | A Display Device that projects images on a surface from in front using a Digital Light Processing Projector. |
109997 | CRT Rear Projection System | A Display Device that projects images on a surface from behind using a Cathode Ray Tube. |
109998 | CRT Front Projection System | A Display Device that projects images on a surface from in front using a Cathode Ray Tube. |
109999 | Other Projection System | A Display Device that projects images on a surface from an unspecified direction using an unspecified means. |
110001 | Image Processing | Image processing work item. |
110002 | Quality Control | Quality control work item. |
110003 | Computer Aided Diagnosis | Computer aided diagnosis work item. |
110004 | Computer Aided Detection | Computer aided detection work item. |
110005 | Interpretation | Interpretation work item. |
110006 | Transcription | Transcription work item. |
110007 | Report Verification | Report verification work item. |
110008 | Print | Print work item. |
110009 | No subsequent Workitems | There will be no more work items scheduled. |
110010 | Film | Film type of output. |
110011 | Dictation | Dictation type of output. |
110012 | Transcription | Transcription type of output. |
110013 | Media Import | The procedure to read DICOM instances from DICOM interchange media, coerce identifying attributes into the local namespace if necessary, and make the instances available. |
110020 | Sheet Film Digitized | Digitization of Sheet Film. |
110021 | Cine Film Digitized | Digitization of Cine Film. |
110022 | Video Tape Digitized | Digitization of Video Tape. |
110023 | Paper Digitized | Digitization of pages of a paper document (Units may be specified as Pages, Documents). |
110024 | CD Imported | Importation of CD. |
110025 | DVD Imported | Importation of DVD. |
110026 | MOD Imported | Importation of MOD. |
110027 | Studies Imported | Importation of DICOM Studies. |
110028 | Instances Imported | Importation of DICOM Composite Instances. |
110030 | USB Disk Emulation | A device that connects using the USB hard drive interface. These may be USB-Sticks, portable hard drives, and other technologies. |
110031 | Email | Email and email attachments used as a media for data transport. |
110032 | CD | CD-R, CD-ROM, and CD-RW media used for data transport. |
110033 | DVD | DVD, DVD-RAM, and other DVD formatted media used for data transport. |
110034 | Compact Flash | Media that comply with the Compact Flash standard. |
110035 | Multi-media Card | Media that comply with the Multi-media Card standard. |
110036 | Secure Digital Card | Media that comply with the Secure Digital Card standard. |
110037 | URI | URI Identifier for network or other resource, see RFC 3968. |
110038 | Paper Document | Any paper or similar document. |
110100 | Application Activity | Audit event: Application Activity has taken place. |
110101 | Audit Log Used | Audit event: Audit Log has been used. |
110102 | Begin Transferring DICOM Instances | Audit event: Storage of DICOM Instances has begun. |
110103 | DICOM Instances Accessed | Audit event: DICOM Instances have been created, read, updated, or deleted -audit event. |
110104 | DICOM Instances Transferred | Audit event: Storage of DICOM Instances has been completed. |
110105 | DICOM Study Deleted | Audit event: Entire Study has been deleted. |
110106 | Export | Audit event: Data has been exported out of the system. |
110107 | Import | Audit event: Data has been imported into the system. |
110108 | Network Entry | Audit event: System has joined or left network. |
110109 | Order Record | Audit event: Order has been created, read, updated or deleted. |
110110 | Patient Record | Audit event: Patient Record has been created, read, updated, or deleted. |
110111 | Procedure Record | Audit event: Procedure Record has been created, read, updated, or deleted. |
110112 | Query | Audit event: Query has been made. |
110113 | Security Alert | Audit event: Security Alert has been raised. |
110114 | User Authentication | Audit event: User Authentication has been attempted. |
110120 | Application Start | Audit event: Application Entity has started. |
110121 | Application Stop | Audit event: Application Entity has stopped. |
110122 | Login | Audit event: User login has been attempted. |
110123 | Logout | Audit event: User logout has been attempted. |
110124 | Attach | Audit event: Node has been attached. |
110125 | Detach | Audit event: Node has been detached. |
110126 | Node Authentication | Audit event: Node Authentication has been attempted. |
110127 | Emergency Override Started | Audit event: Emergency Override has started. |
110128 | Network Configuration | Audit event: Network configuration has been changed. |
110129 | Security Configuration | Audit event: Security configuration has been changed. |
110130 | Hardware Configuration | Audit event: Hardware configuration has been changed. |
110131 | Software Configuration | Audit event: Software configuration has been changed. |
110132 | Use of Restricted Function | Audit event: A use of a restricted function has been attempted. |
110133 | Audit Recording Stopped | Audit event: Audit recording has been stopped. |
110134 | Audit Recording Started | Audit event: Audit recording has been started. |
110135 | Object Security Attributes Changed | Audit event: Security attributes of an object have been changed. |
110136 | Security Roles Changed | Audit event: Security roles have been changed. |
110137 | User security Attributes Changed | Audit event: Security attributes of a user have been changed. |
110138 | Emergency Override Stopped | Audit event: Emergency Override has Stopped. |
110139 | Remote Service Operation Started | Audit event: Remote Service Operation has Begun. |
110140 | Remote Service Operation Stopped | Audit event: Remote Service Operation has Stopped. |
110141 | Local Service Operation Started | Audit event: Local Service Operation has Begun. |
110142 | Local Service Operation Stopped | Audit event: Local Service Operation Stopped. |
110150 | Application | Audit participant role ID of software application. |
110151 | Application Launcher | Audit participant role ID of software application launcher, i.e., the entity that started or stopped an application. |
110152 | Destination Role ID | Audit participant role ID of the receiver of data. |
110153 | Source Role ID | Audit participant role ID of the sender of data. |
110154 | Destination Media | Audit participant role ID of media receiving data during an export. |
110155 | Source Media | Audit participant role ID of media providing data during an import. |
110180 | Study Instance UID | ParticipantObjectID type: Study Instance UID. |
110181 | SOP Class UID | ParticipantObjectID type: SOP Class UID. |
110182 | Node ID | ID of a node that is a participant object of an audit message. |
110190 | Issuer of Identifier | System, organization, agency, or department that has assigned an instance identifier (such as placer or filler number, patient or provider identifier, etc.). |
110500 | Doctor canceled procedure | Procedure order canceled by requesting physician or other authorized physician. |
110501 | Equipment failure | Equipment failure prevented completion of procedure. |
110502 | Incorrect procedure ordered | Procedure discontinued due to incorrect procedure being ordered. |
110503 | Patient allergic to media/contrast | Procedure discontinued due to patient allergy to media/contrast (reported or reaction). |
110504 | Patient died | Procedure discontinued due to death of Patient. |
110505 | Patient refused to continue procedure | Procedure discontinued due to patient refusal to continue procedure. |
110506 | Patient taken for treatment or surgery | Procedure discontinued due to patient being taken for treatment or surgery. |
110507 | Patient did not arrive | Patient did not arrive for procedure. |
110508 | Patient pregnant | Procedure discontinued due to patient pregnancy (reported or determined). |
110509 | Change of procedure for correct charging | Procedure discontinued to restart with new procedure code for correct charging. |
110510 | Duplicate order | Procedure discontinued due to duplicate orders received for same procedure. |
110511 | Nursing unit cancel | Procedure order canceled by nursing unit. |
110512 | Incorrect side ordered | Procedure discontinued due to incorrect side (laterality) being ordered. |
110513 | Discontinued for unspecified reason | Procedure discontinued for unspecified reason. |
110514 | Incorrect worklist entry selected | Procedure discontinued due to incorrect patient or procedure step selected from modality worklist. |
110515 | Patient condition prevented continuing | Patient condition prevented continuation of procedure. |
110516 | Equipment change | Procedure step is discontinued to change to other equipment or modality. |
110518 | Patient Movement | A movement of the patient affecting test quality. |
110519 | Operator Error | An error of the operator affecting test quality. |
110521 | Objects incorrectly formatted | One or more of the objects is malformed. |
110522 | Object Types not supported | Receiving System is unable to accept the object type. |
110523 | Object Set incomplete | One or more objects associated with the object set is missing. |
110524 | Media Failure | The contents of the Media could not be accessed properly. |
110526 | Resource pre-empted | Procedure discontinued due to necessary equipment, staff or other resource becoming (temporarily) unavailable to the procedure. |
110527 | Resource inadequate | Procedure discontinued due to necessary equipment, staff or other resource being inadequate to complete the procedure. |
110528 | Discontinued Procedure Step rescheduled | A new Procedure Step has been scheduled to replace the Discontinued Procedure Step. |
110529 | Discontinued Procedure Step rescheduling recommended | It is recommended that a new Procedure Step be scheduled to replace the Discontinued Procedure Step. |
110700 | Ventral Diencephalon | Ventral structures of the diencephalon that cannot readily be distinguished on MR imaging, including the hypothalamus, mammillary body, subthalamic nuclei, substantia nigra, red nucleus, lateral geniculate nucleus, medial geniculate nucleus, zona incerta, cerebral peduncle, lenticular fasciculus, medial lemniscus, and optic tract. See http://neuromorphometrics.org:8080/Seg/html/segmentation/ventral%20diencephalon.html. |
110701 | White Matter T1 Hypointensity | Area(s) of reduced intensity on T1 weighted images relative to the surrounding white matter. These may be indicative of age-related or neurodegenerative white matter lesions, and may be co-located with areas of white matter T2 hyperintensity, but the concept is specifically confined to the MR appearance on T1 weighted images. |
110702 | White Matter T2 Hyperintensity | Area(s) of increased intensity on T2 weighted images relative to the surrounding white matter. These may be indicative of age-related or neurodegenerative white matter lesions, and may be co-located with areas of white matter T1 hypointensity, but the concept is specifically confined to the MR appearance on T2 weighted images. |
110800 | Spin Tagging Perfusion MR Signal Intensity | Signal intensity of a Spin tagging Perfusion MR image. Spin tagging is a technique for the measurement of blood perfusion, based on magnetically labeled arterial blood water as an endogenous tracer. |
110801 | Contrast Agent Angio MR Signal Intensity | Signal intensity of a Contrast Agent Angio MR image. |
110802 | Time Of Flight Angio MR Signal Intensity | Signal intensity of a Time-of-flight (TOF) MR image. Time-of-flight (TOF) is based on the phenomenon of flow-related enhancement of spins entering into an imaging slice. As a result of being unsaturated, these spins give more signal that surrounding stationary spins. |
110803 | Proton Density Weighted MR Signal Intensity | Signal intensity of a Proton Density Weighted MR image. All MR images have intensity proportional to proton density. Images with very little T1 or T2 weighting are called 'PD-weighted'. |
110804 | T1 Weighted MR Signal Intensity | Signal intensity of T1 Weighted MR image. A T1 Weighted MR image is created typically by using short TE and TR times. |
110805 | T2 Weighted MR Signal Intensity | Signal intensity of a T2 Weighted MR image. T2 Weighted image contrast state is approached by imaging with a TR long compared to tissue T1 (to reduce T1 contribution to image contrast) and a TE between the longest and shortest tissue T2s of interest. |
110806 | T2* Weighted MR Signal Intensity | Signal intensity of a T2* Weighted MR image. The T2* phenomenon results from molecular interactions (spin spin relaxation) and local magnetic field non-uniformities, which cause the protons to precess at slightly different frequencies. |
110807 | Field Map MR Signal Intensity | Signal intensity of a Field Map MR image. A Field Map MR image provides a direct measure of the B 0 inhomogeneity at each point in the image. |
110808 | Fractional Anisotropy | Coefficient reflecting the fractional anisotropy of the tissues, derived from a diffusion weighted MR image. Fractional anisotropy is proportional to the square root of the variance of the Eigen values divided by the square root of the sum of the squares of the Eigen values. |
110809 | Relative Anisotropy | Coefficient reflecting the relative anisotropy of the tissues, derived from a diffusion weighted MR image. |
110810 | Volumetric Diffusion Dxx Component | Dxx Component of the diffusion tensor, quantifying the molecular mobility along the X axis. |
110811 | Volumetric Diffusion Dxy Component | Dxy Component of the diffusion tensor, quantifying the correlation of molecular displacements in the X and Y directions. |
110812 | Volumetric Diffusion Dxz Component | Dxz Component of the diffusion tensor, quantifying the correlation of molecular displacements in the X and Z directions. |
110813 | Volumetric Diffusion Dyy Component | Dyy Component of the diffusion tensor, quantifying the molecular mobility along the Y axis. |
110814 | Volumetric Diffusion Dyz Component | Dyz Component of the diffusion tensor, quantifying the correlation of molecular displacements in the Y and Z directions. |
110815 | Volumetric Diffusion Dzz Component | Dzz Component of the diffusion tensor, quantifying the molecular mobility along the Z axis. |
110816 | T1 Weighted Dynamic Contrast Enhanced MR Signal Intensity | Signal intensity of a T1 Weighted Dynamic Contrast Enhanced MR image. A T1 Weighted Dynamic Contrast Enhanced MR image reflects the dynamics of diffusion of the exogenous contrast media from the blood pool into the extra vascular extracellular space (EES) of the brain at a rate determined by the blood flow to the tissue, the permeability of the Brain Blood Barrier (BBB), and the surface area of the perfusing vessels. |
110817 | T2 Weighted Dynamic Contrast Enhanced MR Signal Intensity | Signal intensity of a T2 Weighted Dynamic Contrast Enhanced MR image. A T2 Weighted Dynamic Contrast Enhanced MR image reflects the T2 of tissue decrease as the Gd contrast agent bolus passes through the brain. |
110818 | T2* Weighted Dynamic Contrast Enhanced MR Signal Intensity | Signal intensity of a T2* Weighted Dynamic Contrast Enhanced MR image. A T2* Weighted Dynamic Contrast Enhanced MR image reflects the T2* of tissue decrease as the Gd contrast agent bolus passes through the brain. |
110819 | Blood Oxygenation Level | Signal intensity of a Blood Oxygenation Level image. BOLD imaging is sensitive to blood oxygenation (but also to cerebral blood flow and volume). This modality is essentially used for detecting brain activation (functional MR). |
110820 | Nuclear Medicine Projection Activity | Accumulated decay event counts in a nuclear medicine projection image. |
110821 | Nuclear Medicine Tomographic Activity | Accumulated decay event counts in a Nuclear Medicine Tomographic image (including PET). |
110822 | Spatial Displacement X Component | Spatial Displacement along axis X of a non linear deformable spatial registration image. The X axis is defined in reference to the patient's orientation, and is increasing to the left hand side of the patient. |
110823 | Spatial Displacement Y Component | Spatial Displacement along axis Y of a non linear deformable spatial registration image. The Y axis is defined in reference to the patient's orientation, and is increasing to the posterior side of the patient. |
110824 | Spatial Displacement Z Component | Spatial Displacement along axis Z of a Non linear deformable spatial registration image. The Z axis is defined in reference to the patient's orientation, and is increasing toward the head of the patient. |
110825 | Hemodynamic Resistance | Measured resistance to the flow of blood. E.g., through the vasculature or through a heart value. |
110826 | Indexed Hemodynamic Resistance | Measured resistance to the flow of blood. E.g., through the vasculature or through a heart value, normalized to a particular indexed scale. |
110827 | Tissue Velocity | Velocity of tissue based on Doppler measurements. |
110828 | Flow Velocity | Velocity of blood flow based on Doppler measurements. |
110829 | Flow Variance | Statistical variance of blood velocity relative to mean. |
110830 | Elasticity | Scalar value related to the elastic properties of the tissue. |
110831 | Perfusion | Scalar value related to the volume of blood perfusing into tissue. |
110832 | Speed of sound | Speed of sound in tissue. |
110833 | Ultrasound Attenuation | Reduction in strength of ultrasound signal as the wave. |
110834 | RGB R Component | Red component of a true color image (RGB). |
110835 | RGB G Component | Green component of a true color image (RGB). |
110836 | RGB B Component | Blue component of a true color image (RGB). |
110837 | YBR FULL Y Component | Y (Luminance) component of a YBR FULL image, as defined in JPEG 2000. |
110838 | YBR FULL CB Component | CB (Blue chrominance) component of a YBR FULL image, as defined in JPEG 2000. |
110839 | YBR FULL CR Component | CR (Red chrominance) component of a YBR FULL image, as defined in JPEG 2000. |
110840 | YBR PARTIAL Y Component | Y (Luminance) component of a YBR PARTIAL image, as defined in JPEG 2000. |
110841 | YBR PARTIAL CB Component | CB (Blue chrominance) component of a YBR PARTIAL image, as defined in JPEG 2000. |
110842 | YBR PARTIAL CR Component | CR (Red chrominance) component of a YBR PARTIAL image, as defined in JPEG 2000. |
110843 | YBR ICT Y Component | Y (Luminance) component of a YBR ICT image (Irreversible Color Transform), as defined in JPEG 2000. |
110844 | YBR ICT CB Component | CB (Blue chrominance) component of a YBR ICT image (Irreversible Color Transform), as defined in JPEG 2000. |
110845 | YBR ICT CR Component | CR (Red chrominance) component of a YBR ICT image (Irreversible Color Transform), as defined in JPEG 2000. |
110846 | YBR RCT Y Component | Y (Luminance) component of a YBR RCT image (Reversible Color Transform), as defined in JPEG 2000. |
110847 | YBR RCT CB Component | CB (Blue chrominance) component of a YBR RCT image (Reversible Color Transform), as defined in JPEG 2000. |
110848 | YBR RCT CR Component | CR (Red chrominance) component of a YBR RCT image (Reversible Color Transform), as defined in JPEG 2000. |
110849 | Echogenicity | The ability of a material to create an ultrasound return echo. |
110850 | X-Ray Attenuation | Decrease in the number of photons in an X-Ray beam due to interactions with the atoms of a material substance. Attenuation is due primarily to two processes, absorption and scattering. |
110851 | X-Ray Attenuation Coefficient | Coefficient that describes the fraction of a beam of X-Rays or gamma rays that is absorbed or scattered per unit thickness of the absorber. This value basically accounts for the number of atoms in a cubic cm volume of material and the probability of a photon being scattered or absorbed from the nucleus or an electron of one of these atoms. | Retired. Replaced by (112031, DCM, "Attenuation Coefficient").
110852 | MR signal intensity | Signal intensity of an MR image, not otherwise specified. |
110853 | Binary Segmentation | Binary value denoting that the segmented property is present. |
110854 | Fractional Probabilistic Segmentation | Probability, defined as a percentage, that the segmented property occupies the spatial area defined by the voxel. |
110855 | Fractional Occupancy Segmentation | Percentage of the voxel area occupied by the segmented property. |
110856 | Linear Displacement | Spatial dimension, denoting a linear displacement. |
110857 | Photon Energy | Dimension denoting the energy (frequency or wavelength) of photons. |
110858 | Time | Dimension used to sequence events, to compare the duration of events and the intervals between events. |
110859 | Angle | Spatial dimension, denoting an angle. |
110860 | Left-Right Axis | A spatial dimension axis running along a line between the patient's left and right side. |
110861 | Head-Foot Axis | A spatial dimension axis running along a line between the patient's head and foot. |
110862 | Anterior-Posterior Axis | A spatial dimension axis running along a line between the patient's anterior and posterior sides. |
110863 | Apex-Base Axis | A spatial dimension axis running along a line between the apex and base of an organ, object, or chamber. |
110864 | Anterior-Inferior Axis | A spatial dimension axis running along a line between the anterior and inferior sides of an organ, object, or chamber. |
110865 | Septum-Wall Axis | A spatial dimension axis running along a line between the septum and wall of a chamber. |
110866 | Right To Left | Orientation of a spatial dimension where increasing values run from the right to the left side of the patient. |
110867 | Left To Right | Orientation of a spatial dimension where increasing values run from the left to the right side of the patient. |
110868 | Head To Foot | Orientation of a spatial dimension where increasing values run from the head to the foot of the patient. |
110869 | Foot To Head | Orientation of a spatial dimension where increasing values run from the foot to the head of the patient. |
110870 | Anterior To Posterior | Orientation of a spatial dimension where increasing values run from the anterior to the posterior side of the patient. |
110871 | Posterior To Anterior | Orientation of a spatial dimension where increasing values run from the posterior to the anterior side of the patient. |
110872 | Apex To Base | Orientation of a spatial dimension where increasing values run from the apex to the base. |
110873 | Base To Apex | Orientation of a spatial dimension where increasing values run from the base to the apex. |
110874 | Anterior To Inferior | Orientation of a spatial dimension where increasing values run from the anterior to the inferior. |
110875 | Inferior To Anterior | Orientation of a spatial dimension where increasing values run from the inferior to the anterior. |
110876 | Septum To Wall | Orientation of a spatial dimension where increasing values run from the septum of a chamber to the opposite wall. |
110877 | Wall To Septum | Orientation of a spatial dimension where increasing values run from the opposite wall to the septum of a chamber. |
110901 | Image Position (Patient) X | The x coordinate of the upper left hand corner (center of the first voxel transmitted) of the image, with respect to the patient-based coordinate system. |
110902 | Image Position (Patient) Y | The y coordinate of the upper left hand corner (center of the first voxel transmitted) of the image, with respect to the patient-based coordinate system. |
110903 | Image Position (Patient) Z | The z coordinate of the upper left hand corner (center of the first voxel transmitted) of the image, with respect to the patient-based coordinate system. |
110904 | Image Orientation (Patient) Row X | The x value of the first row direction cosine with respect to the patient, with respect to the patient-based coordinate system. |
110905 | Image Orientation (Patient) Row Y | The y value of the first row direction cosine with respect to the patient, with respect to the patient-based coordinate system. |
110906 | Image Orientation (Patient) Row Z | The z value of the first row direction cosine with respect to the patient, with respect to the patient-based coordinate system. |
110907 | Image Orientation (Patient) Column X | The x value of the first column direction cosine with respect to the patient, with respect to the patient-based coordinate system. |
110908 | Image Orientation (Patient) Column Y | The y value of the first column direction cosine with respect to the patient, with respect to the patient-based coordinate system. |
110909 | Image Orientation (Patient) Column Z | The z value of the first column direction cosine with respect to the patient, with respect to the patient-based coordinate system. |
110910 | Pixel Data Rows | Number of rows in the pixel data of the image. |
110911 | Pixel Data Columns | Number of columns in the pixel data of the image. |
111001 | Algorithm Name | The name assigned by a manufacturer to a specific software algorithm. |
111002 | Algorithm Parameters | The input parameters used by a manufacturer to configure the behavior of a specific software algorithm. |
111003 | Algorithm Version | The software version identifier assigned by a manufacturer to a specific software algorithm. |
111004 | Analysis Performed | The type of correlation applied to detection results. E.g., temporal, spatial. |
111005 | Assessment Category | Assignment of intermediate or overall interpretation results to a general category. |
111006 | Breast composition | Assessment of annotating tissues in breast; generally including fatty, mixed or dense | Retired. Replaced by (F-01710, SRT, "Breast composition").
111007 | Breast Outline including Pectoral Muscle Tissue | Purpose of reference for an SCOORD content item that is an outline of the breast that includes the pectoral muscle tissue | Purpose of Reference for content item of value type COMPOSITE or SCOORD
111008 | Calcification Distribution | The type of distribution associated with detected calcifications. |
111009 | Calcification Type | Identification of the morphology of detected calcifications. |
111010 | Center | Purpose of reference for an SCOORD content item that identifies the central point of a finding or feature | Purpose of Reference for content item of value type COMPOSITE or SCOORD
111011 | Certainty of Feature | The likelihood that the feature analyzed is in fact the type of feature identified. |
111012 | Certainty of Finding | The likelihood that the finding detected is in fact the type of finding identified. |
111013 | Certainty of Impression | The certainty that a device places on an impression, where 0 equals no certainty and 100 equals certainty. |
111014 | Clockface or region | A location identifier based on clockface numbering or anatomic subregion. |
111015 | Composite Feature | An item that is an inferred correlation relating two or more individual findings or features. |
111016 | Composite type | The inferred relationship between the findings or features making up a composite feature. |
111017 | CAD Processing and Findings Summary | General assessment of whether or not CAD processing was successful, and whether any findings resulted. |
111018 | Content Date | The date the data creation started. |
111019 | Content Time | The time the data creation started. |
111020 | Depth | A location identifier based on a feature's inferred distance from the surface of the associated anatomy. |
111021 | Description of Change | A textual description of the change that occurred over time in a qualitative characteristic of a feature. |
111022 | Detection Performed | The type of finding sought after by a specific algorithm applied to one image. |
111023 | Differential Diagnosis/Impression | A general change that occurred within an imaged area between a prior imaging procedure and the current imaging procedure. |
111024 | Failed Analyses | A group of analysis algorithms that were attempted, but failed. |
111025 | Failed Detections | A group of detection algorithms that were attempted, but failed. |
111026 | Horizontal Pixel Spacing | For projection radiography, the horizontal physical distance measured at the front plane of an Image Receptor housing between the center of each pixel. For tomographic images, the horizontal physical distance in the patient between the center of each pixel. |
111027 | Image Laterality | Laterality of (possibly paired) body part contained in an image. |
111028 | Image Library | A container that references all image data used as evidence to produce a report. |
111029 | Image Quality Rating | A numeric value in the range 0 to 100, inclusive, where 0 is worst quality and 100 is best quality. |
111030 | Image Region | Purpose of reference for an SCOORD content item that identifies a specific region of interest within an image | Purpose of Reference for content item of value type COMPOSITE or SCOORD
111031 | Image View | The projection of the anatomic region of interest on an image receptor. |
111032 | Image View Modifier | Modifier for Image View. |
111033 | Impression Description | Free-form text describing the overall or an individual impression. |
111034 | Individual Impression/Recommendation | A container for a group of related results from interpretation of one or more images and associated clinical information. |
111035 | Lesion Density | The X-Ray attenuation of a lesion relative to the expected attenuation of an equal volume of fibroglandular breast tissue. |
111036 | Mammography CAD Report | A structured report containing the results of computer-aided detection or diagnosis applied to breast imaging and associated clinical information. |
111037 | Margins | The characteristic of the boundary, edges or border of a detected lesion. |
111038 | Number of calcifications | The quantity of calcifications detected within an identified group or cluster. |
111039 | Object type | A non-lesion object identified within one or more images. |
111040 | Original Source | Purpose of reference for a COMPOSITE content item that identifies it as the original source of evidence for another content item in the report | Purpose of Reference for content item of value type COMPOSITE or SCOORD
111041 | Outline | Purpose of reference for an SCOORD content item that identifies the outline or bounding region of a finding or feature | Purpose of Reference for content item of value type COMPOSITE or SCOORD
111042 | Pathology | The inferred type of disease associated with an identified feature. |
111043 | Patient Orientation Column | The patient orientation relative to the image plane, specified by a value that designates the anatomical direction of the positive column axis (top to bottom). |
111044 | Patient Orientation Row | The patient orientation relative to the image plane, specified by a value that designates the anatomical direction of the positive row axis (left to right). |
111045 | Pectoral Muscle Outline | Purpose of reference for an SCOORD content item that is an outline of the pectoral muscle tissue only | Purpose of Reference for content item of value type COMPOSITE or SCOORD
111046 | Percent Fibroglandular Tissue | Percent of breast area that is mammographically dense, excluding pectoralis muscle. |
111047 | Probability of cancer | The likelihood that an identified finding or feature is cancerous. |
111048 | Quadrant location | A location identifier based on the division of an area into four regions. |
111049 | Qualitative Difference | A qualitative characteristic of a feature that has changed over time. |
111050 | Quality Assessment | The effect of the quality of an image on its usability. |
111051 | Quality Control Standard | The quality control standard used to make a quality assessment. |
111052 | Quality Finding | A specific quality related deficiency detected within an image. |
111053 | Recommended Follow-up | Recommended type of follow-up to an imaging procedure, based on interpreted results. |
111054 | Recommended Follow-up Date | Recommended follow-up date to an imaging procedure, based on interpreted results. |
111055 | Recommended Follow-up Interval | Recommended follow-up interval to an imaging procedure, based on interpreted results. |
111056 | Rendering Intent | The recommendation of the producer of a content item regarding presentation of the content item by recipients of the report. |
111057 | Scope of Feature | An indication of how widespread the detection of a feature is within the analyzed image data set. |
111058 | Selected Region Description | A textual description of the contents of a selected region identified within an image. |
111059 | Single Image Finding | An item that was detected on one image. |
111060 | Study Date | Date on which the acquisition of the study information was started. |
111061 | Study Time | Time at which the acquisition of the study information was started. |
111062 | Successful Analyses | A group of analysis algorithms that were attempted and completed successfully. |
111063 | Successful Detections | A group of detection algorithms that were attempted and completed successfully. |
111064 | Summary of Detections | An overall indication of whether the CAD detection algorithms applied were completed successfully. |
111065 | Summary of Analyses | An overall indication of whether the CAD analysis algorithms applied were completed successfully. |
111066 | Vertical Pixel Spacing | For projection radiography, the vertical physical distance measured at the front plane of an Image Receptor housing between the center of each pixel. For tomographic images, the vertical physical distance in the patient between the center of each pixel. |
111069 | Crosstable | A radiographic projection that has been with the patient lying on a table with the X-Ray source on one side of the table and the detector on the other. E.g., may describe a cross-table cervical spine, chest or pelvis X-Ray image. |
111071 | CAD Operating Point | One of a number of discrete points on the Receiver-Operator Characteristics (ROC) curve that reflects the expected sensitivity and specificity of a CAD algorithm, where zero indicates the highest specificity, lowest sensitivity operating point. The value should not exceed the Maximum CAD Operating Point. |
111072 | Maximum CAD Operating Point | The maximum value of CAD Operating Point for the specific CAD algorithm used. |
111081 | CAD Operating Point Description | The intended interpretation of a CAD Operating Point. |
111086 | False Markers per Image | The number of false CAD markers per image. Correlates to inverse of Image Specificity. |
111087 | False Markers per Case | The number of false markers per collection of images that are CAD processed as a group. Correlates to inverse of Case Specificity. |
111088 | Case Sensitivity | The percentage of cancers that should be detected by a CAD algorithm where CAD marks the cancers in at least one view. |
111089 | Lesion Sensitivity | The percentage of cancers that should be detected by a CAD algorithm where CAD marks the cancers in each view. |
111090 | Case Specificity | The percentage of cases (collections of images CAD processed as a group) without cancer that have no CAD findings whatsoever. Correlates to inverse of False Markers per Case. |
111091 | Image Specificity | The percentage of images without cancer that have no CAD findings whatsoever. Correlates to inverse of False Markers per Image. |
111092 | Recommended CAD Operating Point | The CAD operating point that is recommended for initial display by the creator of the structured report. |
111093 | CAD Operating Point Table | A list of CAD operating points including their corresponding characteristics. |
111099 | Selected region | A specific area of interest noted within an image. |
111100 | Breast geometry | The surface shape of all or a portion of breast related anatomy. |
111101 | Image Quality | Image quality incorporates the following clinical image evaluation parameters: assessment of positioning, compression, artifacts, exposure, contrast, sharpness, and labeling. |
111102 | Non-lesion | A finding or feature that is identified as a non-anatomic foreign object. |
111103 | Density | A space-occupying lesion identified in a single image or projection | Retired. Replaced by (F-01796, SRT, "Mammography breast density").
111104 | Individual Calcification | A single identified calcification | Retired. Replaced by (F-01776, SRT, "Individual Calcification").
111105 | Calcification Cluster | Multiple calcifications identified as occupying a small area of tissue (less than 2 cc) | Retired. Replaced by (F-01775, SRT, "Calcification Cluster").
111111 | Cooper's ligament changes | Straightening or thickening of Cooper's ligaments. |
111112 | Mass in the skin | An abnormality noted at imaging within the dermis of the breast. |
111113 | Mass on the skin | An abnormality noted at imaging on the epidermis of the breast. |
111120 | Post Procedure Mammograms for Marker Placement | An assessment category to indicate that images have been acquired to assess marker placement following a breast interventional procedure. |
111121 | Follow-up post biopsy as directed by clinician | An indication that the patient should seek post procedural follow-up directives from a clinical health care provider. |
111122 | Known biopsy proven malignancy - take appropriate action | A recommendation on a patient with known cancer to take steps appropriate to the diagnosis. |
111123 | Marker placement | Positioning of a radiopaque marker. |
111124 | Personal history of breast cancer with mastectomy | Patient has previous diagnosis of breast cancer resulting in mastectomy. |
111125 | Known biopsy proven malignancy | Patient has had biopsy containing proven malignancy. |
111126 | Image detected mass | Patient has a finding of mass reported on a prior imaging exam. |
111127 | Targeted | A breast imaging procedure performed on a specific area of the breast. |
111128 | Survey | A breast imaging procedure performed on the entire breast. |
111129 | Clustered microcysts | A cluster of tiny anechoic foci each smaller than 2-3 mm in diameter with thin (less than 0.5 mm) intervening septations and no discrete solid components. |
111130 | Complicated cyst | A fluid filled mass most commonly characterized by homogeneous low-level internal echoes on ultrasound. |
111135 | Additional projections | Views not inclusive of MLO and CC (BI-RADS®). |
111136 | Spot magnification view(s) | A spot or coned down compression of the breast providing a reduction in the thickness and a magnification of the localized area of interest and improved separation of breast tissue. |
111137 | Ultrasound |  | Retired. Replaced by (P5-B0099, SRT, "Ultrasound procedure").
111138 | Old films for comparison | Obtain previous mammography studies to compare to present study. |
111139 | Ductography | A medical procedure used for the sampling of mammary duct tissue | Retired. Replaced by (P5-40060, SRT, "Mammary ductogram").
111140 | Normal interval follow-up | Follow up study at 12 months for women ≥ 40 years of age having a prior negative study and no mitigating risk factors for breast cancer. |
111141 | Any decision to biopsy should be based on clinical assessment | Any decision to perform tissue acquisition should be based on clinical assessment. |
111142 | Follow-up at short interval (1-11 months) | Follow-up at short interval (1-11 months). |
111143 | Biopsy should be considered | Tissue acquisition should be considered. |
111144 | Needle localization and biopsy | Breast tissue acquisition following the identification of an area of concern with the placement of a needle or needle-wire assembly. |
111145 | Histology using core biopsy | Pathologic analysis of breast tissue and lesions using core tissue samples. |
111146 | Suggestive of malignancy - take appropriate action | Lesions that do not have the characteristic morphologies of breast cancer but have a definite probability of being malignant. There is a sufficient concern to urge a biopsy. |
111147 | Cytologic analysis | Cellular analysis of specimen. |
111148 | Biopsy should be strongly considered | Tissue acquisition should be strongly considered. |
111149 | Highly suggestive of malignancy - take appropriate action | Lesions have a high probability of being cancer, which require additional action. |
111150 | Presentation Required: Rendering device is expected to present | The producer of a report intends for a recipient of the report to present or display the associated content item. |
111151 | Presentation Optional: Rendering device may present | The producer of a report considers the presentation or display of the associated content item by a recipient to be optional. |
111152 | Not for Presentation: Rendering device expected not to present | The producer of a report intends for a recipient of the report NOT to present or display the associated content item. |
111153 | Target content items are related temporally | The associated content items are identified as being the same finding or feature at different points in time. |
111154 | Target content items are related spatially | The associated content items are identified as being the same finding or feature on different projections taken at the same point in time. |
111155 | Target content items are related contra-laterally | The associated content items are identified as being related side-to-side. |
111156 | Feature detected on the only image | There is one image in the interpreted data set. |
111157 | Feature detected on only one of the images | There is more than one image of the same modality in the interpreted data set. |
111158 | Feature detected on multiple images | There is more than one image of the same modality in the interpreted data set. |
111159 | Feature detected on images from multiple modalities | The interpreted data set contains images from multiple modalities. |
111168 | Scar tissue | The fibrous tissue replacing normal tissues destroyed by disease or injury | Retired. Replaced by (M-78060, SRT, "Scar tissue").
111170 | J Wire | A medical appliance used for localization of non palpable breast lesions to insure that the proper area is removed in a surgical biopsy | Retired. Replaced by (A-1016B, SRT, "J Wire").
111171 | Pacemaker | A medical appliance used for regulating cardiac rhythms | Retired. Replaced by (A-11101, SRT, "Cardiac Pacemaker").
111172 | Paddle | A compression device used for obtaining mammographic images | Retired. Replaced by (A-10042, SRT, "Compression paddle").
111173 | Collimator | A device used for restricting an X-Ray beam | Retired. Replaced by (A-10044, SRT, "Collimator").
111174 | ID Plate | An area designated on a radiographic film for facility and patient ID information | Retired. Replaced by (A-16016, SRT, "ID Plate").
111175 | Other Marker | Site specific markers. |
111176 | Unspecified | The value of the concept is not specified | This term may not be used in Context Group Extensions; see Section 7.2.3
111177 | View and Laterality Marker is missing | Image quality deficiency according to MQSA. |
111178 | View and Laterality Marker does not have both view and laterality | Image quality deficiency according to MQCM. |
111179 | View and Laterality Marker does not have approved codes | Image quality deficiency according to MQCM. |
111180 | View and Laterality Marker is not near the axilla | Image quality deficiency according to MQCM. |
111181 | View and Laterality Marker overlaps breast tissue | Image quality deficiency according to MQCM. |
111182 | View and Laterality Marker is partially obscured | Image quality deficiency according to MQCM. |
111183 | View and Laterality Marker is incorrect | Image quality deficiency. |
111184 | View and Laterality Marker is off image | Image quality deficiency. |
111185 | Flash is not near edge of film | Image quality deficiency according to MQCM. |
111186 | Flash is illegible, does not fit, or is lopsided | Image quality deficiency according to MQSA. |
111187 | Flash doesn't include patient name and additional patient id | Image quality deficiency according to MQCM. |
111188 | Flash doesn't include date of examination | Image quality deficiency according to MQCM. |
111189 | Flash doesn't include facility name and location | Image quality deficiency according to MQSA. |
111190 | Flash doesn't include technologist identification | Image quality deficiency according to MQCM. |
111191 | Flash doesn't include cassette/screen/detector identification | Image quality deficiency according to MQCM. |
111192 | Flash doesn't include mammography unit identification | Image quality deficiency according to MQCM. |
111193 | Date sticker is missing | Image quality deficiency according to MQCM. |
111194 | Technical factors missing | Image quality deficiency according to MQCM. |
111195 | Collimation too close to breast | Image quality deficiency according to MQCM. |
111196 | Inadequate compression | Image quality deficiency according to MQCM. |
111197 | MLO Insufficient pectoral muscle | Image quality deficiency according to MQCM. |
111198 | MLO No fat is visualized posterior to fibroglandular tissues | Image quality deficiency according to MQCM. |
111199 | MLO Poor separation of deep and superficial breast tissues | Image quality deficiency according to MQCM. |
111200 | MLO Evidence of motion blur | Image quality deficiency according to MQCM. |
111201 | MLO Inframammary fold is not open | Image quality deficiency according to MQCM. |
111202 | CC Not all medial tissue visualized | Image quality deficiency according to MQCM. |
111203 | CC Nipple not centered on image | Image quality deficiency according to MQCM. |
111204 | CC Posterior nipple line does not measure within 1 cm of MLO | Image quality deficiency according to MQCM. |
111205 | Nipple not in profile | Image quality deficiency. |
111206 | Insufficient implant displacement incorrect | Image quality deficiency according to MQCM. |
111207 | Image artifact(s) | Signals that do not faithfully reproduce actual anatomic structures because of distortion or of addition or deletion of information. |
111208 | Grid artifact(s) | Feature(s) arising from the acquisition unit's anti-scatter grid mechanism. For two-dimensional systems, such features include those of mechanically damaged or incorrectly positioned grids. For moving or Bucky grids, artifacts may result from intentional grid motion that is inadequate in duration or velocity uniformity. |
111209 | Positioning | Inadequate arrangement of the anatomy of interest with respect to the X-Ray field and image detector sensitive area. Examples: 1) positioning is "cutoff" when the projection of anatomy of interest falls outside the sensitive area of the detector; 2) "cone cut", in which the X-Ray field does not adequately cover the anatomy of interest; 3) detector's sensitive surface is too small to cover the projection of the anatomy of interest; 4) improper angular orientation or "rotation" of anatomy of interest with respect to the X-Ray source, or detector; 5) projection of other anatomy or clothing over the anatomy of interest in the image. |
111210 | Motion blur | Unacceptable image blur resulting from motion of the anatomy of interest during exposure or the inadequately compensated motion of X-Ray source with respect to the image detector during exposure. |
111211 | Under exposed | Inadequate number of quanta reached the detector during exposure. Reasons for under exposed images include low kVp, low mAs product, excess Source Image Distance. Under exposed images have inadequate signal and higher noise in the areas of interest. |
111212 | Over exposed | An excess number of quanta reached the detector during exposure. Reasons for over exposed images include high kVp, high mAs product, short Source Image Distance. Over exposed images have high signal and lower noise in the areas of interest. Over exposed area may demonstrate lack of contrast from over saturation of the detector. |
111213 | No image | No evidence of a patient exposure. |
111214 | Detector artifact(s) | Superposed features or flaws of the detector. |
111215 | Artifact(s) other than grid or detector artifact | Features or discontinuities arising from causes other than the anti-scatter grid and image detector. |
111216 | Mechanical failure | Failure of the device to operate according to mechanical design specifications. |
111217 | Electrical failure | Failure of a device to operate according to electrical design specifications. |
111218 | Software failure | Attributable to software used in generation or handling of image. |
111219 | Inappropriate image processing | Images processed inappropriately, not following appropriate protocol. |
111220 | Other failure | Failure that is not mechanical or electrical or otherwise described. |
111221 | Unknown failure | Unidentified or unknown cause of failure. |
111222 | Succeeded | The attempted process was completely successful. |
111223 | Partially Succeeded | The attempted process succeeded in some ways, but failed in others. |
111224 | Failed | The attempted process completely failed. |
111225 | Not Attempted | No process was performed. |
111233 | Individual Impression / Recommendation Analysis | Analysis of a related group of findings or features detected during image data inspection, to produce a summary impression and/or recommendation. |
111234 | Overall Impression / Recommendation Analysis | Analysis of all groups of findings or features, to produce a single impression and/or recommendation. |
111235 | Unusable - Quality renders image unusable | The usability of an image for diagnostic interpretation or CAD, based on a quality control standard. |
111236 | Usable - Does not meet the quality control standard | The usability of an image for diagnostic interpretation or CAD, based on a quality control standard. |
111237 | Usable - Meets the quality control standard | The usability of an image for diagnostic interpretation or CAD, based on a quality control standard. |
111238 | Mammography Quality Control Manual 1999, ACR | An image quality control standard specified by the American College of Radiology. |
111239 | Title 21 CFR Section 900, Subpart B | An image quality control standard in the US Code of Federal Regulations. |
111240 | Institutionally defined quality control standard | An image quality control standard specified or adopted by the institution responsible for the document. |
111241 | All algorithms succeeded; without findings | No findings resulted upon successful completion of all attempted computer-aided detection and/or analysis. |
111242 | All algorithms succeeded; with findings | One or more findings resulted upon successful completion of all attempted computer-aided detection and/or analysis. |
111243 | Not all algorithms succeeded; without findings | No findings resulted from the attempted computer-aided detection and/or analysis, but one or more failures occurred in the process. |
111244 | Not all algorithms succeeded; with findings | One or more findings resulted from the attempted computer-aided detection and/or analysis, but one or more failures occurred in the process. |
111245 | No algorithms succeeded; without findings | All of the attempted computer-aided detection and/or analysis failed, so there could be no findings. |
111248 | Adenolipoma | A benign tumor having glandular characteristics but composed of fat, with the presence of normal mammary ducts | Retired. Replaced by (M-83240, SRT, "Adenolipoma").
111249 | Ductal hyperplasia |  | Retired. Replaced by (M-72170, SRT, "Ductal hyperplasia, Usual").
111250 | Adenomyoepithelioma | Neoplasms composed of myoepithelial cells | Retired. Replaced by (M-89830, SRT, "Adenomyoepithelioma").
111251 | Normal axillary node | Axillary node that is normal in appearance with no associated pathology. |
111252 | Axillary node with calcifications | Axillary node containing calcifications. |
111253 | Axillary node hyperplasia | Excessive proliferation of normal tissue arrangement of the axillary node. |
111254 | Asynchronous involution |  | Retired. Replaced by (F-8A063, SRT, "Asynchronous involution of breast").
111255 | Benign cyst with blood | Cyst with benign morphology containing blood. |
111256 | Benign Calcifications | Calcifications having typically benign morphology. They are not of intermediate or high probability of concern for malignancy. |
111257 | Intracystic papilloma | Growing within a cystic adenoma, filling the cavity with a mass of branching epithelial processes | Retired. Replaced by (M-85040, SRT, "Intracystic papilloma").
111258 | Ductal adenoma | Adenoma located in mammary duct, present as discrete sclerotic nodules, solitary or multiple. |
111259 | Diabetic fibrous mastopathy | The occurrence of fibrous tumor-forming stromal proliferation in patients with diabetes mellitus. |
111260 | Extra abdominal desmoid | A deep seated firm tumor frequently occurring on the chest consisting of collagenous tissue that infiltrates surround muscle; frequently recurs but does not metastasize | Retired. Replaced by (M-88211, SRT, "Extra abdominal desmoid").
111262 | Epidermal inclusion cyst | A cyst formed of a mass of epithelial cells, as a result of trauma has been pushed beneath the epidermis. The cyst is lined with squamous epithelium and contains concentric layers or keratin | Retired. Replaced by (M-33410, SRT, "Epidermal inclusion cyst").
111263 | Fibroadenomatoid hyperplasia | Excessive proliferation of fibroadenoma tissue. |
111264 | Fibroadenolipoma | A lipoma with an abundant stroma of fibrous tissue. |
111265 | Foreign body (reaction) |  | Retired. Replaced by (M-44140, SRT, "Foreign body (reaction) ").
111269 | Galactocele | Retention cyst caused by occlusion of a lactiferous duct | Retired. Replaced by (D7-90364, SRT, "Galactocele").
111271 | Hemangioma - nonparenchymal, subcutaneous | A congenital anomaly that leads to a proliferation of blood vessels leading to a mass that resembles a neoplasm, not located in parenchymal areas but subcutaneous | Retired. Replaced by (D3-F0620, SRT, "Hemangioma of subcutaneous tissue").
111273 | Hyperplasia, usual |  | Retired. Replaced by (M-72000, SRT, "Hyperplasia, usual").
111277 | Juvenile papillomatosis | A form of fibrocystic disease in young woman with florid and sclerosing adenosis that microscopically may suggest carcinoma. |
111278 | Lactating adenoma | Enlarging masses during lactation. A circumscribed benign tumor composed primarily of glandular structures with scanty stroma, with prominent secretory changes in the duct | Retired. Replaced by (M-82040, SRT, "Lactating adenoma").
111279 | Lactational change | Changes related to the process of lactation. |
111281 | Large duct papilloma | A papilloma pertaining to large mammary duct. |
111283 | Myofibroblastoma | Solitary or multiple tumors of muscles and fibrous tissues, or tumors composed of myofibroblasts | Retired. Replaced by (M-88250, SRT, "Myofibroblastoma").
111284 | Microglandular adenosis | Irregular clusters of small tubules are present in adipose or fibrous tissue, resembling tubular carcinoma but lacking stromal fibroblastic proliferation. |
111285 | Multiple Intraductal Papillomas | Papilloma typically involving an aggregate of adjacent ducts in the periphery of the breast, likely representing involvement of several foci of one or two duct systems. |
111286 | No abnormality | No abnormality. |
111287 | Normal breast tissue | Normal breast tissue. |
111288 | Neurofibromatosis | Condition in which there are tumors of various sizes on peripheral nerves. They may be neuromas or fibromas | Retired. Replaced by (M-95401, SRT, "Neurofibromatosis").
111290 | Oil cyst (fat necrosis cyst) | A cyst resulting from the loss of the epithelial lining of a sebaceous dermoid or lacteal cyst. |
111291 | Post reduction mammoplasty | Breast tissue with characteristics of a benign nature, following breast reduction surgery. |
111292 | Pseudoangiomatous stromal hyperplasia | A benign stromal lesion composed of intermixed stromal and epithelial elements. The lobular and duct structures of the breast parenchyma are separated by an increased amount of stroma, non specific proliferative epithelial changes include hyperplasia of duct and lobular epithelium often with accentuation of myoepithelial cells and aprocine metaplasia with or without cyst formation. |
111293 | Radial scar | An nonencapsulated stellate lesion consisting of a fibroelastic core and radiating bands of fibrous connective tissue containing lobules manifesting adenosis and ducts with papillary or diffuse intraductal hyperplasia | Retired. Replaced by (M-78731, SRT, "Radial scar").
111294 | Sclerosing adenosis | Prominent interductal fibrosis of the terminal ductules | Retired. Replaced by (M-74220, SRT, "Sclerosing adenosis").
111296 | Silicone granuloma | Nodular inflammatory lesions due to the presence of silicone in the breast tissue. |
111297 | Nipple Characteristic | The morphologic status of the nipple. |
111298 | Virginal hyperplasia | Spontaneous excessive proliferation of breast tissue, usually found in younger women. |
111299 | Peripheral duct papillomas | Papilloma(s) pertaining the peripheral ducts. |
111300 | Axillary node with lymphoma | Axillary node with lymphoid tissue neoplasm. |
111301 | Axillary nodal metastases | Metastatic disease to the axillary node. |
111302 | Angiosarcoma | A malignant neoplasm occurring most often in breast and skin, believed to originate from endothelial cells of blood vessels, microscopically composed of closely packed round or spindle shaped cells, some of which line small spaces resembling vascular clefts | Retired. Replaced by (M-91203, SRT, "Angiosarcoma").
111303 | Blood vessel (vascular) invasion | Histological changes to the vascular system related to an invasive process. |
111304 | Carcinoma in children | Carcinoma of the breast found in patients less than 20 years of age. |
111305 | Carcinoma in ectopic breast | A carcinoma found in supernumerary breasts and aberrant breast tissue. |
111306 | Carcinoma with endocrine differentiation | A carcinoma that synthesizes substances, including hormones, not considered to be normal products of the breast. |
111307 | Basal cell carcinoma of nipple | A basal cell carcinoma that arises in the nipple of the breast. |
111308 | Carcinoma with metaplasia |  | Retired. Replaced by (M-85733, SRT, "Carcinoma with metaplasia").
111309 | Cartilaginous and osseous change | Tissue changes to bones and cartilage. |
111310 | Carcinoma in pregnancy and lactation | Carcinoma of the breast presenting during pregnancy or lactation. |
111311 | Carcinosarcoma | A malignant neoplasm that contains elements of carcinoma and sarcoma, so extensively intermixed as to indicate neoplasia of epithelial and mesenchymal tissue | Retired. Replaced by (M-89803, SRT, "Carcinosarcoma").
111312 | Intraductal comedocarcinoma with necrosis | Comedocarcinoma of a duct with areas of necrotic tissue. |
111313 | Intraductal carcinoma, low grade | A non-invasive carcinoma restricted to the glandular lumen characterized by less aggressive malignant cytologic features and behavior. |
111314 | Intraductal carcinoma micro-papillary |  | Retired. Replaced by (M-85072, SRT, "Intraductal carcinoma micro-papillary").
111315 | Intracystic papillary carcinoma | A malignant neoplasm characterized by the formation of numerous, irregular, finger-like projections of fibrous stroma that is covered with a surface layer of neoplastic epithelial cells found in a cyst. |
111316 | Invasive and in-situ carcinoma | Carcinoma with both characteristics of localized and spreading disease. |
111317 | Invasive lobular carcinoma |  | Retired. Replaced by (M-85203, SRT, "Invasive lobular carcinoma").
111318 | Leukemic infiltration | Mammary infiltrates as a secondary manifestation in patients with established leukemia. |
111320 | Lympathic vessel invasion | Histological changes to the lymphatic system related to an invasive process. |
111321 | Lymphoma | A heterogeneous group of neoplasms arising in the reticuloendoethelial and lymphatic systems | Retired. Replaced by (M-95903, SRT, "Lymphoma").
111322 | Occult carcinoma presenting with axillary lymph node metastases | A small carcinoma, either asymptomatic or giving rise to metastases without symptoms due to the primary carcinoma presenting with metastatic disease in the axillary lymph nodes. |
111323 | Metastatic cancer to the breast | A malignant lesion in the breast with morphologic patterns not typical of breast carcinoma arising from a non-mammary malignant neoplasm. |
111324 | Metastatic cancer to the breast from the colon | A malignant lesion in the breast with morphologic patterns not typical of breast carcinoma arising from a neoplasm in the colon. |
111325 | Metastatic cancer to the breast from the lung | A malignant lesion in the breast with morphologic patterns not typical of breast carcinoma arising from a neoplasm in the lung. |
111326 | Metastatic melanoma to the breast | A malignant lesion in the breast with morphologic patterns not typical of breast carcinoma arising from a melanoma. |
111327 | Metastatic cancer to the breast from the ovary | A malignant lesion in the breast with morphologic patterns not typical of breast carcinoma arising from a neoplasm in the ovary. |
111328 | Metastatic sarcoma to the breast | A malignant lesion in the breast with morphologic patterns not typical of breast carcinoma arising from a sarcoma. |
111329 | Multifocal intraductal carcinoma | Multiple foci of non-invasive carcinoma restricted to the glandular lumen. |
111330 | Metastatic disease to axillary node | A malignant lesion in an axillary node arising from a non-axillary neoplasm. |
111331 | Malignant fibrous histiocytoma |  | Retired. Replaced by (M-88303, SRT, "Malignant fibrous histiocytoma").
111332 | Multifocal invasive ductal carcinoma | Multiple sites of ductal carcinoma. |
111333 | Metastasis to an intramammary lymph node | A malignant lesion in a intramammary lymph node arising from a non-intramammary lymph node neoplasm. |
111334 | Malignant melanoma of nipple | A malignant melanoma of the skin that arises in the nipple of the breast. |
111335 | Neoplasm of the mammary skin |  | Retired. Replaced by (D0-F035F, SRT, "Neoplasm of the mammary skin").
111336 | Papillary carcinoma in-situ |  | Retired. Replaced by (M-80502, SRT, "Papillary carcinoma in-situ").
111338 | Recurrent malignancy | Recurrent malignancy. |
111340 | Squamous cell carcinoma of the nipple | Squamous cell carcinoma to the terminal portion of the alveolar. |
111341 | Intraductal carcinoma, high grade | A non-invasive carcinoma restricted to the glandular lumen characterized by more aggressive malignant cytologic features and behavior. |
111342 | Invasive cribriform carcinoma |  | Retired. Replaced by (M-82013, SRT, "Invasive cribriform carcinoma").
111343 | Angular margins | An indication that some or all of the margin of a lesion has sharp corners, often forming acute angles. |
111344 | Fine pleomorphic calcification | Calcifications that vary in sizes and shapes and are usually smaller than0.5 mm in diameter. |
111345 | Macrocalcifications | Coarse calcifications that are 0.5 mm or greater in size. |
111346 | Calcifications within a mass | An indicator that calcifications are imbedded within a mass. |
111347 | Calcifications outside of a mass | An indicator that calcifications are imaged outside of a mass finding. |
111350 | Breast background echotexture | Tissue composition of the breast noted on sonography. |
111351 | Homogeneous fat echotexture | Fat lobules and uniformly echogenic bands of supporting structures comprise the bulk of breast tissue. |
111352 | Homogeneous fibroglandular echotexture | A uniformly echogenic layer of fibroglandular tissue is seen beneath a thin layer of subcutaneous fat. |
111353 | Heterogeneous echotexture | The breast texture is characterized by multiple small areas of increased and decreased echogenicity. |
111354 | Orientation | Referential relationship of the finding to the imaging device as noted on sonography. |
111355 | Parallel | The long axis of a lesion parallels the skin line ("wider-than-tall" or in a horizontal orientation). |
111356 | Not parallel | The anterior-posterior or vertical dimension is greater than the transverse or horizontal dimension. |
111357 | Lesion boundary | The lesion boundary describes the transition zone between themas and the surrounding tissue. |
111358 | Abrupt interface | The sharp demarcation between the lesion and surrounding tissue can be imperceptible or a distinct well-defined echogenic rim of any thickness. |
111359 | Echogenic halo | There is no sharp demarcation between the mass and the surrounding tissue, which is bridged by an echogenic transition zone. |
111360 | Echo pattern | An imaging characteristic of resonance noted during sonography. |
111361 | Anechoic | Without internal echoes. |
111362 | Hyperechoic | Having increased echogenicity relative to fat or equal to fibroglandular tissue. |
111363 | Complex | Mass contains both anechoic and echogenic components. |
111364 | Hypoechoic | Defined relative to fat; masses are characterized by low-level echoes throughout. E.g., appearance of a complicated cyst or fibroadenoma. |
111365 | Isoechoic | Having the same echogenicity as fat (a complicated cyst or fibroadenoma may be isoechoic or hypoechoic). |
111366 | Posterior acoustic features | The attenuation characteristics of a mass with respect to its acoustic transmission. |
111367 | No posterior acoustic features | No posterior shadowing or enhancement. |
111368 | Posterior enhancement | Increased posterior echoes. |
111369 | Posterior shadowing | Decreased posterior echoes; edge shadows are excluded. |
111370 | Combined posterior enhancement and shadowing | More than one pattern of posterior attenuation, both shadowing and enhancement. |
111371 | Identifiable effect on surrounding tissues | Sonographic appearance of adjacent structures relative to a mass finding. |
111372 | Vascularity | Characterization of vascularization in region of interest. |
111373 | Vascularity not present | Vascularity not evident, such as on ultrasound. |
111374 | Vascularity not assessed | Vascularity not evaluated. |
111375 | Vascularity present in lesion | Vascularity on imaging is seen within a lesion. |
111376 | Vascularity present immediately adjacent to lesion | Vascularity on imaging is seen immediately adjacent to a lesion. |
111377 | Diffusely increased vascularity in surrounding tissue | Vascularity on imaging is considered diffusely elevated within the surrounding breast tissue. |
111380 | Correlation to other Findings | Relationship of the new anomaly to other clinical or imaging anomalies. |
111381 | Correlates to physical exam findings | An indication that the current imaging finding relates to a finding from a clinical breast exam. |
111382 | Correlates to mammography findings | An indication that the current imaging finding relates to a finding from a mammography exam. |
111383 | Correlates to MRI findings | An indication that the current imaging finding relates to a finding from a breast MRI exam. |
111384 | Correlates to ultrasound findings | An indication that the current imaging finding relates to a finding from a breast ultrasound exam. |
111385 | Correlates to other imaging findings | An indication that the current imaging finding relates to a finding from an imaging exam. |
111386 | No correlation to other imaging findings | An indication that the current imaging finding has no relation to findings from any other imaging exam. |
111387 | No correlation to clinical findings | An indication that the current imaging finding has no relation to any other clinical findings. |
111388 | Malignancy Type | Classification of the cancer as invasive, DCIS, or other. |
111389 | Invasive breast carcinoma | A malignancy that has spread beyond an area of focus. |
111390 | Other malignancy type | A breast cancer with malignant pathology findings that are not classified as invasive or in situ. |
111391 | Menstrual Cycle Phase | A specific timeframe during menses. |
111392 | 1st week | In the first week of the menstrual cycle phase, that is, one week following menses. |
111393 | 2nd week | In the second week of the menstrual cycle phase, that is, two weeks following menses. |
111394 | 3rd week | In the third week of the menstrual cycle phase, that is, three weeks following menses. |
111395 | Estimated Timeframe | An estimated period of time. |
111396 | < 3 months ago | An event occurred less than 3 months ago. |
111397 | 4 months to 1 year ago | An event occurred between 4 months and 1 year ago. |
111398 | > 1 year ago | An event occurred longer than 1 year ago. |
111399 | Timeframe uncertain | The timing of an event is not recalled. |
111400 | Breast Imaging Report | Report title for the diagnostic report for one or more breast imaging or intervention procedures. |
111401 | Reason for procedure | Concept name for the description of why a procedure has been performed. |
111402 | Clinical Finding | A finding during clinical examination (i.e., history and physical examination) such as pain, palpable mass or discharge. |
111403 | Baseline screening mammogram | First screening mammogram taken for patient that is used as a comparison baseline for further examinations. |
111404 | First mammogram ever | First mammogram taken for a patient without regard to whether it was for screening or a diagnostic procedure. |
111405 | Implant type | Concept name for the material of which a breast prosthetic device is constructed. |
111406 | Number of similar findings | A numeric count of findings classified as similar in nature. |
111407 | Implant finding | Concept name for the status of a breast prosthetic device as noted by imaging. |
111408 | Film Screen Mammography | Mammogram using traditional X-Ray film. |
111409 | Digital Mammography | Mammogram using a digital image acquisition system. |
111410 | Surgical consult | Referred for evaluation by a surgeon. |
111411 | Mammography CAD | Computer aided detection and/or computer aided diagnosis for mammography. |
111412 | Narrative Summary | Concept name for a text-based section of a report. |
111413 | Overall Assessment | A title for a report section that summarizes all interpretation results for a report with one overriding assessment. E.g., benign or negative. |
111414 | Supplementary Data | Concept name for a collection of supporting evidence for a report. |
111415 | Additional evaluation requested from prior study | Prior study indicates that additional imaging be performed to further evaluate a suspicious or questionable anatomic region. |
111416 | Follow-up at short interval from prior study | The prior study recommended a follow-up breast imaging exam in 1 to 11 months (generally in 6 months). |
111417 | History of breast augmentation, asymptomatic | Prior breast augmentation (breast enlargement) and is not presenting with any symptoms. |
111418 | Review of an outside study | Review or second opinion made on an image performed outside of the facility. |
111419 | Additional evaluation requested from abnormal screening exam | Additional breast imaging performed at the time of the patient's screening mammogram. |
111420 | History of benign breast biopsy | Patient has had previous benign breast biopsies. |
111421 | Personal history of breast cancer with breast conservation therapy | Patient has had a prior surgery such as a lumpectomy or quadrantectomy to remove malignant breast tissue, but breast tissue remains. |
111423 | Physical Examination Results | The results of a physical examination performed on the patient, possibly including the results of inspection, palpation, auscultation, or percussion. |
111424 | Comparison to previous findings | The result of assessing the current imaging exam in comparison to previous imaging exams. |
111425 | Intraluminal filling defect | An abnormality observed during ductography where the ductal system within the breast fills in an abnormal pattern. Ductography is an imaging exam in which a radio opaque contrast media is introduced into the ductal system of the breast through the nipple and images of the ductal system are obtained. |
111426 | Multiple filling defect | During ductography an observation of more than one filling abnormality within the breast ductal system. |
111427 | Abrupt duct termination | An abnormality observed during ductography where the ductal system within the breast terminates in an unusual fashion. |
111428 | Extravasation | Abnormal flowage of contrast media within the breast noted on ductography. |
111429 | Duct narrowing | An abnormality observed during ductography where the ductal system within the breast appears narrow. |
111430 | Cyst fill | During ductography an observation of the contrast media filling a cyst within the breast. |
111431 | Instrument Approach | The area and line within the anatomy through which a needle or instrument passes during an interventional procedure. |
111432 | Inferolateral to superomedial | The line within the anatomy from the lower outer to the upper inner aspect. E.g., through which a needle or instrument passes in an interventional procedure. |
111433 | Inferomedial to superolateral | The line within the anatomy from the lower inner to the upper outer aspect. E.g., through which a needle or instrument passes in an interventional procedure. |
111434 | Superolateral to inferomedial | The line within the anatomy from the upper outer to the lower inner aspect. E.g., through which a needle or instrument passes in an interventional procedure. |
111435 | Superomedial to inferolateral | The line within the anatomy from the upper inner to the lower outer aspect. E.g., through which a needle or instrument passes in an interventional procedure. |
111436 | Number of passes | The number of times a biopsy instrument is passed through an area of interest. |
111437 | Number of specimens | The number of biopsy specimens obtained from an interventional procedure. |
111438 | Needle in target | An indicator of whether or not a biopsy or localizing needle in an interventional procedure is seen to be in the area of interest. |
111439 | Number of needles around target | The number of localizing needles placed around the area of interest in an interventional procedure. |
111440 | Incision made | An indicator of whether or not an incision was made in the anatomy during an interventional procedure. |
111441 | Microclip placed | An indicator of whether or not a radio opaque microclip was placed in the anatomy during an interventional procedure. | Retired. Replaced by (111123, DCM, "Marker placement")
111442 | Confirmation of target | An indicator of the degree of success of an interventional procedure. |
111443 | Target completely contained in the specimen | An indicator that during an interventional procedure the area of interest was fully excised and is noted in the resultant biopsy specimen. |
111444 | Target partially obtained in the specimen | An indicator that during an interventional procedure the area of interest was partially excised and is noted in the resultant biopsy specimen. |
111445 | Target not in the specimen | An indicator that following an interventional procedure the area of interest is not seen in the resultant biopsy specimen. |
111446 | Calcifications seen in the core | An indicator that following an interventional procedure the targeted calcifications are noted in the resultant biopsy specimen. |
111447 | Lesion completely removed | An indicator that during an interventional procedure the area of interest was fully excised and is noted in the resultant biopsy specimen. |
111448 | Lesion partially removed | An indicator that during an interventional procedure the area of interest was partially excised and is noted in the resultant biopsy specimen. |
111449 | Fluid obtained | An indicator that during an interventional procedure fluid was successfully aspirated. |
111450 | Light brown color | Color that is a light shade of brown. |
111451 | Dark red color | Color that is a dark shade of red. |
111452 | Dark brown color | Color that is a dark shade of brown. |
111453 | Bright red color | Color that is a bright shade of red. |
111454 | Blood tinged color | Color that is tinged with the color of blood. |
111455 | Occult blood test result | An indicator of whether or not the fluid obtained during an interventional procedure contains red blood cells. |
111456 | Action on fluid | An indicator of whether or not fluid during an interventional procedure was sent for cytological analysis or simply discarded. |
111457 | Sent for analysis | An indicator that fluid obtained during an interventional procedure was sent to a laboratory for analysis. |
111458 | Discarded | An indicator that fluid obtained during an interventional procedure was discarded. |
111459 | Mass with calcifications | A radiopaque density noted during diagnostic imaging that has associated calcific densities. |
111460 | Complex cyst | A fluid-filled sac with greater than normal characteristics. |
111461 | Intracystic lesion | A tumor within a cyst. |
111462 | Solid mass | A tumor or lesion. |
111463 | Supplementary Data for Intervention | Supporting evidence for interpretation results of an interventional procedure. |
111464 | Procedure Modifier | A descriptor that further qualifies or characterizes a type of procedure. |
111465 | Needle Gauge | Needle size (diameter) characterization. E.g., of a biopsy needle. |
111466 | Severity of Complication | An indicator of the gravity of a problem experienced by a patient, related to a procedure that was performed. |
111467 | Needle Length | Distance from the hub or bushing to the tip of the needle. |
111468 | Pathology Results | The collection of observations and findings from pathologic analysis. |
111469 | Sampling DateTime | The date and time that the sample was collected from the patient. |
111470 | Uninvolved | Indicates that the margin of the biopsy specimen was not involved with the tumor. |
111471 | Involved | Indicates that the margin of the biopsy specimen was involved with the tumor. |
111472 | Nipple involved | Indicates whether the nipple was involved in an interventional procedure or pathologic analysis. |
111473 | Number of nodes removed | Indicates the number of lymph nodes removed. |
111474 | Number of nodes positive | Indicates the number of lymph nodes removed that contain cancer cells. |
111475 | Estrogen receptor | The result of a test for the presence of a protein that binds with estrogen. |
111476 | Progesterone receptor | The result of a test for the presence of a protein that binds with progesterone. |
111477 | S Phase | Indicates the percentage of cells in S phase. Cell division is defined by phases; the S phase is the stage during which DNA replicates. |
111478 | Non-bloody discharge (from nipple) | The visible emission of non-bloody fluid from the nipple. |
111479 | Difficult physical/clinical examination | The inability to discern normal versus abnormal breast tissue during palpation. |
111480 | Cancer elsewhere | An indication that a patient has or had a malignant occurrence in an area of the body other than the breast. |
111481 | Saline implant | A salt water filled prosthetic device implanted in the breast. |
111482 | Polyurethane implant | A polymer based (plastic) prosthetic device implanted in the breast. |
111483 | Percutaneous silicone injection | The introduction of polymeric organic silicon based material through the skin, as for breast augmentation or reconstruction. |
111484 | Combination implant | A prosthetic device that contains more than one material implanted in the breast. |
111485 | Pre-pectoral implant | A breast implant placed in front of the pectoralis major muscle. |
111486 | Retro-pectoral implant | A breast implant placed behind the pectoralis major muscle. |
111487 | Mammographic (crosshair) | Using X-Ray technique and a superimposed set of crossed lines for detection or placement. |
111488 | Mammographic (grid) | Using X-Ray technique and a superimposed aperture for detection or placement. |
111489 | Palpation guided | Using physical touch for detection or placement. |
111490 | Vacuum assisted | The performance of a biopsy procedure using a vacuum device attached to the biopsy needle. |
111491 | Abnormal discharge | Unusual or unexpected emission of fluid. |
111492 | No complications | Having experienced no adverse medical conditions related to or resulting from an interventional procedure. |
111494 | Stage 0 | TNM grouping of tumor stage, from AJCC, where primary tumor is Tis, regional lymph node is N0, and distant metastasis is M0. |
111495 | Stage I | TNM grouping of tumor stage, from AJCC, where primary tumor is T1, regional lymph node is N0, and distant metastasis is M0. |
111496 | Stage IIA | TNM grouping of tumor stage, from AJCC, where primary tumor is T0 or T1, with regional lymph node N1 and distant metastasis is M0, or T2 with N0 and M0. |
111497 | Stage IIB | TNM grouping of tumor stage, from AJCC, where primary tumor is T2, with regional lymph node N1 and distant metastasis is M0, or T3 with N0 and M0. |
111498 | Stage IIIA | TNM grouping of tumor stage, from AJCC, where primary tumor is T0, T1 or T2, with regional lymph node N2 and distant metastasis is M0, or T3 with N1 or N2 and M0. |
111499 | Stage IIIB | TNM grouping of tumor stage, from AJCC, where primary tumor is T4, regional lymph node is N0, N1 or N2, and distant metastasis is M0. |
111500 | Stage IIIC | TNM grouping of tumor stage, from AJCC, where primary tumor is any T value, regional lymph node is N3, and distant metastasis is M0. |
111501 | Stage IV | TNM grouping of tumor stage, from AJCC, where primary tumor is any T value, regional lymph node is any N value, and distant metastasis is M1. |
111502 | Bloom-Richardson Grade | Histologic tumor grade (sometimes called Scarff-Bloom-Richardson grade) is based on the arrangement of the cells in relation to each other -- whether they form tubules, how closely they resemble normal breast cells (nuclear grade) and how many of the cancer cells are in the process of dividing (mitotic count). |
111503 | Normal implants | Breast prosthetic devices are intact, not leaking, and are in a normal shape and form. |
111504 | Asymmetric implants | Breast prosthetic devices are not symmetric, equal, corresponding in form, or are in one breast (unilateral). |
111505 | Calcified implant | Fibrous or calcific contracture of the tissue capsule that forms around a breast prosthetic device. |
111506 | Distorted implant | Breast prosthetic device is twisted out of normal shape or form. |
111507 | Silicone-laden lymph nodes | Silicone from breast prosthetic device found in lymphatic tissue. |
111508 | Free silicone | Silicone found in breast tissue outside of the prosthetic capsule or implant membrane. |
111509 | Herniated implant | Protrusion of part of the structure normally encapsulating the content of the breast prosthetic device. |
111510 | Explantation | Evidence of removal of a breast prosthetic device. |
111511 | Relevant Patient Information for Breast Imaging | Historical patient health information of interest to the breast health clinician. |
111512 | Medication History | Information regarding usage by the patient of certain medications, such as hormones. | Retired. Replaced by (10160-0, LN, "History Of Medication Use")
111513 | Relevant Previous Procedures | Interventional or non-interventional procedures previously performed on the patient, such as breast biopsies. |
111514 | Relevant Indicated Problems | Abnormal conditions experienced by the patient that serve as the reason for performing a procedure, such as a breast exam. | Retired. Replaced by (11450-4, LN, "Problem List")
111515 | Relevant Risk Factors | Personal, familial, and other health factors that may indicate an increase in the patient's chances of developing a health condition or disease, such as breast cancer. |
111516 | Medication Type | A classification of a medicinal substance, such as hormonal contraceptive or antibiotic. |
111517 | Relevant Patient Information | Historical patient health information for general purpose use. |
111518 | Age when first menstrual period occurred | The age of the patient at the first occurrence of menses. |
111519 | Age at First Full Term Pregnancy | The age of the patient at the time of her first full term pregnancy. |
111520 | Age at Menopause | The age of the patient at the cessation of menses. |
111521 | Age when hysterectomy performed | The age of the patient at the time her uterus was removed. |
111522 | Age when left ovary removed | The age of the patient at the time she had her left ovary removed. |
111523 | Age when right ovary removed | The age of the patient at the time she had her right ovary removed. |
111524 | Age Started | The age of a patient on the first occurrence of an event, such as the first use of a medication. |
111525 | Age Ended | The age of a patient on the last occurrence of an event, such as the last use of a medication. |
111526 | DateTime Started | The date and time of the first occurrence of an event, such as the first use of a medication. |
111527 | DateTime Ended | The date and time of the last occurrence of an event, such as the last use of a medication. |
111528 | Ongoing | An indicator of whether an event is still in progress, such as the use of a medication or substance, or environmental exposure. |
111529 | Brand Name | Product name of a device or substance, such as medication, to identify it as the product of a single firm or manufacturer. |
111530 | Risk Factor modifier | A descriptor that further qualifies or characterizes a risk factor. |
111531 | Previous Procedure | A prior non-interventional exam or interventional procedure performed on a patient. |
111532 | Pregnancy Status | Describes the pregnancy state of a referenced subject. |
111533 | Indicated Problem | A symptom experienced by a patient that is used as the reason for performing an exam or procedure. |
111534 | Role of person reporting | The function of the individual who is reporting information on a patient, which could be a specific health care related profession, the patient him/herself, or a relative or friend. |
111535 | DateTime problem observed | The date and time that a symptom was noted. |
111536 | DateTime of last evaluation | The date and time of the most recent evaluation of an indicated problem. |
111537 | Family Member with Risk Factor | A patient's biological relative who exhibits a health factor that may indicate an increase in the patient's chances of developing a particular disease or medical problem. |
111538 | Age at Occurrence | The age at which an individual experienced a specific event, such as breast cancer. |
111539 | Menopausal phase | The current stage of an individual in her gynecological development. |
111540 | Side of Family | An indicator of paternal or maternal relationship. |
111541 | Maternal | Relating to biological female parentage. |
111542 | Unspecified gynecological hormone | A gynecological hormone for which the specific type is not specified. E.g., contraceptive, estrogen, Tamoxifen. |
111543 | Breast feeding history | An indicator of whether or not a patient ever provided breast milk to her offspring. |
111544 | Average breast feeding period | The average length of time that a patient provided breast milk to her offspring. |
111545 | Substance Use History | Information regarding usage by the patient of certain legal or illicit substances. |
111546 | Used Substance Type | A classification of a substance, such as alcohol or a legal or illicit drug. |
111547 | Environmental Exposure History | Information regarding exposure of the patient to potentially harmful environmental factors. |
111548 | Environmental Factor | A classification of a potentially harmful substance or gas in a subject's environment, such as asbestos, lead, or carcinogens. |
111549 | Previous Reports | Previous Structured Reports that could have relevant information for a current imaging service request. |
111550 | Personal breast cancer history | An indication that a patient has had a previous malignancy of the breast. |
111551 | History of endometrial cancer | Indicates a previous occurrence of cancer of the lining of the uterus. |
111552 | History of ovarian cancer | Indicates a previous occurrence of cancer of the lining of the ovary. |
111553 | History of high risk lesion on previous biopsy | Indicates a prior diagnosis of pre-cancerous cells or tissue removed for pathologic evaluation. |
111554 | Post menopausal patient | A female patient whose menstrual periods have ceased. |
111555 | Late child bearing (after 30) | A female patient whose first child was born after the patient was 30 years old. |
111556 | BRCA1 breast cancer gene | The first level genetic marker indicating risk for breast cancer. |
111557 | BRCA2 breast cancer gene | The second level genetic marker indicating risk for breast cancer. |
111558 | BRCA3 breast cancer gene | The third level genetic marker indicating risk for breast cancer. |
111559 | Weak family history of breast cancer | A patient's biological aunt, grandmother, or female cousin was diagnosed with breast cancer. Definition from BI-RADS®. |
111560 | Intermediate family history of breast cancer | A patient's biological mother or sister was diagnosed with breast cancer after they had gone through menopause. Definition from BI-RADS®. |
111561 | Very strong family history of breast cancer | A patient's biological mother or sister was diagnosed with breast cancer before they had gone through menopause, or more than one of the patient's first-degree relatives (biological mother or sister) were diagnosed with breast cancer after they had gone through menopause. Definition from BI-RADS®. |
111562 | Family history of prostate cancer | Previous diagnosis of a malignancy of the prostate gland in a biological relative. |
111563 | Family history unknown | The health record of a patient's biological relatives is not known. |
111564 | Nipple discharge cytology | The study of cells obtained from fluid emitted from the breast. |
111565 | Uterine malformations | A developmental abnormality resulting in an abnormal shape of the uterus. |
111566 | Spontaneous Abortion | A naturally occurring premature expulsion from the uterus of the products of conception - the embryo or a nonviable fetus. |
111567 | Gynecologic condition | An ailment/abnormality or state of the female reproductive tract. |
111568 | Gynecologic surgery | A surgical operation performed on any portion of the female reproductive tract. |
111569 | Previous LBW or IUGR birth | Prior pregnancy with a low birth weight baby or a fetus with Intrauterine Growth Restriction or Retardation. |
111570 | Previous fetal malformation/syndrome | History of at least one prior pregnancy with fetal anatomic abnormality(s). |
111571 | Previous RH negative or blood dyscrasia at birth | History of delivering a Rhesis Isoimmunization affected child(ren) or a child(ren) with another blood disorder. |
111572 | History of multiple fetuses | History of at least one pregnancy that contained more than one fetus. E.g., twins, triplets, etc.. |
111573 | Current pregnancy, known or suspected malformations/syndromes | At least one fetus of this pregnancy has an anatomic abnormality(s) that is known to exist, or a "marker" is present that suggests the abnormality(s) may be present. |
111574 | Family history, fetal malformation/syndrome | Biological relatives have previously conceived a fetus with an anatomic abnormality(s). |
111575 | High | A subjective descriptor for an elevated amount of exposure, use, or dosage, incurring high risk of adverse effects. |
111576 | Medium | A subjective descriptor for a moderate amount of exposure, use, or dosage, incurring medium risk of adverse effects. |
111577 | Low | A subjective descriptor for a limited amount of exposure, use, or dosage, incurring low risk of adverse effects. |
111578 | Dose frequency | A measurement of the rate of occurrence of which a patient takes a certain medication. |
111579 | Rate of exposure | The quantity per unit of time that a patient was or is being exposed to an environmental irritant. |
111580 | Volume of use | The quantity per unit of time that a medication or substance was or is being used. |
111581 | Relative dose amount | A qualitative descriptor for the amount of a medication that was or is being taken. |
111582 | Relative amount of exposure | A qualitative descriptor for the amount of present or past exposure to an environmental irritant. |
111583 | Relative amount of use | A qualitative descriptor for the amount of a medication or substance that was or is being used. |
111584 | Relative dose frequency | A qualitative descriptor for the frequency with which a medication was or is being taken. |
111585 | Relative frequency of exposure | A qualitative descriptor for the frequency of present or past exposure to an environmental irritant. |
111586 | Relative frequency of use | A qualitative descriptor for the frequency with which a medication or substance was or is being used. |
111587 | No known exposure | Patient is not known to have been exposed to or used the substance or medication. |
111590 | Recall for technical reasons | Patient returns for additional images to improve the quality of the most recent exam. |
111591 | Recall for imaging findings | Patient returns for additional images to clarify findings from the most recent exam. |
111592 | Recall for patient symptoms/ clinical findings | Patient returns for additional images to clarify symptoms or signs reported by the patient or a healthcare professional at the time of the most recent exam. |
111593 | LBW or IUGR | Number of births with low birth weight or intrauterine growth restriction. |
111601 | Green filter | Filter that transmits green light while blocking the other colors, typically centered at 510-540 nm | Retired. Replaced by (A-010E2, SRT, "Green optical filter")
111602 | Red filter | Filter that transmits red light while blocking the other colors, typically centered at 630-680 nm | Retired. Replaced by (A-010DF, SRT, "Red optical filter")
111603 | Blue filter | Filter that transmits blue while blocking the other colors, typically centered at 460-480 nm | Retired. Replaced by (A-010DA, SRT, "Blue optical filter")
111604 | Yellow-green filter | A filter of 560nm that is used for retinal imaging and can provide good contrast and good visibility of the retinal vasculature | Retired. Replaced by (A-010E0, SRT, "Yellow-green optical filter")
111605 | Blue-green filter | A filter of 490nm that is used for retinal imaging because of excessive scattering of some retinal structures at very short wavelengths | Retired. Replaced by (A-010D8, SRT, "Blue-green optical filter")
111606 | Infrared filter | Filter that transmits the infrared spectrum, which is light that lies outside of the visible spectrum, with wavelengths longer than those of red light, while blocking visible light | Retired. Replaced by (A-010DC, SRT, "Infrared optical filter")
111607 | Polarizing filter | A filter that reduces reflections from non-metallic surfaces such as glass or water by blocking light waves that are vibrating at selected angles to the filter. | Retired. Replaced by (A-010E1, SRT, "Polarizing optical filter")
111609 | No filter | No filter used. |
111621 | Field 1 for Joslin 3 field | Joslin NM-1 is a 45 degree field focused centrally between the temporal margin of optic disc and the center of the macula: Center the camera on the papillomacular bundle midway between the temporal margin of the optic disc and the center of the macula. The horizontal centerline of the image should pass directly through the center of the disc. |
111622 | Field 2 for Joslin 3 field | Joslin NM-2 is a 45 degree field focused superior temporal to the optic disc: Center the camera laterally approximately one-half disc diameter temporal to the center of the macula. The lower edge of the field is tangent to a horizontal line passing through the upper edge of the optic disc. The image is taken temporal to the macula but includes more retinal nasal and superior to the macula than standard field 2. |
111623 | Field 3 for Joslin 3 field | Joslin NM-3 is a 45 degree field focused nasal to the optic disc: This field is nasal to the optic disc and may include part of the optic disc. The horizontal centerline of the image should pass tangent to the lower edge of the optic disc. |
111625 | Diffuse direct illumination | A broad or "soft" light supplied from a single source. |
111626 | Scheimpflug Camera | A slit reflected light microscope, which has the ability to form an image of the back scattered light from the eye in a sagittal plane. Scheimpflug cameras are able to achieve a wide depth of focus by employing the "Sheimpflug principle" where the lens and image planes are not parallel with each other. Rotating Sheimplug cameras are able to generate three-dimensional images and calculate measurements of the anterior chamber of the eye. |
111627 | Scotopic light | Lighting condition approximately 0.04 lux. |
111628 | Mesopic light | Lighting condition approximately 4 lux. |
111629 | Photopic light | Lighting condition approximately 40 lux. |
111630 | Dynamic light | Acquisition preceded by intense light. |
111631 | Average Glandular Dose | Calculated from values of entrance exposure in air, the X-Ray beam quality (half-value layer), and compressed breast thickness, is the energy deposited per unit mass of glandular tissue averaged over all the glandular tissue in the breast. |
111632 | Anode Target Material | The primary material in the anode of an X-Ray source. |
111633 | Compression Thickness | The average thickness of the body part examined when compressed, if compression has been applied during X-Ray exposure. |
111634 | Half Value Layer | Thickness of Aluminum required to reduce the X-Ray output at the patient entrance surface by a factor of two. |
111635 | X-Ray Grid | An anti-scatter device based on radiation absorbing strips above the detector. E.g., in the patient support. |
111636 | Entrance Exposure at RP | Exposure measurement in air at the reference point that does not include back scatter, according to MQCM 1999. |
111637 | Accumulated Average Glandular Dose | Average Glandular Dose to a single breast accumulated over multiple images. |
111638 | Patient Equivalent Thickness | Value of the control variable used to parametrize the Automatic Exposure Control (AEC) closed loop. E.g., "Water Value". |
111641 | Fixed grid | An X-Ray Grid that does not move during exposure. |
111642 | Focused grid | An X-Ray Grid with radiation absorbing strips that are focused toward the focal spot, to eliminate grid cutoff. |
111643 | Reciprocating grid | An X-Ray Grid that is designed to move during exposure, to eliminate the appearance of grid lines on the image. |
111644 | Parallel grid | An X-Ray Grid with radiation absorbing strips that are parallel to each other and that is used only with long source to image distances. |
111645 | Crossed grid | An X-Ray Grid with crossed radiation absorbing strips used for more complete cleanup of scatter radiation. |
111646 | No grid | No X-Ray Grid was used due to low scatter conditions. |
111671 | Spectacle Prescription Report | The spectacle prescription for a patient. |
111672 | Add Near | Refractive measurements of the eye to correct for inability to focus at near while wearing the distance prescription. |
111673 | Add Intermediate | Refractive measurements of the eye to correct for inability to focus at intermediate distance while wearing the distance prescription. |
111674 | Add Other | Refractive measurements of the eye to correct for inability to focus at the specified distance while wearing the distance prescription. |
111675 | Horizontal Prism Power | The power of a prism to bend light in the horizontal direction, in prism diopters. |
111676 | Horizontal Prism Base | Direction of the base of a horizontal prism -- either in (toward the nose), or out (away from the nose). |
111677 | Vertical Prism Power | The power of a prism to bend light in the vertical direction, in prism diopters. |
111678 | Vertical Prism Base | Direction of the base of a vertical prism -- either up, or down. |
111679 | Distance Pupillary Distance | Distance in mm between the pupils when the patient's object of regard is in the distance. |
111680 | Near Pupillary Distance | Distance in mm between the pupils when the patient's object of regard is at near. |
111685 | Autorefraction Visual Acuity | A patient's vision with the correction measured by an autorefractor in place. |
111686 | Habitual Visual Acuity | A patient's vision with whichever vision correction the patient customarily wears. |
111687 | Prescription Visual Acuity | A patient's vision with the final spectacle prescription in place. |
111688 | Right Eye Rx | The spectacle prescription for the right eye. |
111689 | Left Eye Rx | The spectacle prescription for the left eye. |
111690 | Macular Grid Thickness and Volume Report | A macular grid thickness and volume report for a patient. The macular grid is an analytic tool described in PS3.1. |
111691 | Number of Images Used for Macular Measurements | Number of images used for the macular grid measurement. |
111692 | Number of Samples Used per Image | Number of samples used per Image for analysis. |
111693 | Analysis Quality Rating | A numeric rating of the quality of the entire analysis with respect to grading and diagnostic purposes. Higher numbers indicate greater quality. |
111694 | Image Set Quality Rating | A numeric rating of the quality of an entire image set with respect to grading and diagnostic purposes. Higher numbers indicate greater quality. |
111695 | Interfering Tears or Drops | Tear film or drops affecting test quality. |
111696 | Visual Fixation Quality During Acquisition | The assessment of the centricity and persistence of the visual fixation (direction of gaze) during the acquisition. |
111697 | Visual Fixation Quality Problem | The reason why the patient's visual fixation was not steady or was indeterminate. |
111698 | Ophthalmic Macular Grid Problem | The reason why the macular grid measurements may be questionable. |
111700 | Specimen Container Identifier | Identifier of container (box, block, microscope slide, etc.) for the specimen under observation. |
111701 | Processing type | Type of processing that tissue specimen underwent. |
111702 | DateTime of processing | Date and time of processing step. |
111703 | Processing step description | Description of the individual step in the tissue processing sequence. |
111704 | Sampling Method | Method of sampling used to derive specimen from its parent. |
111705 | Parent Specimen Identifier | Identifier of the parent specimen that gave rise to the current specimen. |
111706 | Issuer of Parent Specimen Identifier | Assigning authority for parent specimen's identifier. |
111707 | Parent specimen type | Parent specimen type that gave rise to current specimen. |
111708 | Position Frame of Reference | Description of coordinate system and origin reference point on parent specimen, or parent specimen container, or image used for localizing the sampling site or location within container or image. |
111709 | Location of sampling site | Reference to image of parent specimen localizing the sampling site; may include referenced Presentation State object. |
111710 | Location of sampling site X offset | Location of sampling site of specimen (nominal center) relative to the Position Frame of Reference. |
111711 | Location of sampling site Y offset | Location of sampling site of specimen (nominal center) relative to the Position Frame of Reference. |
111712 | Location of sampling site Z offset | Location of sampling site of specimen (nominal center) relative to the Position Frame of Reference. |
111718 | Location of Specimen | Description of specimen location, either in absolute terms or relative to the Position Frame of Reference. |
111719 | Location of Specimen X offset | Location of specimen (nominal center) relative to the Position Frame of Reference in the X dimension. |
111720 | Location of Specimen Y offset | Location of specimen (nominal center) relative to the Position Frame of Reference in the Y dimension. |
111721 | Location of Specimen Z offset | Location of specimen (nominal center) relative to the Position Frame of Reference in the Z dimension. |
111723 | Visual Marking of Specimen | Description of visual distinguishing identifiers. E.g., ink, or a particular shape of the specimen. |
111724 | Issuer of Specimen Identifier | Assigning authority for specimen identifier. |
111726 | Dissection with entire specimen submission | Dissection of specimen with submission of all its sections for further processing or examination. |
111727 | Dissection with representative sections submission | Dissection of specimen with submission of representative sections for further processing or examination. |
111729 | Specimen storage | A workflow step, during which tissue specimens are stored in a climate-controlled environment. |
111741 | Transmission illumination | Transmission illumination method for specimen microscopy. |
111742 | Reflection illumination | Reflection illumination method for specimen microscopy. |
111743 | Epifluorescence illumination | Epifluorescence illumination method for specimen microscopy. |
111744 | Brightfield illumination | Brightfield illumination method for specimen microscopy. |
111745 | Darkfield illumination | Darkfield illumination method for specimen microscopy. |
111746 | Oblique illumination | Oblique illumination method for specimen microscopy. |
111747 | Phase contrast illumination | Phase contrast illumination method for specimen microscopy. |
111748 | Differential interference contrast | Differential interference contrast method for specimen microscopy. |
111749 | Total internal reflection fluorescence | Total internal reflection fluorescence method for specimen microscopy. |
111750 | Ultrasound Contact | A method of obtaining ophthalmic axial measurements that uses ultrasound, and that requires applanation of the cornea. |
111751 | Ultrasound Immersion | A method of obtaining ophthalmic axial measurements that uses ultrasound, and that requires immersion of the patient's eye in fluid as he lies in a supine position. |
111752 | Optical | A method of obtaining ophthalmic axial measurements that uses light. |
111753 | Manual Keratometry | Measurements taken of the corneal curvature using a manual keratometer. |
111754 | Auto Keratometry | Measurements taken of the corneal curvature using an automated keratometer. |
111755 | Simulated Keratometry | Simulated Keratometry measurements derived from corneal topography. |
111756 | Equivalent K-reading | Corneal power measurements using Scheimpflug camera. |
111760 | Haigis | The Haigis IOL calculation formula. Haigis W, Lege B, Miller N, Schneider B. Comparison of immersion ultrasound biometry and partial coherence interferometry for intraocular lens calculation according to Haigis. Graefes Arch Clin Exp Ophthalmol 2000;238:765-73. |
111761 | Haigis-L | The Haigis-L IOL calculation formula. Haigis W. Intraocular lens calculation after refractive surgery for myopia: Haigis-L formula. J Cataract Refract Surg. 2008 Oct;34(10):1658-63. |
111762 | Holladay 1 | The Holladay 1 IOL calculation formula. Holladay JT, Prager TC, Chandler TY, Musgrove KH, Lewis JW, Ruiz RS. A three-part system for refining intraocular lens power calculations. J Cataract Refract Surg. 1988; 14:17-24. |
111763 | Holladay 2 | The Holladay 2 IOL calculation formula. |
111764 | Hoffer Q | The Hoffer Q IOL calculation formula. Hoffer KJ. The Hoffer Q formula: a comparison of theoretic and regression formulas. J Cataract Refract Surg 1993;19:700-12. Errata. J Cataract Refract Surg 1994;20:677 and 2007;33:2-3. |
111765 | Olsen | The Olsen IOL calculation formula. Olsen T. Calculation of intraocular lens power: a review. Acta Ophthalmol. Scand. 2007: 85: 472-485. |
111766 | SRKII | The SRKII IOL calculation formula. Sanders DR, Retzlaff J, Kraff MC. Comparison of the SRK II formula and other second generation formulas. J Cataract Refract Surg. 1988 Mar;14(2):136-41. |
111767 | SRK-T | The SRK-T IOL calculation formula. Retzlaff JA, Sanders DR, Kraff MC. Development of the SRK/T intraocular lens implant power calculation formula. J Cataract Refract Surg 1990;16:333-40. Erratum 1990;16:528. |
111768 | ACD Constant | The "ACD Constant" used in IOL calculation. |
111769 | Haigis a0 | The "Haigis a0" constant used in IOL calculation. |
111770 | Haigis a1 | The "Haigis a1" constant used in IOL calculation. |
111771 | Haigis a2 | The "Haigis a2" constant used in IOL calculation. |
111772 | Hoffer pACD Constant | The "Hoffer pACD Constant" used in IOL calculation. |
111773 | Surgeon Factor | The "Surgeon Factor" constant used in IOL calculation. |
111776 | Front Of Cornea To Front Of Lens | Anterior chamber depth defined as the front of the cornea to the front of the lens. |
111777 | Back Of Cornea To Front Of Lens | Anterior chamber depth defined as the back of the cornea to the front of the lens. |
111778 | Single or Anterior Lens | Refers to the anterior lens when there are two lenses in the eye. The distance, in mm, from the anterior surface of the lens to the posterior surface of the lens. |
111779 | Posterior Lens | Refers to the posterior lens when there are two lenses in the eye. The distance, in mm, from the anterior surface of the lens to the posterior surface of the lens. |
111780 | Measurement From This Device | Value obtained from measurements taken by the device creating this SOP Instance. |
111781 | External Data Source | Value obtained by data transfer from an external source - not from measurements taken by the device providing the value. |
111782 | Axial Measurements SOP Instance | Axial Measurements DICOM SOP Instance. |
111783 | Refractive Measurements SOP Instance | Refractive Measurements DICOM SOP Instance. |
111786 | Standard Deviation of measurements used | Standard Deviation is a simple measure of the variability of a data set. |
111787 | Signal to Noise Ratio | Signal to Noise Ratio of the data samples taken to create a measurement. |
111800 | Visual Field 24-2 Test Pattern | Test pattern, nominally covering an area within 24° of fixation. Consists of 54 test points a minimum of 3° from each meridian and placed 6° apart. The "-2" distinguishes this from another 24° pattern (no longer supported). |
111801 | Visual Field 10-2 Test Pattern | Test pattern, nominally covering an area within 10° of fixation. Consists of 68 test points a minimum of 1° from each meridian and placed 2° apart. The "-2" in this case indicates its point layout to be similar to the 30-2 and 24-2. |
111802 | Visual Field 30-2 Test Pattern | Test pattern consisting of test point locations within 30° of fixation. Consists of 76 test points a minimum of 3° from each meridian and placed 6° apart. The "-2" distinguishes this from another 30° pattern (no longer supported). |
111803 | Visual Field 60-4 Test Pattern | Test pattern consisting of 60 test point locations between 30° and 60° of fixation a minimum of 6° from each meridian and placed 12° apart. The "-4" distinguishes this from a similar 60° pattern having 4 additional points. |
111804 | Visual Field Macula Test Pattern | Test pattern consisting of 16 test point locations within 10° of fixation a minimum of 1° from each meridian and placed 2° apart. |
111805 | Visual Field Central 40 Point Test Pattern | Test pattern consisting of 40 test point locations within 30° of fixation that spread out radially from fixation. |
111806 | Visual Field Central 76 Point Test Pattern | Test pattern consisting of 76 test point locations within 30° of fixation a minimum of 3° from each meridian and placed 6° apart. |
111807 | Visual Field Peripheral 60 Point Test Pattern | Test pattern consisting of 60 test point locations between 30° and 60° of fixation a minimum of 6° from each meridian and placed 12° apart. |
111808 | Visual Field Full Field 81 Point Test Pattern | Test pattern consisting of 81 test point locations within 60° of fixation that spread out radially from fixation. |
111809 | Visual Field Full Field 120 Point Test Pattern | Test pattern consisting of 120 test point locations within 60° of fixation that spread out radially from fixation, concentrated in the nasal hemisphere. |
111810 | Visual Field G Test Pattern | Test pattern for Glaucoma and general visual field assessment with 59 test locations of which 16 test locations are in the macular area (up to 10° eccentricity) and where the density of test location is reduced with eccentricity. The test can be extended with the inclusion of 14 test locations between 30° and 60° eccentricity, 6 of which are located at the nasal step. |
111811 | Visual Field M Test Pattern | Test pattern for the macular area. Orthogonal test pattern with 0.7° spacing within the central 4° of eccentricity and reduced density of test locations between 4 and 10, 5° of eccentricity. 81 test locations over all. The test can be extended to include the test locations of the Visual Field G Test Pattern between 10, 5° and 60°. |
111812 | Visual Field 07 Test Pattern | Full field test pattern with 48 test locations from 0-30° and 82 test locations from 30-70°. Reduced test point density with increased eccentricity. Can be combined with screening and threshold strategies. |
111813 | Visual Field LVC Test Pattern | Low Vision Central. Orthogonal off-center test pattern with 6° spacing. 75 test locations within the central 30°. Corresponds with the 32/30-2 excluding the 2 locations at the blind spot, including a macular test location. The LVC is linked with a staircase threshold strategy starting at 0 dB intensity and applies stimulus area V. |
111814 | Visual Field Central Test Pattern | General test corresponding to the 30-2 but excluding the 2 test locations in the blind spot area, hence with 74 instead of 76 test locations. |
111815 | Visual Field SITA-Standard Test Strategy | Swedish Interactive Thresholding Algorithm (SITA). Strategy gains testing efficiency through use of visual field and information theory models. In: Bengtsson B, Olsson J, Heijl A, Rootzen H. A new generation of algorithms for computerized threshold perimetry, SITA. Acta Ophthalmologica Scandinavica, 1997, 75: 368-375. |
111816 | Visual Field SITA-SWAP Test Strategy | Adaptation of SITA testing methods to Blue-Yellow testing. |
111817 | Visual Field SITA-Fast Test Strategy | Similar to SITA-Standard but with less strict criteria for closing test points. Intended for patients who must be tested in the shortest possible time. In: Bengtsson B, Hejl A. SITA Fast, a new rapid perimetric threshold test. Description of methods and evaluation in patients with manifest and suspect glaucoma. Acta Ophthalmologica Scandinavica, 1998, 76: 431-437. |
111818 | Visual Field Full Threshold Test Strategy | Threshold test algorithm that determines a patient's sensitivity at each test point in the threshold test pattern by adjusting intensity by 4 dB steps until the patient changes their response, and then adjusts the intensity in the opposite direction by 2 dB steps until the patient changes their response again. The last stimulus seen by the patient is recognized as the threshold for that point. The starting values are determined by first thresholding a "primary" point in each quadrant then using the results of each primary point to determine the starting values for neighboring points. |
111819 | Visual Field FastPac Test Strategy | Similar to the Full Threshold algorithm except that it steps by 3 dB and only crosses the threshold only once. In: Flanagan JG, Wild JM, Trope GE. Evaluation of FASTPAC, a new strategy for threshold estimation with the Humphrey Field Analyzer, in a glaucomatous population. Ophthalmology, 1993, 100: 949-954. |
111820 | Visual Field Full From Prior Test Strategy | Identical to Full Threshold except that starting values are determined by the results of a previous test performed using the same test pattern and the Full Threshold test strategy. |
111821 | Visual Field Optima Test Strategy | Similar to FastPac except that the steps are pseudo-dynamic (differ based on the intensity of the last presentation). |
111822 | Visual Field Two-Zone Test Strategy | Suprathreshold testing strategy, in which each point is initially tested using stimulus that is 6 dB brighter than the expected hill of vision. If the patient does not respond, the stimulus is presented a second time at the same brightness. If the patient sees either presentation, the point is marked as "seen"; otherwise it is marked as "not seen". |
111823 | Visual Field Three-Zone Test Strategy | An extension of the two-zone strategy in which test points where the second stimulus is not seen are presented with a third stimulus at maximum brightness. |
111824 | Visual Field Quantify-Defects Test Strategy | An extension of the two-zone strategy, in which test points where the second stimulus is not seen receive threshold testing to quantify the depth of any detected scotomas. |
111825 | Visual Field TOP Test Strategy | Tendency Oriented Perimetry. Fast thresholding algorithm. Test strategy makes use of the interaction between neighboring test locations to reduce the test time compared to normal full threshold strategy by 60-80%. In: Morales J, Weitzman ML, Gonzalez de la Rosa M. Comparison between Tendency-Oriented Perimetry (TOP) and octopus threshold perimetry. Ophthalmology, 2000, 107: 134-142. |
111826 | Visual Field Dynamic Test Strategy | Dynamic strategy is a fast thresholding strategy reducing test duration by adapting the dB step sizes according to the frequency-of-seeing curve of the threshold. Reduction of test time compared to normal full threshold strategy 30-50%. |
111827 | Visual Field Normal Test Strategy | Traditional full threshold staircase strategy. Initial intensities are presented, based on anchor point sensitivities in each quadrant and based on already known neighboring sensitivities. In a first run, thresholds are changed in 4dB steps until the first response reversal. Then the threshold is changed in 2 dB steps until the second response reversal. The threshold is calculated as the average between the last seen and last not-seen stimulus, supposed to correspond with the 50% point in the frequency-of-seeing curve. |
111828 | Visual Field 1-LT Test Strategy | One level screening test: Each test location is tested with a single intensity. The result is shown as seen or not-seen. The intensity can either be a 0 dB stimulus or a predefined intensity. |
111829 | Visual Field 2-LT Test Strategy | Two level screening test: Each test location is initially tested 6 dB brighter than the age corrected normal value. |
111830 | Visual Field LVS Test Strategy | Low Vision Strategy is a full threshold normal strategy with the exception that it starts at 0 dB intensity and applies stimulus area V. |
111831 | Visual Field GATE Test Strategy | German Adaptive Threshold Estimation is a fast strategy based on a modified 4-2 staircase algorithm, using prior visual fields to calculate the starting intensity. In: Chiefer U, Pascual JP, Edmunds B, Feudner E, Hoffmann EM, Johnson CA, Lagreze WA, Pfeiffer N, Sample PA, Staubach F, Weleber RG, Vonthein R, Krapp E, Paetzold J. Comparison of the new perimetric GATE strategy with conventional full-threshold and SITA standard strategies. Investigative Ophthalmology and Visual Science, 2009, 51: 488-494. |
111832 | Visual Field GATEi Test Strategy | Similar to GATE. The i stands for initial. If there was no prior visual field test to calculate the starting values, an anchor point method is used to define the local start values. |
111833 | Visual Field 2LT-Dynamic Test Strategy | A test started as two level screening test. In the course of the test, the threshold of relative defects and/or normal test locations has been quantified using the dynamic threshold strategy. |
111834 | Visual Field 2LT-Normal Test Strategy | A test started as two level screening test. In the course of the test, the threshold of relative defects and/or normal test locations has been quantified using the normal full threshold strategy. |
111835 | Visual Field Fast Threshold Test Strategy | Takes neighborhood test point results into account and offers stimuli with an adapted value to save time. |
111836 | Visual Field CLIP Test Strategy | Continuous Luminance Incremental Perimetry, which measures at first the individual reaction time of the patient and threshold values in every quadrant. The starting value for the main test is slightly below in individual threshold. |
111837 | Visual Field CLASS Strategy | A supra threshold screening strategy. The starting stimuli intensities depend on the classification of the patient's visual hill by measuring the central (fovea) or peripheral (15° meridian) threshold. The result of each dot slightly underestimates the sensitivity value (within 5 dB). |
111838 | Age corrected | Mode for determining the starting luminance for screening test points - the starting luminance s is chosen based on the age of the patient. |
111839 | Threshold related | Mode for determining the starting luminance for screening test points - the starting luminance is chosen based on the results of thresholding a set of "primary" test points (one in each quadrant). |
111840 | Single luminance | Mode for determining the starting luminance for screening test points - in this case, all starting luminance is set to the same value. |
111841 | Foveal sensitivity related | Mode for determining the starting luminance for screening test points - the starting luminance is chosen based on the result of the foveal threshold value. |
111842 | Related to non macular sensitivity | Mode for determining the starting luminance for screening test points - the starting luminance is chosen based on the result of four threshold values measured near the 15° meridian (one in each quadrant). |
111843 | Automated Optical | Real time evaluation of the camera image to recognize blinks and fixation losses with influence on the test procedure. Blinks that interfere with stimuli presentation cause the automated repetition of such stimulus presentations. Fixation losses can be used to delay the stimulus presentation until correct fixation is regained. |
111844 | Blind Spot Monitoring | A method of monitoring the patient's fixation by periodically presenting stimulus in a location on the background surface that corresponds to the patient's blind spot. |
111845 | Macular Fixation Testing | A method of monitoring the patient's fixation by presenting the stimulus to the patient's macula. |
111846 | Observation by Examiner | A method of monitoring the patient's fixation by observation from the examiner of the patient. |
111847 | Outside normal limits | Analysis Results are outside normal limits. |
111848 | Borderline | Analysis Results are borderline. |
111849 | Abnormally high sensitivity | Analysis Results identify abnormally high sensitivity. |
111850 | General reduction in sensitivity | Analysis Results identify general reduction in sensitivity. |
111851 | Borderline and general reduction in sensitivity | Analysis Results identify Borderline and general reduction in sensitivity. |
111852 | Visual Field Index | Index of a patient's remaining visual field normalized for both age and generalized defect. |
111853 | Visual Field Loss Due to Diffuse Defect | Estimate of the portion of a patient's visual field loss that is diffuse (i.e., spread evenly across all portions of the visual field). |
111854 | Visual Field Loss Due to Local Defect | Estimate of the portion of a patient's visual field loss that is local (i.e., not spread evenly across all portions of the visual field). |
111855 | Glaucoma Hemifield Test Analysis | An analysis of asymmetry between zones of the superior and inferior visual field. It is designed to be specific for defects due to glaucoma. |
111856 | Optical Fixation Measurements | The data output of an optical fixation monitoring process, consisting of a list of positive and negative numbers indicating the quality of patient fixation over the course of a visual field test. The value 0 represents the initial fixation. Negative numbers indicate a measuring error (i.e., the patient blinked). Positive numbers quantify the degree of eccentricity from initial fixation. |
111900 | Macula centered | An image of at least 15° angular subtend that is centered on the macula; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111901 | Disc centered | An image of at least 15° angular subtend that is centered on the optic disc; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111902 | Lesion centered | An image of any angular subtend that is centered on a lesion located in any region of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111903 | Disc-macula centered | An image of at least 15° angular subtend centered midway between the disc and macula and containing at least a portion of the disc and both the disc and the macula; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111904 | Mid-peripheral-superior | An image of at least 15° angular subtend positioned between the central zone and the equator, and spanning both the superior-temporal and superior-nasal quadrants of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111905 | Mid-peripheral-superior temporal | An image of at least 15° angular subtend positioned between the central zone and the equator in the superior-temporal quadrant of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111906 | Mid-peripheral-temporal | An image of at least 15° angular subtend positioned between the central zone and the equator, and spanning both the superior-temporal and inferior-temporal quadrants of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111907 | Mid-peripheral-inferior temporal | An image of at least 15° angular subtend positioned between the central zone and the equator in the inferior-temporal quadrant of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111908 | Mid-peripheral-inferior | An image of at least 15° angular subtend positioned between the central zone and the equator, and spanning both the inferior-temporal and inferior-nasal quadrants of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111909 | Mid-peripheral-inferior nasal | An image of at least 15° angular subtend positioned between the central zone and the equator in the inferior-nasal quadrant of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111910 | Mid-peripheral-nasal | An image of at least 15° angular subtend positioned between the central zone and the equator, and spanning both the superior-nasal and inferior-nasal quadrants of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111911 | Mid-peripheral-superior nasal | An image of at least 15° angular subtend positioned between the central zone and the equator in the superior-nasal quadrant of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111912 | Peripheral-superior | An image of at least 15° angular subtend positioned between the equator and the ora serrata, and spanning both the superior temporal and superior nasal quadrants of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111913 | Peripheral-superior temporal | An image of at least 15° angular subtend positioned between the equator and ora serrata in the superior-temporal quadrant of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111914 | Peripheral-temporal | An image of at least 15° angular subtend positioned between the equator and ora serrata, and spanning both the superior-temporal and inferior-temporal quadrants of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111915 | Peripheral-inferior temporal | An image of at least 15° angular subtend positioned between the equator and ora serrata in the inferior-temporal quadrant of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111916 | Peripheral-inferior | An image of at least 15° angular subtend positioned between the equator and ora serrata, and spanning both the inferior-temporal and inferior-nasal quadrants of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111917 | Peripheral-inferior nasal | An image of at least 15° angular subtend positioned between the equator and ora serrata in the inferior-nasal quadrant of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111918 | Peripheral-nasal | An image of at least 15° angular subtend positioned between the equator and ora serrata, and spanning both the superior-nasal and inferior-nasal quadrants of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111919 | Peripheral-superior nasal | An image of at least 15° angular subtend positioned between the equator and ora serrata in the superior-nasal quadrant of the fundus; see Section U.1.8 “Relative Image Position Definitions” in PS3.17 . |
111920 | Time domain | Identifies the use of physical signals with respect to time to capture information. |
111921 | Spectral domain | Identifies the use of physical signals with respect to multiple frequencies to capture information. |
111922 | No corneal compensation | No compensation algorithm for corneal birefringence. |
111923 | Corneal birefringence compensation | Algorithm to compensate for variability in corneal birefringence. |
111924 | Retinal topography | Measurement of the retinal surface contour relative to an assigned datum plane. |
111925 | Retinal nerve fiber layer thickness | Measurement approximating the distance related to the structure between the internal limiting membrane (ILM) and the outer boarder of the retinal nerve fiber layer (RNFL); see Section III.6 “Retinal Thickness Definition” in PS3.17 . |
111926 | Ganglion cell complex thickness | Measurement approximating the distance related to the structure between the ILM and the outer border of the inner plexiform layer (IPL), called the ganglion cell complex (GCC); see Section III.6 “Retinal Thickness Definition” in PS3.17 . |
111927 | Total retinal thickness (ILM to IS-OS) | Measurement approximating the distance related to the structure between the ILM and the inner-outer segment junction (IS-OS); see Section III.6 “Retinal Thickness Definition” in PS3.17 . |
111928 | Total retinal thickness (ILM to RPE) | Measurement approximating the distance related to the structure between the ILM and the retinal pigment epithelium (RPE); see Section III.6 “Retinal Thickness Definition” in PS3.17 . |
111929 | Total retinal thickness (ILM to BM) | Measurement approximating the distance related to the structure between the ILM and the Bruch's membrane (BM); see Section III.6 “Retinal Thickness Definition” in PS3.17 . |
111930 | Absolute ophthalmic thickness | Thickness of a component of the posterior segment of the eye. E.g., thickness of retina, choroid, etc. |
111931 | Thickness deviation category from normative data | Ophthalmic Thickness map based upon statistical significance category (such as percentile) from a normative data set. |
111932 | Thickness deviation from normative data | Ophthalmic Thickness map based upon deviation (such as microns) from a normative data set. |
111933 | Related ophthalmic thickness map | Ophthalmic Thickness Map related to another Ophthalmic Thickness Map or another SOP Instance. |
111934 | Disc-Fovea | An anatomic point centered midway between the disc and fovea centralis. |
111935 | p>5% | Assuming the null hypothesis is true, the conditional percent probability of observing this result is not statistically significant. |
111936 | p<5% | Assuming the null hypothesis is true, the conditional percent probability of observing this result is statistically significant, 95% unlikely to happen by chance. |
111937 | p<2% | Assuming the null hypothesis is true, the conditional percent probability of observing this result is statistically significant, 98% unlikely to happen by chance. |
111938 | p<1% | Assuming the null hypothesis is true, the conditional percent probability of observing this result is statistically significant, 99% unlikely to happen by chance. |
111939 | p<0.5% | Assuming the null hypothesis is true, the conditional percent probability of observing this result is statistically significant, 99.5% unlikely to happen by chance. |
111940 | Corneal axial power map | A two dimensional representation of the axial curvature of the cornea. Axial curvature is calculated from the reciprocal of the distance from a point on a meridian normal at the point to the corneal topographer axis. Also known as sagittal curvature. |
111941 | Corneal instantaneous power map | A two dimensional representation of the instantaneous curvature of the cornea. Instantaneous curvature is calculated from the reciprocal of the distance from a point on a meridian normal at the point to the center of curvature of that point. Also called tangential curvature. |
111942 | Corneal refractive power map | A two dimensional representation of the refractive power of the cornea. Corneal refractive power is calculated using Snell's Law. |
111943 | Corneal elevation map | A two dimensional representation of the elevation of the cornea. Elevation is calculated as the distance from a point on the corneal surface to a point on a reference surface along a line parallel to the corneal topographer axis. For the purpose of visualization the reference surface is usually a sphere or an ellipse. |
111944 | Corneal wavefront map | A two dimensional representation of a wavefront aberration surface of the cornea. Wavefront aberration surface is calculated from the corneal elevation data fit with either the Zernike polynomial series or the Fourier Series. Maps generally display total aberrations and selectable higher order aberrations. |
111945 | Elevation-based corneal tomographer | A device that measures corneal anterior surface shape using elevation-based methods (stereographic and light slit-based). Rasterstereography images a grid pattern illuminating the fluorescein dyed tear film with 2 cameras to produce 3D. Slit-based devices scan the cornea, usually by rotation about the instrument axis centered on the cornea vertex. |
111946 | Reflection-based corneal topographer | A reflection-based device that projects a pattern of light onto the cornea and an image of the reflection of that pattern from the tear film is recorded in one video frame. Light patterns include the circular mire pattern (Placido disc) and spot matrix patterns. Sequential scanning of light spots reflected from the corneal surface is also used requiring multiple video frames for recording. |
111947 | Interferometry-based corneal tomographer | An Interference-based device that projects a beam of light onto and through the cornea. Light reflected from within the cornea is combined with a reference beam giving rise to an interference pattern. Appropriately scanned, this imaging is used to construct 3-dimensional images of the cornea from anterior to posterior surfaces. E.g., swept source OCT. |
112000 | Chest CAD Report | A structured report containing the results of computer-aided detection or diagnosis applied to chest imaging and associated clinical information. |
112001 | Opacity | The shadow of an absorber that attenuates the X-Ray beam more effectively than do surrounding absorbers. In a radiograph, any circumscribed area that appears more nearly white (of lesser photometric density) than its surround [Fraser and Pare]. |
112002 | Series Instance UID | A unique identifier for a series of DICOM SOP instances. |
112003 | Associated Chest Component | A named anatomic region within the chest cavity. |
112004 | Abnormal interstitial pattern | A collection of opacities detected within the continuum of loose connective tissue throughout the lung, that is not expected in a diagnostically normal radiograph. |
112005 | Radiographic anatomy | A type of anatomy that is expected to be detectable on a radiographic (X-Ray based) image. |
112006 | Distribution Descriptor | Characteristic of the extent of spreading of a finding or feature. |
112007 | Border definition | Characteristic of the clarity of the boundary or edges of a finding or feature. |
112008 | Site involvement | The part(s) of the anatomy affected or encompassed by a finding or feature. |
112009 | Type of Content | Characteristic of the matter or substance within a finding or feature. |
112010 | Texture Descriptor | Characteristic of the surface or consistency of a finding or feature. |
112011 | Positioner Primary Angle | Position of the X-Ray beam about the patient from the RAO to LAO direction where movement from RAO to vertical is positive. |
112012 | Positioner Secondary Angle | Position of the X-Ray beam about the patient from the caudal to cranial direction where movement from caudal to vertical is positive. |
112013 | Location in Chest | The zone, lobe or segment within the chest cavity in which a finding or feature is situated. |
112014 | Orientation Descriptor | Vertical refers to orientation parallel to the superior-inferior (cephalad-caudad) axis of the body, with horizontal being perpendicular to this, and an oblique orientation having projections in both the horizontal and vertical. |
112015 | Border shape | Characteristic of the shape formed by the boundary or edges of a finding or feature. |
112016 | Baseline Category | Indicates whether a finding was considered a target lesion, non-target lesion, or non-lesion during evaluation of a baseline series, according to a method such as RECIST. |
112017 | Cavity extent as percent of volume | The extent of a detected cavity, represented as the percent of the surrounding volume that it occupies. |
112018 | Calcification extent as percent of surface | The extent of a detected calcification, represented as the percent of the surrounding surface that it occupies. |
112019 | Calcification extent as percent of volume | The extent of a detected calcification, represented as the percent of the surrounding volume that it occupies. |
112020 | Response Evaluation | A heading for the reporting of response evaluation for treatment of solid tumors. |
112021 | Response Evaluation Method | The system applied in the reporting of response evaluation for treatment of solid tumors. |
112022 | RECIST | Response Evaluation Criteria In Solid Tumors; see Normative References. |
112023 | Composite Feature Modifier | A term that further specifies the name of an item that is an inferred correlation relating two or more individual findings or features. |
112024 | Single Image Finding Modifier | A term that further specifies the name of an item that was detected on one image. |
112025 | Size Descriptor | A qualitative descriptor for the extent of a finding or feature. |
112026 | Width Descriptor | A qualitative descriptor for the thickness of tubular structures, such as blood vessels. |
112027 | Opacity Descriptor | A characteristic that further describes the nature of an opacity. |
112028 | Abnormal Distribution of Anatomic Structure | The type of adverse affect that a finding or feature is having on the surrounding anatomy. |
112029 | WHO | Response evaluation method as defined in chapter 5, "Reporting of Response" of the WHO Handbook for Reporting Results for Cancer Treatment; see Normative References. |
112030 | Calcification Descriptor | Identification of the morphology of detected calcifications. |
112031 | Attenuation Coefficient | A quantitative numerical statement of the relative attenuation of the X-Ray beam at a specified point. Coefficient that describes the fraction of a beam of X-Rays or gamma rays that is absorbed or scattered per unit thickness of the absorber. This value basically accounts for the number of atoms in a cubic cm volume of material and the probability of a photon being scattered or absorbed from the nucleus or an electron of one of these atoms. Usually expressed in Hounsfield units [referred to as CT Number in Fraser and Pare]. |
112032 | Threshold Attenuation Coefficient | An X-Ray attenuation coefficient that is used as a threshold. E.g., in calcium scoring. |
112033 | Abnormal opacity | An opacity that is not expected in a diagnostically normal radiograph. |
112034 | Calculation Description | A textual description of the mathematical method of calculation that resulted in a calculated value. |
112035 | Performance of Pediatric and Adult Chest Radiography, ACR | American College of Radiology. ACR Standard for the Performance of Pediatric and Adult Chest Radiography. In: Standards. Reston, Va: 2001:95-98. |
112036 | ACR Position Statement | American College of Radiology. ACR Position Statement for Quality Control and Improvement, Safety, Infection Control, and Patient Concerns. In: Practice Guidelines and Technical Standards. Reston, Va: 2001:iv. |
112037 | Non-lesion Modifier | A descriptor for a non-lesion object finding or feature, used to indicate whether the object was detected as being internal or external to the patient's body. |
112038 | Osseous Modifier | A concept modifier for an Osseous Anatomy, or bone related, finding. |
112039 | Tracking Identifier | A text label used for tracking a finding or feature, potentially across multiple reporting objects, over time. This label shall be unique within the domain in which it is used. |
112040 | Tracking Unique Identifier | A unique identifier used for tracking a finding or feature, potentially across multiple reporting objects, over time. |
112041 | Target Lesion Complete Response | Disappearance of all target lesions. |
112042 | Target Lesion Partial Response | At least a 30% decrease in the sum of the Longest Diameter of target lesions, taking as reference the baseline sum Longest Diameter. |
112043 | Target Lesion Progressive Disease | At least a 20% increase in the sum of the Longest Diameter of target lesions, taking as reference the smallest sum Longest Diameter recorded since the treatment started, or the appearance of one or more new lesions. |
112044 | Target Lesion Stable Disease | Neither sufficient shrinkage to qualify for Partial Response nor sufficient increase to qualify for Progressive Disease, taking as reference the smallest sum Longest Diameter since the treatment started. |
112045 | Non-Target Lesion Complete Response | Disappearance of all non-target lesions and normalization of tumor marker level. |
112046 | Non-Target Lesion Incomplete Response or Stable Disease | Persistence of one or more non-target lesions and/or maintenance of tumor marker level above the normal limits. |
112047 | Non-Target Lesion Progressive Disease | Appearance of one or more new lesions and/or unequivocal progression of existing non-target lesions. |
112048 | Current Response | The current response evaluation for treatment of solid tumors, according to a method such as RECIST. |
112049 | Best Overall Response | Best response recorded from the start of the treatment until disease progression/recurrence, taking as reference for Progressive Disease the smallest measurements recorded since the treatment started, according to a method such as RECIST. |
112050 | Anatomic Identifier | A text identifier of an anatomic feature when a multiplicity of features of that type may be present, such as "Rib 1", "Rib 2" or thoracic vertebrae "T1" or "T2". |
112051 | Measurement of Response | A measured or calculated evaluation of response. E.g., according to a method such as RECIST, the value would be the calculated sum of the lengths of the longest axes of a set of target lesions. |
112052 | Bronchovascular | Of or relating to a bronchial (lung) specific channel for the conveyance of a body fluid. |
112053 | Osseous | Of, relating to, or composed of bone. |
112054 | Secondary pulmonary lobule | The smallest unit of lung surrounded by connective tissue septa; the unit of lung subtended by any bronchiole that gives off three to five terminal bronchioles [Fraser and Pare]. |
112055 | Agatston scoring method | A method of calculating an overall calcium score, reflecting the calcification of coronary arteries, based on the maximum X-Ray attenuation coefficient and the area of calcium deposits. |
112056 | Volume scoring method | A method of calculating an overall calcium score, reflecting the calcification of coronary arteries, based on the volume of each calcification, typically expressed in mm3. |
112057 | Mass scoring method | A method of calculating an overall calcium score, reflecting the calcification of coronary arteries, based on the total mass of calcification, typically expressed in mg. |
112058 | Calcium score | A measure often arrived at through calculation of findings from CT examination, which is a common predictor of significant stenosis of the coronary arteries. |
112059 | Primary complex | The combination of a focus of pneumonia due to a primary infection with granulomas in the draining hilar or mediastinal lymph nodes [Fraser and Pare]. |
112060 | Oligemia | General or local decrease in the apparent width of visible pulmonary vessels, suggesting less than normal blood flow (reduced blood flow) [Fraser and Pare]. |
112061 | Abnormal lines (1D) | Linear opacity of very fine width, i.e., a nearly one dimensional opacity. |
112062 | Abnormal lucency | Area of abnormal very low X-Ray attenuation, typically lower than aerated lung when occurring in or projecting over lung, or lower than soft tissue when occurring in or projecting over soft tissue. |
112063 | Abnormal calcifications | A calcific opacity within the lung that may be organized, but does not display the trabecular organization of true bone [Fraser and Pare]. |
112064 | Abnormal texture | Relatively homogeneous, extended, pattern of abnormal opacity in the lung, typically low in contrast. |
112065 | Reticulonodular pattern | A collection of innumerable small, linear, and nodular opacities that together produce a composite appearance resembling a net with small superimposed nodules. The reticular and nodular elements are dimensionally of similar magnitude [Fraser and Pare]. |
112066 | Beaded septum sign | Irregular septal thickening that suggests the appearance of a row of beads; usually a sign of lymphangitic carcinomatosis, but may also occur rarely in sarcoidosis [Fraser and Pare]. |
112067 | Nodular pattern | A collection of innumerable, small discrete opacities ranging in diameter from 2-10 mm, generally uniform in size and widespread in distribution, and without marginal spiculation [Fraser and Pare]. |
112068 | Pseudoplaque | An irregular band of peripheral pulmonary opacity adjacent to visceral pleura that simulates the appearance of a pleural plaque and is formed by coalescence of small nodules [Fraser and Pare] . |
112069 | Signet-ring sign | A ring of opacities (usually representing a dilated, thick-walled bronchus) in association with a smaller, round, soft tissue opacity (the adjacent pulmonary artery) suggesting a "signet ring" [Fraser and Pare]. |
112070 | Air bronchiologram | Equivalent of air bronchogram, but in airways assumed to be bronchioles because of peripheral location and diameter [Fraser and Pare]. |
112071 | Air bronchogram | Radiographic shadow of an air-containing bronchus; presumed to represent an air-containing segment of the bronchial tree (identity often inferred) [Fraser and Pare]. |
112072 | Air crescent | Air in a crescentic shape in a nodule or mass, in which the air separates the outer wall of the lesion from an inner sequestrum, which most commonly is a fungus ball of Aspergillusspecies [Fraser and Pare]. |
112073 | Halo sign | Ground-glass opacity surrounding the circumference of a nodule or mass. May be a sign of invasive aspergillosis or hemorrhage of various causes [Fraser and Pare]. |
112074 | Target Lesion at Baseline | Flag denoting that this lesion was identified, at baseline, as a target lesion intended for tracking over time [RECIST]. |
112075 | Non-Target Lesion at Baseline | Flag denoting that this lesion was not identified, at baseline, as a target lesion, and was not intended for tracking over time [RECIST]. |
112076 | Non-Lesion at Baseline | Flag denoting that this finding was identified, at baseline, as a category other than a lesion, and was not intended for tracking over time [RECIST]. |
112077 | Vasoconstriction | Local or general reduction in the caliber of visible pulmonary vessels, presumed to result from decreased flow occasioned by contraction of muscular pulmonary arteries [Fraser and Pare]. |
112078 | Vasodilation | Local or general increase in the width of visible pulmonary vessels resulting from increased pulmonary blood flow [Fraser and Pare]. |
112079 | Architectural distortion | A manifestation of lung disease in which bronchi, pulmonary vessels, a fissure or fissures, or septa of secondary pulmonary lobules are abnormally displaced [Fraser and Pare]. |
112080 | Mosaic perfusion | A patchwork of regions of varied attenuation, interpreted as secondary to regional differences in perfusion [Fraser and Pare]. |
112081 | Pleonemia | Increased blood flow to the lungs or a portion thereof, manifested by a general or local increase in the width of visible pulmonary vessels [Fraser and Pare]. |
112082 | Interface | The common boundary between the shadows of two juxtaposed structures or tissues of different texture or opacity (edge, border) [Fraser and Pare]. |
112083 | Line | A longitudinal opacity no greater than 2 mm in width [Fraser and Pare]. |
112084 | Lucency | The shadow of an absorber that attenuates the primary X-Ray beam less effectively than do surrounding absorbers. In a radiograph, any circumscribed area that appears more nearly black (of greater photometric density) than its surround [Fraser and Pare]. |
112085 | Midlung window | A midlung region, characterized by the absence of large blood vessels and by a paucity of small blood vessels, that corresponds to the minor fissure and adjacent peripheral lung [Fraser and Pare]. |
112086 | Carina angle | The angle formed by the right and left main bronchi at the tracheal bifurcation [Fraser and Pare]. |
112087 | Centrilobular structures | The pulmonary artery and its immediate branches in a secondary lobule; HRCT depicts these vessels in certain cases; a.k.a. core structures or lobular core structures [Fraser and Pare]. |
112088 | Anterior junction line | A vertically oriented linear or curvilinear opacity approximately 1-2 mm wide, commonly projected on the tracheal air shadow [Fraser and Pare]. |
112089 | Posterior junction line | A vertically oriented, linear or curvilinear opacity approximately 2 mm wide, commonly projected on the tracheal air shadow, and usually slightly concave to the right [Fraser and Pare]. |
112090 | Azygoesophageal recess interface | A space in the right side of the mediastinum into which the medial edge of the right lower lobe extends [Fraser and Pare]. |
112091 | Paraspinal line | A vertically oriented interface usually seen in a frontal chest radiograph to the left of the thoracic vertebral column [Fraser and Pare]. |
112092 | Posterior tracheal stripe | A vertically oriented linear opacity ranging in width from 2-5 mm, extending from the thoracic inlet to the bifurcation of the trachea, and visible only on lateral radiographs of the chest [Fraser and Pare]. |
112093 | Right tracheal stripe | A vertically oriented linear opacity approximately 2-3 mm wide extending from the thoracic inlet to the right tracheobronchial angle [Fraser and Pare]. |
112094 | Stripe | A longitudinal composite opacity measuring 2-5 mm in width; acceptable when limited to anatomic structures within the mediastinum [Fraser and Pare]. |
112095 | Hiatus | A gap or passage through an anatomical part or organ; especially : a gap through which another part or organ passes. |
112096 | Rib Scalene Tubercle | A small rounded elevation or eminence on the first rib for the attachment of the scalenus anterior. |
112097 | Vertebral Intervertebral Notch | A groove that serves for the transmission of the vertebral artery. |
112098 | Subscapular Fossa | The concave depression of the anterior surface of the scapula. |
112099 | Scapular Spine | A sloping ridge dividing the dorsal surface of the scapula into the supraspinatous fossa (above), and the infraspinatous fossa (below). |
112100 | Scapular Supraspinatus Fossa | The portion of the dorsal surface of the scapula above the scapular spine. |
112101 | Scapular Infraspinatus Fossa | The portion of the dorsal surface of the scapula below the scapular spine. |
112102 | Aortic knob | The portion of the aortic arch that defines the transition between its ascending and descending limbs. |
112103 | Arch of the Azygos vein | Section of Azygos vein near the fourth thoracic vertebra, where it arches forward over the root of the right lung, and ends in the superior vena cava, just before that vessel pierces the pericardium. |
112104 | Air-fluid level | A local collection of gas and liquid that, when traversed by a horizontal X-Ray beam, creates a shadow characterized by a sharp horizontal interface between gas density above and liquid density below [Fraser and Pare]. |
112105 | Corona radiata | A circumferential pattern of fine linear spicules, approximately 5 mm long, extending outward from the margin of a solitary pulmonary nodule through a zone of relative lucency [Fraser and Pare]. |
112106 | Honeycomb pattern | A number of closely approximated ring shadows representing air spaces 5-10 mm in diameter with walls 2-3 mm thick that resemble a true honeycomb; implies "end-stage" lung [Fraser and Pare]. |
112107 | Fleischner's line(s) | A straight, curved, or irregular linear opacity that is visible in multiple projections; usually situated in the lower half of the lung; vary markedly in length and width [Fraser and Pare]. |
112108 | Intralobular lines | Fine linear opacities present in a lobule when the intralobular interstitium is thickened. When numerous, they may appear as a fine reticular pattern [Fraser and Pare]. |
112109 | Kerley A line | Essentially straight linear opacity 2-6 cm in length and 1-3 mm in width, usually in an upper lung zone [Fraser and Pare]. |
112110 | Kerley B line | A straight linear opacity 1.5-2 cm in length and 1-2 mm in width, usually at the lung base [Fraser and Pare]. |
112111 | Kerley C lines | A group of branching, linear opacities producing the appearing of a fine net, at the lung base [Fraser and Pare]. |
112112 | Parenchymal band | Elongated opacity, usually several millimeters wide and up to about 5 cm long, often extending to the pleura, which may be thickened and retracted at the site of contact [Fraser and Pare]. |
112113 | Reticular pattern | A collection of innumerable small linear opacities that together produce an appearance resembling a net [Fraser and Pare]. |
112114 | Septal line(s) | Usually used in the plural, a generic term for linear opacities of varied distribution produced when the interstitium between pulmonary lobules is thickened [Fraser and Pare]. |
112115 | Subpleural line | A thin curvilinear opacity, a few millimeters or less in thickness, usually less than 1 cm from the pleural surface and paralleling the pleura [Fraser and Pare]. |
112116 | Tramline shadow | Parallel or slightly convergent linear opacities that suggest the planar projection of tubular structures and that correspond in location and orientation to elements of the bronchial tree [Fraser and Pare]. |
112117 | Tubular shadow | Paired, parallel, or slightly convergent linear opacities presumed to represent the walls of a tubular structure seen en face; used if the anatomic nature of a shadow is obscure [Fraser and Pare]. |
112118 | Density | The opacity of a radiographic shadow to visible light; film blackening; the term should never be used to mean an "opacity" or "radiopacity" [Fraser and Pare]. |
112119 | Dependent opacity | Subpleural increased attenuation in dependent lung. The increased attenuation disappears when the region of lung is nondependent; a.k.a. dependent increased attenuation [Fraser and Pare]. |
112120 | Ground glass opacity | Hazy increased attenuation of lung, but with preservation of bronchial and vascular margins; caused by partial filling of air spaces, interstitial thickening, partial collapse of alveoli, normal expiration, or increased capillary blood volume [Fraser and Pare]. |
112121 | Infiltrate | Any ill-defined opacity in the lung [Fraser and Pare]. |
112122 | Micronodule | Discrete, small, round, focal opacity of at least soft tissue attenuation and with a diameter no greater than 7 mm [Fraser and Pare]. |
112123 | Phantom tumor (pseudotumor) | A shadow produced by a local collection of fluid in one of the interlobar fissures, usually elliptic in one radiographic projection and rounded in the other, resembling a tumor [Fraser and Pare]. |
112124 | Shadow | Any perceptible discontinuity in film blackening attributed to the attenuation of the X-Ray beam by a specific anatomic absorber or lesion on or within the body of the patient; to be employed only when more specific identification is not possible [Fraser and Pare]. |
112125 | Small irregular opacities | Term used to define a reticular pattern specific to pneumoconioses [Fraser and Pare]. |
112126 | Small rounded opacities | Term used to define a nodular pattern specific to pneumoconioses [Fraser and Pare]. |
112127 | Tree-in-bud sign | Nodular dilation of centrilobular branching structures that resembles a budding tree and represents exudative bronchiolar dilation [Fraser and Pare]. |
112128 | Granular pattern | Any extended, finely granular pattern of pulmonary opacity within which normal anatomic details are partly obscured [Fraser and Pare]. |
112129 | Miliary pattern | A collection of tiny discrete opacities in the lungs, each measuring 2 mm or less in diameter, generally uniform in size and widespread in distribution [Fraser and Pare]. |
112130 | Mosaic pattern | Generalized pattern of relatively well defined areas in the lung having different X-Ray attenuations due to a longstanding underlying pulmonary disease. |
112131 | Extremely small | A qualitative descriptor of a size that is dramatically less than typical. |
112132 | Very small | A qualitative descriptor of a size that is considerably less than typical. |
112133 | Too small | A qualitative descriptor of a size that is so small as to be abnormal versus expected size. |
112134 | Elliptic | Shaped like an ellipse (oval). |
112135 | Lobulated | A border shape that is made up of, provided with, or divided into lobules (small lobes, curved or rounded projections or divisions). |
112136 | Spiculated | Radially orientated border shape. |
112137 | Sharply defined | The border of a shadow (opacity) is sharply defined [Fraser and Pare]. |
112138 | Distinctly defined | The border of a shadow (opacity) is distinctly defined [Fraser and Pare]. |
112139 | Well demarcated | The border of a shadow (opacity) is well distinct from adjacent structures [Fraser and Pare]. |
112140 | Sharply demarcated | The border of a shadow (opacity) is sharply distinct from adjacent structures [Fraser and Pare]. |
112141 | Poorly demarcated | The border of a shadow (opacity) is poorly distinct from adjacent structures [Fraser and Pare]. |
112142 | Circumscribed | A shadow (opacity) possessing a complete or nearly complete visible border [Fraser and Pare]. |
112143 | Air | Inspired atmospheric gas. The word is sometimes used to describe gas within the body regardless of its composition or site [Fraser and Pare]. |
112144 | Soft tissue | Material having X-Ray attenuation properties similar to muscle. |
112145 | Calcium | Material having X-Ray attenuation properties similar to calcium, a silver-white bivalent metallic element occurring in plants and animals. |
112146 | Acinar | A pulmonary opacity 4-8 mm in diameter, presumed to represent anatomic acinus, or a collection of opacities in the lung, each measuring 4-8 mm in diameter, and together producing an extended, homogeneous shadow [Fraser and Pare]. |
112147 | Air space | The gas-containing portion of the lung parenchyma, including the acini and excluding the interstitium [Fraser and Pare]. |
112148 | Fibronodular | Sharply defined, approximately circular opacities occurring singly or in clusters, usually in the upper lobes [Fraser and Pare]. |
112149 | Fluffy | A shadow (opacity) that is ill-defined, lacking clear-cut margins [Fraser and Pare]. |
112150 | Linear | A shadow resembling a line; any elongated opacity of approximately uniform width [Fraser and Pare]. |
112151 | Profusion | The number of small opacities per unit area or zone of lung. In the International Labor Organization (ILO) classification of radiographs of the pneumoconioses, the qualifiers 0 through 3 subdivide the profusion into 4 categories. The profusion categories may be further subdivided by employing a 12-point scale [Fraser and Pare]. |
112152 | Silhouette sign | The effacement of an anatomic soft tissue border by either a normal anatomic structure or a pathologic state such as airlessness of adjacent lung or accumulation of fluid in the contiguous pleural space; useful in detecting and localizing an opacity along the axis of the X-Ray beam [Fraser and Pare]. |
112153 | Subpleural | Situated or occurring between the pleura and the body wall. |
112154 | Bat's wing distribution | Spatial arrangement of opacities that bears vague resemblance to the shape of a bat in flight; bilaterally symmetric [Fraser and Pare]. |
112155 | Butterfly distribution | Spatial arrangement of opacities that bears vague resemblance to the shape of a butterfly in flight; bilaterally symmetric [Fraser and Pare]. |
112156 | Centrilobular | Referring to the region of the bronchioloarteriolar core of a secondary pulmonary lobule [Fraser and Pare]. |
112157 | Coalescent | The joining together of a number of opacities into a single opacity [Fraser and Pare]. |
112158 | Lobar | Of or relating to a lobe (a curved or rounded projection or division). E.g., involving an entire lobe of the lung. |
112159 | Hyper-acute | Extremely or excessively acute, as a qualitative measure of severity. |
112160 | Homogeneous (uniform opacity) | Of uniform opacity or texture throughout [Fraser and Pare]. |
112161 | Inhomogeneous | Lack of homogeneity in opacity or texture. |
112162 | Target | Discrete opacity centrally within a larger opacity, as a calcification descriptor. |
112163 | Fibrocalcific | Pertaining to sharply defined, linear, and/or nodular opacities containing calcification(s) [Fraser and Pare]. |
112164 | Flocculent | Calcifications made up of loosely aggregated particles, resembling wool. |
112165 | Difference in border shape | A change in the shape formed by the boundary or edges of a finding or feature. | Retired. Replaced by (F-0517E, SRT, "Difference in border shape")
112166 | Difference in border definition | A change in the clarity of the boundary or edges of a finding or feature. | Retired. Replaced by (F-05166, SRT, "Difference in border definition")
112167 | Difference in distribution | A change in the extent of spreading of a finding or feature. | Retired. Replaced by (F-0516C, SRT, "Difference in distribution")
112168 | Difference in site involvement | A change in the part(s) of the anatomy affected or encompassed by a finding or feature. | Retired. Replaced by (F-05170, SRT, "Difference in site involvement")
112169 | Difference in Type of Content | A change in the matter or substance within a finding or feature. | Retired. Replaced by (F-05167, SRT, "Difference in substance")
112170 | Difference in Texture | A change in the surface or consistency of a finding or feature. | Retired. Replaced by (F-0516A, SRT, "Difference in texture")
112171 | Fiducial mark | A location in image space, which may or may not correspond to an anatomical reference, which is often used for registering data sets. |
112172 | Portacath | Connected to an injection chamber placed under the skin in the upper part of the chest. When it is necessary to inject some drug, a specific needle is put in the chamber through the skin and a silicon membrane. The advantage of a portacath is that it may be left in place several months contrarily of "classical" catheters. |
112173 | Chest tube | A tube inserted into the chest wall from outside the body, for drainage. Sometimes used for collapsed lung. Usually connected to a receptor placed lower than the insertion site. |
112174 | Central line | A tube placed into the subclavian vein to deliver medication directly into the venous system. |
112175 | Kidney stent | A stent is a tube inserted into another tube. Kidney stent is a tube that is inserted into the kidney, ureter, and bladder, to help drain urine. Usually inserted through a scoping device presented through the urethra. |
112176 | Pancreatic stent | A stent is a tube inserted into another tube. Pancreatic stent is inserted through the common bile duct to the pancreatic duct, to drain bile. |
112177 | Nipple ring | A non-lesion object that appears to be a circular band, attached to the body via pierced nipple. |
112178 | Coin | A non-lesion object that appears to be a flat round piece of metal. |
112179 | Minimum Attenuation Coefficient | The least quantity assignable, admissible, or possible; the least of a set of X-Ray attenuation coefficients. |
112180 | Maximum Attenuation Coefficient | The greatest quantity or value attainable or attained; the largest of a set of X-Ray attenuation coefficients. |
112181 | Mean Attenuation Coefficient | The value that is computed by dividing the sum of a set of X-Ray attenuation coefficients by the number of values . |
112182 | Median Attenuation Coefficient | The value in an ordered set of X-Ray attenuation coefficients, below and above which there is an equal number of values. |
112183 | Standard Deviation of Attenuation Coefficient | For a set of X-Ray attenuation coefficients: 1) a measure of the dispersion of a frequency distribution that is the square root of the arithmetic mean of the squares of the deviation of each of the class frequencies from the arithmetic mean of the frequency distribution; 2) a parameter that indicates the way in which a probability function or a probability density function is centered around its mean and that is equal to the square root of the moment in which the deviation from the mean is squared. |
112184 | Performance of Pediatric and Adult Thoracic CT | American College of Radiology. ACR Standard for the Performance of Pediatric and Adult Thoracic Computed Tomography (CT). In: Standards. Reston, Va: 2001:103-107. |
112185 | Performance of CT for Detection of Pulmonary Embolism in Adults | American College of Radiology. ACR Standard for the Performance of Computed Tomography for the Detection of Pulmonary Embolism in Adults. In: Standards. Reston, Va: 2001:109-113. |
112186 | Performance of High-Resolution CT of the Lungs in Adults | American College of Radiology. ACR Standard for the Performance of High-Resolution Computed Tomography (HRCT) of the Lungs in Adults. In: Standards. Reston, Va: 2001:115-118. |
112187 | Unspecified method of calculation | The method of calculation of a measurement or other type of numeric value is not specified. |
112188 | Two-dimensional method | The calculation method was performed in two-dimensional space. |
112189 | Three-dimensional method | The calculation method was performed in three-dimensional space. |
112191 | Breast tissue density | The relative density of parenchymal tissue as a proportion of breast volume. |
112192 | Volume of parenchymal tissue | The volume of parenchymal tissue. |
112193 | Volume of breast | The volume of the breast. |
112194 | Mass of parenchymal tissue | The mass of parenchymal tissue. |
112195 | Mass of breast | The mass of the breast. |
112196 | Area of Vascular Calcification | A measured or calculated area of vascular calcification. |
112197 | Volume of Vascular Calcification | A measured or calculated volume of vascular calcification. |
112198 | Percentage of Vascular Calcification | A measured or calculated percentage of vascular calcification. |
112199 | Mass of Vascular Calcification | A measured or calculated mass of vascular calcification. |
112200 | Average calcification distance in a calcification cluster | The average nearest neighbor distance of all individual microcalcifications in a cluster. |
112201 | Standard deviation distance of calcifications in a cluster | The standard deviation of nearest neighbor distance of all individual microcalcifications in a cluster. |
112220 | Colon CAD Report | A structured report containing the results of computer-aided detection or diagnosis applied to colon imaging and associated clinical information. |
112222 | Colon Overall Assessment | Overall interpretation of the colon using C-RADS categorization system. |
112224 | Image Set Properties | Characteristics of a set of images. |
112225 | Slice Thickness | Nominal slice thickness, in mm. |
112226 | Spacing between slices | Distance between contiguous images, measured from the center-to-center of each image. |
112227 | Frame of Reference UID | Uniquely identifies groups of composite instances that have the same coordinate system that conveys spatial and/or temporal information. |
112228 | Recumbent Patient Position with respect to gravity | Patient orientation with respect to downward direction (gravity). |
112229 | Identifying Segment | Distinguishes a part of a segmentation. |
112232 | Polyp stalk width | The diameter of a polyp stalk measured perpendicular to the axis of the stalk. |
112233 | Distance from anus | The length of the path following the centerline of the colon from the anus to the area of interest. |
112238 | Anatomic non-colon | A location in the body that is outside the colon. |
112240 | C0 - Inadequate Study/Awaiting Prior Comparisons | An inadequate study or a study that is awaiting prior comparisons. The study may have inadequate preparation and cannot exclude lesions greater than or equal to ten millimeters owing to presence of fluid or feces. The study may have inadequate insufflation where one or more colonic segments collapsed on both views. Based on "CT Colonography Reporting and Data System: A Consensus Proposal", Radiology, July 2005; 236:3-9. |
112241 | C1 - Normal Colon or Benign Lesion | The study has a normal colon or benign lesion, with the recommendation to continue routine screening. The study has no visible abnormalities of the colon. The study has no polyps greater than six millimeters. The study may have lipoma, inverted diverticulum, or nonneoplastic findings, such as colonic diverticula. Based on "CT Colonography Reporting and Data System: A Consensus Proposal", Radiology, July 2005; 236:3-9. |
112242 | C2 - Intermediate Polyp or Indeterminate Finding | The study has an intermediate polyp or indeterminate finding and surveillance or colonoscopy is recommended. There may be intermediate polyps between six and nine millimeters and there are less than three in number. The study may have an intermediate finding and cannot exclude a polyp that is greater than or equal to six millimeters in a technically adequate exam. Based on "CT Colonography Reporting and Data System: A Consensus Proposal", Radiology, July 2005; 236:3-9. |
112243 | C3 - Polyp, Possibly Advanced Adenoma | The study has a polyp, possibly advanced adenoma, and a follow-up colonoscopy is recommended. The study has a polyp greater than or equal to ten millimeters or the study has three or more polyps that are each between six to nine millimeters. Based on "CT Colonography Reporting and Data System: A Consensus Proposal", Radiology, July 2005; 236:3-9. |
112244 | C4 - Colonic Mass, Likely Malignant | The study has a colonic mass, likely malignant, and surgical consultation is recommended. The lesion compromises bowel lumen and demonstrates extracolonic invasion. Based on "CT Colonography Reporting and Data System: A Consensus Proposal", Radiology, July 2005; 236:3-9. |
112248 | ACR Guideline, Performance of Adult CT Colonography | American College of Radiology Practice Guideline for the Performance of Computed Tomography (CT) Colonography in Adults. In: Practice Guidelines and Technical Standards.Reston, Va: American College of Radiology;2006:371-376. |
112249 | ACR Standard, CT medical physics performance monitoring | American College of Radiology Technical Standard for Diagnostic Medical Physics Performance Monitoring of Computed Tomography (CT) Equipment. In: Practice Guidelines and Technical Standards.Reston, Va: American College of Radiology;2006:945-948. |
112300 | AP+45 | View Orientation Modifier indicates that the view orientation of the imaging plane is rotated +45° along the cranial-caudal axis. |
112301 | AP-45 | View Orientation Modifier indicates that the view orientation of the imaging plane is rotated -45° along the cranial-caudal axis. |
112302 | Anatomical axis of femur | The axis following the shaft of the femur. |
112303 | Acetabular Center of Rotation | Center of Rotation of the natural Acetabulum. |
112304 | Femur Head Center of Rotation | Center of Rotation of the natural femur head. |
112305 | Acetabular Cup Shell | Prosthetic component implanted into the acetabulum. Provides hold for the insert that is mounted inside the cup. |
112306 | Acetabular Cup Insert | Prosthetic pelvic joint component. Inserted into the cup, takes in the femoral head replacement. |
112307 | Acetabular Cup Monoblock | Prosthetic pelvic joint cup including insert. |
112308 | Femoral Head Ball Component | Component for Femoral Head Prosthesis where the conic intake for the stem neck can be exchanged. Combined with a Femoral Head Cone Taper Component. |
112309 | Femoral Head Cone Taper Component | Exchangeable neck intake for composite femoral head prosthesis. Combined with a Femoral Head Ball Component. |
112310 | Femoral Stem | Prosthesis Implanted into the femoral bone to provide force transmission between joint replacement and bone. On the proximal end a conic neck holds the femoral head replacement. |
112311 | Femoral Stem Distal Component | Distal half of a modular stem prosthesis system. Combined with a Stem Proximal Component. |
112312 | Femoral Stem Proximal Component | Proximal half of a modular stem prosthesis system. Combined with a Stem Distal Component. |
112313 | Femoral Stem Component | Stem prosthetic component with a modular insert for an exchangeable neck component. Combined with a Neck Component. |
112314 | Neck Component | Prosthetic Neck to be combined with a Stem Component. |
112315 | Monoblock Stem | Prosthetic Stem and Femoral Head in one piece. |
112316 | Prosthetic Shaft Augment | A proximal attachment to the shaft used to compensate for bone deficiencies or bone loss. |
112317 | Femoral Head Resurfacing Component | Artificial femur head surface needed for the partial replacement of the femoral head where only the surface is replaced. |
112318 | Pinning | Fixation using a pin. |
112319 | Sewing | Fixation sewing several objects together. |
112320 | Bolting | Fixation using a bolt. |
112321 | Wedging | Fixation due to forcing an object into a narrow space. |
112325 | Distal Centralizer | Attachment to the distal end of a cemented stem assuring that the stem is in a central position inside the drilled femoral canal before cementation. |
112340 | Generic 2D Planning | Planning by an unspecified 2D method. |
112341 | Generic 3D Planning | Planning by an unspecified 3D method. |
112342 | Generic Planning for Hip Replacement | Planning of a Hip Replacement, by an unspecified method. |
112343 | Generic Planning for Knee Replacement | Planning of Knee Replacement, by an unspecified method. |
112344 | Müller Method Planning for Hip Replacement | Planning of Hip Replacement according to the procedure of M. E. Müller [Eggli et. al.1998]. |
112345 | Implantation Plan | A Report containing the results of an Implantation Planning Activity. |
112346 | Selected Implant Component | A selection of one Implant Component. |
112347 | Component ID | Identification ID of an Implant Component. |
112348 | Implant Template | An implant template describing the properties (2D/3D geometry and other data) of one Implant Component. |
112350 | Component Connection | A connection of two Connected Implantation Plan Components. |
112351 | Mating Feature Set ID | ID of a Mating Feature Set in an Implant Component. |
112352 | Mating Feature ID | ID of the Mating Feature in a Mating Feature Set in an Implant Component. |
112353 | Spatial Registration | The Spatial Registration of one or more Implant Components. |
112354 | Patient Image | Patient Images used for an implantation planning activity. |
112355 | Assembly | A collection of Component Connections of Implant Components. |
112356 | User Selected Fiducial | Fiducials that are selected by the user and may or may not belong to anatomical landmarks. |
112357 | Derived Fiducial | Fiducials that represent geometric characteristics, such as center of rotation, and are derived from other fiducials. |
112358 | Information used for planning | All parameters and data that were used for the planning activity. |
112359 | Supporting Information | A description of the plan as encapsulated PDF SOP Instance. |
112360 | Implant Component List | A list of all Implant Components selected for an implantation. |
112361 | Patient Data Used During Planning | Reference to objects containing patient data that is used for planning. |
112362 | Degrees of Freedom Specification | A specification of the values from one or more Degrees of Freedom. |
112363 | Degree of Freedom ID | ID of one Degree of Freedom. |
112364 | Related Patient Data Not Used During Planning | Reference to objects containing patient data that were not used for planning but are somehow related. |
112365 | Related Implantation Reports | Implantation Reports that are somehow related. E.g., contemporaneous implantations that are independent. |
112366 | Implant Assembly Template | Implant Assembly Template. |
112367 | Planning Information for Intraoperative Usage | Information that is intended to be used intra-operatively. |
112368 | Implantation Patient Positioning | Position of the patient on the operating room table. |
112369 | Fiducial Intent | Intended use of the fiducial. |
112370 | Component Type | Type of an Implant Component. |
112371 | Manufacturer Implant Template | Implant Template released by the Manufacturer. |
112372 | Derived Planning Images | Images that are created by a planning application. |
112373 | Other Derived Planning Data | Data that is created by a planning application. |
112374 | Connected Implantation Plan Component | One Implant Component that is connected to another Implant Component. |
112375 | Planning Method | The method used for planning. |
112376 | Degree of Freedom Exact Translational Value | Defines the exact value that was planned for translation. |
112377 | Degree of Freedom Minimum Translational Value | Defines the minimum value that was planned for translation. |
112378 | Degree of Freedom Maximum Translational Value | Defines the maximum value that was planned for translation. |
112379 | Degree of Freedom Exact Rotational Translation Value | Defines the exact value that was planned for rotation. |
112380 | Degree of Freedom Minimum Rotational Value | Defines the minimum value that was planned for rotation. |
112381 | Degree of Freedom Maximum Rotational Value | Defines the maximum value that was planned for rotation. |
112700 | Peri-operative Photographic Imaging | Procedure step protocol for photographic imaging of surgical procedures, including photography of specimens collected. |
112701 | Gross Specimen Imaging | Procedure step protocol for imaging gross specimens, typically with a photographic camera (modality XC), and planning further dissection. |
112702 | Slide Microscopy | Procedure step protocol for imaging slide specimens. |
112703 | Whole Slide Imaging | Procedure step protocol for imaging slide specimens using a whole slide scanner. |
112704 | WSI 20X RGB | Procedure step protocol for imaging slide specimens using a whole slide scanner with a 20X nominal objective lens, in full color, with a single imaging focal plane across the image. |
112705 | WSI 40X RGB | Procedure step protocol for imaging slide specimens using a whole slide scanner with a 40X nominal objective lens, in full color, with a single imaging focal plane across the image. |
112706 | Illumination Method | Technique of illuminating specimen. |
112707 | Number of focal planes | Number of focal planes for a microscopy image acquisition. |
112708 | Focal plane Z offset | Nominal distance above a reference plane (typically a slide glass substrate top surface) of the focal plane. |
112709 | Magnification selection | Microscope magnification based on nominal objective lens power. |
112710 | Illumination wavelength | Nominal center wavelength for an imaging spectral band. |
112711 | Illumination spectral band | Name (coded) for an imaging spectral band. |
112712 | Optical filter type | Type of filter inserted into the optical imaging path. |
112713 | Tissue selection method | Technique for identifying tissue to be imaged versus area of slide not to be imaged. |
112714 | Multiple planes | Imaging performed at multiple imaging (focal) planes. |
112715 | 5X | Nominal 5 power objective lens, resulting in a digital image at approximately 2 um/pixel spacing. |
112716 | 10X | Nominal 10 power objective lens, resulting in a digital image at approximately 1 um/pixel spacing. |
112717 | 20X | Nominal 20 power microscope objective lens, resulting in a digital image at approximately 0.5 um/pixel spacing. |
112718 | 40X | Nominal 40 power microscope objective lens, with a combined condenser and objective lens numerical aperture of approximately 1.3, resulting in a digital image at approximately 0.25 um/pixel spacing. |
112719 | Nominal empty tile suppression | Equipment-specific nominal or default method for identifying tiles without tissue imaged for suppression from inclusion in image object. |
112720 | High threshold empty tile suppression | Equipment-specific high threshold method for identifying tiles without tissue imaged for suppression from inclusion in image object. |
112721 | No empty tile suppression | Tiles without tissue imaged are not suppressed from inclusion in image object. |
113000 | Of Interest | Of Interest. |
113001 | Rejected for Quality Reasons | Rejected for Quality Reasons. |
113002 | For Referring Provider | For Referring Provider. |
113003 | For Surgery | For Surgery. |
113004 | For Teaching | For Teaching. |
113005 | For Conference | For Conference. |
113006 | For Therapy | For Therapy. |
113007 | For Patient | For Patient. |
113008 | For Peer Review | For Peer Review. |
113009 | For Research | For Research. |
113010 | Quality Issue | Quality Issue. |
113011 | Document Title Modifier | Document Title Modifier. |
113012 | Key Object Description | Key Object Description. |
113013 | Best In Set | A selection that represents the "best" chosen from a larger set of items. E.g., the best images within a Study or Series. The criteria against which "best" is measured is not defined. Contrast this with the more specific term "Best illustration of finding". |
113014 | Study | A study is a collection of one or more series of medical images, presentation states, and/or SR documents that are logically related for the purpose of diagnosing a patient. A study may include composite instances that are created by a single modality, multiple modalities or by multiple devices of the same modality. [From Section A.1.2.2 “Study IE” in PS3.3 ] |
113015 | Series | A distinct logical set used to group composite instances. All instances within a Series are of the same modality, in the same Frame of Reference (if any), and created by the same equipment. [See Section A.1.2.3 “Series IE” in PS3.3 ] |
113016 | Performed Procedure Step | An arbitrarily defined unit of service that has actually been performed (not just scheduled). [From Section 7.3.1.9 “Modality Performed Procedure Step” in PS3.3 ] |
113017 | Stage-View | An image or set of images illustrating a specific stage (phase in a stress echo exam protocol) and view (combination of the transducer position and orientation at the time of image acquisition). |
113018 | For Printing | For Printing. |
113020 | For Report Attachment | Selection of information objects for attachment to the clinical report of the Current Requested Procedure. |
113021 | For Litigation | List of objects that are related to litigation and should be specially handled. E.g., may apply if a complaint has been received regarding a patient, or a specific set of images has been the subject of a subpoena, and needs to be sequestered or excluded from automatic purging according to retention policy. |
113026 | Double exposure | Double exposure. |
113030 | Manifest | A list of objects that have been exported out of one organizational domain into another domain. Typically, the first domain has no direct control over what the second domain will do with the objects. |
113031 | Signed Manifest | A signed list of objects that have been exported out of one organizational domain into another domain, referenced securely with either Digital Signatures or MACs. Typically, the first domain has no direct control over what the second domain will do with the objects. |
113032 | Complete Study Content | The list of objects that constitute a study at the time that the list was created. |
113033 | Signed Complete Study Content | The signed list of objects that constitute a study at the time that the list was created, referenced securely with either Digital Signatures or MACs. |
113034 | Complete Acquisition Content | The list of objects that were generated in a single procedure step. |
113035 | Signed Complete Acquisition Content | The signed list of objects that were generated in a single procedure step, referenced securely with either Digital Signatures or MACs. |
113036 | Group of Frames for Display | A list of frames or single-frame or entire multi-frame instances that together constitute a set for some purpose, such as might be displayed together in the same viewport, as distinct from another set that might be displayed in a separate viewport. |
113037 | Rejected for Patient Safety Reasons | List of objects whose use is potentially harmful to the patient. E.g., an improperly labeled image could lead to dangerous surgical decisions. |
113038 | Incorrect Modality Worklist Entry | List of objects that were acquired using an incorrect modality worklist entry, and that should not be used, since they may be incorrectly identified. |
113039 | Data Retention Policy Expired | List of objects that have expired according to a defined data retention policy. |
113040 | Lossy Compression | Lossy compression has been applied to an image. |
113041 | Apparent Diffusion Coefficient | Values are derived by calculation of the apparent diffusion coefficient. |
113042 | Pixel by pixel addition | Values are derived by the pixel by pixel addition of two images. |
113043 | Diffusion weighted | Values are derived by calculation of the diffusion weighting. |
113044 | Diffusion Anisotropy | Values are derived by calculation of the diffusion anisotropy. |
113045 | Diffusion Attenuated | Values are derived by calculation of the diffusion attenuation. |
113046 | Pixel by pixel division | Values are derived by the pixel by pixel division of two images. |
113047 | Pixel by pixel mask | Values are derived by the pixel by pixel masking of one image by another. |
113048 | Pixel by pixel Maximum | Values are derived by calculating the pixel by pixel maximum of two or more images. |
113049 | Pixel by pixel mean | Values are derived by calculating the pixel by pixel mean of two or more images. |
113050 | Metabolite Maps from spectroscopy data | Values are derived by calculating from spectroscopy data pixel values localized in two dimensional space based on the concentration of specific metabolites (i.e, at specific frequencies). |
113051 | Pixel by pixel Minimum | Values are derived by calculating the pixel by pixel minimum of two or more images. |
113052 | Mean Transit Time | The time required for blood to pass through a region of tissue. |
113053 | Pixel by pixel multiplication | Values are derived by the pixel by pixel multiplication of two images. |
113054 | Negative Enhancement Integral | Values are derived by calculating negative enhancement integral values. |
113055 | Regional Cerebral Blood Flow | The flow rate of blood perfusing a region of the brain as volume per mass per unit of time. |
113056 | Regional Cerebral Blood Volume | The volume of blood perfusing a region of brain as as volume per mass. |
113057 | R-Coefficient | Correlation Coefficient, r. |
113058 | Proton Density | Values are derived by calculating proton density values. |
113059 | Signal Change | Values are derived by calculating signal change values. |
113060 | Signal to Noise | Values are derived by calculating the signal to noise ratio. |
113061 | Standard Deviation | Values are derived by calculating the standard deviation of two or more images. |
113062 | Pixel by pixel subtraction | Values are derived by the pixel by pixel subtraction of two images. |
113063 | T1 | Values are derived by calculating T1 values. |
113064 | T2* | Values are derived by calculating T2* values. |
113065 | T2 | Values are derived by calculating T2 values. |
113066 | Time Course of Signal | Values are derived by calculating values based on the time course of signal. |
113067 | Temperature encoded | Values are derived by calculating values based on temperature encoding. |
113068 | Student's T-Test | Values are derived by calculating the value of the Student's T-Test statistic from multiple image samples. |
113069 | Time To Peak | The time from the start of the contrast agent injection to the maximum enhancement value. |
113070 | Velocity encoded | Values are derived by calculating values based on velocity encoded. E.g., phase contrast. |
113071 | Z-Score | Values are derived by calculating the value of the Z-Score statistic from multiple image samples. |
113072 | Multiplanar reformatting | Values are derived by reformatting in a flat plane other than that originally acquired. |
113073 | Curved multiplanar reformatting | Values are derived by reformatting in a curve plane other than that originally acquired. |
113074 | Volume rendering | Values are derived by volume rendering of acquired data. |
113075 | Surface rendering | Values are derived by surface rendering of acquired data. |
113076 | Segmentation | Values are derived by segmentation (classification into tissue types) of acquired data. |
113077 | Volume editing | Values are derived by selectively editing acquired data (removing values from the volume), such as in order to remove obscuring structures or noise. |
113078 | Maximum intensity projection | Values are derived by maximum intensity projection of acquired data. |
113079 | Minimum intensity projection | Values are derived by minimum intensity projection of acquired data. |
113080 | Glutamate and glutamine | For single-proton MR spectroscopy, the resonance peak corresponding to glutamate and glutamine. |
113081 | Choline/Creatine Ratio | For single-proton MR spectroscopy, the ratio between the Choline and Creatine resonance peaks. |
113082 | N-acetylaspartate /Creatine Ratio | For single-proton MR spectroscopy, the ratio between the N-acetylaspartate and Creatine resonance peaks. |
113083 | N-acetylaspartate /Choline Ratio | For single-proton MR spectroscopy, the ratio between the N-acetylaspartate and Choline resonance peaks. |
113085 | Spatial resampling | Values are derived by spatial resampling of acquired data. |
113086 | Edge enhancement | Values are derived by edge enhancement. |
113087 | Smoothing | Values are derived by smoothing. |
113088 | Gaussian blur | Values are derived by Gaussian blurring. |
113089 | Unsharp mask | Values are derived by unsharp masking. |
113090 | Image stitching | Values are derived by stitching two or more images together. |
113091 | Spatially-related frames extracted from the volume | Spatially-related frames in this image are representative frames from the referenced 3D volume data set. |
113092 | Temporally-related frames extracted from the set of volumes | Temporally-related frames in this image are representative frames from the referenced 3D volume data set. |
113093 | Polar to Rectangular Scan Conversion | Conversion of a polar coordinate image to rectangular (Cartesian) coordinate image. |
113094 | Creatine and Choline | For single-proton MR spectroscopy, the resonance peak corresponding to creatine and choline. |
113095 | Lipid and Lactate | For single-proton MR spectroscopy, the resonance peak corresponding to lipid and lactate. |
113096 | Creatine+Choline/ Citrate Ratio | For single-proton MR spectroscopy, the ratio between the Choline and Creatine resonance peak and the Citrate resonance peak. |
113097 | Multi-energy proportional weighting | Image pixels created through proportional weighting of multiple acquisitions at distinct X-Ray energies. |
113100 | Basic Application Confidentiality Profile | De-identification using a profile defined in PS3.15 that requires removing all information related to the identity and demographic characteristics of the patient, any responsible parties or family members, any personnel involved in the procedure, the organizations involved in ordering or performing the procedure, additional information that could be used to match instances if given access to the originals, such as UIDs, dates and times, and private attributes, when that information is present in the non-Pixel Data Attributes, including graphics or overlays. |
113101 | Clean Pixel Data Option | Additional de-identification according to an option defined in PS3.15 that requires any information burned in to the Pixel Data corresponding to the Attribute information specified to be removed by the Profile and any other Options specified also be removed. |
113102 | Clean Recognizable Visual Features Option | Additional de-identification according to an option defined in PS3.15 that requires that sufficient removal or distortion of the Pixel Data shall be applied to prevent recognition of an individual from the instances themselves or a reconstruction of a set of instances. |
113103 | Clean Graphics Option | Additional de-identification according to an option defined in PS3.15 that requires that any information encoded in graphics, text annotations or overlays corresponding to the Attribute information specified to be removed by the Profile and any other Options specified also be removed. |
113104 | Clean Structured Content Option | Additional de-identification according to an option defined in PS3.15 that requires that any information encoded in SR Content Items or Acquisition Context Sequence Items corresponding to the Attribute information specified to be removed by the Profile and any other Options specified also be removed. |
113105 | Clean Descriptors Option | Additional de-identification according to an option defined in PS3.15 that requires that any information that is embedded in text or string Attributes corresponding to the Attribute information specified to be removed by the Profile and any other Options specified also be removed. |
113106 | Retain Longitudinal Temporal Information Full Dates Option | Retention of information that would otherwise be removed during de-identification according to an option defined in PS3.15 that requires that any dates and times be retained,. |
113107 | Retain Longitudinal Temporal Information Modified Dates Option | Retention of information that would otherwise be removed during de-identification according to an option defined in PS3.15 that requires that any dates and times be modified in a manner that preserves temporal relationships. E.g., Study Date and Time. |
113108 | Retain Patient Characteristics Option | Retention of information that would otherwise be removed during de-identification according to an option defined in PS3.15 that requires that any physical characteristics of the patient, which are descriptive rather than identifying information per se, be retained. E.g., Patient's Age, Sex, Size (height) and Weight. |
113109 | Retain Device Identity Option | Retention of information that would otherwise be removed during de-identification according to an option defined in PS3.15 that requires that any information that identifies a device be retained. E.g., Device Serial Number. |
113110 | Retain UIDs Option | Retention of information that would otherwise be removed during de-identification according to an option defined in PS3.15 that requires that UIDs be retained. E.g., SOP Instance UID. |
113111 | Retain Safe Private Option | Retention of information that would otherwise be removed during de-identification according to an option defined in PS3.15 that requires that private attributes that are known not to contain identity information be retained. E.g., private SUV scale factor. |
113500 | Radiopharmaceutical Radiation Dose Report | The procedure report is a Radiopharmaceutical Radiation Dose report |
113502 | Radiopharmaceutical Administration | Information pertaining to the administration of a radiopharmaceutical |
113503 | Radiopharmaceutical Administration Event UID | Unique identification of a single radiopharmaceutical administration event. |
113505 | Intravenous Extravasation Symptoms | Initial signs or symptoms of extravasation |
113506 | Estimated Extravasation Activity | The estimated percentage of administered activity lost at the injection site. The estimation includes extravasation, paravenous administration and leakage at the injection site. |
113507 | Administered activity | The calculated activity at the Radiopharmaceutical Start Time when the radiopharmaceutical is administered to the patient. The residual activity (i.e. radiopharmaceutical not administered) , if measured, is reflected in the calculated value. The estimated extravasation is not reflected in the calculated value. |
113508 | Pre-Administration Measured Activity | Radioactivity measurement of radiopharmaceutical before or during the administration. |
113509 | Post-Administration Measured Activity | Radioactivity measurement of radiopharmaceutical after the administration. |
113510 | Drug Product Identifier | Registered drug establishment code for product, coding scheme example is NDC or RxNorm |
113511 | Radiopharmaceutical Dispense Unit Identifier | The human readable identification of the specific radiopharmaceutical dispensed quantity or dose ("dose" as unit of medication delivery, not radiation dose measure) to be administered to the patient. |
113512 | Radiopharmaceutical Lot Identifier | Identifies the vial, batch or lot number from which the individual dispense radiopharmaceutical quantity (dose) is produced. The Radiopharmaceutical Dispense Unit Identifier records the identification for each individual dose. |
113513 | Reagent Vial Identifier | Identifies the lot or unit serial number for the reagent component for the radiopharmaceutical. |
113514 | Radionuclide Vial Identifier | Identifies the lot or unit serial number for the radionuclide component for the radiopharmaceutical. |
113516 | Prescription Identifier | Administered Product's Prescription Number |
113517 | Organ Dose Information | Information pertaining to the estimated absorbed radiation dose to an organ. |
113518 | Organ Dose | The absorbed radiation dose to organ |
113520 | MIRD Pamphlet 1 | Reference authority MIRD Pamphlet No.1 (rev) ,Society of Nuclear Medicine, 1976 |
113521 | ICRP Publication 53 | Reference authority ICRP, 1988. Radiation Dose to Patients from Radiopharmaceuticals. ICRP Publication 53. Ann. ICRP 18 (1-4). |
113522 | ICRP Publication 80 | Reference authority ICRP, 1998. Radiation Dose to Patients from Radiopharmaceuticals (Addendum to ICRP Publication 53). ICRP Publication 80. Ann. ICRP 28 (3). |
113523 | ICRP Publication 106 | Reference authority ICRP, 2008. Radiation Dose to Patients from Radiopharmaceuticals - Addendum 3 to ICRP Publication 53. ICRP Publication 106. Ann. ICRP 38 (1-2). |
113526 | MIRDOSE | Reference authority Stabin MG, Sparks RB, Crowe E (1994) MIRDOSE: personal computer software for internal dose assessment in nuclear medicine [Computer program] |
113527 | OLINDA-EXM | Reference authority Stabin MG, Sparks RB, Crowe E (2005) OLINDA/EXM: The Second-Generation Personal Computer Software for Internal Dose Assessment in Nuclear Medicine [Computer program] |
113528 | Package Insert | Reference authority The reported organ dose is based on radiopharmaceutical's package insert. |
113529 | Institutionally Approved Estimates | Reference authority The reported organ dose is based on Institutionally approved estimates from the Radioactive Drug Research Committee (RDRC) of the institution itself. |
113530 | Investigational New Drug | Reference authority The reported organ dose is based on an Investigation new drug. |
113540 | Activity Measurement Device | The type of device that performed the activity measurement. |
113541 | Dose Calibrator | The device that measures the radiation activity of the radiopharmaceutical |
113542 | Infusion System | Radiopharmaceutical Infusion System |
113543 | Generator | Radioisotope Generator |
113550 | Fasting Duration | The number hours the patient has gone without food. |
113551 | Hydration Volume | The amount of fluids the patient has consumed before the procedure. |
113552 | Recent Physical Activity | A description of physical activity the patient performed before the start of the procedure, such as that which may affect imaging agent biodistribution. |
113560 | Acute unilateral renal blockage | Blockage in one of the tubes (ureters) that drain urine from the kidneys |
113561 | Low Thyroid Uptake | 5% or less Thyroid Uptake of Iodine |
113562 | High Thyroid Uptake | 25% or higher Thyroid Uptake of Iodine |
113563 | Severely Jaundiced | The patient exhibits symptoms severe of jaundice and/or has a Bilirubin >10 mg/dL. |
113568 | Extravasation visible in image | Extravasation or paravenous administration of the product is visible in the images. |
113570 | Cockroft-Gault Formula estimation of GFR | The measurement method of the Glomerular Filtration Rate is Cockroft-Gault Formula |
113571 | CKD-EPI Formula estimation of GFR | The measurement method of the Glomerular Filtration Rate is CKD-EPI Formula |
113572 | Glomerular Filtration Rate (MDRD) | The measurement method of the Glomerular Filtration Rate is MDRD |
113573 | Glomerular Filtration Rate non-black (MDRD) | The measurement method of the Glomerular Filtration Rate is non-black MDRD |
113574 | Glomerular Filtration Rate black (MDRD) | The measurement method of the Glomerular Filtration Rate is black (MDRD) |
113575 | Glomerular Filtration Rate female (MDRD) | The measurement method of the Glomerular Filtration Rate is female (MDRD) |
113576 | Glomerular Filtration Rate Cystatin-based formula | The measurement method of the Glomerular Filtration Rate is Cystatin-based formula |
113577 | Glomerular Filtration Rate Creatinine-based formula (Schwartz) | The measurement method of the Glomerular Filtration Rate is Creatinine-based formula (Schwartz) |
113601 | Small: < 32.0 cm lateral thickness | Small body thickness for calcium scoring adjustment. Lateral thickness is measured from skin-to-skin, at the level of the proximal ascending aorta, from an A/P localizer image. |
113602 | Medium: 32.0-38.0 cm lateral thickness | Medium body thickness for calcium scoring adjustment. Lateral thickness is measured from skin-to-skin, at the level of the proximal ascending aorta, from an A/P localizer image. |
113603 | Large: > 38.0 cm lateral thickness | Large body thickness for calcium scoring adjustment. Lateral thickness is measured from skin-to-skin, at the level of the proximal ascending aorta, from an A/P localizer image. |
113605 | Irradiation Event Label | A human-readable label identifying an irradiation event. |
113606 | Label Type | The type of a human-readable label. |
113607 | Series Number | A number that identifies a Series. Corresponds to (0020,0011) in PS3.3. |
113608 | Acquisition Number | A number that identifies an Acquisition. Corresponds to (0020,0012) in PS3.3. |
113609 | Instance Number | A number that identifies an Instance. Corresponds to (0020,0013) in PS3.3. |
113611 | Stationary Acquisition | Acquisition where the X-Ray source does not move in relation to the patient. |
113612 | Stepping Acquisition | Acquisition where the X-Ray source moves laterally in relation to the patient. |
113613 | Rotational Acquisition | Acquisition where the X-Ray source moves angularly in relation to the patient. |
113620 | Plane A | Primary plane of a Biplane acquisition equipment. |
113621 | Plane B | Secondary plane of a Biplane acquisition equipment. |
113622 | Single Plane | Single plane acquisition equipment. |
113630 | Continuous | Continuous X-Ray radiation is applied during an irradiation event. |
113631 | Pulsed | Pulsed X-Ray radiation is applied during an irradiation event. |
113650 | Strip filter | Filter with uniform thickness. |
113651 | Wedge filter | Filter with variation in thickness from one edge to the opposite edge. |
113652 | Butterfly filter | Filter with two triangular sections. |
113653 | Flat filter | Filter with uniform thickness that is for spectral filtering only. E.g., filter out low energy portion of the X-Ray that would only contribute to skin dose, but not to image. |
113661 | Outline of lobulations | A polyline defining the outline of a lobulated finding. |
113662 | Inner limits of fuzzy margin | A polyline defining the inner limits of a finding with fuzzy margin. |
113663 | Outer limits of fuzzy margin | A polyline defining the outer limits of a finding with fuzzy margin. |
113664 | Outline of spiculations | A polyline defining the outline of the spiculations of a finding. |
113665 | Linear spiculation | A polyline segment graphically indicating the location and direction of a spiculation of a finding. |
113666 | Pixelated spiculations | A collection of points indicating the pixel locations of the spiculations of a finding. |
113669 | Orthogonal location arc | Connected line segments indicating the center of location of a finding on an orthogonal view. |
113670 | Orthogonal location arc inner margin | Connected line segments indicating the inner margin of the location of a finding on an orthogonal view. |
113671 | Orthogonal location arc outer margin | Connected line segments indicating the outer location of a finding on an orthogonal view. |
113680 | Quality Control Intent | This procedure is intended to gather data that is used for calibration or other quality control purposes. |
113681 | Phantom | An artificial subject of an imaging study. |
113682 | ACR Accreditation Phantom - CT | A phantom acceptable for the ACR Computed Tomography Accreditation program. |
113683 | ACR Accreditation Phantom - MR | A phantom acceptable for the ACR Magnetic Resonance Imaging Accreditation program. |
113684 | ACR Accreditation Phantom - Mammography | A phantom acceptable for the ACR Mammography Accreditation program. |
113685 | ACR Accreditation Phantom - Stereotactic Breast Biopsy | A phantom acceptable for the ACR Stereotactic Breast Biopsy Accreditation program. |
113686 | ACR Accreditation Phantom - ECT | A phantom acceptable for the ACR SPECT Accreditation program (but not for PET). |
113687 | ACR Accreditation Phantom - PET | A phantom acceptable for the ACR PET Accreditation program (but not for SPECT). |
113688 | ACR Accreditation Phantom - ECT/PET | A SPECT phantom with a PET faceplate acceptable for both the ACR SPECT and PET Accreditation programs. |
113689 | ACR Accreditation Phantom - PET Faceplate | A PET faceplate (made to fit an existing flangeless or flanged ECT phantom) acceptable for the ACR PET Accreditation program. |
113690 | IEC Head Dosimetry Phantom | A phantom used for CTDI measurement in head modes according to IEC 60601-2-44, Ed.2.1 (Head 16 cm diameter Polymethyl methacrylate PMMA). |
113691 | IEC Body Dosimetry Phantom | A phantom used for CTDI measurement in body modes according to IEC 60601-2-44, Ed.2.1 (Body 32cm diameter Polymethyl methacrylate PMMA). |
113692 | NEMA XR21-2000 Phantom | A phantom in accordance with NEMA standard XR-21-2000. |
113701 | X-Ray Radiation Dose Report | X-Ray Radiation Dose Report. |
113702 | Accumulated X-Ray Dose Data | X-Ray dose data accumulated over multiple irradiation events. E.g., for a study or a performed procedure step. |
113704 | Projection X-Ray | Imaging using a point X-Ray source with a diverging beam projected onto a 2 dimensional detector. |
113705 | Scope of Accumulation | Entity over which dose accumulation values are integrated. |
113706 | Irradiation Event X-Ray Data | X-Ray dose data for a single Irradiation Event. |
113710 | Niobium or Niobium compound | Material containing Niobium or a Niobium compound | Retired. Replaced by (C-1190E, SRT, "Niobium or Niobium compound")
113711 | Europium or Europium compound | Material containing Europium or a Europium compound | Retired. Replaced by (C-1190F, SRT, "Europium or Europium compound")
113720 | Calibration Protocol | Describes the method used to derive the calibration factor. |
113721 | Irradiation Event Type | Denotes the type of irradiation event recorded. |
113722 | Dose Area Product Total | Total calculated Dose Area Product (in the scope of the including report). |
113723 | Calibration Date | Last calibration Date for the integrated dose meter or dose calculation. |
113724 | Calibration Responsible Party | Individual or organization responsible for calibration. |
113725 | Dose (RP) Total | Total Dose related to Reference Point (RP). (in the scope of the including report). |
113726 | Fluoro Dose Area Product Total | Total calculated Dose Area Product applied in Fluoroscopy Modes (in the scope of the including report). |
113727 | Acquisition Dose Area Product Total | Total calculated Dose Area Product applied in Acquisition Modes (in the scope of the including report). |
113728 | Fluoro Dose (RP) Total | Dose applied in Fluoroscopy Modes, related to Reference Point (RP). (in the scope of the including report). |
113729 | Acquisition Dose (RP) Total | Dose applied in Acquisition Modes, related to Reference Point (RP). (in the scope of the including report). |
113730 | Total Fluoro Time | Total accumulated clock time of Fluoroscopy in the scope of the including report (i.e., the sum of the Irradiation Duration values for accumulated fluoroscopy irradiation events). |
113731 | Total Number of Radiographic Frames | Accumulated Count of exposure pulses (single or multi-frame encoded) created from irradiation events performed with high dose (acquisition). |
113732 | Fluoro Mode | Mode of application of X-Rays during Fluoroscopy. |
113733 | KVP | Applied X-Ray Tube voltage at peak of X-Ray generation, in kilovolts; Mean value if measured over multiple peaks (pulses). |
113734 | X-Ray Tube Current | Mean value of applied Tube Current. |
113735 | Exposure Time | Cumulative time the patient has received X-Ray exposure during the irradiation event | Retired. Replaced by (113824, DCM, "Exposure Time").
113736 | Exposure | Mean value of X-Ray Current Time product. |
113737 | Distance Source to Reference Point | Distance to the Reference Point (RP) defined according to IEC 60601-2-43 or equipment defined. |
113738 | Dose (RP) | Dose applied at the Reference Point (RP). |
113739 | Positioner Primary End Angle | Positioner Primary Angle at the end of an irradiation event. For further definition see (112011, DCM, "Positioner Primary Angle"). |
113740 | Positioner Secondary End Angle | Positioner Secondary Angle at the end of an irradiation event. For further definition see (112012, DCM, "Positioner Secondary Angle"). |
113742 | Irradiation Duration | Clock time from the start of loading time of the first pulse until the loading time trailing edge of the final pulse in the same irradiation event. ### Note Loading time is defined in IEC 60601-1-3:2008, 3.37, and described in IEC 60601-2-54:2009, 203.4.101.3. |
113743 | Patient Orientation | Orientation of the Patient with respect to Gravity. |
113744 | Patient Orientation Modifier | Enhances or modifies the Patient orientation specified in Patient Orientation. |
113745 | Patient Table Relationship | Orientation of the Patient with respect to the Head of the Table. |
113748 | Distance Source to Isocenter | Distance from the X-Ray Source to the Equipment C-Arm Isocenter.(Center of Rotation). |
113750 | Distance Source to Detector | Measured or calculated distance from the X-Ray source to the detector plane in the center of the beam. |
113751 | Table Longitudinal Position | Table Longitudinal Position with respect to an arbitrary chosen reference by the equipment. Table motion towards LAO is positive assuming that the patient is positioned supine and its head is in normal position. |
113752 | Table Lateral Position | Table Lateral Position with respect to an arbitrary chosen reference by the equipment. Table motion towards CRA is positive assuming that the patient is positioned supine and its head is in normal position. |
113753 | Table Height Position | Table Height Position with respect to an arbitrary chosen reference by the equipment in (mm). Table motion downwards is positive. |
113754 | Table Head Tilt Angle | Angle of the head-feet axis of the table in degrees relative to the horizontal plane. Positive values indicate that the head of the table is upwards. |
113755 | Table Horizontal Rotation Angle | Rotation of the table in the horizontal plane (clockwise when looking from above the table). |
113756 | Table Cradle Tilt Angle | Angle of the left-right axis of the table in degrees relative to the horizontal plane. Positive values indicate that the left of the table is upwards. |
113757 | X-Ray Filter Material | X-Ray absorbing material used in the filter. |
113758 | X-Ray Filter Thickness Minimum | The minimum thickness of the X-Ray absorbing material used in the filters. |
113759 | Table Longitudinal End Position | Table Longitudinal Position at the end of an irradiation event; see (113751, DCM, "Table Longitudinal Position"). |
113760 | Table Lateral End Position | Table Lateral Position at the end of an irradiation event; see (113752, DCM, "Table Lateral Position"). |
113761 | Table Height End Position | Table Height Position at the end of an irradiation event; see (113753, DCM, "Table Height Position"). |
113763 | Calibration Uncertainty | Uncertainty of the 'actual' value. |
113764 | Acquisition Plane | Identification of Acquisition Plane with Biplane systems. |
113766 | Focal Spot Size | Nominal Size of Focal Spot of X-Ray Tube. |
113767 | Average X-Ray Tube Current | Average X-Ray Tube Current averaged over time for pulse or for continuous Fluoroscopy. |
113768 | Number of Pulses | Number of pulses applied by X-Ray systems during an irradiation event (acquisition run or pulsed fluoro). |
113769 | Irradiation Event UID | Unique identification of a single irradiation event. |
113770 | Column Angulation | Angle of the X-Ray beam in degree relative to an orthogonal axis to the detector plane. |
113771 | X-Ray Filters | Devices used to modify the energy or energy distribution of X-Rays. |
113772 | X-Ray Filter Type | Type of filter(s) inserted into the X-Ray beam. E.g., wedges. |
113773 | X-Ray Filter Thickness Maximum | The maximum thickness of the X-Ray absorbing material used in the filters. |
113780 | Reference Point Definition | System provided definition of the Reference Point used for Dose calculations. |
113788 | Collimated Field Height | Distance between the collimator blades in pixel column direction as projected at the detector plane. |
113789 | Collimated Field Width | Distance between the collimator blades in pixel row direction as projected at the detector plane. |
113790 | Collimated Field Area | Collimated field area at image receptor. Area for compatibility with IEC 60601-2-43. |
113791 | Pulse Rate | Pulse rate applied by equipment during Fluoroscopy. |
113792 | Distance Source to Table Plane | Measured or calculated distance from the X-Ray source to the table plane in the center of the beam. |
113793 | Pulse Width | (Average) X-Ray pulse width. |
113794 | Dose Measurement Device | Calibrated device to perform dose measurements. |
113795 | Acquired Image | Image acquired during a specified event. |
113800 | DLP to E conversion via MC computation | Effective Dose evaluation from the product of Dose Length Product (DLP) and the Effective Dose Conversion Factor (E/DLP in units of mSv/mGy-cm), where the ratio is derived by means of Monte Carlo computations. |
113801 | CTDIfreeair to E conversion via MC computation | Effective Dose evaluation from the product of the Mean CTDIfreeair and the ratio E/CTDIfreeair (mSv/mGy), where the ratio is derived by means of Monte Carlo computations. |
113802 | DLP to E conversion via measurement | Effective Dose evaluation from the product of Dose Length Product (DLP) and the Effective Dose Conversion Factor (E/DLP in units of mSv/mGy-cm), where the ratio is derived by means of dosimetric measurements with an anthropomorphic phantom. |
113803 | CTDIfreeair to E conversion via measurement | Effective Dose evaluation from the product of the Mean CTDIfreeair and the ratio E/CTDIfreeair (mSv/mGy), where the ratio is derived by means of dosimetric measurements with an anthropomorphic phantom. |
113804 | Sequenced Acquisition | The CT acquisition was performed by acquiring single or multi detector data while rotating the source about the gantry while the table is not moving. Additional slices are acquired by incrementing the table position and again rotating the source about the gantry while the table is not moving. |
113805 | Constant Angle Acquisition | The CT acquisition was performed by holding the source at a constant angle and moving the table to obtain a projection image. E.g., localizer. |
113806 | Stationary Acquisition | The CT acquisition was performed by holding the table at a constant position and acquiring multiple slices over time at the same location. |
113807 | Free Acquisition | The CT acquisition was performed while rotating the source about the gantry while the table movement is under direct control of a human operator or under the control of an analysis application. E.g., fluoro. |
113808 | ICRP Pub 60 | Reference authority 1990 Recommendations of the International Commission on Radiological Protection (ICRP Publication 60, published as the Annals of the ICRP Vol. 21, No. 1-3, Pergamon Press,1991). |
113809 | Start of X-Ray Irradiation | Start, DateTime of the first X-Ray Irradiation Event of the accumulation within a Study. |
113810 | End of X-Ray Irradiation | End, DateTime of the last X-Ray Irradiation Event of the accumulation within a Study. |
113811 | CT Accumulated Dose Data | X-Ray dose accumulated over multiple CT irradiation events. E.g., for a study or a performed procedure step. |
113812 | Total Number of Irradiation Events | Total number of events during the defined scope of accumulation. |
113813 | CT Dose Length Product Total | The total dose length product defined scope of accumulation. |
113814 | CT Effective Dose Total | The total Effective Dose at the defined scope of accumulation. |
113815 | Patient Model | Identification of the reference-patient model used when Effective Dose is evaluated via Monte Carlo calculations or from a Dose Length Product conversion factor based on Monte Carlo calculations. |
113816 | Condition Effective Dose measured | References the physical phantom and the type of dosimeter used when measurements are done to establish Effective Dose Conversion Factors (E/DLP) or ratios E/CTDIfreeair. |
113817 | Effective Dose Phantom Type | Type of Effective Dose phantom used. |
113818 | Dosimeter Type | Type of dosimeter used. |
113819 | CT Acquisition | General description of the CT Irradiation event. |
113820 | CT Acquisition Type | Method of the CT acquisition. |
113821 | X-Ray Filter Aluminum Equivalent | Thickness of an equivalent filter in mm in Aluminum. |
113822 | CT Acquisition Parameters | General description of the acquisition parameters. |
113823 | Number of X-Ray Sources | Number of X-Ray sources. |
113824 | Exposure Time | Total time the patient has received X-Ray exposure during the irradiation event. |
113825 | Scanning Length | Length of the table travel during the entire tube loading, according to IEC 60601-2-44 ### Note Scanning Length might be longer than the programmed acquisition length (Length of Reconstructable Volume) |
113826 | Nominal Single Collimation Width | The value of the nominal width referenced to the location of the isocenter along the z axis of a single row of acquired data in mm. |
113827 | Nominal Total Collimation Width | The value of the nominal width referenced to the location of the isocenter along the z axis of the total collimation in mm over the area of active X-Ray detection. |
113828 | Pitch Factor | For Spiral scanning: Pitch Factor = (Table Feed per Rotation (mm)) /(Nominal Total Collimation Width (mm)) For Sequenced scanning: Pitch Factor = (Table Feed per single Sequenced scan (mm)) /(Nominal Total Collimation Width (mm)). |
113829 | CT Dose | General description of CT dose values. |
113830 | Mean CTDIvol | "Mean CTDIvol" refers to the average value of the CTDIvol associated with this acquisition. |
113831 | CT X-Ray Source Parameters | Identification, tube-potential, tube-current, and exposure-time parameters associated with an X-Ray source during an acquisition. |
113832 | Identification of the X-Ray Source | Identifies the particular X-Ray source (in a multi-source CT system) for which the set of X-Ray source parameter values is reported. |
113833 | Maximum X-Ray Tube Current | Maximum X-Ray tube current. |
113834 | Exposure Time per Rotation | The exposure time for one rotation of the source around the object in s. |
113835 | CTDIw Phantom Type | A label describing the type of phantom used for CTDIW measurement according to IEC 60601-2-44 (Head 16 cm diameter PMMA, Body 32 cm diameter PMMA). |
113836 | CTDIfreeair Calculation Factor | The CTDIfreeair Calculation Factor is the CTDIfreeair per mAs, expressed in units of mGy/mAs. The CTDIfreeair Calculation Factor may be used in one method calculating Dose. |
113837 | Mean CTDIfreeair | The average value of the free-in-air CTDI associated with this acquisition. |
113838 | DLP | Dose Length Product (DLP), expressed in mGy-cm, is an index characterizing the product of the CTDIvol and the length scanned. For Spiral scanning, DLP = CTDIvol ✕ Scanning Length. For Sequenced scanning, DLP = CTDIvol ✕ Nominal Total Collimation Width ✕ Cumulative Exposure Time / Exposure Time per Rotation. For Stationary and Free scanning, DLP = CTDIvol ✕ Nominal Total Collimation Width. |
113839 | Effective Dose | Effective dose in mSv. |
113840 | Effective Dose Conversion Factor | Effective Dose per DLP, reference value for Effective Dose calculation, expressed in mSv/mGY.cm. |
113841 | ICRP Pub 103 | Effective Dose Reference authority 2007 Recommendations of the International Commission on Radiological Protection (ICRP Publication 103, published as the Annals of the ICRP Vol. 37, No. 2-4, Elsevier, 2007). |
113842 | X-Ray Modulation Type | The type of exposure modulation used for the purpose of limiting the dose. |
113845 | Exposure Index | Measure of the detector response to radiation in the relevant image region of an image acquired with a digital X-Ray imaging system as defined in IEC 62494-1; see PS3.3 definition of Exposure Index Macro. |
113846 | Target Exposure Index | The target value used to calculate the Deviation Index as defined in IEC 62494-1; see PS3.3 definition of Exposure Index Macro. |
113847 | Deviation Index | A scaled representation of the accuracy of the Exposure Index compared to the Target Exposure Index as defined in IEC 62494-1; see PS3.3 definition of Exposure Index Macro. |
113850 | Irradiation Authorizing | The clinician responsible for determining that the irradiating procedure was appropriate for the indications. |
113851 | Irradiation Administering | The person responsible for the administration of radiation. |
113852 | Irradiation Event | An irradiation event is the loading of X-Ray equipment caused by a single continuous actuation of the equipment's irradiation switch, from the start of the loading time of the first pulse until the loading time trailing edge of the final pulse. Any automatic on-off switching of the irradiation source during the event is not treated as separate events, rather the event includes the time between start and stop of irradiation as triggered by the user. E.g., a pulsed fluoro X-Ray acquisition shall be treated as a single irradiation event. |
113853 | Irradiation Event UID | Unique Identifier of an Irradiation Event. |
113854 | Source of Dose Information | Method by which dose-related details of an Irradiation Event were obtained. |
113855 | Total Acquisition Time | Total accumulated acquisition clock time in the scope of the including report (i.e., the sum of the Irradiation Duration values for accumulated acquisition irradiation events). |
113856 | Automated Data Collection | Direct recording of data by a relevant system. |
113857 | Manual Entry | Recording of data by a human operator, including manual transcription of electronic data. |
113858 | MPPS Content | The data is taken from an MPPS SOP Instance. |
113859 | Irradiating Device | A device exposing a patient to ionizing radiation. |
113860 | 15cm from Isocenter toward Source | 15cm from the isocenter towards the X-Ray source; See IEC 60601-2-43. |
113861 | 30cm in Front of Image Input Surface | 30cm in front (towards the tube) of the input surface of the image receptor; See FDA Federal Performance Standard for Diagnostic X-Ray Systems §1020.32(d) (7). |
113862 | 1cm above Tabletop | 1cm above the patient tabletop or cradle; See FDA Federal Performance Standard for Diagnostic X-Ray Systems §1020.32(d) (7). |
113863 | 30cm above Tabletop | 30cm above the patient tabletop of cradle; See FDA Federal Performance Standard for Diagnostic X-Ray Systems §1020.32(d) (7). |
113864 | 15cm from Table Centerline | 15cm from the centerline of the X-Ray table and in the direction of the X-Ray source; See FDA Federal Performance Standard for Diagnostic X-Ray Systems §1020.32(d) (7). |
113865 | Entrance exposure to a 4.2 cm breast thickness | Standard breast means a 4.2 centimeter (cm) thick compressed breast consisting of 50 percent glandular and 50 percent adipose tissue. See Department of Health and Human Services, Food and Drug Administration. Mammography quality standards; final rule. Federal Register. Oct. 28, 1997; 68(208):55852-55994; see 900.2(uu). |
113866 | Copied From Image Attributes | The data is copied from information present in the image attributes. E.g., dose attributes such as CTDIvol (0018,9345). |
113867 | Computed From Image Attributes | The data is computed from information present in the image attributes. E.g., by using dosimetry information for the specific irradiating device make and model, applied to technique information such as KVP and mAs. |
113868 | Derived From Human-Readable Reports | The data is derived from human-readable reports. E.g., by natural language parsing of text reports, or optical character recognition from reports saved as images by the irradiating device. |
113870 | Person Name | The name of a specific person. |
113871 | Person ID | An identification number or code for a specific person. |
113872 | Person ID Issuer | The organization that issued a Person ID. |
113873 | Organization Name | The name of an organization. |
113874 | Person Role in Organization | The role played by a person in an organization. |
113875 | Person Role in Procedure | The role played by a person in a procedure. |
113876 | Device Role in Procedure | The role played by a device in a procedure. |
113877 | Device Name | The name used to refer to a device; usually locally unique. |
113878 | Device Manufacturer | Manufacturer of a device. |
113879 | Device Model Name | Model Name of a device. |
113880 | Device Serial Number | Serial Number of a device. |
113890 | All Planes | All planes of a multi-plane acquisition equipment. |
113893 | Length of Reconstructable Volume | The length from which images may be reconstructed (i.e., excluding any overranging performed in a spiral acquisition that is required for data interpolation). Value is distinct from (1113825, DCM, "Scanning Length"), which is the actual length of the table travel during the entire tube loading, according to IEC 60601-2-44, and includes overranging. Also distinct from any actual Reconstructed Volume, which may depend on the slice thickness chosen for a particular reconstruction. |
113895 | Top Z Location of Reconstructable Volume | The Z location that is the top (highest Z value) of the Reconstructable Volume. Specified as the Z component within the Patient Coordinate System defined by a specified Frame of Reference. |
113896 | Bottom Z Location of Reconstructable Volume | The Z location that is the bottom (lowest Z value) of the Reconstructable Volume. Specified as the Z component within the Patient Coordinate System defined by a specified Frame of Reference. |
113897 | Top Z Location of Scanning Length | The Z location that is the top (highest Z value) of the scanning length. Specified as the Z component within the Patient Coordinate System defined by a specified Frame of Reference. |
113898 | Bottom Z Location of Scanning Length | The Z location that is the bottom (lowest Z value) of the scanning length. Specified as the Z component within the Patient Coordinate System defined by a specified Frame of Reference. |
113899 | Exposed Range | The range along the z axis of the total volume irradiated, per IEC 60601-2-44, Ed. 3, 203.115(b). The start and stop of loading corresponding to the outer edge of the full width half maximum of the free-in-air dose profile for the beam collimation used. |
113900 | Dose Check Alert Details | Report section about cumulative dose alerts during an examination. |
113901 | DLP Alert Value Configured | Flag denoting whether a DLP Alert Value was configured. |
113902 | CTDIvol Alert Value Configured | Flag denoting whether a CTDIvol Alert Value was configured. |
113903 | DLP Alert Value | Cumulative Dose Length Product value configured to trigger an alert; see NEMA XR 25-2010 Dose Check Standard. |
113904 | CTDIvol Alert Value | Cumulative CTDIvol value configured to trigger an alert; see NEMA XR 25-2010 Dose Check Standard. |
113905 | Accumulated DLP Forward Estimate | A forward estimate of the accumulated DLP plus the estimated DLP for the next Protocol Element Group; see NEMA XR 25-2010 Dose Check Standard. |
113906 | Accumulated CTDIvol Forward Estimate | A forward estimate at a given location of the accumulated CTDIvol plus the estimated CTDIvol for the next Protocol Element Group; see NEMA XR 25-2010 Dose Check Standard. |
113907 | Reason for Proceeding | Reason provided for proceeding with a procedure that is projected to exceed a configured dose value. |
113908 | Dose Check Notification Details | Report section about dose notifications during a protocol element. |
113909 | DLP Notification Value Configured | Flag denoting whether a DLP Notification Value was configured. |
113910 | CTDIvol Notification Value Configured | Flag denoting whether a CTDIvol Notification Value was configured. |
113911 | DLP Notification Value | Dose Length Product value configured to trigger a notification for a given protocol element. |
113912 | CTDIvol Notification Value | CTDIvol value configured to trigger a notification for a given protocol element. |
113913 | DLP Forward Estimate | A forward estimate of the DLP for the next Protocol Element Group; see NEMA XR 25-2010 Dose Check Standard. |
113914 | CTDIvol Forward Estimate | A forward estimate of the CTDIvol for the next Protocol Element Group; see NEMA XR 25-2010 Dose Check Standard. |
113921 | Radiation Exposure | The amount of ionizing radiation to which the patient was exposed. |
113922 | Radioactive Substance Administered | Type, amount and route of radioactive substance administered. |
113923 | Radiation Exposure and Protection Information | Exposure to ionizing radiation and associated preventive measures used to reduce the exposure of parts of the body like lead apron or eye, thyroid gland or gonad protection. | Retired. Replaced by (73569-6, LN, "Radiation Exposure and Protection Information")
113930 | Size Specific Dose Estimation | The Size-Specific Dose Estimate is a patient dose estimate that takes into account the size of the patient, such as described in AAPM Report 204 by using linear dimensions measured on the patient or patient images or estimated from patient age. |
113931 | Measured Lateral Dimension | The side-to-side (left to right) dimension of the body part being scanned (per AAPM Report 204). |
113932 | Measured AP Dimension | The thickness of the body part being scanned, in the antero-posterior dimension (per AAPM Report 204). |
113933 | Derived Effective Diameter | The diameter of the patient at a given location along the Z-axis of the patient, assuming that the patient has a circular cross-section (per AAPM Report 204). |
113934 | AAPM 204 Lateral Dimension | The Size Specific Dose Estimation is computed using Table 1B (32cm phantom) or Table 2B (16cm phantom) of AAPM Report 204. |
113935 | AAPM 204 AP Dimension | The Size Specific Dose Estimation is computed using Table 1C (32cm phantom) or Table 2C (16cm phantom) of AAPM Report 204. |
113936 | AAPM 204 Sum of Lateral and AP Dimension | The Size Specific Dose Estimation is computed using Table 1A (32cm phantom) or Table 2A (16cm phantom) of AAPM Report 204. |
113937 | AAPM 204 Effective Diameter Estimated From Patient Age | The Size Specific Dose Estimation is computed using Table 1D (32cm phantom) or Table 2D (16cm phantom) using an effective diameter estimated from the patient's age using Table 3 of AAPM Report 204. |
113940 | System Calculated | Values calculated from other existing parameters. |
113941 | In Detector Plane | A segmented region of the detector surface within the irradiated area (but might not be near the center of the detector). |
113942 | X-Ray Reading Device | A device that creates digital images from X-Ray detectors (Direct, Indirect or Storage). |
113943 | X-Ray Source Data Available | Parameters related to the X-Ray source (generator, tube, etc) are available to the recording application. |
113944 | X-Ray Mechanical Data Available | Parameters related to the X-Ray Mechanical System (Stand, Table) are available to the recording application. |
113945 | X-Ray Detector Data Available | Parameters related to the X-Ray Detector are available to the recording application. |
113946 | Projection Eponymous Name | Describes the radiographic method of patient, tube and detector positioning to achieve a well described projection or view. |
113947 | Detector Type | Type of Detector used to acquire data. E.g., Images. |
113948 | Direct Detector | Detector that directly transforms the input signal to pixel values. |
113949 | Indirect Detector | Detector that transforms an intermediate signal into pixel values. E.g., a scintillator-based detector. |
113950 | Storage Detector | Storage detector that stores a signal that is later transformed by a reader into pixel values. E.g., a phosphor-based detector. |
113951 | Film | Film that is scanned to create pixel values. |
113952 | Table Mount | The cassette/detector is mounted in the patient table. |
113953 | Unmounted Detector | The cassette/detector is not mounted.. E.g., a cassette placed underneath the patient. |
113954 | Upright Stand Mount | The cassette/detector is mounted in an upright stand. |
113955 | C-Arm Mount | The cassette/detector is mounted on a c-arm. |
113956 | CR/DR Mechanical Configuration | Method of mounting or positioning a CR/DR cassette or detector. |
113957 | Fluoroscopy-Guided Projection Radiography System | An integrated projection radiography system capable of fluoroscopy. |
113958 | Integrated Projection Radiography System | A projection radiography system where the X-Ray detector, X-Ray Source and gantry components are integrated and the managing system is able to access details of each component. |
113959 | Cassette-based Projection Radiography System | A projection radiography system where the X-Ray detector, X-Ray Source and gantry components are not integrated. E.g., cassette-based CR and DR systems. |
113961 | Reconstruction Algorithm | Description of the algorithm used when reconstructing the image from the data acquired during the acquisition process. |
113962 | Filtered Back Projection | An algorithm for reconstructing an image from multiple projections by back-projecting the measured values along the line of the projection and filtering the result to reduce blurring. |
113963 | Iterative Reconstruction | An algorithm for reconstructing an image from multiple projections by starting with an assumed reconstructed image, computing projections from the image, comparing the original projection data and updating the reconstructed image based upon the difference between the calculated and the actual projections. |
113970 | Procedure Step To This Point | The period of time from the start of a Procedure Step until the time point established by the context of the reference. |
114000 | Not a number | Measurement not available: Not a number (per IEEE 754). |
114001 | Negative Infinity | Measurement not available: Negative Infinity (per IEEE 754). |
114002 | Positive Infinity | Measurement not available: Positive Infinity (per IEEE 754). |
114003 | Divide by zero | Measurement not available: Divide by zero (per IEEE 754). |
114004 | Underflow | Measurement not available: Underflow (per IEEE 754). |
114005 | Overflow | Measurement not available: Overflow (per IEEE 754). |
114006 | Measurement failure | Measurement not available: Measurement failure. |
114007 | Measurement not attempted | Measurement not available: Measurement not attempted. |
114008 | Calculation failure | Measurement not available: Calculation failure. |
114009 | Value out of range | Measurement not available: Value out of range. |
114010 | Value unknown | Measurement not available: Value unknown. |
114011 | Value indeterminate | Measurement not available: Value indeterminate. |
114201 | Time of flight | Measures the time-of-flight of a light signal between the camera and the subject for each point of the image. |
114202 | Interferometry | Interferometry is a family of techniques in which waves are superimposed in order to extract depth information about the scanned object. |
114203 | Laser scanning | Laser scanning describes the general method to sample or scan a surface using laser technology. |
114204 | Pattern projection | Projecting a narrow band of light onto a three-dimensionally shaped surface produces a line of illumination that appears distorted from other perspectives than that of the projector. It can be used for an exact geometric reconstruction of the surface shape. |
114205 | Shape from shading | A technique for estimating the surface normal of an object by observing that object under different lighting conditions. |
114206 | Shape from motion | A technique for estimating the surface normal of an object by observing that object under different motions. |
114207 | Confocal imaging | An optical imaging technique used to increase optical resolution and contrast of a micrograph by using point illumination and a spatial pinhole to eliminate out-of-focus light in specimens that are thicker than the focal plane. It enables the reconstruction of 3D structures from the obtained images. |
114208 | Point Cloud Algorithmic | Point cloud that was calculated by an algorithm. |
114209 | Turntable Scan Method | Scanning the object from different views by placing it on a rotating table. |
114210 | High resolution | Higher resolution with a longer acquisition time. |
114211 | Fast mode | Lower resolution with a shorter acquisition time. |
114213 | Iterative Closest Point | An algorithm employed to minimize the difference between two clouds of points. It iteratively revises the transformation (translation, rotation) needed to minimize the distance between the points of two point clouds. |
114215 | Freehand | Human controlled minimization of the distance between the points of two point clouds. |
114216 | Checkerboard | Scanning the object from different views by placing it in front of a checkerboard pattern. |
121001 | Quotation Mode | Type of source for observations quoted from an external source. |
121002 | Quoted Source | Reference to external source of quoted observations. |
121003 | Document | Documentary source of quoted observations. |
121004 | Verbal | Verbal source of quoted observations. |
121005 | Observer Type | Type of observer that created the observations. |
121006 | Person | Human observer created the observations. |
121007 | Device | Automated device created the observations. |
121008 | Person Observer Name | Name of human observer that created the observations. |
121009 | Person Observer's Organization Name | Organization or institution with which the human observer is affiliated for the context of the current observation. |
121010 | Person Observer's Role in the Organization | Organizational role of human observer for the context of the current observation. |
121011 | Person Observer's Role in this Procedure | Procedural role of human observer for the context of the current observation. |
121012 | Device Observer UID | Unique identifier of automated device that created the observations. |
121013 | Device Observer Name | Institution-provided identifier of automated device that created the observations. |
121014 | Device Observer Manufacturer | Manufacturer of automated device that created the observations. |
121015 | Device Observer Model Name | Manufacturer-provided model name of automated device that created the observations. |
121016 | Device Observer Serial Number | Manufacturer-provided serial number of automated device that created the observations. |
121017 | Device Observer Physical Location During Observation | Location of automated device that created the observations whilst the observations were being made. |
121018 | Procedure Study Instance UID | Unique identifier for the Study or Requested Procedure. |
121019 | Procedure Study Component UID | Unique identifier for the Performed Procedure Step. |
121020 | Placer Number | Identifier for the Order (or Service Request) assigned by the order placer system. |
121021 | Filler Number | Identifier for the Order (or Service Request) assigned by the order filler system. |
121022 | Accession Number | Identifier for the Order (or Service Request) assigned by the department information system. |
121023 | Procedure Code | Type of procedure scheduled or performed. |
121024 | Subject Class | Type of observation subject. |
121025 | Patient | A patient is the subject of observations. |
121026 | Fetus | Fetus of patient is the subject of observations. |
121027 | Specimen | Specimen is the subject of observations. |
121028 | Subject UID | Unique Identifier of patient or fetus who is the subject of observations. |
121029 | Subject Name | Name of patient who is the subject of observations. |
121030 | Subject ID | Identifier of patient or fetus who is the subject of observations. |
121031 | Subject Birth Date | Birth Date of patient who is the subject of observations. |
121032 | Subject Sex | Sex of patient who is the subject of observations. |
121033 | Subject Age | Age of patient who is the subject of observations. |
121034 | Subject Species | Species of patient who is the subject of observations. |
121035 | Subject Breed | The breed of the subject. |
121036 | Mother of fetus | Name of mother of fetus that is the subject of observations. |
121037 | Fetus number |  | Retired. Replaced by (11951-1, LN, "Fetus ID").
121038 | Number of Fetuses |  | Retired. Replaced by (55281-0, LN, "Number of Fetuses").
121039 | Specimen UID | Unique Identifier of specimen that is the subject of observations. |
121040 | Specimen Accession Number | Accession Number of specimen that is the subject of observations | Retired.
121041 | Specimen Identifier | Identifier of specimen that is the subject of observations. |
121042 | Specimen Type | Coded category of specimen that is the subject of observations | Retired. Replaced by (R-00254, SRT, "Specimen Type")
121043 | Slide Identifier | Identifier of specimen microscope slide that is the subject of observations | Retired. Replaced by (111700, DCM, "Specimen Container Identifier")
121044 | Slide UID | Unique Identifier of specimen microscope slide that is the subject of observations | Retired.
121045 | Language | The language of the content, being a language that is primarily used for human communication. E.g., English, French. |
121046 | Country of Language | The country-specific variant of language. E.g., Canada for Candadian French. |
121047 | Language of Value | The language of the value component of a name-value pair. |
121048 | Language of Name and Value | The language of both the name component and the value component of a name-value pair. |
121049 | Language of Content Item and Descendants | The language of the current content item (node in a tree of content) and all its descendants. |
121050 | Equivalent Meaning of Concept Name | The human readable meaning of the name component of a name-value pair that is equivalent to the post-coordinated meaning conveyed by the coded name and its concept modifier children. |
121051 | Equivalent Meaning of Value | The human readable meaning of the value component of a name-value pair that is equivalent to the post-coordinated meaning conveyed by the coded value and its concept modifier children. |
121052 | Presence of property | Whether or not the property concept being modified is present or absent. |
121053 | Present |  | Retired. Replaced by (G-A203, SRT, "Present")
121054 | Absent |  | Retired. Replaced by (R-4089B, SRT, "Absent")
121055 | Path | A set of points on an image, that when connected by line segments, provide a polyline from which a linear measurement was inferred. |
121056 | Area outline | A set of points on an image, that when connected by line segments, provide a closed polyline that is the border of a defined region from which an area, or two-dimensional measurement, was inferred. |
121057 | Perimeter outline | A set of points on an image, that when connected by line segments, provide a closed polyline that is a two-dimensional border of a three-dimensional region's intersection with, or projection into the image. |
121058 | Procedure reported | The imaging procedure whose results are reported. |
121059 | Presence Undetermined | Presence or absence of a property is undetermined | Retired. Replaced by (R-4089B, SRT, "Undetermined")
121060 | History |  | Retired. Replaced by (11329-0, LN, "History")
121062 | Request |  | Retired. Replaced by (55115-0, LN, "Request")
121064 | Current Procedure Descriptions |  | Retired. Replaced by (55111-9, LN, "Current Procedure Descriptions")
121065 | Procedure Description | A description of the imaging procedure. |
121066 | Prior Procedure Descriptions |  | Retired. Replaced by (55114-3, LN, "Prior Procedure Descriptions")
121068 | Previous Findings |  | Retired. Replaced by (18834-2, LN, "Previous Findings")
121069 | Previous Finding | An observation found on a prior imaging study. |
121070 | Findings |  | Retired. Replaced by (18782-3, LN, "Findings")
121071 | Finding | An observation found on an imaging study. |
121072 | Impressions |  | Retired. Replaced by (19005-8, LN, "Impressions")
121073 | Impression | An interpretation in the clinical context of the finding(s) on an imaging study. |
121074 | Recommendations |  | Retired. Replaced by (18783-1, LN, "Recommendations")
121075 | Recommendation | A recommendation for management or investigation based on the findings and impressions of an imaging study. |
121076 | Conclusions |  | Retired. Replaced by (55110-1, LN, "Conclusions")
121077 | Conclusion | An interpretation in the clinical context of the finding(s) on an imaging study. |
121078 | Addendum |  | Retired. Replaced by (55107-7, LN, "Addendum")
121079 | Baseline | Initial images used to esyablish a beginning condition that is used for comparison over time to look for changes. [Paraphrases NCI-PT (C1442488, UMLS, "Baseline"), which is defined as "An initial measurement that is taken at an early time point to represent a beginning condition, and is used for comparison over time to look for changes. For example, the size of a tumor will be measured before treatment (baseline) and then afterwards to see if the treatment had an effect. A starting point to which things may be compared."] |
121080 | Best illustration of finding | A selection of composite instances that best illustrates a particular finding. E.g., an image slice at the location of the largest extent of a tumor. |
121081 | Physician |  | Retired. Replaced by (J-004E8, SRT, "Physician")
121082 | Nurse |  | Retired. Replaced by (J-07100, SRT, "Nurse")
121083 | Technologist |  | Retired. Replaced by (J-00187, SRT, "Radiologic Technologist")
121084 | Radiographer |  | Retired. Replaced by (J-00187, SRT, "Radiographer")
121085 | Intern |  | Retired. Replaced by (C1144859, UMLS, "Intern")
121086 | Resident |  | Retired. Replaced by (J-005E6, SRT, "Resident")
121087 | Registrar |  | Retired. Replaced by (J-00172, SRT, "Registrar")
121088 | Fellow | A medical practitioner undergoing sub-specialty training. E.g., during the period after specialty training (residency). |
121089 | Attending [Consultant] |  | Retired. Replaced by (J-005E8, SRT, "Attending")
121090 | Scrub nurse |  | Retired. Replaced by (J-0714A, SRT, "Scrub nurse")
121091 | Surgeon |  | Retired. Replaced by (J-00556, SRT, "Surgeon")
121092 | Sonologist | A medical practitioner with sub-specialty training in Ultrasound. |
121093 | Sonographer |  | Retired. Replaced by (C1954848, UMLS, "Sonographer")
121094 | Performing | The person responsible for performing the procedure. |
121095 | Referring | The person responsible for referring the patient for the procedure. | Retired. Replaced by (C1709880, UMLS, "Referring physician").
121096 | Requesting | The person responsible for requesting the procedure. |
121097 | Recording | The person responsible for recording the procedure or observation. |
121098 | Verifying | The person responsible for verifying the recorded procedure or observation. |
121099 | Assisting | The person responsible for assisting with the procedure. |
121100 | Circulating | The person responsible for making preparations for and monitoring the procedure. | Retired. Replaced by (J-0714B, SRT, "Circulating Nurse").
121101 | Standby | The person responsible for standing by to assist with the precedure if required. |
121102 | Other sex | Other sex. |
121103 | Undetermined sex | Sex of subject undetermined at time of encoding. |
121104 | Ambiguous sex | Ambiguous sex. |
121105 | Radiation Physicist | Radiation Physicist. | Retired. Replaced by (C2985483, UMLS, "Radiation Physicist").
121106 | Comment | Comment. |
121109 | Indications for Procedure | Indications for Procedure | Retired. Replaced by (18785-6, LN, "Indications for Procedure")
121110 | Patient Presentation | Patient condition at the beginning of a healthcare encounter | Retired. Replaced by (55108-5, LN, "Patient Presentation")
121111 | Summary | Summary of a procedure, including most significant findings | Retired. Replaced by (55112-7, LN, "Summary")
121112 | Source of Measurement | Image or waveform used as source for measurement. |
121113 | Complications | Complications from a procedure | Retired. Replaced by (55109-3, LN, "Complications")
121114 | Performing Physician | Physician who performed a procedure. |
121115 | Discharge Summary | Summary of patient condition upon Discharge from a healthcare facility. |
121116 | Proximal Finding Site | Proximal Anatomic Location for a differential measurement; may be considered subtype of term (G-C0E3, SRT, "Finding Site"). E.g., distance or pressure gradient. |
121117 | Distal Finding Site | Distal Anatomic Location for a differential measurement; may be considered subtype of term (G-C0E3, SRT, "Finding Site"). E.g., distance or pressure gradient. |
121118 | Patient Characteristics | Patient Characteristics (findings). |
121120 | Cath Lab Procedure Log | Time-stamped record of events that occur during a catheterization procedure. |
121121 | Room identification | Room identification. |
121122 | Equipment Identification | Equipment identification. |
121123 | Patient Status or Event | A recorded Patient Status or an event involving a patient. |
121124 | Procedure Action Item ID | Identification of a step, action, or phase of a procedure. |
121125 | DateTime of Recording of Log Entry | DateTime of Recording of an Entry in an Event Log. |
121126 | Performed Procedure Step SOP Instance UID | SOP Instance UID of a Performed Procedure Step. |
121127 | Performed Procedure Step SOP Class UID | SOP Class UID of a Performed Procedure Step. |
121128 | Procedure Action Duration | Duration of a step, action, or phase of a procedure. |
121130 | Start Procedure Action Item | Beginning of a step, action, or phase of a procedure. |
121131 | End Procedure Action Item | End of a step, action, or phase of a procedure. |
121132 | Suspend Procedure Action Item | Suspension of a step, action, or phase of a procedure. |
121133 | Resume Procedure Action Item | Resumption of a step, action, or phase of a procedure. |
121135 | Observation DateTime Qualifier | Concept modifier for the DateTime of Recording of an Entry in an Event Log. |
121136 | DateTime Unsynchronized | Recorded DateTime had its source in a system clock not synchronized to other recorded DateTimes. |
121137 | DateTime Estimated | Recorded DateTime is estimated. |
121138 | Image Acquired | Event of the acquisition of an image. |
121139 | Modality | Type of data acquisition device. |
121140 | Number of Frames | Number of Frames in a multi-frame image. |
121141 | Image Type | Descriptor of an Image. |
121142 | Acquisition Duration | Duration of the acquisition of an image or a waveform. |
121143 | Waveform Acquired | Event of the acquisition of an image. |
121144 | Document Title | Document Title. |
121145 | Description of Material | Description of Material used in a procedure. |
121146 | Quantity of Material | Quantity of Material used in a procedure. |
121147 | Billing Code | Billing Code for materials used in a procedure. |
121148 | Unit Serial Identifier | Unit or Device Serial Identifier. |
121149 | Lot Identifier | Lot Identifier. |
121150 | Device Code | Vendor or local coded value identifying a device. |
121151 | Lesion Identifier | Identification of a Lesion observed during an imaging procedure. |
121152 | Person administering drug/contrast | Person administering drug/contrast. |
121153 | Lesion Risk | Assessment of the risk a coronary lesion presents to the health of a patient. |
121154 | Intervention attempt identifier | Identifier for an attempted Intervention. |
121155 | Deployment | Use of a device to deploy another device. |
121156 | Percutaneous Entry Action | Action of a clinical professional at the site of percutaneous access to a patient's cardiovascular system. |
121157 | Begin Circulatory Support | The action or event of beginning circulatory support for a patient. |
121158 | End Circulatory Support | The action or event of ending circulatory support for a patient. |
121160 | Oxygen Administration Rate | Rate of Oxygen Administration. |
121161 | Begin Oxygen Administration | The action or event of beginning administration of oxygen to a patient. |
121162 | End oxygen administration | The action or event of ending administration of oxygen to a patient. |
121163 | By ventilator | Method of administration of oxygen to a patient by ventilator. |
121165 | Patient Assessment Performed | The action or event of assessing the clinical status of a patient. |
121166 | Begin Pacing | The action or event of beginning pacing support for a patient. |
121167 | End Pacing | The action or event of ending pacing support for a patient. |
121168 | Begin Ventilation | The action or event of beginning ventilation support for a patient. |
121169 | End Ventilation | The action or event of ending ventilation support for a patient. |
121171 | Tech Note | Procedural note originated by a technologist. |
121172 | Nursing Note | Procedural note originated by a nurse. |
121173 | Physician Note | Procedural note originated by a Physician. |
121174 | Procedure Note | General procedural note. |
121180 | Key Images | List of references to images considered significant | Retired. Replaced by (55113-5, LN, "Key Images")
121181 | DICOM Object Catalog | List of references to DICOM SOP Instances. |
121190 | Referenced Frames | Individual frames selected as a subset of a multi-frame image. |
121191 | Referenced Segment | Segment selected as a subset of a segmentation image, specifically the pixels/voxels identified as belonging to the classification of the identified segment. |
121192 | Device Subject | A device is the subject of observations. |
121193 | Device Subject Name | Name or other identifier of a device that is the subject of observations. |
121194 | Device Subject Manufacturer | Manufacturer of a device that is the subject of observations. |
121195 | Device Subject Model Name | Model Name of a device that is the subject of observations. |
121196 | Device Subject Serial Number | Serial Number of a device that is the subject of observations. |
121197 | Device Subject Physical Location during observation | Physical Location of a device that is the subject of observations during those observations. |
121198 | Device Subject UID | Unique Identifier of a device that is the subject of observations. |
121200 | Illustration of ROI | Illustration of a region of interest. |
121201 | Area Outline |  | Retired. Replaced by (121056, DCM, "Area Outline").
121202 | Area of Defined Region |  | Retired. Replaced by (G-A16A, SRT, "Area of defined region").
121206 | Distance | A one dimensional, or linear, numeric measurement. |
121207 | Height | Vertical measurement value. |
121208 | Inter-Marker Distance | Distance between marks on a device of calibrated size. E.g., a ruler. |
121210 | Path |  | Retired. Replaced by (121055, DCM, "Path").
121211 | Path length | A one dimensional, or linear, numeric measurement along a polyline. |
121213 | Perimeter Outline |  | Retired. Replaced by (121057, DCM, "Perimeter Outline").
121214 | Referenced Segmentation Frame | Frame selected from a segmentation image, specifically the pixels/voxels identified as belonging to the classification of the segment encompassing the identified frame. |
121216 | Volume estimated from single 2D region | A three-dimensional numeric measurement that is approximate, based on a two-dimensional region in a single image. |
121217 | Volume estimated from three or more non-coplanar 2D regions | A three-dimensional numeric measurement that is approximate, based on three or more non-coplanar two-dimensional image regions. |
121218 | Volume estimated from two non-coplanar 2D regions | A three-dimensional numeric measurement that is approximate, based on two non-coplanar two-dimensional image regions. |
121219 | Volume of bounding three dimensional region | A three-dimensional numeric measurement of the bounding region of a three-dimensional region of interest in an image set. |
121220 | Volume of circumscribed sphere | A three-dimensional numeric measurement of the bounding sphere of a three-dimensional region of interest in an image set. |
121221 | Volume of ellipsoid | A three-dimensional numeric measurement of an ellipsoid shaped three-dimensional region of interest in an image set. |
121222 | Volume of sphere | A three-dimensional numeric measurement of a sphere shaped three-dimensional region of interest in an image set. |
121230 | Path Vertex | Coordinates of a point on a defined path. |
121231 | Volume Surface | Surface of an identified or measured volume. |
121232 | Source series for segmentation | Series of image instances used as source data for a segmentation. |
121233 | Source image for segmentation | Image instances used as source data for a segmentation. |
121242 | Distance from nipple | Indicates the location of the area of interest as measured from the nipple of the breast. |
121243 | Distance from skin | Indicates the location of the area of interest as measured from the most direct skin point of the breast. |
121244 | Distance from chest wall | Indicates the location of the area of interest as measured from the chest wall. |
121301 | Simultaneous Doppler | Reference is to a Doppler waveform acquired simultaneously with an image. |
121302 | Simultaneous Hemodynamic | Reference is to a Hemodynamic waveform acquired simultaneously with an image. |
121303 | Simultaneous ECG | Reference is to a ECG waveform acquired simultaneously with an image. |
121304 | Simultaneous Voice Narrative | Reference is to a voice narrative recording acquired simultaneously with an image. |
121305 | Simultaneous Respiratory Waveform | A waveform representing chest expansion and contraction due to respiratory activity, measured simultaneously with the acquisition of this Image. |
121306 | Simultaneous Arterial Pulse Waveform | Arterial pulse waveform obtained simultaneously with acquisition of a referencing image. |
121307 | Simultaneous Phonocardiographic Waveform | Phonocardiographic waveform obtained simultaneously with acquisition of a referencing image. |
121311 | Localizer | Image providing an anatomical reference on the patient under examination, for the purpose of defining the location of the ensuing imaging. |
121312 | Biopsy localizer | Image providing an anatomical reference on the patient under examination, for the purpose of planning or documenting a biopsy. |
121313 | Other partial views | Image providing a partial view of the target anatomy, when the target anatomy is too large for a single image. |
121314 | Other image of biplane pair | Image providing a view of the target anatomy in a different imaging plane, typically from a near perpendicular angle. |
121315 | Other image of stereoscopic pair | Image providing a view of the target anatomy in a different imaging plane, typically with a small angular difference. |
121316 | Images related to standalone object | Image related to a non-image information object. |
121317 | Spectroscopy | Image where signals are identified and separated according to their frequencies. E.g., to identify individual chemicals, or individual nuclei in a chemical compound. |
121318 | Spectroscopy Data for Water Phase Correction | MR spectroscopy data acquired to correct the phase of the diagnostic data for the phase signal of the Water. |
121320 | Uncompressed predecessor | An image that has not already been lossy compressed that is used as the source for creation of a lossy compressed image. |
121321 | Mask image for image processing operation | Image used as the mask for an image processing operation, such as subtraction. |
121322 | Source image for image processing operation | Image used as the source for an image processing operation. |
121323 | Source series for image processing operation | Series used as the source for an image processing operation. |
121324 | Source Image | Image used as the source for a derived or compressed image. |
121325 | Lossy compressed image | Image encoded with a lossy compression transfer syntax. |
121326 | Alternate SOP Class instance | SOP Instance encoded with a different SOP Class but otherwise equivalent data. |
121327 | Full fidelity image | Full fidelity image, uncompressed or lossless compressed. |
121328 | Alternate Photometric Interpretation image | Image encoded with a different photometric interpretation. |
121329 | Source image for montage | Image used as a source for a montage (stitched) image. |
121330 | Lossy compressed predecessor | An image that has previously been lossy compressed that is used as the source for creation of another lossy compressed image. |
121331 | Equivalent CDA Document | HL7 Document Architecture (CDA) Document that contains clinical content equivalent to the referencing Instance. |
121332 | Complete Rendering for Presentation | Instance that contains a displayable complete rendering of the referencing Instance. |
121333 | Partial Rendering for Presentation | Instance that contains a displayable partial rendering of the referencing Instance. |
121334 | Extended Rendering for Presentation | Instance that contains a displayable complete rendering of the referencing Instance, plus additional content such as inline rendering of referenced images. |
121335 | Source Document | Document whose content has been wholly or partially transformed to create the referencing document. |
121338 | Anatomic image | Image showing structural anatomic features. |
121339 | Functional image | Image showing physical or chemical activity. |
121340 | Spectral filtered image | Image providing the same view of the target anatomy acquired using only a specific imaging wavelength, frequency or energy. |
121341 | Device localizer | Image providing an anatomical reference on the patient under examination, for the purpose of documenting the location of device such as a diagnostic or therapeutic catheter. |
121342 | Dose Image | Image providing a graphic view of the distribution of radiation dose. |
121346 | Acquisition frames corresponding to volume | The referenced image is the source of spatially-related frames from which the referencing 3D volume data set was derived. |
121347 | Volume corresponding to spatially-related acquisition frames | 3D Volume containing the spatially-related frames in the referencing instance. |
121348 | Temporal Predecessor | Instance acquired prior to the referencing instance in a set of consecutively acquired instances. |
121349 | Temporal Successor | Instance acquired subsequent to the referencing instance in a set of consecutively acquired instances. |
121350 | Same acquisition at lower resolution | Image of the same target area at lower resolution acquired in the same acquisition process. |
121351 | Same acquisition at higher resolution | Image of the same target area at higher resolution acquired in the same acquisition process. |
121352 | Same acquisition at different focal depth | Image of the same target area at different focal depth (Z-plane) acquired in the same acquisition process. |
121353 | Same acquisition at different spectral band | Image of the same target area at different spectral band acquired in the same acquisition process. |
121354 | Imaged container label | Image specifically targeting the container label. |
121358 | For Processing predecessor | Source image from which FOR PRESENTATION images were created. |
121360 | Replaced report | The reference is to a predecessor report that has been replaced by the current report. |
121361 | Addended report | The reference is to a predecessor report to which the current report provides and addendum. |
121362 | Preliminary report | A report that precedes the final report and may contain only limited information; it may be time sensitive, and it is not expected to contain all the reportable findings. |
121363 | Partial report | A report that is not complete. |
121370 | Composed from prior doses | The dose object created was calculated by summation of existing, previously calculated, RT Dose instances. |
121371 | Composed from prior doses and current plan | The dose object created was calculated by summation of existing, previously calculated, RT Dose instances and dose newly calculated by the application. The newly calculated dose may or may not exist as an independent object. |
121372 | Source dose for composing current dose | RT Dose Instances used as source for calculated dose. |
121380 | Active Ingredient Undiluted Concentration | Concentration of the chemically or physically interesting (active) ingredient of a drug or contrast agent as delivered in product form from the manufacturer, typically in mg/ml. |
121381 | Contrast/Bolus Ingredient Opaque | X-Ray absorption of the active ingredient of a contrast agent ingredient is greater than the absorption of water (tissue). |
121382 | Quantity administered | Number of units of substance administered to a patient. E.g., tablets. |
121383 | Mass administered | Mass of substance administered to a patient. |
121401 | Derivation | Method of deriving or calculating a measured value. E.g., mean, or maximum of set. |
121402 | Normality | Assessment of a measurement relative to a normal range of values; may be considered subtype of term (G-C0F2, SRT, "has interpretation"). |
121403 | Level of Significance | Significance of a measurement. |
121404 | Selection Status | Status of selection of a measurement for further processing or use. |
121405 | Population description | Description of a population of measurements. |
121406 | Reference Authority | Bibliographic or clinical reference for a Description of a population of measurements. |
121407 | Normal Range description | Description of a normal range of values for a measurement concept. |
121408 | Normal Range Authority | Bibliographic or clinical reference for a Description of a normal range of values. |
121410 | User chosen value | Observation value selected by user for further processing or use, or as most representative. |
121411 | Most recent value chosen | Observation value is the recently obtained, and has been selected for further processing or use. |
121412 | Mean value chosen | Observation value is the mean of several measurements, and has been selected for further processing or use. |
121414 | Standard deviation of population | Standard deviation of a measurement in a reference population. |
121415 | Percentile Ranking of measurement | Percentile Ranking of an observation value with respect a reference population. |
121416 | Z-Score of measurement | Z-score of an observation value with respect a reference population, expressed as the dimensionless quantity (x-m) /s, where (x-m) is the deviation of the observation value (x) from the population mean (m), and s is the standard deviation of the population. |
121417 | 2 Sigma deviation of population | 2 Sigma deviation of a measurement in a reference population. |
121420 | Equation | Formula used to compute a derived measurement. |
121421 | Equation Citation | Bibliographic reference to a formula used to compute a derived measurement; reference may be to a specific equation in a journal article. |
121422 | Table of Values Citation | Bibliographic reference to a Table of Values used to look up a derived measurement. |
121423 | Method Citation | Bibliographic reference to a method used to compute a derived measurement. |
121424 | Table of Values | A Table of Values used to look up a derived measurement. |
121425 | Index | Factor (divisor or multiplicand) for normalizing a measurement. E.g., body surface area used for normalizing hemodynamic measurements. |
121427 | Estimated | Measurement obtained by observer estimation, rather than with a measurement tool or by calculation | Retired. Replaced by (R-10260, SRT, "Estimated")
121428 | Calculated | Measurement obtained by calculation | Retired. Replaced by (R-41D2D, SRT, "Calculated")
121430 | Concern | Identified issue about a state or process that has the potential to require intervention or management. |
121431 | DateTime Concern Noted | DateTime concern noted (noted by whom is determined by context of use). |
121432 | DateTime Concern Resolved | DateTime the concern was resolved. |
121433 | DateTime Problem Resolved | DateTime the problem was resolved. |
121434 | Service Delivery Location | Place at which healthcare services may be provided. |
121435 | Service Performer | Identification of a healthcare provider who performed a healthcare service; may be either a person or an organization. |
121436 | Medical Device Used | Type or identifier of a medical device used on, in, or by a patient. |
121437 | Pharmacologic and exercise stress test | Cardiac stress test using pharmacologic and exercise stressors | Retired. Replaced by (P2-31011, SRT, "Pharmacologic and exercise stress test")
121438 | Paced stress test | Cardiac stress test using an implanted or external cardiac pacing device | Retired. Replaced by (P2-3110B, SRT, "Stress test using cardiac pacing")
121439 | Correction of congenital cardiovascular deformity | Procedure for correction of congenital cardiovascular deformity | Retired. Replaced by (P1-080B4, SRT, "Correction of congenital cardiovascular deformity")
121701 | RT Patient Setup | Process of placing patient in the anticipated treatment position, including specification and location of positioning aids, and other treatment delivery accessories. |
121702 | RT Patient Position Acquisition, single plane MV | Acquisition of patient positioning information prior to treatment delivery, using single-plane megavoltage imaging. |
121703 | RT Patient Position Acquisition, dual plane MV | Acquisition of patient positioning information prior to treatment delivery, using dual-plane megavoltage imaging. |
121704 | RT Patient Position Acquisition, single plane kV | Acquisition of patient positioning information prior to treatment delivery, using single-plane kilovoltage imaging. |
121705 | RT Patient Position Acquisition, dual plane kV | Acquisition of patient positioning information prior to treatment delivery, using dual-plane kilovoltage imaging. |
121706 | RT Patient Position Acquisition, dual plane kV/MV | Acquisition of patient positioning information prior to treatment delivery, using dual-plane combination kilovoltage and megavoltage imaging. |
121707 | RT Patient Position Acquisition, CT kV | Acquisition of patient positioning information prior to treatment delivery, using kilovoltage CT imaging. |
121708 | RT Patient Position Acquisition, CT MV | Acquisition of patient positioning information prior to treatment delivery, using megavoltage CT imaging. |
121709 | RT Patient Position Acquisition, Optical | Acquisition of patient positioning information prior to treatment delivery, using optical imaging. |
121710 | RT Patient Position Acquisition, Ultrasound | Acquisition of patient positioning information prior to treatment delivery, using ultrasound imaging. |
121711 | RT Patient Position Acquisition, Spatial Fiducials | Acquisition of patient positioning information prior to treatment delivery, using spatial fiducials. |
121712 | RT Patient Position Registration, single plane | Registration of intended and actual patient position prior to treatment delivery, using single-plane images. |
121713 | RT Patient Position Registration, dual plane | Registration of intended and actual patient position prior to treatment delivery, using dual-plane images. |
121714 | RT Patient Position Registration, 3D CT general | Registration of intended and actual patient position prior to treatment delivery, using 3D CT images and an unspecified registration approach. |
121715 | RT Patient Position Registration, 3D CT marker-based | Registration of intended and actual patient position prior to treatment delivery, using 3D CT images and a marker-based registration approach. |
121716 | RT Patient Position Registration, 3D CT volume-based | Registration of intended and actual patient position prior to treatment delivery, using 3D CT images and a volume-based registration approach. |
121717 | RT Patient Position Registration, 3D on 2D reference | Registration of intended and actual patient position prior to treatment delivery, using 3D verification images and 2D reference images. |
121718 | RT Patient Position Registration, 2D on 3D reference | Registration of intended and actual patient position prior to treatment delivery, using 2D verification images and 3D reference images. |
121719 | RT Patient Position Registration, Optical | Registration of intended and actual patient position prior to treatment delivery, using optical images. |
121720 | RT Patient Position Registration, Ultrasound | Registration of intended and actual patient position prior to treatment delivery, using ultrasound images. |
121721 | RT Patient Position Registration, Spatial Fiducials | Registration of intended and actual patient position prior to treatment delivery, using spatial fiducials. |
121722 | RT Patient Position Adjustment | Adjustment of patient position such that the patient is correctly positioned for treatment. |
121723 | RT Patient Position In-treatment-session Review | Review of patient positioning information in the process of delivering a treatment session. |
121724 | RT Treatment Simulation with Internal Verification | Simulated radiotherapy treatment delivery using verification integral to the Treatment Delivery System. |
121725 | RT Treatment Simulation with External Verification | Simulated radiotherapy treatment delivery using verification by a external Machine Parameter Verifier. |
121726 | RT Treatment with Internal Verification | Radiotherapy treatment delivery using verification integral to the Treatment Delivery System. |
121727 | RT Treatment with External Verification | Radiotherapy treatment delivery using verification by a external Machine Parameter Verifier. |
121728 | RT Treatment QA with Internal Verification | Quality assurance of a radiotherapy treatment delivery using verification integral to the Treatment Delivery System. |
121729 | RT Treatment QA with External Verification | Quality assurance of a radiotherapy treatment delivery using verification by a external Machine Parameter Verifier. |
121730 | RT Machine QA | Quality assurance of a Treatment Delivery Device. |
121731 | RT Treatment QA by RT Plan Dose Check | Perform Quality Assurance on an RT Plan by evaluating dosimetric content of the current RT Plan. |
121732 | RT Treatment QA by RT Plan Difference Check | Perform Quality Assurance on an RT Plan by comparing the content of previously quality-assessed RT Plans with the current RT Plan. |
121733 | RT Treatment QA by RT Ion Plan Dose Check | Perform Quality Assurance on an RT Ion Plan by evaluating dosimetric content of the current RT Ion Plan. |
121734 | RT Treatment QA with RT Ion Plan Difference Check | Perform Quality Assurance on an RT Ion Plan by comparing the content of previously quality-assessed RT Ion Plans by the current RT Ion Plan. |
121740 | Treatment Delivery Type | Indicates whether the treatment to be delivered is a complete fraction or a continuation of previous incompletely treated fraction. |
122001 | Patient called to procedure room | Patient called to procedure room. |
122002 | Patient admitted to procedure room | Patient admitted to procedure room. |
122003 | Patient given pre-procedure instruction | Patient given pre-procedure instruction. |
122004 | Patient informed consent given | Patient informed consent given. |
122005 | Patient advance directive given | Patient advance directive given. |
122006 | Nil Per Os (NPO) status confirmed | Nil Per Os (NPO) status confirmed. |
122007 | Patient assisted to table | Patient assisted to table. |
122008 | Patient prepped and draped | Patient prepped and draped. |
122009 | Patient connected to continuous monitoring | Patient connected to continuous monitoring. |
122010 | Patient transferred to holding area | Patient transferred to holding area. |
122011 | Patient transferred to surgery | Patient transferred to surgery. |
122012 | Patient transferred to CCU | Patient transferred to CCU. |
122020 | Patient disoriented | Patient disoriented. |
122021 | Patient reports nausea | Patient reports nausea. |
122022 | Patient reports discomfort | Patient reports discomfort. |
122023 | Patient reports chest pain | Patient reports chest pain. |
122024 | Patient reports no pain | Patient reports no pain. |
122025 | Patient alert | Patient alert. |
122026 | Patient restless | Patient restless. |
122027 | Patient sedated | Patient sedated. |
122028 | Patient asleep | Patient asleep. |
122029 | Patient unresponsive | Patient unresponsive. |
122030 | Patient has respiratory difficulty | Patient has respiratory difficulty. |
122031 | Patient coughed | Patient coughed. |
122032 | Patient disconnected from continuous monitoring | Patient disconnected from continuous monitoring. |
122033 | Hemostasis achieved | Hemostasis achieved. |
122034 | Hemostasis not achieved - oozing | Hemostasis not achieved - oozing. |
122035 | Hemostasis not achieved - actively bleeding | Hemostasis not achieved - actively bleeding. |
122036 | Patient given post-procedure instruction | Patient given post-procedure instruction. |
122037 | Patient discharged from department | Patient discharged from department or laboratory. |
122038 | Patient pronounced dead | Patient pronounced dead. |
122039 | Patient transferred to morgue | Patient transferred to morgue. |
122041 | Personnel Arrived | Identified personnel or staff arrived in procedure room. |
122042 | Personnel Departed | Identified personnel or staff departed procedure room. |
122043 | Page Sent To | Page sent to identified personnel or staff. |
122044 | Consultation With | Consultation with identified personnel or staff. |
122045 | Office called | Office of identified personnel or staff was called. |
122046 | Equipment failure | Equipment failure | Retired. Replaced by (110501, DCM, "Equipment failure")
122047 | Equipment brought to procedure room | Equipment brought to procedure room. |
122048 | Equipment ready | Equipment ready for procedure. |
122049 | Equipment removed | Equipment removed from procedure room. |
122052 | Bioptome | Device for obtaining biopsy sample. |
122053 | Valvular Intervention | Valvular Intervention. |
122054 | Aortic Intervention | Aortic Intervention. |
122055 | Septal Defect Intervention | Septal Defect Intervention. |
122056 | Vascular Intervention | Vascular Intervention. |
122057 | Myocardial biopsy | Myocardial biopsy. |
122058 | Arterial conduit angiography | Arterial conduit angiography. |
122059 | Single plane Angiography | Single plane Angiography. |
122060 | Bi-plane Angiography | Bi-plane Angiography. |
122061 | Percutaneous Coronary Intervention | Percutaneous Coronary Intervention. |
122062 | 15-Lead ECG | 15-Lead electrocardiography | Retired. Replaced by (P2-3120E, SRT, "15-Lead ECG")
122072 | Pre-procedure log | Log of events occurring prior to the current procedure. |
122073 | Current procedure evidence | Analysis or measurements for current procedure (purpose of reference to evidence document). |
122075 | Prior report for current patient | Prior report for current patient. |
122076 | Consumable taken from inventory | Identifier of Consumable taken from inventory. |
122077 | Consumable returned to inventory | Identifier of Consumable returned to inventory. |
122078 | Remaining consumable disposed | Identifier of consumable whose remaining content has been disposed. |
122079 | Consumable unusable | Identifier of Consumable determined to be unusable. |
122081 | Drug start | Identifier of Drug whose administration has started. |
122082 | Drug end | Identifier of Drug whose administration has ended. |
122083 | Drug administered | Identifier of Drug administered as part of procedure. |
122084 | Contrast start | Identifier of Contrast agent whose administration has started. |
122085 | Contrast end | Identifier of Contrast agent whose administration has ended. |
122086 | Contrast administered | Identifier of Contrast agent administered. |
122087 | Infusate start | Identifier of Infusate whose administration has started. |
122088 | Infusate end | Identifier of Infusate whose administration has ended. |
122089 | Device crossed lesion | Action of a device traversing a vascular lesion. |
122090 | Intervention Action | Action of a clinical professional performed on a patient for therapeutic purpose. |
122091 | Volume administered | Volume of Drug, Contrast agent, or Infusate administered. |
122092 | Undiluted dose administered | Undiluted dose of Drug, Contrast agent, or Infusate administered. |
122093 | Concentration | Concentration of Drug, Contrast agent, or Infusate administered. |
122094 | Rate of administration | Rate of Drug, Contrast agent, or Infusate administration. |
122095 | Duration of administration | Duration of Drug, Contrast agent, or Infusate administration. |
122096 | Volume unadministered or discarded | Volume of Drug, Contrast agent, or Infusate unadministered or discarded. |
122097 | Catheter Curve | Numeric parameter of Curvature of Catheter. |
122098 | Transmit Frequency | Transmit Frequency. |
122099 | ST change from baseline | Measured change of patient electrocardiographic ST level relative to baseline measurement. |
122101 | Aneurysm on cited vessel | Anatomic term modifier indicating aneurysm on cited vessel is the subject of the finding. |
122102 | Graft to cited segment, proximal section | Anatomic term modifier indicating proximal section of graft to cited vessel is the subject of the finding. |
122103 | Graft to cited segment, mid section | Anatomic term modifier indicating mid section of graft to cited vessel is the subject of the finding. |
122104 | Graft to cited segment, distal section | Anatomic term modifier indicating distal section of graft to cited vessel is the subject of the finding. |
122105 | DateTime of Intervention | DateTime of Intervention. |
122106 | Duration of Intervention | Duration of Intervention. |
122107 | Baseline Stenosis Measurement | Lesion stenosis measured prior to any interventional procedure | Retired. Replaced by (R-101BB, SRT, "Lumen Diameter Stenosis"), post-coordinated with (G-7293, SRT, "Baseline Phase")
122108 | Post-Intervention Stenosis Measurement | Lesion stenosis measured after an interventional procedure | Retired. Replaced by (R-101BB, SRT, "Lumen Diameter Stenosis"), post-coordinated with (G-7298, SRT, "Post-intervention Phase")
122109 | Baseline TIMI Flow | Assessment of perfusion across a coronary lesion measured prior to any interventional procedure. |
122110 | Post-Intervention TIMI Flow | Assessment of perfusion across a coronary lesion measured after an interventional procedure. |
122111 | Primary Intervention Device | Indication that device is the primary (first and/or most significant) device used for interventional therapy of a particular pathology. E.g., lesion. |
122112 | Normal Myocardium | Normal Myocardium. |
122113 | Sacrred Myocardial | Sacrred Myocardial. |
122114 | Thinning Myocardium | Thinning Myocardium. |
122120 | Hemodynamics Report | Hemodynamics Report. |
122121 | Atrial pressure measurements | Atrial pressure measurements, report section. |
122122 | Ventricular pressure measurements | Ventricular pressure measurements, report section. |
122123 | Gradient assessment | Gradient assessment, report section. |
122124 | Blood velocity measurements | Blood velocity measurements, report section. |
122125 | Blood lab measurements | Blood lab measurements, report section. |
122126 | Derived Hemodynamic Measurements | Derived Hemodynamic Measurements, report section. |
122127 | Clinical Context | Clinical Context, report section. |
122128 | Patient Transferred From | Location from which the patient was transferred. |
122129 | PCI during this procedure | Indication that the procedure includes a percutaneous coronary intervention. |
122130 | Dose Area Product | Radiation dose times area of exposure. |
122131 | Degree of Thrombus | Finding of probability and/or severity of thrombus. |
122132 | Severity of Calcification | Severity of Calcification, property of lesion. |
122133 | Lesion Morphology | Lesion Morphology; form and/or structural properties of lesion. |
122134 | Vessel Morphology | Vessel Morphology; form and/or structural properties of vessel. |
122138 | Circulatory Support | Technique (device or procedure) of support for patient circulatory system; hemodynamic support. |
122139 | Reason for Exam | Reason for Exam. |
122140 | Comparison with Prior Exam Done | Indication that the current exam data has been compared with prior exam data. |
122141 | Electrode Placement | Electrocardiographic electrode placement technique. |
122142 | Acquisition Device Type | Acquisition Device Type. |
122143 | Acquisition Device ID | Acquisition Device ID. |
122144 | Quantitative Analysis | Quantitative Analysis, report section. |
122145 | Qualitative Analysis | Qualitative Analysis, report section. |
122146 | Procedure DateTime | The date and time on which a procedure was performed on a patient. |
122147 | Clinical Interpretation | Clinical Interpretation, report section. |
122148 | Lead ID | ECG Lead Identifier. |
122149 | Beat Number | Beat Number; ordinal of cardiac cycle within an acquisition. |
122150 | Compound Statement | Complex coded semantic unit, consisting of several coded components. |
122151 | Trend | Trend (temporal progression) of a clinical condition, finding, or disease. |
122152 | Statement | Coded semantic unit. |
122153 | Statement Modifier | Coded modifier for a semantic unit. |
122154 | Conjunctive Term | Conjunctive term between semantic units. |
122157 | Probability | Probability. |
122158 | ECG Global Measurements | ECG Global Measurements, report section. |
122159 | ECG Lead Measurements | ECG Lead Measurements, report section. |
122160 | Derived Area, Non-Valve | Derived cross-sectional area of a vessel or anatomic feature, other than a cardiac valve. |
122161 | Pulmonary Flow | Rate of blood flow through Pulmonary artery. |
122162 | Systemic Flow | Rate of blood flow through the aorta. |
122163 | Discharge DateTime | DateTime of patient discharge from hospital admission. |
122164 | Coronary Artery Bypass During This Admission | Indication that a Coronary Artery Bypass operation was performed during the current hospital admission. |
122165 | Date of Death | Date of Death. |
122166 | Death During This Admission | Indication that the patient died during the current hospital admission. |
122167 | Death During Catheterization | Indication that the patient died during the current Catheterization procedure. |
122170 | Type of Myocardial Infarction | Finding of type of Myocardial Infarction. |
122171 | Coronary lesion > = 50% stenosis | Finding of Coronary lesion with greater than 50% stenosis. |
122172 | Acute MI Present | Finding of Acute Myocardial Infarction Presence as indication for interventional procedure. |
122173 | ST Elevation Onset DateTime | DateTime of first determination of elevated ECG ST segment, as indication of Myocardial Infarction. |
122175 | Number of lesion interventions attempted | Number of lesion interventions attempted during current procedure. |
122176 | Number of lesion interventions successful | Number of lesion interventions successful during current procedure, where the residual post intervention stenosis is less than or equal to 50% of the arterial luminal diameter, TIMI Flow is 3 and the minimal decrease in stenosis was 20%. |
122177 | Procedure Result | Overall success of interventional procedure. |
122178 | Lesion Intervention Information | Lesion Intervention Information, report section. |
122179 | Peri-procedural MI occurred | Indication that Myocardial Infarction occurred during current procedure. |
122180 | CK-MB baseline | Creatine Kinase-MB value at baseline (start of procedure). |
122181 | CK-MB peak | Creatine Kinase-MB highest value measured during procedure. |
122182 | R-R interval | Time interval between ECG R-wave peaks in subsequent cardiac cycles. |
122183 | Blood temperature | Blood temperature. |
122185 | Blood Oxygen content | Blood Oxygen content. |
122187 | Blood Carbon dioxide saturation | Blood Carbon dioxide saturation. |
122188 | Pulmonary Arterial Content (FCpa) | Pulmonary Arterial Content (FCpa). |
122189 | Pulmonary Venous Content (FCpv) | Pulmonary Venous Content (FCpv). |
122190 | Max dp/dt/P | Max dp/dt/P. |
122191 | Ventricular End Diastolic pressure | Ventricular End Diastolic pressure. |
122192 | Indicator appearance time | Elapsed time from injection of an indicator bolus until it is observed at another location. |
122193 | Maximum pressure acceleration | Maximum pressure acceleration. |
122194 | Ventricular Systolic blood pressure | Ventricular Systolic blood pressure. |
122195 | Pulse Strength | Pulse Strength; palpable strength of systolic flow. |
122196 | C wave pressure | The secondary peak pressure in the atrium during atrial contraction. |
122197 | Gradient pressure, average | Gradient pressure, average. |
122198 | Gradient pressure, peak | Gradient pressure, peak. |
122199 | Pressure at dp/dt max | Pressure at dp/dt max. |
122201 | Diastolic blood velocity, mean | Diastolic blood velocity, mean. |
122202 | Diastolic blood velocity, peak | Diastolic blood velocity, peak. |
122203 | Systolic blood velocity, mean | Systolic blood velocity, mean. |
122204 | Systolic blood velocity, peak | Systolic blood velocity, peak. |
122205 | Blood velocity, mean | Blood velocity, mean. |
122206 | Blood velocity, minimum | Blood velocity, minimum. |
122207 | Blood velocity, peak | Blood velocity, peak. |
122208 | x-descent pressure | Venous or atrial pressure minimum during ventricular systole, after A-wave. |
122209 | y-descent pressure | Venous or atrial pressure minimum when tricuspid valve opens during diastole, after V-wave. |
122210 | z-point pressure | Atrial pressure upon closure of tricuspid and mitral valves. |
122211 | Left Ventricular ejection time | Left Ventricular ejection time. |
122212 | Left Ventricular filling time | Left Ventricular filling time. |
122213 | Right Ventricular ejection time | Right Ventricular ejection time. |
122214 | Right Ventricular filling time | Right Ventricular filling time. |
122215 | Total Pulmonary Resistance | Total Pulmonary Resistance. |
122216 | Total Vascular Resistance | Total Vascular Resistance. |
122217 | Coronary Flow reserve | Coronary Flow reserve. |
122218 | Diastolic/Systolic velocity ratio | Diastolic/Systolic velocity ratio. |
122219 | Hyperemic ratio | Hyperemic ratio. |
122220 | Hemodynamic Resistance Index | Hemodynamic Resistance Index. |
122221 | Thorax diameter, sagittal | Thorax diameter, sagittal. |
122222 | Procedure Environmental Characteristics | Environmental characteristics in the procedure room. |
122223 | Room oxygen concentration | Oxygen concentration in the procedure room. |
122224 | Room temperature | Temperature in the procedure room. |
122225 | Room Barometric pressure | Barometric pressure in the procedure room. |
122227 | Left to Right Flow | Left to Right Flow. |
122228 | Right to Left Flow | Right to Left Flow. |
122229 | Arteriovenous difference | Arteriovenous oxygen content difference. |
122230 | 10 Year CHD Risk | Framingham Study 10 Year CHD Risk. |
122231 | Comparative Average10 Year CHD Risk | Framingham Study Comparative Average10 Year CHD Risk. |
122232 | Comparative Low10 Year CHD Risk | Framingham Study Comparative Low10 Year CHD Risk. |
122233 | LDL Cholesterol Score Sheet for Men | Framingham Study LDL Cholesterol Score Sheet for Men. |
122234 | LDL Cholesterol Score Sheet for Women | Framingham Study LDL Cholesterol Score Sheet for Women. |
122235 | Total Cholesterol Score Sheet for Men | Framingham Study Total Cholesterol Score Sheet for Men. |
122236 | Total Cholesterol Score Sheet for Women | Framingham Study Total Cholesterol Score Sheet for Women. |
122237 | Corrected Sinus Node Recovery Time | Corrected Sinus Node Recovery Time. |
122238 | Max volume normalized to 50mmHg pulse pressure | Max volume normalized to 50mmHg pulse pressure. |
122239 | Oxygen Consumption | Oxygen Consumption. |
122240 | BSA = 3.207*WT^(0.7285-0.0188 log (WT)) *HT^0.3 | Body Surface Area computed from patient height and weight: BSA = 3.207*WT[g]^(0.7285-0.0188 log (WT[g])) *HT[cm] ^ 0.3 [Boyd E, The growth of the surface area of the human body. Minneapolis: University of Minnesota Press, 1935, eq. (36) ]. |
122241 | BSA = 0.007184*WT^ 0.425*HT^0.725 | Body Surface Area computed from patient height and weight: BSA = 0.007184* WT[kg] ^ 0.425*HT[cm] ^ 0.725 [Dubois and Dubois, Arch Int Med 1916 17:863-71]. |
122242 | BSA = 0.0235*WT^0.51456*HT^ 0.42246 | Body Surface Area computed from patient height and weight: BSA = 0.0235* WT[kg] ^0.51456*HT[cm]^ 0.42246 [Gehan EA, George SL, 'Estimation of human body surface area from height and weight', Cancer Chemother Rep 1970 54:225-35]. |
122243 | BSA = 0.024265*WT^0.5378*HT^0.3964 | Body Surface Area computed from patient height and weight: BSA = 0.024265 * WT[kg] ^ 0.5378 * HT[cm] ^ 0.3964 [Haycock G.B., Schwartz G.J., Wisotsky D.H. 'Geometric method for measuring body surface area: A height weight formula validated in infants, children and adults.' The Journal of Pediatrics 1978 93:1:62-66]. |
122244 | BSA = (HT * WT/36) ^0.5 | Body Surface Area computed from patient height and weight: BSA = (HT[m] * WT[kg] / 36) ^ 0.5 [Mosteller, R.D. 'Simplified Calculation of Body Surface Area.' N Engl J Med 1987 Oct 22;317(17):1098]. |
122245 | BSA = 1321+0.3433*WT | Body Surface Area computed from patient weight:BSA = 1321 + 0.3433 * WT[kg] (for pediatrics 3-30 kg) [Current, J.D. 'A Linear Equation For Estimating The Body Surface Area In Infants And Children', The Internet Journal of Anesthesiology. 1998. 2:2]. |
122246 | BSA = 0.0004688 * WT ^ (0.8168 - 0.0154 * log(WT)) | BSA = 0.0004688 * (1000 * WT) ^ (0.8168 - 0.0154 * log(1000 * WT)) Where (WT is weight in kilogram) Units = m2 Boyd, Edith. The Growth of the Surface Area of the Human Body(originally published in 1935 by the University of Minnesota Press), Greenwood Press, Westport, Connecticut, 1975, p. 102.Equation (35). |
122247 | VO2male = BSA (138.1 - 11.49 * loge(age) + 0.378 * HRf) | Equation for estimated oxygen consumption: VO2male = BSA (138.1 - 11.49 * loge(age) + 0.378 * HRf). |
122248 | VO2female = BSA (138.1 - 17.04 * loge(age) + 0.378 * HRf) | Equation for estimated oxygen consumption: VO2female = BSA (138.1 - 17.04 * loge(age) + 0.378 * HRf). |
122249 | VO2 = VeSTPD * 10 * (FIO2 - FE02) | Equation for estimated oxygen consumption: VO2 = VeSTPD * 10 * (FIO2 - FE02). |
122250 | VO2 = 152 * BSA | Equation for estimated oxygen consumption: VO2 = 152 * BSA. |
122251 | VO2 = 175 * BSA | Equation for estimated oxygen consumption: VO2 = 175 * BSA. |
122252 | VO2 = 176 * BSA | Equation for estimated oxygen consumption: VO2 = 176 * BSA. |
122253 | Robertson & Reid table | Robertson & Reid Table for estimated oxygen consumption. |
122254 | Fleisch table | Fleisch table for estimated oxygen consumption. |
122255 | Boothby table | Boothby table for estimated oxygen consumption. |
122256 | if (prem age< 3days) P50 = 19.9 | Estimate of Oxygen partial pressure at 50% saturation for premature infants less than 3 days old: P50 = 19.9. |
122257 | if (age < 1day) P50 = 21.6 | Estimate of Oxygen partial pressure at 50% saturation for infants less than 1 day old: P50 = 21.6. |
122258 | if (age < 30day) P50 = 24.6 | Estimate of Oxygen partial pressure at 50% saturation for infants less than 30 days old: P50 = 24.6. |
122259 | if (age < 18y) P50 = 27.2 | Estimate of Oxygen partial pressure at 50% saturation for patients less than 18 years old: P50 = 27.2. |
122260 | if (age < 40y) P50 = 27.4 | Estimate of Oxygen partial pressure at 50% saturation for patients less than 40 years old: P50 = 27.4. |
122261 | if (age > 60y) P50 = 29.3 | Estimate of Oxygen partial pressure at 50% saturation for patients more than 60 years old: P50 = 29.3. |
122262 | Area = Flow / 44.5 * sqrt(Gradient[mmHg]) | Cardiac valve area computed from flow and pressure gradient: Area = Flow / 44.5 * sqrt(Gradient[mmHg]) [Gorlin and Gorlin, Am Heart J, 1951]. |
122263 | MVA = Flow / 38.0 * sqrt(Gradient[mmHg]) | Mitral valve area computed from flow and pressure gradient: Mitral valve Area = Flow / 38.0 * sqrt(Gradient[mmHg]) [Gorlin and Gorlin, Am Heart J, 1951]. |
122265 | BMI = Wt / Ht ^ 2 | Body Mass Index computed from weight and height: BMI = Wt/Ht^2. |
122266 | BSA = 0.007358 * WT ^ 0.425 * HT ^ 0.725 | Body Surface Area computed from patient height and weight: BSA = 0.007358 * WT[kg] ^ 0.425 * HT[cm] ^ 0.725 (for East Asian adult, aged 15+ years) [Kanai Izumi, Masamitsu Kanai, 'Clinical examination method summary']. |
122267 | BSA = 0.010265 * WT ^ 0.423 * HT ^ 0.651 | Body Surface Area computed from patient height and weight: BSA = 0.010265 * WT[kg] ^ 0.423 * HT[cm] ^ 0.651 (For East Asian child aged 12-14 years). |
122268 | BSA = 0.008883 * WT ^ 0.444 * HT ^ 0.663 | Body Surface Area computed from patient height and weight: BSA = 0.008883* WT[kg] ^ 0.444 * HT[cm] ^ 0.663 (For East Asian child aged 6-11 years). |
122269 | BSA = 0.038189 * WT ^ 0.423 * HT ^ 0.362 | Body Surface Area computed from patient height and weight: BSA = 0.038189 * WT[kg] ^ 0.423 * HT[cm] ^ 0.362 (For East Asian child aged 1-5 years). |
122270 | BSA = 0.009568 * WT ^ 0.473 * HT ^ 0.655 | Body Surface Area computed from patient height and weight: BSA = 0.009568* WT[kg] ^ 0.473 * HT[cm] ^ 0.655 (For East Asian child aged 0-12 months). |
122271 | Skin Condition Warm | Skin Condition Warm. |
122272 | Skin Condition Cool | Skin Condition Cool. |
122273 | Skin Condition Cold | Skin Condition Cold. |
122274 | Skin Condition Dry | Skin Condition Dry. |
122275 | Skin Condition Clammy | Skin Condition Clammy. |
122276 | Skin Condition Diaphoretic | Skin Condition Diaphoretic. |
122277 | Skin Condition Flush | Skin Condition Flush. |
122278 | Skin Condition Mottled | Skin Condition Mottled. |
122279 | Skin Condition Pale | Skin Condition Pale. |
122281 | Airway unobstructed | Airway unobstructed. |
122282 | Airway partially obstructed | Airway partially obstructed. |
122283 | Airway severely obstructed | Airway severely obstructed. |
122288 | Not Visualized | Anatomy could not be visualized for the purpose of evaluation. |
122291 | Quantitative Arteriography Report | Quantitative Arteriography Report. |
122292 | Quantitative Ventriculography Report | Quantitative Ventriculography Report. |
122301 | Guidewire crossing lesion unsuccessful | Guidewire crossing lesion unsuccessful. |
122302 | Guidewire crossing lesion successful | Guidewire crossing lesion successful. |
122303 | Angioplasty balloon inflated | Angioplasty balloon inflated. |
122304 | Angioplasty balloon deflated | Angioplasty balloon deflated. |
122305 | Device deployed | Device deployed. |
122306 | Stent re-expanded | Stent re-expanded. |
122307 | Object removed | Object removed. |
122308 | Radiation applied | Radiation applied. |
122309 | Radiation removed | Radiation removed. |
122310 | Interventional device placement unsuccessful | Interventional device placement unsuccessful. |
122311 | Interventional device placed | Interventional device placed. |
122312 | Intervention performed | Intervention performed. |
122313 | Interventional device withdrawn | Interventional device withdrawn. |
122319 | Catheter Size | Catheter Size. |
122320 | Injectate Temperature | Injectate Temperature. |
122321 | Injectate Volume | Injectate Volume. |
122322 | Calibration Factor | Factor by which a measured or calculated value is multiplied to obtain the estimated real-world value. |
122325 | IVUS Report | Intravascular Ultrasound Report. |
122330 | EEM Diameter | External Elastic Membrane (EEM) diameter measured through the center point of the vessel. Center point of the vessel is defined as the center of gravity of the EEM area. The EEM is a discrete interface at the border between the media and the adventitia. |
122331 | Plaque Plus Media Thickness | The distance from intimal leading edge to the external elastic membrane along any line passing through the luminal center, which is defined as the center of gravity of the lumen area. |
122332 | Lumen Perimeter | Planimetered perimeter of the lumen. |
122333 | EEM Cross-Sectional Area | Vessel area measured at the External Elastic Membrane (EEM), a discrete interface at the border between the media and the adventitia. |
122334 | Plaque plus Media Cross-Sectional Area | Area within the EEM occupied by atheroma, regardless of lumen compromise. Plaque plus Media Area = EEM cross-sectional area - vessel lumen cross-sectional area. |
122335 | In-Stent Neointimal Cross-Sectional Area | Measurement of in-stent restenosis. In-Stent Intimal Area = Stent cross-sectional area - vessel lumen cross-sectional area. |
122336 | Vascular Volume measurement length | Longitudinal extent of the Vascular Volume Measurement. This is the distance from the distal edge to the proximal edge of the Volume measurement. |
122337 | Relative position | Longitudinal distance from the closest edge of a fiducial feature or reference location to the start of the vascular measurement. This value will be a positive if the measurement is distal to the fiducial feature or reference location, or negative if the measurement is proximal to the fiducial feature or reference location. |
122339 | Stent Volume Obstruction | In-Stent Neointimal Volume / Stent Volume. |
122340 | Fiducial feature | Reference, normally anatomical, which is used for locating the position of a measurement. |
122341 | Calcium Length | Longitudinal calcium length measurement. |
122343 | Lumen Eccentricity Index | Measurement of vessel lumen eccentricity. Lumen Eccentricity Index = (maximum vessel lumen diameter - minimum vessel lumen diameter) / maximum vessel lumen diameter. Lumen diameters are measured through the center point of the lumen, which is defined as the center of gravity of the lumen area. |
122344 | Plaque plus Media Eccentricity Index | Plaque plus Media Eccentricity Index = (maximum Plaque plus media thickness - minimum Plaque plus media thickness) / maximum Plaque plus media thickness. |
122345 | Remodeling Index | Measurement of increase or decrease in EEM area that occurs during the development of atherosclerosis. Remodeling Index = Lesion EEM area / reference EEM area. |
122346 | Stent Symmetry Index | Measurement of stent circularity. Stent Symmetry Index = (maximum stent diameter - minimum stent diameter) / maximum stent diameter. |
122347 | Stent Expansion Index | Measurement of stent area relative to the reference lumen area. Stent Expansion Index = Minimum stent area / reference vessel lumen cross-sectional area. |
122348 | Lumen Shape Index | Measurement of vessel lumen eccentricity. Lumen Shape Index = (2p * sqrt(Vessel lumen cross-sectional area / p) / Lumen Perimeter) 2 Reference: Tobis & Yock, "Intravascular Ultrasound Imaging", Chapter 7. |
122350 | Lumen Diameter Ratio | Lumen diameter ratio = minimum vessel lumen diameter / maximum vessel lumen diameter, measured at the same cross section in the vessel. Lumen diameters are measured through the center point of the lumen, which is defined as the center of gravity of the lumen area. |
122351 | Stent Diameter Ratio | Stent diameter ratio = Minimum stent diameter / Maximum stent diameter, measured at the same cross section in the vessel. Stent diameters are measured through the center point of the stent, which is defined as the center of gravity of the stent area. |
122352 | EEM Diameter Ratio | EEM diameter ratio = minimum EEM diameter / maximum EEM diameter. Measured at the same cross section in the vessel. |
122354 | Plaque Burden | Fractional area within the External Elastic Membrane (EEM) occupied by atheroma. Plaque Burden = (EEM area - vessel lumen cross-sectional area) / EEM area. |
122355 | Arc of Calcium | Angular measurement of a Calcium deposit with the apex located at the center of the lumen, which is defined as the center of gravity of the lumen area. |
122356 | Soft plaque | Plaque characterized by low density or echogenicity. |
122357 | In-Stent Neointima | Abnormal thickening of the intima within the stented segment. |
122360 | True Lumen | Lumen surrounded by all three layers of the vessel-intima, media, and adventitia. |
122361 | False Lumen | A channel, usually parallel to the true lumen, which does not communicate with the true lumen over a portion of its length. |
122363 | Plaque Rupture | Plaque ulceration with a tear detected in a fibrous cap. |
122364 | Stent Gap | Length of gap between two consecutive stents. |
122367 | T-1 Worst | Worst stenosis - the stenosis with the smallest lumen size within a vessel segment. |
122368 | T-2 Secondary | 2nd most severe stenosis within a vessel segment. |
122369 | T-3 Secondary | 3rd most severe stenosis within a vessel segment. |
122370 | T-4 Secondary | 4th most severe stenosis within a vessel segment. |
122371 | EEM Volume | External Elastic Membrane (EEM) volume measured within a specified region. The EEM is a discrete interface at the border between the media and the Adventitia. |
122372 | Lumen Volume | Lumen volume measured within a specified region. |
122374 | In-Stent Neointimal Volume | The amount of plaque between the lumen and stent, within the stent region; In-stent restenosis. In-Stent Neointimal Volume = Stent Volume - Lumen Volume. |
122375 | Native Plaque Volume | The amount of plaque between the stent and the EEM, within the stent region. Native Plaque Volume = EEM Volume - Stent Volume. |
122376 | Total Plaque Volume | Total amount of plaque between the EEM and the Lumen, over the entire region that is measured. Total Plaque Volume = EEM Volume - Lumen Volume. |
122380 | Proximal Reference | Proximal reference segment measurement site. Typically the site with the largest lumen proximal to a stenosis but within the same segment (usually within 10 mm of the stenosis with no major intervening branches). |
122381 | Distal Reference | Distal reference segment measurement site. Typically the site with the largest lumen distal to a stenosis but within the same segment (usually within 10 mm of the stenosis with no major intervening branches). |
122382 | Site of Lumen Minimum | Site of the smallest lumen in a vessel. E.g., due to a stenotic lesion. |
122383 | Entire Pullback | Measurement region that encompasses the entire vessel imaged in a single pullback acquisition. |
122384 | Stented Region | Measurement region occupied by the stent. |
122385 | Proximal Stent Margin | Region starting at the proximal edge of the Stent and extending several millimeters (usually 5 mm) proximal to the Stent edge. |
122386 | Distal Stent Margin | Region starting at the distal edge of the Stent and extending several millimeters (usually 5 mm) distal to the Stent edge. |
122387 | Dissection Classification | Classification of dissections in a vessel. |
122388 | Intra-stent Dissection | Separation of neointimal hyperplasia from stent struts, usually seen only after treatment of in-stent restenosis. |
122389 | Vulnerable Plaque | Plaque with a thin cap fibrous atheroma that is at increased risk of rupture and thrombosis (or re-thrombosis) and rapid stenosis progression. |
122390 | Eroded Plaque | Plaque erosions with no structural defect (beyond endothelial injury) or gap in the plaque. |
122391 | Relative Stenosis Severity | Stenosis severity classifications of multiple lesions in a vessel. |
122393 | Restenotic Lesion | A finding of a previously treated lesion in which stenosis has reoccurred. |
122394 | Fibro-Lipidic Plaque | Loosely packed bundles of collagen fibers with regions of lipid deposition present. Region is cellular and no cholesterol clefts or necrosis are present. Some macrophage infiltration. Increase in extra cellular matrix. |
122395 | Necrotic-Lipidic Plaque | Area within the plaque with very low echogenicity separated from the lumen and surrounded by more echogenic structures (fibrous cap). Highly lipidic necrotic region with remnants of foam cells and dead lymphocytes present. No collagen fibers are visible and mechanical integrity is poor. Cholesterol clefts and micro calcifications are visible. |
122398 | Intimal Dissection | Dissection limited to the intima or atheroma, and not extending to the media. |
122399 | Medial Dissection | Dissection in the arterial Media, extending into the media. |
122400 | Simultaneously Acquired | The referenced information was acquired simultaneously with the information in the object in which the reference occurs. |
122401 | Same Anatomy | Information acquired for the same anatomic region. |
122402 | Same Indication | Information acquired for the same indication. E.g., to elucidate the same diagnostic question. |
122403 | For Attenuation Correction | The referenced information was used to correct the data for differential attenuation through different anatomic tissue. |
122404 | Reconstructed | Value estimated for a vessel in the absence of a stenosis. |
122405 | Algorithm Manufacturer | Manufacturer of application used. |
122406 | Left Atrial Ejection Fraction by Angiography | Left Atrial Ejection Fraction by Angiography. |
122407 | Left Atrial ED Volume | Left Atrial End Diastolic Volume. |
122408 | Left Atrial ES Volume | Left Atrial End Systolic Volume. |
122410 | Contour Realignment | Contour repositioning of End Diastolic relative to End Systolic contour. |
122411 | Threshold Value | The minimum standard deviation to define the hypokinesis and hyperkinesis. |
122417 | Regional Abnormal Wall Motion | Report of differentiation of wall motion compared to normal. |
122421 | Calibration Object | Object used for Calibration. |
122422 | Calibration Method | Method used for Calibration. |
122423 | Calibration Object Size | Size of calibration object. |
122428 | Area Length Method | Method how long axis is positioned. |
122429 | Volume Method | Model for cardiac chamber volume calculation. |
122430 | Reference Method | Method to define original diameter of the artery. |
122431 | Regression Slope ED | Relation between calculated End Diastolic volume and ventricular End Diastolic volume. The specific meaning is dependent on volume method used. |
122432 | Regression Offset ED | Correction factor for the calculated End Diastolic volume and ventricular End Diastolic volume. The specific meaning is dependent on volume method used. |
122433 | Regression Slope ES | Relation between calculated End Systolic volume and ventricular End Systolic volume. The specific meaning is dependent on volume method used. |
122434 | Regression Offset ES | Correction factor for the calculated End Systolic volume and ventricular End Systolic volume. The specific meaning is dependent on volume method used. |
122435 | Regression Volume Exponent | Exponent of volume in regression formula. |
122438 | Reference Points | Container for spatial locations or coordinates used for calculation. |
122445 | Wall Thickness | Average thickness of the chamber wall in the current view. |
122446 | Wall Volume | Volume of the chamber wall estimated from the current view. |
122447 | Wall Mass | Mass of the chamber wall (myocardium). |
122448 | Wall Stress | Peak systolic stress of chamber wall. |
122449 | Centerline Wall Motion Analysis | Method to calculate wall motion [example: Sheehan, 1986]. |
122450 | Normalized Chord Length | The length between End Diastolic and End Systolic contour perpendicular on the centerline normalized by a method dependent ventricular perimeter length. The centerline is the line equidistant between the End Diastolic and End Systolic contour [example: Sheehan, 1986]. |
122451 | Abnormal Region | The report of the boundaries of the abnormal (hyperkinetic, hypokinetic, a-kinetic) regions associated with the territory of the artery [example: Sheehan, 1986]. |
122452 | First Chord of Abnormal Region | The chord number specifying the begin of abnormal region [example: Sheehan, 1986]. |
122453 | Last Chord of Abnormal Region | The chord number specifying the end of abnormal region [example: Sheehan, 1986]. |
122459 | Territory Region Severity | Severity at the regional abnormality extent [example: Sheehan, 1986]. |
122461 | Opposite Region Severity | Severity at the opposite regional abnormality extent [example: Sheehan, 1986]. |
122464 | LAD Region in RAO Projection | Based on a total number of chords of 100 and RAO project the range of chords belonging to this circumferential extent lies between 5 - 85. [Sheehan, 1986]. |
122465 | RCA Region in ROA Projection | Based on a total number of chords of 100 and RAO project the range of chords belonging to this circumferential extent lies between 25 - 85. [Sheehan, 1986]. |
122466 | Single LAD Region in RAO Projection | Based on a total number of chords of 100 and RAO projection the range of chords belonging to this regional extent lies between 10 - 66 (hypokinetic) and 67 - 80 (hyperkinetic). [Sheehan, 1986]. |
122467 | Single RCA Region in RAO Projection | Based on a total number of chords of 100 and RAO projection the range of chords belonging to this regional extent lies between 51 - 80 (hypokinetic) and 10 - 50 (hyperkinetic). [Sheehan, 1986]. |
122468 | Multiple LAD Region in RAO Projection | Based on a total number of chords of 100 and RAO projection the range of chords belonging to this regional extent lies between 10 - 58 (hypokinetic) and 59 -80 (hyperkinetic). [Sheehan, 1986]. |
122469 | Multiple RCA Region in RAO Projection | Based on a total number of chords of 100 and RAO projection the range of chords belonging to this regional extent lies between 59 - 80 (hypokinetic) and 10 - 58 (hyperkinetic). [Sheehan, 1986]. |
122470 | LAD Region in LAO Projection | Based on a total number of chords of 100 and LAO projection the range of chords belonging to this regional extent lies between 50 -100 (hypokinetic) and 20 - 49 (hyperkinetic). [Sheehan, 1986]. |
122471 | RCA Region in LAO Projection | Based on a total number of chords of 100 and LAO projection the range of chords belonging to this regional extent lies between 19 - 67 (hypokinetic) and 68 - 100 (hyperkinetic). [Sheehan, 1986]. |
122472 | CFX Region in LAO Projection | Based on a total number of chords of 100 and LAO projection the range of chords belonging to this regional extent lies between 38 -74 (hypokinetic) and 75 - 100 (hyperkinetic). [Sheehan, 1986]. |
122473 | Circular Method | Method based on assumption that the image object is circular. |
122474 | Densitometric Method | Method based on the gray value distribution of the image. |
122475 | Center of Gravity | End Systolic contour realigned to End Diastolic contour based on the center of gravity. |
122476 | Long Axis Based | End Systolic contour realigned to End Diastolic contour based on the mid point of the long axis. The long axis end points are defined as the posterior and apex. |
122477 | No Realignment | No Contour Realignment applied. |
122480 | Vessel Lumen Cross-Sectional Area | Calculated Vessel Lumen Cross-Sectional Area based on the referenced method. |
122481 | Contour Start | Location of the beginning of a contour. |
122482 | Contour End | Location of the end of a contour. |
122485 | Sphere | Sphere is used as calibration object. |
122486 | Geometric Isocenter | Object of interest in isocenter of image and pixel separation is calculated from geometric data. |
122487 | Geometric Non-Isocenter | Object of interest not in isocenter of image and pixel separation is calculated from geometric data and out of isocenter distances. |
122488 | Calibration Object Used | Object used for calibration. E.g., sphere. |
122489 | Curve Fitted Reference | Application dependent method to calculate the reference diameter based on the multiple diameter values. |
122490 | Interpolated Local Reference | Application dependent method to calculate reference by interpolation, based on the diameter of two or more user defined reference positions. |
122491 | Mean Local Reference | Application dependent method to calculate by averaging the reference, based on the diameter of one or more user defined reference positions. |
122493 | Radial Based Wall Motion Analysis | Method to calculate wall motion based on the lengths of the radials in the predefined regions [Ingels]. |
122495 | Regional Contribution to Ejection Fraction | Contribution of Region to global Ejection factor based on radial or landmark based wall motion method. |
122496 | Radial Shortening | The reduction of area between End Systolic and End Diastolic based on radial wall motion analysis. |
122497 | Landmark Based Wall Motion Analysis | Method to calculate wall motion based on the move of landmarks on the wall [Slager]. |
122498 | Slice Contribution to Ejection Fraction | Contribution to the ejection fraction of a specific slice region [Slager]. |
122499 | Frame to Frame Analysis | Method to calculate volumes of heart chambers for every image in a range. |
122501 | Area of closed irregular polygon | The area is derived by considering a set of coordinates as a closed irregular polygon, accounting for inner angles. The exact method, such as by decomposition into triangles or quadrilaterals, is not specified, since it does not affect the numeric result, apart from the effect of numeric precision during computation of intermediate results. |
122502 | Area of a closed NURBS | The area is derived by using a set of coordinates as control points for a Non Uniform Rational B-Spline (NURBS). |
122503 | Integration of sum of closed areas on contiguous slices | The volume derived by integrating the sum of the areas on adjacent slices across the slice interval; each area is defined by a regular planar shape or by considering a set of coordinates as a closed irregular polygon, accounting for inner angles. |
122505 | Calibration | Procedure used to calibrate measurements or measurement devices. |
122507 | Left Contour | Left contour of lumen (direction proximal to distal). |
122508 | Right Contour | Right contour of lumen (direction proximal to distal). |
122509 | Diameter Graph | Ordered set of diameters values derived from contours (direction proximal to distal). |
122510 | Length Luminal Segment | Length Luminal Segment. |
122511 | Graph Increment | Increment value along X-axis in Diameter Graph. |
122516 | Site of Maximum Luminal | Location of the maximum lumen area in a lesion or vessel. |
122517 | Densitometric Luminal Cross-sectional Area Graph | Ordered set of cross-sectional Vessel Lumen Cross-Sectional Area values derived from contours (direction proximal to distal) based on densitometric method. |
122528 | Position of Proximal Border | Position of proximal border of segment relative to the contour start (proximal end of analysis area). |
122529 | Position of Distal Border | Position of distal border of segment relative to the contour start (proximal end of analysis area). |
122542 | Plaque Area | Longitudinal cross sectional area of plaque. |
122544 | Diameter Symmetry | Symmetry of stenosis (0 = complete asymmetry, 1 = complete symmetry); see Section T.2 “Definition of Diameter Symmetry with Arterial Plaques” in PS3.17 . |
122545 | Area Symmetry | Symmetry of plaque (0 = complete asymmetry, 1 = complete symmetry); see Section T.2 “Definition of Diameter Symmetry with Arterial Plaques” in PS3.17 . |
122546 | Inflow Angle | The average slope of the diameter function between the position of the minimum luminal diameter and the position of the proximal border of the segment. |
122547 | Outflow Angle | The average slope of the diameter function between the position of the minimum luminal diameter and the position of the distal border of the segment. |
122548 | Stenotic Flow Reserve | The relation between coronary pressure and coronary flow. |
122549 | Poiseuille Resistance | Poiseuille Resistance at the location of the stenosis. |
122550 | Turbulence Resistance | Turbulence Resistance at the location of the stenosis. |
122551 | Pressure Drop at SFR | Pressure drop over the stenosis at maximum heart output. |
122554 | Segmentation Method | Method for selection of vessel sub-segments. |
122555 | Estimated Normal Flow | Estimate of the volume of blood flow in the absence of stenosis. |
122558 | Area Length Kennedy | Area Length method defined by Kennedy [Kennedy, 1970]. |
122559 | Area Length Dodge | Area Length method defined by Dodge [Dodge, 1960]. |
122560 | Area Length Wynne | Area Length method defined by Wynne [Wynne]. |
122562 | Multiple Slices | Volume method based on multiple slice. |
122563 | Boak | Volume method defined by Boak [Boak]. |
122564 | TS Pyramid | Volume method defined by Ferlinz [Ferlinz]. |
122565 | Two Chamber | Volume method defined by Graham [Graham]. |
122566 | Parallelepiped | Volume method defined by Arcilla [Arcilla]. |
122572 | BSA^1.219 | Corrected Body Surface area for indexing the hemodynamic measurements for a pediatric patient. |
122574 | Equidistant method | Method for selecting sub-segments that are all of the same length. |
122575 | User selected method | Manually selected start and end of sub-segment. |
122582 | Left ventricular posterobasal segment | Left ventricular posterobasal segment. |
122600 | Cardiovascular Analysis Report | Report of a Cardiovascular Analysis, typically from a CT or MR study. |
122601 | Ventricular Analysis | Ventricular Analysis. |
122602 | Myocardial Perfusion Analysis | Myocardial Perfusion Analysis. |
122603 | Calcium Scoring Analysis | Calcium Scoring Analysis. |
122604 | Flow Quantification | Flow Quantification Analysis. |
122605 | Vascular Morphological Analysis | Vascular Morphological Analysis. |
122606 | Vascular Functional Analysis | Vascular Functional Analysis. |
122607 | Thickening Analysis | Analysis of myocardial wall thickening. |
122608 | Absolute Values Of Ventricular Measurements | Section Heading for absolute values of ventricular measurements. |
122609 | Normalized Values Of Ventricular Measurements | Results of normalizing ventricular measurements. |
122611 | Reference Point | Reference Point of a measurement. |
122612 | Central breathing position | Central breathing position between inspiration and expiration. |
122616 | Peak Ejection Rate | Peak of the ventricular ejection rate. |
122617 | Peak Ejection Time | Time of the peak of ventricular ejection. |
122618 | Peak Filling Rate | Peak of the fluid filling rate. |
122619 | Peak Filling Time | Time interval until time of peak filling from a given reference point. E.g., end systole. |
122620 | Papillary Muscle Excluded | Papillary muscle was excluded from the measurement. |
122621 | Papillary Muscle Included | Papillary muscle was included in the measurement. |
122624 | Wall Thickness Ratio end-systolic to end-diastolic | The ratio of the end-systolic wall thickness compared to the end-diastolic wall thickness. |
122627 | Curve Fit Method | The method to smooth a ventricular volume as a function of time. |
122628 | Baseline Result Correction | Baseline correction used in the calculation of the results. |
122631 | Signal Earliest Peak Time | The time in a dynamic set of images at which the first peak of the signal is observed for the analyzed myocardial wall segments. |
122633 | Signal Increase Start Time | This is the time at which the signal begins to increase. |
122634 | Signal Time to Peak | Time interval between the beginning of the signal increase to the time at which the signal intensity reaches its first maximum in a dynamic set of images. |
122635 | MR Perfusion Peak | Peak of the MR perfusion signal. |
122636 | MR Perfusion Slope | Signal intensity as a function of time. It is the change in the signal intensity divided by the change in the time. |
122637 | MR Perfusion Time Integral | MR perfusion time integral from baseline (foot time) to earliest peak. |
122638 | Signal Baseline Start | First time point in a dynamic set of images used in the calculation of the baseline signal intensity for each myocardial wall segment. |
122639 | Signal Baseline End | Last time point in a dynamic set of images used in the calculation of the baseline signal intensity for each myocardial wall segment. |
122640 | Image Interval | The time delta between images in a dynamic set of images. |
122642 | Velocity Encoding Minimum Value | The minimum velocity encoded by the phase encoding gradient. |
122643 | Velocity Encoding Maximum Value | The maximum velocity encoded by the phase encoding gradient. |
122645 | Net Forward Volume | Forward volume-reverse volume. |
122650 | Area Based Method | Area Based Method for estimating volume or area. |
122651 | Diameter Based Method | Diameter Based Method for estimating volume, area or diameter. |
122652 | Volume Based Method | Volume Based Method for estimating volume. |
122655 | NASCET | A method of diameter measurements according to NASCET (North American Symptomatic Carotid Endarterectomy Trial). |
122656 | ECST | A method of diameter measurements according to ECST (European Carotid Surgery Trial). |
122657 | Agatston Score Threshold | Agatston Score Threshold. |
122658 | Calcium Mass Threshold | Calcium Mass Threshold. |
122659 | Calcium Scoring Calibration | Calcium Scoring Calibration. |
122660 | Calcium Volume | Calcium Volume. |
122661 | Calcium Mass | Calcium Mass. |
122664 | Late Contrast Enhancement | Delayed hyperenhancement of a tissue observed in an image acquired after injection of contrast media. |
122665 | Time interval since injection of contrast media | Time interval since injection of contrast media. |
122666 | Time relative to R-wave peak | Time relative to R-wave peak. |
122667 | Blood velocity vs. time of cardiac cycle | Relationship between blood velocity and time relative to R-wave peak. |
122668 | Time interval since detection of contrast bolus | Time interval since detection of contrast bolus. |
122670 | Papillary Muscle Included/Excluded | Indicates if the papillary muscle was included or excluded in the measurement. |
122675 | Anterior-Posterior | Anterior to Posterior direction. |
122680 | endoleak | Persistent flow of blood into the stent-grafting. |
122683 | Stent Fracture | Fracture of a stent. |
122684 | Stent Disintegration | Disintegration of a stent. |
122685 | Stent Composition | Material that a stent consists of. |
122686 | Parent Vessel Finding | Finding about the characteristics of the parent vessel of a vessel. |
122687 | Site of Lumen Maximum | Site of Maximal lumen diameter of a vessel. |
122698 | X-Concept | The physical domain (time, space, etc.) to the horizontal axis of the graphical presentation. |
122699 | Y-Concept | The physical domain (time, space, etc.) to the vertical axis of the graphical presentation. |
122700 | Indications for Pharmacological Stress | Indications for Pharmacological Stress. |
122701 | Procedure time base | Reference time for measurement of elapsed time in a procedure. |
122702 | Treadmill speed | Treadmill speed. |
122703 | Treadmill gradient | Treadmill gradient. |
122704 | Ergometer power | Ergometer power. |
122705 | Pharmacological Stress Agent Dose Rate | Pharmacological Stress Agent Dose Rate. |
122706 | Rating of Perceived Exertion | Rating of Perceived Exertion. |
122707 | Number of Ectopic Beats | Number of ectopic beats during a period of collection. |
122708 | Double Product | Heart rate times systolic blood pressure. |
122709 | Activity workload | Physical activity workload (intensity) measurement. |
122710 | Time since start of stage | Elapsed time at stage. |
122711 | Exercise duration after stress agent injection | Exercise duration after stress agent injection. |
122712 | Imaging Start Time | Imaging Start Time. |
122713 | Attenuation correction method | Attenuation correction method. |
122715 | Pharmacological Stress Agent Dose | Pharmacological Stress Agent Dose. |
122716 | Maximum Power Output Achieved | Maximum power output achieved during course of procedure. |
122717 | Peak activity workload | Peak physical activity intensity measurement during course of procedure. |
122718 | Peak Double Product | Peak Double Product measurement during course of procedure. |
122720 | OSEM algorithm | Ordered subsets expectation maximization reconstruction algorithm. |
122721 | Chang method | Chang attenuation correction method. |
122726 | Algorithmic attenuation correction | Attenuation correction not based on image-based attenuation maps. |
122727 | NM transmission attenuation correction | NM transmission attenuation correction. |
122728 | CT-based attenuation correction | CT-based attenuation correction. |
122729 | No Attenuation Correction | No attenuation correction. |
122730 | Bazett QTc Algorithm | Bazett QT Correction Algorithm; QT/(RR ^ 0.5); Bazett HC. "An analysis of the time-relations of electrocardiograms" Heart7:353-370 (1920). |
122731 | Hodges QTc Algorithm | Hodges QT Correction Algorithm; QT + 1.75 (heart rate-60); Hodges M, Salerno Q, Erlien D. "Bazett's QT correction reviewed. Evidence that a linear QT correction for heart rate is better." J Am Coll Cardiol1:694 (1983). |
122732 | Fridericia QTc Algorithm | Fridericia QT Correction Algorithm; QT/(RR ^ 0.333); Fridericia LS. "The duration of systole in the electrocardiogram of normal subjects and of patients with heart disease" Acta Med Scand53:469-486 (1920). |
122733 | Framingham QTc Algorithm | Framingham QT Correction Algorithm; QT + 0.154 (1- RR); Sagie A, Larson MG, Goldberg RJ, et al. "An improved method for adjusting the QT interval for heart rate (the Framingham Heart Study)." Am J Cardiol70:797-801(1992). |
122734 | Borg RPE Scale | Borg Rating of Perceived Exertion Scale, range 6:20. |
122735 | Borg CR10 Scale | Borg category ratio scale, open ended range with nominal range 0:10. |
122739 | Overall study quality | Overall study quality. |
122740 | Excellent image quality | Excellent image quality. |
122741 | Good image quality | Good image quality. |
122742 | Poor image quality | Poor image quality. |
122743 | Body habitus attenuation | Image attenuation due to body physique (overweight). |
122744 | Breast attenuation | Image attenuation due to breast tissue. |
122745 | Diaphragmatic attenuation | Image attenuation due to diaphragm. |
122748 | False positive defect finding | Finding of a defect is incorrect. E.g., from automated analysis. |
122750 | Non-diagnostic - low heart rate | ECG is non-diagnostic due to low heart rate. |
122751 | Non-diagnostic - resting ST abnormalities | ECG is non-diagnostic due to resting ST abnormalities. |
122752 | Non-diagnostic - ventricular pacing or LBBB | ECG is non-diagnostic due to ventricular pacing or Left Bundle Branch Block. |
122753 | Non-diagnostic ECG | ECG is non-diagnostic for presence of acute coronary syndrome. |
122755 | Strongly positive | Strongly positive finding. |
122756 | Strongly positive - ST elevation | Strongly positive finding - ST elevation. |
122757 | ST Depression - Horizontal | Finding of ST segment depression with no slope. |
122758 | ST Depression - Upsloping | Finding of ST segment depression with upslope. |
122759 | ST Depression - Downsloping | Finding of ST segment depression with downslope. |
122760 | Stress test score | Stress test score. |
122762 | Number of diseased vessel territories | Number of diseased vessel territories. |
122764 | Weight exceeds equipment limit | Patient weight exceeds equipment limit. |
122768 | Difference in Ejection Fraction | Difference in Ejection Fraction. |
122769 | Difference in ED LV Volume | Difference in End Diastolic Left Ventricular Volume. |
122770 | Ratio of achieved to predicted maximal oxygen consumption | Ratio of achieved to predicted maximal oxygen consumption. |
122771 | Ratio of achieved to predicted functional capacity | Ratio of achieved to predicted functional capacity. |
122772 | Aerobic index | Workload (Watts) at target heart rate divided by body weight. |
122773 | ST/HR Index | ST depression at peak exercise divided by the exercise-induced increase in heart rate [Kligfield P, Ameisen O, Okin PM. "Heart rate adjustment of ST segment depression for improved detection of coronary artery disease." Circulation 1989;79:245-55.]. |
122775 | Agreement with prior findings | Agreement with prior findings. |
122776 | Disagreement with prior findings | Disagreement with prior findings. |
122781 | Rest thallium/stress technetium procedure | Nuclear Medicine Rest thallium/stress technetium procedure. |
122782 | Rest technetium/stress technetium 1 day procedure | Nuclear Medicine Rest technetium/stress technetium 1 day procedure. |
122783 | Rest technetium/stress technetium 2 day procedure | Nuclear Medicine Rest technetium/stress technetium 2 day procedure. |
122784 | Stress technetium/rest technetium 1 day procedure | Nuclear Medicine Stress technetium/rest technetium 1 day procedure. |
122785 | NM Myocardial Viability procedure | Nuclear Medicine Myocardial Viability procedure. |
122791 | PET Myocardial Perfusion, Rest only | Positron Emission Tomography Perfusion Imaging procedure, rest only. |
122792 | PET Myocardial Perfusion, Stress only | Positron Emission Tomography Perfusion Imaging procedure, stress only. |
122793 | PET Myocardial Perfusion, Rest and Stress | Positron Emission Tomography Perfusion Imaging procedure, rest and stress. |
122795 | PET Myocardial Viability, Rest only | Positron Emission Tomography Myocardial Viability procedure, rest only. |
122796 | PET Myocardial Viability, Stress only | Positron Emission Tomography Myocardial Viability procedure, stress only. |
122797 | PET Myocardial Viability, Rest and Stress | Positron Emission Tomography Myocardial Viability procedure, rest and stress. |
122799 | Anginal Equivalent | Group of symptoms heralding angina pectoris that does not include chest pain (dyspnea, diaphoresis, profuse vomiting in a diabetic patient, or arm or jaw pain). |
123001 | Radiopharmaceutical | Active ingredient (molecular) used for radioactive tracing. | Retired. Replaced by (F-61FDB, SRT, "Radiopharmaceutical agent").
123003 | Radiopharmaceutical Start Time | Time of radiopharmaceutical administration to the patient for imaging purposes. |
123004 | Radiopharmaceutical Stop Time | Ending time of radiopharmaceutical administration to the patient for imaging purposes. |
123005 | Radiopharmaceutical Volume | Volume of radiopharmaceutical administered to the patient. |
123006 | Radionuclide Total Dose | Total amount of radionuclide administered to the patient at Radiopharmaceutical Start Time. |
123007 | Radiopharmaceutical Specific Activity | Activity per unit mass of the radiopharmaceutical at Radiopharmaceutical Start Time. |
123009 | Radionuclide Syringe Counts | Pre-injection syringe acquisition count rate. |
123010 | Radionuclide Residual Syringe Counts | Syringe acquisition count rate following patient injection. |
123011 | Contrast/Bolus Agent | Contrast or bolus agent. |
123012 | Pre-Medication | Medication to be administered at the beginning of the Scheduled Procedure Step. |
123014 | Target Region | Anatomic Region to be imaged. |
123015 | Imaging Direction | Direction of imaging (includes view, transducer orientation, patient orientation, and/or projection). |
123016 | Imaging Conditions | Imaging condition for refinement of protocol (includes secondary posture, instruction, X-Ray / electron beam energy or nuclide, and ultrasound modes), as used in JJ1017 v3.0. |
123019 | Caudal 10 degree distal-cranioproximal oblique | Caudal 10 degree distal-cranioproximal oblique radiographic projection, defined per Smallwood et al. |
123101 | Neighborhood Analysis | Surface processing utilizing predefined weighting factors (i.e., kernels) applied to different data values depending on their location relative to other data values within the data domain. Includes Low Pass, High Pass, Gaussian, Laplacian, etc. |
123102 | Adaptive Filtering | Surface processing applied non-uniformly utilizing a priori knowledge of the system and/or relative locations of the data values within the data domain. Example: Neighborhood analysis where weighting factors are modified continuously based on predefined criteria. |
123103 | Edge Detection | Surface processing through the exploitation of discontinuities in the data values within their domain. Includes Gradient filters. |
123104 | Morphological Operations | Surface processing based on the connectivity of values based on the shape or structure of the data values within their domain. Includes erode, dilate, etc. |
123105 | Histogram Analysis | Surface processing applied to the distribution of the data values. Includes thresholding, Bayesian Classification, etc. |
123106 | Multi-Scale/Resolution Filtering | Surface processing accomplished through varying the data domain size. Include deformable models. |
123107 | Cluster Analysis | Surface processing accomplished by combining data values based on their relative location within their domain or value distribution. Includes K- and C-means, Fuzzy Analysis, Watershed, Seed Growing, etc. |
123108 | Multispectral Processing | Surface processing accomplished through the weighted combination of multiple data sets. Includes Principle Component Analysis, linear and non-linear weighed combinations, etc. |
123109 | Manual Processing | Surface processing accomplished through human interaction. Region drawing. |
123110 | Artificial Intelligence | Surface processing using Artificial Intelligence techniques, such as Machine Learning, Neural Networks, etc. |
123111 | Deformable Models | Surface processing using Deformable Model techniques, such as Point Distribution Models, Level Sets, Simplex Meshes, etc. |
125000 | OB-GYN Ultrasound Procedure Report | Document Title of OB-GYN procedure report. |
125001 | Fetal Biometry Ratios | Report section for assessment of fetal growth using ratios and indexes. |
125002 | Fetal Biometry | Report section for assessment of fetal growth. |
125003 | Fetal Long Bones | Report section for assessment of fetal growth by long bone measurements. |
125004 | Fetal Cranium | Report section for assessment of fetal cranium growth. |
125005 | Biometry Group | Biometric assessment of. |
125006 | Biophysical Profile | Report section for assessment of biophysical observations that evaluate fetal well-being according to Manning, Antepartum Fetal Evaluation: Development of a Fetal Biophysical Profile Score, Am. J Obstet Gynecol, 1980;136:787. |
125007 | Measurement Group | A grouping of related measurements and calculations that share a common context. |
125008 | Fetus Summary | Report section for fetus specific procedure summary observations. |
125009 | Early Gestation | Report section for assessment of early gestation fetus. |
125010 | Identifier | A name to differentiate between multiple instances of some item. |
125011 | Pelvis and Uterus | Report section for assessment of pelvis and uterus. |
125012 | Growth Percentile rank | The rank of a measured growth indicator relative to a normal distribution expressed as a percentage. |
125013 | Growth Z-score | The rank of a measured growth indicator relative to a normal distribution expressed as the dimensionless quantity z = (x-m) /s where (x-m) is the deviation of the value x, from the distribution mean, m, and s is the standard deviation of the distribution. |
125015 | Fetus Characteristics | Fetus characteristics (findings section title). |
125016 | Fetal Measurements | Fetal Measurements (findings section title). |
125021 | Frame of Reference Identity | There is a defined equivalence between the Frame of Reference of the Registration SOP instance and the Frame of Reference of the referenced images. |
125022 | Fiducial Alignment | The registration is based on fiducials that represent patient or specimen features identified in each data set. |
125023 | Acquisition Equipment Alignment | Registration based on a-priori knowledge of the acquisition geometry. This is not an object registration as in fiducial registration. Rather, it specifies a known spatial relationship. |
125024 | Image Content-based Alignment | Computed registration based on global image information. |
125025 | Visual Alignment | Registration by visually guided manipulation. |
125030 | Inter-Hemispheric Plane | A plane fiducial that specifies the location of the plane separating the two hemispheres of the brain. |
125031 | Right Hemisphere Most Anterior | A point fiducial that specifies the location in the plane perpendicular to the Anterior- Posterior-Commissure axis and tangential to the anterior limit of the right brain hemisphere. |
125032 | Right Hemisphere Most Posterior | A point fiducial that specifies the location in the plane perpendicular to the Anterior- Posterior-Commissure axis and tangential to the posterior limit of the right brain hemisphere. |
125033 | Right Hemisphere Most Superior | A point fiducial that specifies the location in the plane perpendicular to the Anterior- Posterior-Commissure axis and tangential to the superior limit of the right brain hemisphere. |
125034 | Right Hemisphere Most Inferior | A point fiducial that specifies the location in the plane perpendicular to the Anterior- Posterior-Commissure axis and tangential to the inferior limit of the Right brain hemisphere. |
125035 | Left Hemisphere Most Anterior | A point fiducial that specifies the location in the plane perpendicular to the Anterior- Posterior-Commissure axis and tangential to the anterior limit of the left brain hemisphere. |
125036 | Left Hemisphere Most Posterior | A point fiducial that specifies the location in the plane perpendicular to the Anterior- Posterior-Commissure axis and tangential to the posterior limit of the left brain hemisphere. |
125037 | Left Hemisphere Most Superior | A point fiducial that specifies the location in the plane perpendicular to the Anterior- Posterior-Commissure axis and tangential to the superior limit of the left brain hemisphere. |
125038 | Left Hemisphere Most Inferior | A point fiducial that specifies the location in the plane perpendicular to the Anterior- Posterior-Commissure axis and tangential to the inferior limit of the left brain hemisphere. |
125040 | Background | That which is not part of an object. | E.g., background of an image (that which might be encoded with Pixel Padding Value, or a Segmentation Property Type.
125041 | Registration Input | A segment for use as an input to an image registration process. E.g., to specify the bounding region for determining a Frame of Reference Transformation Matrix. |
125100 | Vascular Ultrasound Procedure Report | Root Document Title for ultrasound vascular evidence reports (worksheets). |
125101 | Vessel Branch | The particular vessel branch, such as the inferior, medial or lateral. |
125102 | Graft Type | A descriptor or elaboration of the type of graft. |
125105 | Measurement Orientation | A modifier to a 2D distance measurement to describe its orientation. E.g., a vascular distance measurement for a vessel plague could have a modifier Transverse or Longitudinal. |
125106 | Doppler Angle | The angle formed between the Doppler beam line and the direction of blood flow within a region of interest in the body defined by the sample volume. |
125107 | Sample Volume Depth | The depth of the center of the Doppler sample volume measured from skin line along the Doppler line. |
125195 | Pediatric Cardiac Ultrasound Report | Pediatric Cardiac Ultrasound Report (document title). |
125196 | Fetal Cardiac Ultrasound Report | Fetal Cardiac Ultrasound Report (document title). |
125197 | Adult Congenital Cardiac Ultrasound Report | Adult Congenital Cardiac Ultrasound Report (document title). |
125200 | Adult Echocardiography Procedure Report | Document title of adult echocardiography procedure (evidence) report. |
125201 | Illustration of Finding | An image that is a pictorial representation of findings. The concept is typically used as a purpose of reference to an image, such as a depiction of myocardium segments depicting wall motion function. |
125202 | LV Wall Motion Score Index | The average of all scored (non-zero) Left Ventricle segment wall motion scores. |
125203 | Acquisition Protocol | A type of clinical acquisition protocol for creating images or image-derived measurements. Acquisition protocols may be specific to a manufacturer's product. |
125204 | Area-length biplane | Method for calculating left ventricular volume from two orthogonal views containing the true long axis (usually the apical 4 and 2 chamber views). Volume = [pL1 / 6] * [(4A1) ¸ (pL1) ] * [(4A2) ¸ (pL2) ]. |
125205 | Area-Length Single Plane | Method for calculating left ventricular volume from a view containing the true long axis (usually the apical 4-chamber view). Volume = [8(A)2]¸[3pL]. |
125206 | Cube | Method (formula) for calculating left ventricle volumes and function derivatives (EF, SV, SI, etc.) that estimates the volume as the cube of diameter. |
125207 | Method of Disks, Biplane | Method of calculating volume based on the summation of disk volumes. The disk axis is parallel to the left ventricular long axis and using a disk diameter averaged from the two chamber and four chamber views. |
125208 | Method of Disks, Single Plane | Method of calculating volume based on the summation of disk volumes. The disk axis is parallel to the left ventricular long axis with disk diameter taken from the four-chamber view. |
125209 | Teichholz | Method (formula) for calculating left ventricle volumes and function derivatives (EF, SV, SI, etc.) Volume = [7.0/(2.4+D) ]*D3. |
125210 | Area by Pressure Half-Time | Mitral valve area (cm2) by Pressure Half-time = 220 (cm2.ms) / PHT (ms). |
125211 | Biplane Ellipse | Area = P/4 X d1 X d2 d1 = anterior/posterior axis d2 = medial/lateral axis Hagen-Ansert, Sandra L., Textbook of Diagnostic Ultrasound, ed. 3, The C.V.Mosby Co., 1989, p. 73. . |
125212 | Continuity Equation | For conduits in series ("in continuity"), volume flow is equal: A1*V1 = A2*V2. where V is the velocity. |
125213 | Continuity Equation by Mean Velocity | For conduits in series ("in continuity"), volume flow is equal: A1*V1 = A2*V2. where V is the mean velocity. |
125214 | Continuity Equation by Peak Velocity | For conduits in series ("in continuity"), volume flow is equal: A1*V1 = A2*V2. where V is the peak velocity. |
125215 | Continuity Equation by Velocity Time Integral | For conduits in series ("in continuity"), volume flow is equal: A1*V1 = A2*V2. where V is the velocity time integral. |
125216 | Proximal Isovelocity Surface Area | Utilizes aliasing velocity (by color Doppler) of flow into an orifice (often regurgitant or stenotic) to measure instantaneous flow rate, orifice area, and flow volume. The instantaneous flow rate = (2πr2vav ) * (απ) where vav is the constant velocity known as aliasing velocity at radius r, vp is the peak velocity at the orifice, and α is the angle in radians of the constant velocity surface. Estimated Orifice area = Flow rate / vp , where vp is the peak velocity at the orifice and the flow rate is the PISA peak flow rate. The volume flow is then the product of the orifice area and Velocity Time Integral. |
125217 | Full Bernoulli | ΔP = 4*(V12 - V22). |
125218 | Simplified Bernoulli | ΔP = 4*V2. |
125219 | Doppler Volume Flow | Volume flow = Conduit CSA * (Velocity Time Integral). |
125220 | Planimetry | Direct measurement of an area by tracing an irregular perimeter. |
125221 | Left Ventricle Mass by M-mode | Mass = 1.04 * [(ST+LVID+PWT)3 - LVID3] * 0.8+ 0.6. Mass unit is grams and length in cm. |
125222 | Left Ventricle Mass by Truncated Ellipse | Mass = 1.05P ((b + t)2 X (2/3 (a + t) + d - d3 /3(a + t)2) - b2 (2/3a + d - d3 /3a2)) a = Semi-major axis from widest minor axis radius to apex. b = Short axis radius calculated from short axis cavity area t = Myocardial thickness calculated from short axis epicardial and cavity areas d = Truncated semi-major axis from widest short axis diameter to plane of mitral annulus. Mass unit is grams and length in cm. Schiller NB et al: Recommendations for quantification of the left ventricle by two-dimensional echocardiography, American Society of Echocardiography 2:364, 1989. . |
125223 | 4 Point Segment Finding Scale | A four point, echocardiographic numeric scoring scheme of myocardium segments based on evaluation of wall motion and ventricle morphology. Recommendations for Quantitation of the Left Ventricle by Two-Dimensional Echocardiography, Journal of the American Society of Echocardiography, 2:358-367, 1989. |
125224 | 5 Point Segment Finding Scale | A five point, echocardiographic numeric scoring scheme of myocardium segments based on evaluation of wall motion and ventricle morphology. Recommendations for Quantitation of the Left Ventricle by Two-Dimensional Echocardiography, Journal of the American Society of Echocardiography, 2:358-367, 1989. |
125225 | 5 Point Segment Finding Scale With Graded Hypokinesis | A five point, echocardiographic numeric scoring scheme of myocardium segments based on evaluation of wall motion and ventricle morphology, with severity of hypokinesis graded. Recommendations for Quantitation of the Left Ventricle by Two-Dimensional Echocardiography, Journal of the American Society of Echocardiography, 2:358-367, 1989. |
125226 | Single Plane Ellipse | Method of estimating volume from a planar ellipse. Equivalent to Biplane Ellipse with an assumption that the ellipse in the orthogonal plane has identical major and minor diameters. |
125227 | Modified Simpson | Modified Simpson's Method of estimating ventricular volume, based on the method of disks with paired apical views. Schiller NB, et al. "Recommendations for quantitation of the left ventricle by two-dimensional echocardiography. American Society of Echocardiography Committee on Standards, Subcommittee on Quantitation of Two-Dimensional Echocardiograms". J Am Soc Echocardiogr.1989 2(5):358-367. Sep-Oct. |
125228 | Bullet Method | Bullet method of estimating ventricular volume. Volume = 5/6 * L * S L: Left ventricle long axis length S: Left ventricle area, SAX view at level of Mitral Valve. |
125230 | Power Doppler | Color coded ultrasound images of blood flow, which depict the amplitude, or power, of Doppler signals. | Retired Replaced by (P0-02241, SRT, "Power Doppler")
125231 | 3D mode | Volumetric ultrasound imaging | Retired Replaced by (P0-02242, SRT, "3D mode")
125233 | Start of drug dose administration | Onset of administration of dose of a drug. |
125234 | Start of contrast agent administration | Onset of contrast agent administration. |
125235 | Destruction of microbubbles | Destruction of ultrasonic contrast microbubbles by a high-energy ultrasound pulse. |
125236 | Onset of exercise | Instant at which exercise begins. |
125237 | Cessation of exercise | Instant at which exercise ends. |
125238 | Onset of stimulation | Instant at which stimulation begins. |
125239 | Cessation of stimulation | Instant at which stimulation ends. |
125240 | Line scan pattern | Ultrasound transducer scan pattern in which information is gathered along a line. |
125241 | Plane scan pattern | Ultrasound transducer scan pattern in which information is gathered within a plane. |
125242 | Volume scan pattern | Ultrasound transducer scan pattern in which information is gathered within a volume. |
125251 | Non-imaging Doppler ultrasound transducer geometry | Ultrasound transducer geometry characterized by a single scan line used for PW or CW Doppler scanning. |
125252 | Linear ultrasound transducer geometry | Ultrasonic transducer geometry characterized by parallel lines. |
125253 | Curved linear ultrasound transducer geometry | Ultrasonic transducer geometry characterized by radial lines normal to the outside of a curved surface. |
125254 | Sector ultrasound transducer geometry | Ultrasonic transducer geometry characterized by lines originating from a common apex. |
125255 | Radial ultrasound transducer geometry | Ultrasonic transducer geometry characterized by lines emanating radially from a single point. |
125256 | Ring ultrasound transducer geometry | Ultrasonic transducer geometry characterized by a circular ring of transducer elements. |
125257 | Fixed beam direction | Ultrasonic steering technique consisting of a single beam normal to the transducer face steered by the orientation of the probe. |
125258 | Mechanical beam steering | Ultrasonic steering technique consisting of mechanically directing the beam. |
125259 | Phased beam steering | Ultrasonic steering technique consisting of electronically-steered beams. |
125261 | External Transducer | Transducer is designed to be placed onto the surface of the subject. |
125262 | Transesophageal Transducer | Transducer is designed for insertion into the esophagus. |
125263 | Endovaginal Transducer | Transducer is designed for insertion into the vagina. |
125264 | Endorectal Transducer | Transducer is designed for insertion into the rectum. |
125265 | Intravascular Transducer | Transducer is designed for insertion via a catheter. |
125270 | Left Ventricle Mass by Area Length | method to measure the mass of the Left Ventricle via the ASE area-length method at end diastole. LV Mass = 1.05*(5/6*(A1*(L+t)) - 5/6*(A2*L)) A1 = Left Ventricle epicardial SAX area at the level of the papillary muscle tips at end diastole. A2 = Left Ventricle endocardial SAX area cavity area at the level of the papillary muscle tips at end diastole. L = Left Ventricle apical view long axis length at end diastole. t = Myocardial thickness can be computed as: t = sqrt (A1/3.14) - sqrt (A2/3.14) Reference: 1) Schiller, N.B., et al. "Recommendations for Quantification of the LV by Two-dimensional Echocardiography." J Am Soc Echo, Vol. 2, No. 5: 358-367, Sep-Oct 1989. 2) Reichek, N., et al. "Anatomic Validation of Left Ventricular Mass Estimates from Clinical Two-dimensional Echocardiography: Initial Results." Circulation, Vol. 67, No. 2: 348-52, February 1983. |
125271 | Left Ventricle Mass by M-mode - adjusted by Height | Equation = Left Ventricle Mass by M-mode (in gram) / (Height (in meter)) ^2.7 Reference: Giovanni De Simone, et al. "Effect of Growth on Variability of Left Ventricular Mass: Assessment of Allometric Signals in Adults and Children and Their Capacity to Predict Cardiovascular Risk". New York, New York and Cincinnati, Ohio. |
125272 | Left Ventricle Mass by Truncated Ellipse - adjusted by Height | Equation = Left Ventricle Mass by Truncated Ellipse / Height^2.7 Reference: Giovanni De Simone, et al. "Effect of Growth on Variability of Left Ventricular Mass: Assessment of Allometric Signals in Adults and Children and Their Capacity to Predict Cardiovascular Risk". New York, New York and Cincinnati, Ohio. |
125273 | Left Ventricle Mass by Area Length - adjusted by Height | Equation = Left Ventricle Mass by Area Length / Height^2.7 Reference: Giovanni De Simone, et al. "Effect of Growth on Variability of Left Ventricular Mass: Assessment of Allometric Signals in Adults and Children and Their Capacity to Predict Cardiovascular Risk". New York, New York and Cincinnati, Ohio. |
125901 | CARDIOsphere | CARDIOsphere™ ultrasonic contrast agent produced by POINT Biomedical. |
125902 | Echovist | Echovist® ultrasonic contrast agent produced by Schering AG. |
125903 | Imagify | Imagify™ ultrasonic contrast agent produced by Accusphere Inc. |
125904 | Levovist | Levovist® ultrasonic contrast agent produced by Schering AG. |
125905 | Sonazoid | Sonazoid™ ultrasonic contrast agent produced by Daiichi Pharmaceutical / General Electric. |
125906 | SonoVue | SonoVue™ ultrasonic contrast agent produced by Bracco Diagnostics. |
125907 | Targestar-B | Targestar™-B ultrasonic contrast agent produced by Targeson LLC. |
125908 | Targestar-P | Targestar™-P ultrasonic contrast agent produced by Targeson LLC. |
126000 | Imaging Measurement Report | A structured report containing the quantitative results of human or machine analysis of images. |
126001 | Oncology Measurement Report | A structured report containing the quantitative results of human or machine analysis of images for oncology evaluation. |
126002 | Dynamic Contrast MR Measurement Report | A structured report containing the quantitative results of human or machine analysis of DCE-MR. |
126003 | PET Measurement Report | A structured report containing the quantitative results of human or machine analysis of PET images. |
126010 | Imaging Measurements | Measurements made on images |
126011 | Derived Imaging Measurements | Measurements derived from measurements made on images. |
126020 | Multiparametric MRI | An MRI procedure in which multiple parameters including diffusion, dynamic contrast and T2 are measured. |
126021 | Multiparametric MRI of prostate | An MRI procedure of the prostate in which multiple parameters including diffusion, dynamic contrast and T2 are measured. |
126022 | Multiparametric MRI of whole body | An MRI procedure of the whole body in which multiple parameters including diffusion, dynamic contrast and T2 are measured. |
126030 | Sum of segmented voxel volumes | The volume derived by summing the volumes of all the voxels (and partial voxels if the segment contains partially occupied voxels) included in the segment |
126031 | Peak Value Within ROI | Maximum average gray value that is calculated from a 1 cubic centimeter sphere placed within the region of interest. Reference Wahl et al PERCIST article. |
126032 | Metabolic Volume | The volume of a lesion (e.g., a tumor) ascertained through information about its metabolic activity (e.g., SUV on PET). Abbreviated "MV". Synonymous with Metabolic Tumor Volume (MTV). |
126033 | Total Lesion Glycolysis | The total activity of a lesion obtained as the product of its volume and its glycolytic activity (on FDG-PET). The volume may be defined on the same modality (e.g., the MV on FDG-PET by some thresholding or other technique) or on another spatially registered modality (e.g., the lesion outline segmented on CT or MR). Does not apply to other radiopharmaceuticals than those involved in glucose metabolism. Abbreviated TLG. Synonymnous with "Tumor Lesion Glycolysis". |
126034 | Glycolysis | The amount glycolytic activity summed across all voxels in a defined region or within a defined range of SUV (on FDG-PET). |
126035 | Total Lesion Proliferation | The total activity of a lesion obtained as the product of its volume and its proliferative activity (on FLT-PET). The volume may be defined on the same modality (e.g., the MV on FDG-PET by some thresholding or other technique) or on another spatially registered modality (e.g., the lesion outline segmented on CT or MR). Does not apply to other radiopharmaceuticals than those involved in cellular proliferation. Abbreviated TLP. Synonymnous with "Tumor Lesion Proliferation". |
126036 | Proliferative Activity | The amount proliferative activity summed across all voxels in a defined region or within a defined range of SUV (on FLT-PET). |
126037 | Standardized Added Metabolic Activity (SAM) | A background-corrected, partial volume independent version of TLG. SAM is calculated by drawing a volume of interest (VOI1) around the tumour and a larger VOI (VOI2) around VOI1. Subtracting the background activity in VOI2-VOI1 from VOI1 yields SAM. See Mertens et al. "Standardized added metabolic activity (SAM): a partial volume independent marker of total lesion glycolysis in liver metastases". Eur J Nucl Med Mol Imaging (2012) 39:1441–1448. |
126038 | Standardized Added Metabolic Activity (SAM) Background | The background value (VOI2-VOI1) used to calculate Standardized Added Metabolic Activity (SAM). SAM is calculated by drawing a volume of interest (VOI1) around the tumour and a larger VOI (VOI2) around VOI1. Subtracting the background activity in VOI2-VOI1 from VOI1 yields SAM. See Mertens et al. "Standardized added metabolic activity (SAM): a partial volume independent marker of total lesion glycolysis in liver metastases". Eur J Nucl Med Mol Imaging (2012) 39:1441–1448. |
126039 | Lesion to Background SUV Ratio | The ratio of the SUV within a tumor to the SUV of a pre-defined background region. Need reference(s). A more general concept than Tumor to Background Ratio (TBR). |
126040 | Background for Lesion to Background SUV Ratio | The SUV of a pre-defined background region used to compute Lesion to Background SUV Ratio. Need reference(s). |
126050 | Fractal Dimension | A statistical index of complexity comparing how detail in a fractal pattern changes with the scale at which it is measured; a ratio of the change in detail to the change in scale. |
126051 | Skewness | Measure of the asymmetry of the probability distribution of a real-valued random variable about its mean. |
126052 | Kurtosis | Measure of the peakedness of the probability distribution of a real-valued random variable. |
126060 | Entropy of GLCM | The zero order entropy of a Gray Level Co-occurrence Matrix (GLCM). A measure of disorder. See http://www.fp.ucalgary.ca/mhallbey/equations.htm. |
126061 | Energy of GLCM | The energy (uniformity) (square root of the Angular Second Moment (ASM)) of a Gray Level Co-occurrence Matrix (GLCM). A measure of orderliness. See http://www.fp.ucalgary.ca/mhallbey/equations.htm. |
126062 | Homogeneity of GLCM | The Inverse Difference Moment of a Gray Level Co-occurrence Matrix (GLCM). See http://www.fp.ucalgary.ca/mhallbey/equations.htm. |
126063 | Contrast of GLCM | The sum of squares variance of a Gray Level Co-occurrence Matrix (GLCM). See http://www.fp.ucalgary.ca/mhallbey/equations.htm. |
126064 | Dissimilarity of GLCM | The dissimilarity of a Gray Level Co-occurrence Matrix (GLCM). See http://www.fp.ucalgary.ca/mhallbey/equations.htm. |
126065 | ASM of GLCM | The Angular Second Moment of a Gray Level Co-occurrence Matrix (GLCM). See http://www.fp.ucalgary.ca/mhallbey/equations.htm. |
126066 | Correlation of GLCM | A measure of the linear dependency of grey levels on those of neighbouring pixels of a Gray Level Co-occurrence Matrix (GLCM). See http://www.fp.ucalgary.ca/mhallbey/equations.htm. |
126067 | Gray Level Co-occurrence Matrix (GLCM) | A tabulation of how often different combinations of pixel values (grey levels) occur in an image. See http://www.fp.ucalgary.ca/mhallbey/the_glcm.htm. |
126070 | Subject Time Point Identifier | An identifier of a specific time point in a continuum, which is unique within an appropriate local context (such as an entire organization, system or treatment protocol), which identifies the time point for a specific patient. |
126071 | Protocol Time Point Identifier | An identifier of a specific time point in a continuum, which is unique within an appropriate local context (such as an entire organization, system or treatment protocol), which identifies the time point “slot” within a treatment protocol using the same value for all patients in the protocol. |
126072 | Time Point Type | A pre-defined type of a specific time point in a continuum. |
126073 | Time Point Order | A number indicating the order of a time point relative to other time points in the same continuum. |
126074 | Posttreatment | The time after the treatment of interest. |
126075 | Eligibility | For the purpose of determining eligibility for a protocol. | Similar but not identical to (21954-3, LN, "Protocol eligibility status Cancer"), since not constrained to cancer, etc.
126080 | RECIST 1.0 | Response Evaluation Criteria in Solid Tumors version 1.0. See [RECIST] in Normative References. | More specific than (112022, DCM, "RECIST") or (C1709926, UMLS, "RECIST") or (C49164, NCIt, "RECIST") in that a specific version is specified.
126081 | RECIST 1.1 | Response Evaluation Criteria in Solid Tumors Version 1.1. See Eisenhauer et al. "New Response Evaluation Criteria in Solid Tumours: Revised RECIST Guideline (version 1.1)." European Journal of Cancer 45, no. 2 (n.d.): 228–47. doi:10.1016/j.ejca.2008.10.026.. | More specific than (112022, DCM, "RECIST") or (C1709926, UMLS, "RECIST") or (C49164, NCIt, "RECIST") in that a specific version is specified.
126100 | Real World Value Map used for measurement | A reference to the Real World Value Map applied to the stored image pixel values before their use for a measurement |
126200 | Image Library Group | A container that groups common information about a set of images used as evidence to produce a report. |
126201 | Acquisition Date | The date the acquisition of data started |
126202 | Acquisition Time | The time the acquisition of data started |
126203 | PET Radionuclide Incubation Time | The time between the start of injection of the PET radionuclide and the start of acquisition of the PET data. |
126220 | R2-Coefficient | Coefficient of determination, R2. An indication of goodness of fit. |
126300 | Perfusion analysis by Stable Xenon CT technique | Perfusion analysis by Stable Xenon CT technique |
126301 | Perfusion analysis by IV Iodinated Contrast CT technique | Perfusion analysis by IV Iodinated Contrast CT technique |
126302 | Perfusion analysis by Arterial Spin Labeling MR technique | Perfusion analysis by Arterial Spin Labeling (ASL) MR technique |
126303 | Perfusion analysis by Susceptibility MR technique | Perfusion analysis by Susceptibility (T2*) MR technique |
126310 | Least Mean Square (LMS) deconvolution | Least Mean Square (LMS) deconvolution |
126311 | Singular Value Decomposition (SVD) deconvolution | Singular Value Decomposition (SVD) deconvolution |
126312 | Ktrans | Ktrans, the volume transfer constant of a tracer diffusion kinetic model, specifically the volume transfer constant between blood plasma and extravascular extracellular space (EES) See Tofts et al, "Estimating Kinetic Parameters From Dynamic Contrast-Enhanced T1-Weighted MRI of a Diffusable Tracer: Standardized Quantities and Symbols", Journal of Magnetic Resonance Imaging, vol. 10, pp. 223–232, 1999. |
126313 | kep | kep, the rate constant between extravascular extracellular space (EES) and blood plasma See Tofts et al, "Estimating Kinetic Parameters From Dynamic Contrast-Enhanced T1-Weighted MRI of a Diffusable Tracer: Standardized Quantities and Symbols", Journal of Magnetic Resonance Imaging, vol. 10, pp. 223–232, 1999. |
126314 | ve | ve, the fractional (not absolute) volume of extravascular extracellular space (EES) per unit volume of tissue See Tofts et al, "Estimating Kinetic Parameters From Dynamic Contrast-Enhanced T1-Weighted MRI of a Diffusable Tracer: Standardized Quantities and Symbols", Journal of Magnetic Resonance Imaging, vol. 10, pp. 223–232, 1999. |
126320 | IAUC | The intial area under the contrast agent concentration–time curve |
126321 | IAUC60 | The intial area under the contrast agent concentration–time curve at 60 seconds after the onset time |
126322 | IAUC90 | The intial area under the contrast agent concentration–time curve at 90 seconds after the onset time |
126330 | tau_m | τm. The mean intracellular water lifetime (τi). Used in the Shutter-Speed Model (SSM) of tracer kinetics. |
126331 | vp | vp. The fractional (not absolute) blood plasma volume per unit volume of tissue. See Tofts et al, "Estimating Kinetic Parameters From Dynamic Contrast-Enhanced T1-Weighted MRI of a Diffusable Tracer: Standardized Quantities and Symbols", Journal of Magnetic Resonance Imaging, vol. 10, pp. 223–232, 1999. |
126340 | Standard Tofts Model | A tracer diffusion kinetic model in which the permeability is assumed to be isodirectional. See P. Tofts, "Modeling tracer kinetics in dynamic Gd-DTPA MR imaging", Journal of Magnetic Resonance Imaging, vol. 7, pp. 91–101, 1997. |
126341 | Extended Tofts Model | A tracer diffusion kinetic model in which the permeability is not assumed to be isodirectional, and which includes the contribution of tracer in the blood plasma to the total tissue concentration. See P. Tofts, "Modeling tracer kinetics in dynamic Gd-DTPA MR imaging", Journal of Magnetic Resonance Imaging, vol. 7, pp. 91–101, 1997. |
126342 | Model-free concentration-time quantitification | A semiquantitative analysis of the contrast-enhancement concentration versus time curve that avoids the use of a pharmacokinetic model. E.g., integration to compute the initial area under the curve. |
126343 | First Pass Leakage Profile (FPLP) | A tracer diffusion kinetic model that accounts for the tumor leakage profile during the first pass of contrast. See Li, Ka-Loh, Xiao Ping Zhu, John Waterton, and Alan Jackson. "Improved 3D Quantitative Mapping of Blood Volume and Endothelial Permeability in Brain Tumors." Journal of Magnetic Resonance Imaging 12, no. 2 (2000): 347–357. doi:10.1002/1522-2586(200008)12:2<347::AID-JMRI19>3.0.CO;2-7. |
126344 | Shutter-Speed Model (SSM) | A tracer diffusion kinetic model that does not assume that intercompartmental water molecule exchange is infinitely fast. See Li, Xin, Wei Huang, Thomas E. Yankeelov, Alina Tudorica, William D. Rooney, and Charles S. Springer. “Shutter-Speed Analysis of Contrast Reagent Bolus-Tracking Data: Preliminary Observations in Benign and Malignant Breast Disease.” Magnetic Resonance in Medicine 53, no. 3 (2005): 724–29. doi:10.1002/mrm.20405. |
126350 | T1 by Multiple Flip Angles | T1 measurement by Multiple Flip Angles (MFA) (variable saturation) method |
126351 | T1 by Inversion Recovery | T1 measurement by Inversion Recovery (IR) method |
126352 | T1 by Fixed Value | Calculation was performed using a fixed value of T1 rather than a measured value. The value could be encoded as the value of (126353, DCM, "T1 Used For Calculation"). |
126353 | T1 Used For Calculation | The fixed value of T1 used for a calculation. |
126360 | AIF Ignored | No Arterial Input Function was used. |
126361 | Population Averaged AIF | A population-averaged Arterial Input Function. |
126362 | User-defined AIF ROI | An Arterial Input Function computed from a user-defined Region of Interest. |
126363 | Automatically Detected AIF ROI | An Arterial Input Function computed from an automatically detected Region of Interest. |
126364 | Blind Estimation of AIF | A data-driven blind source separation (BSS) algorithm that estimates AIF from individuals without any presumed AIF model and initialization. See Lin, Yu-Chun, Tsung-Han Chan, Chong-Yung Chi, Shu-Hang Ng, Hao-Li Liu, Kuo-Chen Wei, Yau-Yau Wai, Chun-Chieh Wang, and Jiun-Jie Wang. "Blind Estimation of the Arterial Input Function in Dynamic Contrast-Enhanced MRI Using Purity Maximization." Magnetic Resonance in Medicine 68, no. 5 (November 1, 2012): 1439–49. doi:10.1002/mrm.24144. |
126370 | Time of Peak Concentration | The time at which the concentration-time curve achieves its peak for the first time. Used as a concept name for a value or as a method. E.g., used as a method of calculation for BAT. See Shpilfoygel Med Phys 2008. doi: 10.1118/1.1288669 |
126371 | Bolus Arrival Time | The nominal time at which arrival of a contrast bolus is detected, which is used as a reference point for subsequent calculations. Used as a concept name for a value or as a method. No specific computational method is implied by this general definition. Abbreviated BAT. |
126372 | Time of Leading Half-Peak Concentration | The time at which the concentration-time curve achieves half of its peak density for the first time. Used as a concept name for a value or as a method. E.g., used as a method of calculation for BAT. See Shpilfoygel Med Phys 2008. doi: 10.1118/1.1288669 |
126373 | Temporal Derivative Exceeds Threshold | A method of determining BAT that involves computing the temporal derivative of the concentration-time curve and selecting the time when the temporal derivative exceeds a specified threshold. See Shpilfoygel Med Phys 2008. doi: 10.1118/1.1288669 |
126374 | Temporal Derivative Threshold | A threshold applied to the temporal derivative of the concentration-time curve. E.g., used to establish BAT. See Shpilfoygel Med Phys 2008. doi: 10.1118/1.1288669 |
126375 | Maximum Slope | The maximum rate of signal intensity change within a measured region of a time-activity curve. See Boonsirikamchai, Piyaporn, Harmeet Kaur, Deborah A. Kuban, Edward Jackson, Ping Hou, and Haesun Choi. “Use of Maximum Slope Images Generated From Dynamic Contrast-Enhanced MRI to Detect Locally Recurrent Prostate Carcinoma After Prostatectomy: A Practical Approach.” American Journal of Roentgenology 198, no. 3 (March 1, 2012): W228–W236. doi:10.2214/AJR.10.6387. |
126376 | Maximum Difference | The maximum degree of signal intensity change within a measured region of a time-activity curve. See Boonsirikamchai, Piyaporn, Harmeet Kaur, Deborah A. Kuban, Edward Jackson, Ping Hou, and Haesun Choi. “Use of Maximum Slope Images Generated From Dynamic Contrast-Enhanced MRI to Detect Locally Recurrent Prostate Carcinoma After Prostatectomy: A Practical Approach.” American Journal of Roentgenology 198, no. 3 (March 1, 2012): W228–W236. doi:10.2214/AJR.10.6387. |
126377 | Tracer Concentration | Tracer concentration in tissue. E.g., in a DCE-MR experiment, the concentration of contrast agent in mmol/l. |
126380 | Contrast Longitudinal Relaxivity | The degree to which a paramagnetic contrast agent can enhance the proton longitudinal relaxation rate constant (R1, 1/T1), normalized to the concentration of the contrast agent. Also referred to as r1. Typically expressed in units of l/mmol/s. |
126390 | Regional Blood Flow | The flow rate of blood perfusing a region as volume per mass per unit of time. |
126391 | Regional Blood Volume | The volume of blood perfusing a region as as volume per mass. |
126392 | Oxygen Extraction Fraction | The percent of the oxygen removed from the blood by tissue during its passage through the capillary network. For example, as measured by blood oxygenation level dependent (BOLD) MR. See He, Xiang, and Dmitriy A. Yablonskiy. “Quantitative BOLD: Mapping of Human Cerebral Deoxygenated Blood Volume and Oxygen Extraction Fraction: Default State.” Magnetic Resonance in Medicine 57, no. 1 (2007): 115–26. |
126393 | R1 | The longitiudinal relaxation rate constant. The inverse of longitudinal relaxation time, i.e., R1 = 1/T1. |
126394 | R2 | The transverse relaxation rate constant. The inverse of transverse relaxation time, i.e., R2 = 1/T2. |
126400 | Standardized Uptake Value | A ratio of locally measured radioactivity concentration versus the injected radioactivity distibuted evenly throughout the whole body. This general concept encompasses all specific methods of calculating the whole body volume of distribution, such as using body weight, lean body mass, body surface area, etc. |
126401 | SUVbw | Standardized Uptake Value calculated using body weight. The patient size correction factor for males and females is body weight. Defined in Sugawara et al. Reevaluation of the Standardized Uptake Value for FDG: Variations with Body Weight and Methods for Correction.Radiology, 1999 at http://radiology.rsna.org/content/213/2/521 |
126402 | SUVlbm | Standardized Uptake Value calculated using lean body mass. The patient size correction factor for males is 1.10 * weight - 120 * (weight/height) ^2, and for females is 1.07 * weight - 148 * (weight/height) ^2. Defined in Sugawara et al. Reevaluation of the Standardized Uptake Value for FDG: Variations with Body Weight and Methods for Correction.Radiology, 1999 at http://radiology.rsna.org/content/213/2/521 |
126403 | SUVbsa | Standardized Uptake Value calculated using body surface area. The patient size correction factor for males and females is weight^ 0.425 * height^0.725 * 0.007184. Defined in Sugawara et al. Reevaluation of the Standardized Uptake Value for FDG: Variations with Body Weight and Methods for Correction.Radiology, 1999 at http://radiology.rsna.org/content/213/2/521 |
126404 | SUVibw | Standardized Uptake Value calculated using ideal body weight. The patient size correction factor for males is 48.0 + 1.06 * (height - 152) and for females is 45.5 + 0.91 * (height - 152). Defined in Sugawara et al. Reevaluation of the Standardized Uptake Value for FDG: Variations with Body Weight and Methods for Correction.Radiology, 1999 at http://radiology.rsna.org/content/213/2/521 |
126410 | SUV body weight calculation method | Method of calculating Standardized Uptake Value using body weight. The patient size correction factor for males and females is body weight. Defined in Sugawara et al. Reevaluation of the Standardized Uptake Value for FDG: Variations with Body Weight and Methods for Correction.Radiology, 1999 at http://radiology.rsna.org/content/213/2/521 |
126411 | SUV lean body mass calculation method | Method of calculating Standardized Uptake Value using lean body mass. The patient size correction factor for males is 1.10 * weight - 120 * (weight/height) ^2, and for females is 1.07 * weight - 148 * (weight/height) ^2. Defined in Sugawara et al. Reevaluation of the Standardized Uptake Value for FDG: Variations with Body Weight and Methods for Correction.Radiology, 1999 at http://radiology.rsna.org/content/213/2/521 |
126412 | SUV body surface area calculation method | Method of calculating Standardized Uptake Value using body surface area. The patient size correction factor for males and females is weight^ 0.425 * height^0.725 * 0.007184. Defined in Sugawara et al. Reevaluation of the Standardized Uptake Value for FDG: Variations with Body Weight and Methods for Correction.Radiology, 1999 at http://radiology.rsna.org/content/213/2/521 |
126413 | SUV ideal body weight calculation method | Method of calculating Standardized Uptake Value using ideal body weight. The patient size correction factor for males is 48.0 + 1.06 * (height - 152) and for females is 45.5 + 0.91 * (height - 152). Defined in Sugawara et al. Reevaluation of the Standardized Uptake Value for FDG: Variations with Body Weight and Methods for Correction.Radiology, 1999 at http://radiology.rsna.org/content/213/2/521 |
126500 | Pittsburgh compound B C^11^ | A beta-amyloid PET radiotracer that is an analog of thioflavin T. |
126501 | Florbetaben F^18^ | A beta-amyloid PET radiotracer. |
126502 | T807 F^18^ | A PHF-tau PET radiotracer. |
126503 | Flubatine F^18^ | A nicotinic α4β2 receptor (nAChR) PET radiotracer. |
126510 | Monoclonal Antibody (mAb) ^64^Cu | A Cu 64 Monoclonal Antibody (mAb) PET Radiotracer. |
126511 | Monoclonal Antibody (mAb) ^89^Zr | A Zr 89 Monoclonal Antibody (mAb) PET Radiotracer. |
126512 | Trastuzumab ^89^Zr | A Zr 89 Trastuzumab PET Radiotracer. |
126513 | Cetuximab ^89^Zr | A Zr 89 Cetuximab PET Radiotracer. |
126514 | J591 ^89^Zr | A Zr 89 J591 PET Radiotracer. |
126515 | cU36 ^89^Zr | A Zr 89 cU36 PET Radiotracer. |
126516 | Bevacizumab ^89^Zr | A Zr 89 Bevacizumab PET Radiotracer. |
126517 | cG250-F(ab')(2) ^89^Zr | A Zr 89 cG250-F(ab')(2) PET Radiotracer. |
126518 | R1507 ^89^Zr | A Zr 89 R1507 PET Radiotracer. |
126519 | E4G10 ^89^Zr | A Zr 89 E4G10 PET Radiotracer. |
126520 | Df-CD45 ^89^Zr | A Zr 89 Df-CD45 PET Radiotracer. |
126600 | ^44^Scandium | ^44^Scandium |
126601 | ^51^Manganese | ^51^Manganese |
126602 | ^70^Arsenic | ^70^Arsenic |
126603 | ^90^Niobium | ^90^Niobium |
126604 | ^191m^Iridium | ^191m^Iridium |
126605 | ^43^Scandium | ^43^Scandium |
126606 | ^152^Terbium | ^152^Terbium |
126700 | ATSM Cu^60^ | A Cu 60 ATSM PET radiotracer. |
126701 | ATSM Cu^61^ | A Cu 61 ATSM PET radiotracer. |
126702 | ATSM Cu^62^ | A Cu 62 ATSM PET radiotracer. |
126703 | Choline C^11^ | A C 11 Choline PET radiotracer. |
126704 | Fallypride C^11^ | A C 11 Fallypride PET radiotracer. |
126705 | Fallypride F^18^ | An F 18 Fallypride PET radiotracer. |
126706 | FLB 457 C^11^ | A C 11 FLB 457 PET radiotracer. |
126707 | Fluorotriopride F^18^ | An F 18 Fluorotriopride PET radiotracer. |
126708 | Fluoromisonidazole (FMISO) F^18^ | An F 18 Fluoromisonidazole PET radiotracer. |
126709 | Glutamine C^11^ | A C 11 Glutamine PET radiotracer. |
126710 | Glutamine C^14^ | A C 14 Glutamine PET radiotracer. |
126711 | Glutamine F^18^ | An F 18 Glutamine PET radiotracer. |
126712 | Flubatine F^18^ | An F 18 Flubatine PET radiotracer. |
126713 | 2FA F^18^ | An F 18 2FA PET radiotracer. |
126714 | Nifene F^18^ | An F 18 Nifene PET radiotracer. |
126715 | CLR1404 I^124^ | An I 124 cancer targeted phospholipid ether PET radiotracer. |
126716 | CLR1404 I^131^ | An I 131 cancer targeted phospholipid ether PET radiotracer. |

# E French Language Meanings of Selected Codes Used in the DCMR (Normative)

This Annex defines the French language code meanings for selected codes used in the DCMR.

Table E-1. French Language Meanings of Selected Codes

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning English Language | Code Meaning French Language
--- | --- | --- | --- | ---
BI | 3.0 | II.AC.a | 0 - Need additional imaging evaluation | 0. L'évaluation nécessite des compléments d'imagerie
BI | 3.0 | II.AC.b.1 | 1 - Negative | 1. Négatif
SRT |  | F-01781 | 1 o'clock position | Situé à 1 heure
SRT |  | F-0178A | 10 o'clock position | Situé à 10 heures
SRT |  | F-0178B | 11 o'clock position | Situé à 11 heures
SRT |  | F-0178C | 12 o'clock position | Situé à 12 heures
BI | 3.0 | II.AC.b.2 | 2 - Benign Finding | 2. Constatations bénignes
SRT |  | F-01782 | 2 o'clock position | Situé à 2 heures
BI | 3.0 | II.AC.b.3 | 3 - Probably Benign Finding - short interval follow-up | 3. Anomalie probablement bénigne - proposition d'une surveillance à court terme
SRT |  | F-01783 | 3 o'clock position | Situé à 3 heures
BI | 3.0 | II.AC.b.4 | 4 - Suspicious abnormality, biopsy should be considered | 4. Anomalie suspecte, une biopsie doit être envisagée
SRT |  | F-01784 | 4 o'clock position | Situé à 4 heures
BI | 3.0 | II.AC.b.5 | 5 - Highly suggestive of malignancy, take appropriate action | 5. Haute probabilité de malignité, une action appropriée doit être entreprise
SRT |  | F-01785 | 5 o'clock position | Situé à 5 heures
SRT |  | F-01786 | 6 o'clock position | Situé à 6 heures
SRT |  | F-01787 | 7 o'clock position | Situé à 7 heures
SRT |  | F-01788 | 8 o'clock position | Situé à 8 heures
SRT |  | F-01789 | 9 o'clock position | Situé à 9 heures
DCM |  | 112063 | Abnormal calcifications | Calcifications anormales
DCM |  | 112028 | Abnormal Distribution of Anatomic Structure | Distribution anormale des structures anatomiques
DCM |  | 112004 | Abnormal interstitial pattern | Opacité interstitielle
DCM |  | 112061 | Abnormal lines (1D) | Lignes anormales (1D)
DCM |  | 112062 | Abnormal lucency | Clarté anormale
DCM |  | 112033 | Abnormal opacity | Opacité anormale
DCM |  | 112064 | Abnormal texture | Texture anormale ### Note If the term refers to a localized lesion use " Texture anormale " but if the term refers to the entire lung it is more appropriate to use " Trame anormale ".
SRT |  | M-41610 | Abscess | Abcès
DCM |  | 112146 | Acinar | Acinaire
DCM |  | 112036 | ACR Position Statement | Position de l'ACR
SRT |  | T-15420 | Acromioclavicular Joint | Articulation acromioclaviculaire
SRT |  | T-12281 | Acromion process of scapula | Acromion
SRT |  | G-A231 | Acute | Aigu
DCM |  | 121078 | Addendum | Addendum
DCM |  | 111135 | Additional projections | Incidence complémentaire
SRT |  | M-82003 | Adenoid cystic carcinoma | Carcinome adénoïde kystique (cylindrome)
SRT |  | M-83240 | Adenolipoma | Adénolipome
SRT |  | M-81400 | Adenoma | Adénome
SRT |  | M-89830 | Adenomyoepithelioma | Adénomyoépithéliome
SRT |  | M-74200 | Adenosis | Adénose
SRT |  | G-A127 | Afferent | Afférent
DCM |  | 112055 | Agatston scoring method | Score de calcification cororaire par la méthode d'Agatston
DCM |  | 112143 | Air | Air
DCM |  | 112070 | Air bronchiologram | Bronchiologramme aérique
DCM |  | 112071 | Air bronchogram | Bronchogramme aérique
DCM |  | 112072 | Air crescent | Croissant aérique
DCM |  | 112147 | Air space | Espace aérique
DCM |  | 112104 | Air-fluid level | Niveau hydro-aérique
SRT |  | F-20240 | Air-trapping | Piégeage
SRT |  | T-20001 | Airway structure | Structure des voies aériennes
DCM |  | 111001 | Algorithm Name | Nom de l'algorithme
DCM |  | 111002 | Algorithm Parameters | Paramètres de l'algorithme
DCM |  | 111003 | Algorithm Version | Version de l'algorithme
DCM |  | 111242 | All algorithms succeeded; with findings | Tous les algorithmes ont réussi; avec élément découvert
DCM |  | 111241 | All algorithms succeeded; without findings | Tous les algorithmes ont réussi; sans élément découvert
SRT |  | F-01711 | Almost entirely fat | Presque entièrement graisseux
SRT |  | G-A174 | Along edge | Au bord
SRT |  | F-0176C | Amorphous calcification | Calcification amorphe
SRT |  | M-55160 | Amyloid (tumor) | (Tumeur) amyloïde
DCM |  | 111004 | Analysis Performed | Analyse effectuée
DCM |  | 112050 | Anatomic Identifier | Identificateur anatomique
SRT |  | F-10326 | anatomical | Anatomique
SRT |  | M-88610 | Angiolipoma | Angiolipome
SRT |  | M-76100 | Angiomatosis | Angiomatose
SRT |  | M-91203 | Angiosarcoma | Angiosarcome (hémangiosarcome)
SRT |  | T-11307 | Angle of rib | Angle de la côte
SRT |  | G-A105 | Anterior | Antérieur
DCM |  | 112088 | Anterior junction line | Ligne médiastinale antérieure
SRT |  | T-28630 | Anterior segment of left upper lobe | Segment antérieur du lobe supérieur gauche
SRT |  | T-28230 | Anterior segment of right upper lobe | Segment antérieur du lobe supérieur droit
SRT |  | G-A180 | Anterolateral | Antéro-latéral
DCM |  | 111141 | Any decision to biopsy should be based on clinical assessment | Une éventuelle décision de biopsie doit être basée sur l'évaluation clinique
SRT |  | T-42000 | Aorta | Aorte
SRT |  | T-42300 | Aortic arch | Crosse de l'aorte
SRT |  | T-42310 | Aortic isthmus | Isthme aortique
DCM |  | 112102 | Aortic knob | Bouton aortique
SRT |  | T-35400 | Aortic Valve | Valve aortique
SRT |  | G-A122 | Apical | Apical
SRT |  | G-A122 | Apical | Apical
SRT |  | M-84013 | Apocrine adenocarcinoma | Carcinome apocrine
SRT |  | M-73310 | Apocrine Metaplasia | Métaplasie apocrine
DCM |  | 112103 | Arch of the Azygos vein | Crosse de la veine Azygos
SRT |  | T-11511 | Arch of vertebra | Arc vertébral
DCM |  | 112079 | Architectural distortion | Modification des rapports anatomiques
SRT |  | F-01795 | Architectural distortion of breast | Distorsion architecturale du sein
SRT |  | G-A166 | Area | Surface
SRT |  | G-A16A | Area of defined region | Surface de la région définie
DCM |  | 121056 | Area Outline | Tracé de la surface
DCM |  | 111215 | Artifact(s) other than grid or detector artifact | Artéfacts autres qu'artéfacts de grille ou du détecteur
SRT |  | T-42100 | Ascending aorta | Aorte thoracique ascendante
DCM |  | 111005 | Assessment Category | Catégorie d'évaluation
DCM |  | 112003 | Associated Chest Component | Structure anatomique du thorax
SRT |  | F-01793 | Asymmetric breast tissue | Tissu mammaire asymétrique
SRT |  | P5-B3412 | Asymmetric breast tissue analysis | Analyse de l'asymétrie du tissu mammaire
SRT |  | F-8A063 | Asynchronous involution of breast | Involution asynchrone du sein
SRT |  | D4-31220 | Atrial Septal Defect | Communication inter atriale
SRT |  | T-32100 | Atrium | Atrium ou Oreillette
SRT |  | J-005E8 | Attending (syn. Consultant) | Consultant
DCM |  | 112031 | Attenuation Coefficient | Coefficient d'atténuation
SRT |  | M-72175 | Atypical intraductal hyperplasia | Hyperplasie intracanalaire atypique
SRT |  | M-72105 | Atypical lobular hyperplasia | Hyperplasie lobulaire atypique
SRT |  | G-A147 | Axial | Axial
SRT |  | F-01794 | Axilla position | Situation axillaire
BI | 3.0 | I.E.6 | Axillary adenopathy | Adénopathie axillaire
SRT |  | T-47100 | Axillary Artery | Artère axillaire
SRT |  | T-18774 | Axillary Fascia | Fascia axillaire
DCM |  | 111301 | Axillary nodal metastases | Métastases ganglionnaires axillaire
DCM |  | 111253 | Axillary node hyperplasia | Hyperplasie dans ganglion axillaire
DCM |  | 111252 | Axillary node with calcifications | Ganglion axillaire avec calcifications
DCM |  | 111300 | Axillary node with lymphoma | Lympphome dans ganglion axillaire
SRT |  | R-102D1 | Axillary Tail | Prolongement axillaire
SRT |  | F-0178E | Axillary tail position | Situé dans le prolongement axillaire du sein
SRT |  | T-49110 | Axillary vein | Veine axillaire
DCM |  | 112090 | Azygoesophageal recess interface | Ligne para-azygo-oesophagienne
SRT |  | T-48340 | Azygos vein | Grande veine Azygos
SRT |  | G-A123 | Basal | Basal
DCM |  | 111307 | Basal cell carcinoma of the nipple | Carcinome basocellulaire du mamelon
DCM |  | 121079 | Baseline | Référence
DCM |  | 112016 | Baseline Category | Catégorie à T0
DCM |  | 112154 | Bat's wing distribution | Aspect en aile de papillon ### Note In France, the two concepts as described in Annex D 112154 and 112155 are not distinguished. For this reason both "Bat's wing" and "Butterfly distribution" have a code meaning of " Aspect en aile de papillon ".
SRT |  | A-32475 | BB shot (Lead Pellet) | Marque de plomb (Grain de plomb)
DCM |  | 112066 | Beaded septum sign | Septa perlés
DCM |  | 111256 | Benign Calcifications | Calcifications bénignes
DCM |  | 111255 | Benign cyst with blood | Kyste bénin hémorragique
SRT |  | D7-F0810 | Benign neoplasm of nipple of female breast (Nipple adenoma) | Adénomatose (papillomatose) érosive du mamelon
DCM |  | 121080 | Best illustration of finding | Meilleure illustration des résultats
DCM |  | 112049 | Best Overall Response | La meilleure réponse
SRT |  | G-A102 | Bilateral | Bilatéral
DCM |  | 111143 | Biopsy should be considered | Une biopsie doit être envisagée
DCM |  | 111148 | Biopsy should be strongly considered | Une biopsie doit être absolument envisagée
DCM |  | 111303 | Blood vessel (vascular) invasion | Embole vasculaire
SRT |  | T-11220 | Body of sternum | Corps du sternum
DCM |  | 112007 | Border definition | Définition des bords
DCM |  | 112015 | Border shape | Forme des bords
SRT |  | T-04080 | Both breasts | Les deux seins
SRT |  | T-47160 | Brachial artery | Artère brachiale
SRT |  | T-A9090 | Brachial plexus | Plexus brachial
SRT |  | T-46010 | Brachiocephalic trunk | Tronc artériel brachio-céphalique
SRT |  | T-48620 | Brachiocephalic vein | Tronc veineux brachio-céphalique
SRT |  | T-04000 | Breast | Sein
SRT |  | F-01710 | Breast composition | Composition du sein (des seins)
SRT |  | P5-B3414 | Breast composition analysis | Analyse de la composition du sein (des seins)
DCM |  | 111100 | Breast geometry | Morphologie du sein (des seins)
SRT |  | D7-90428 | Breast lobular hyperplasia | Hyperplasie lobulaire mammaire
DCM |  | 111007 | Breast Outline including Pectoral Muscle Tissue | Limites du sein incluant le muscle pectoral
SRT |  | T-04000 | Breast, NOS | Sein, SAI
SRT |  | R-40939 | Bronchial | Bronchique
SRT |  | T-46310 | Bronchial artery | Artère bronchique
DCM |  | 112052 | Bronchovascular | Broncho-vasculaire
SRT |  | T-26000 | Bronchus | Bronche
SRT |  | A-32110 | Bullet | Balle
DCM |  | 112155 | Butterfly distribution | Aspect en ailes de papillon
DCM |  | 111017 | CAD Processing and Findings Summary | Résumé du traitement et des résultats du système de DAO
SRT |  | F-01775 | Calcification Cluster | Foyer de microcalcifications
DCM |  | 112030 | Calcification Descriptor | Descripteur des calcifications
DCM |  | 111008 | Calcification Distribution | Distribution des calcifications
DCM |  | 112018 | Calcification extent as percent of surface | % de surface calcifiée
DCM |  | 112019 | Calcification extent as percent of volume | % de volume calcifié
DCM |  | 111009 | Calcification Type | Type de calcification
SRT |  | F-01769 | Calcified skin of breast | Calcification cutanée
SRT |  | F-0176A | Calcified suture material | Fils de suture calcifiés
DCM |  | 112145 | Calcium | Calcium
DCM |  | 112058 | Calcium score | Score de calcification
DCM |  | 112034 | Calculation Description | Description du calcul
SRT |  | G-A171 | Capsular | Capsulaire
DCM |  | 111304 | Carcinoma in children | Carcinome de l'enfant
DCM |  | 111305 | Carcinoma in ectopic breast | Carcinome sur glande mammaire ectopique
DCM |  | 111310 | Carcinoma in pregnancy and lactation | Carcinome au cours de la grossesse et de la lactation
SRT |  | D7-F0902 | Carcinoma in situ of male breast | Carcinome de l'homme
DCM |  | 111306 | Carcinoma with endocrine differentiation | Carcinome avec différenciation endocrine
SRT |  | M-85733 | Carcinoma with metaplasia | Carcinome métaplasique
SRT |  | M-89803 | Carcinosarcoma | Carcinosarcome
SRT |  | A-040CB | Cardiac pacemaker lead | Electrode de pace-maker cardiaque
SRT |  | T-25201 | Carina | Carène
DCM |  | 112086 | Carina angle | Angle carinaire
SRT |  | T-B4000 | Carotid Body | Corpuscule carotidien
DCM |  | 111309 | Cartilaginous and osseous change | Métaplasie cartilagineuse ou osseuse
SRT |  | A-26800 | Catheter | Cathéter
SRT |  | G-A108 | Caudal | Caudal
SRT |  | G-A108 | Caudal | Caudal
SRT |  | G-A107 | Caudal-cranial | Pieds-tête
SRT |  | R-10244 | caudo-cranial (from below) | Face caudo-craniale
DCM |  | 112017 | Cavity extent as percent of volume | Taille de la cavité en % du volume
DCM |  | 111203 | CC Nipple not centered on image | Cranio-caudal: mamelon non centré sur l'image
DCM |  | 111202 | CC Not all medial tissue visualized | Cranio-caudal: le tissu interne n'est pas totalement visible
DCM |  | 111204 | CC Posterior nipple line does not measure within 1 cm of MLO | Cranio-caudal:longueur de la ligne rétroaréolaire sur la face plus courte de plus d'un centimètre que sur l'oblique
DCM |  | 111010 | Center | Centre
UCUM |  | cm | centimeter | Centimètre
SRT |  | G-A110 | Central | Central
SRT |  | G-A110 | Central | Central
DCM |  | 112174 | Central line | Cathéter central
SRT |  | F-0178F | Central portion of breast position | Situé dans la partie centrale du sein
DCM |  | 112156 | Centrilobular | Centro-lobulaire
DCM |  | 112087 | Centrilobular structures | Structures centro-lobulaires
SRT |  | G-A107 | Cephalic | Céphalique
SRT |  | G-A107 | Cephalic | Céphalique
DCM |  | 111011 | Certainty of Feature | Certitude concernant la caractéristique
DCM |  | 111012 | Certainty of Finding | Certitude concernant le résultat
DCM |  | 111013 | Certainty of Impression | Certitude concernant l'impression
SRT |  | A-12210 | Cervical collar | Minerve
DCM |  | 112000 | Chest CAD Report | Compte-rendu de la DAO du thorax
DCM |  | 112173 | Chest tube | Drain thoracique
SRT |  | T-D3050 | Chest wall | Paroi thoracique
SRT |  | M-92200 | Chondroma | Chondrome
SRT |  | M-92203 | Chondrosarcoma | Chondrosarcome
SRT |  | T-35020 | Chordae tendineae cordis | Cordage
SRT |  | G-A270 | Chronic | Chronique
SRT |  | M-02560 | Circumference | Circonférence
DCM |  | 112142 | Circumscribed | Circonscrit
SRT |  | F-01741 | Circumscribed lesion | Lésion circonscrite (bien définie ou à contour net)
SRT |  | T-12310 | Clavicle | Clavicule
SRT |  | T-11219 | Clavicular notch of sternum | Incisure claviculaire du sternum
SRT |  | R-102D2 | Cleavage | Sillon inter-mammaire
SRT |  | A-12062 | Clip | Clip
DCM |  | 111014 | Clockface or region | Quadrant ou région
DCM |  | 112157 | Coalescent | Confluent
SRT |  | F-01761 | Coarse (popcorn-like) calcification | Grossière (en popcorn ou coralliforme)
DCM |  | 112178 | Coin | Pièce de monnaie
SRT |  | F-20172 | Coin lesion | Lésion nodulaire
DCM |  | 111195 | Collimation too close to breast | Collimation trop proche du sein
SRT |  | A-10044 | Collimator | Collimateur
SRT |  | M-85012 | Comedocarcinoma (intraductal) | Carcinome intracanalaire de type comédo
SRT |  | T-45100 | Common carotid artery | Artère carotide commune
DCM |  | 111015 | Composite Feature | Caractéristique composite
DCM |  | 112023 | Composite Feature Modifier | Modificateur lié à une anomalie à caractéristiques composites
DCM |  | 111016 | Composite type | Type composite
DCM |  | 110004 | Computer Aided Detection | Système de Détection Assistée par Ordinateur
DCM |  | 110003 | Computer Aided Diagnosis | Système de Diagnostic Assisté par Ordinateur
DCM |  | 121077 | Conclusion | Conclusion
DCM |  | 121076 | Conclusions | Conclusions
DCM |  | 111018 | Content Date | Date du contenu
DCM |  | 111019 | Content Time | Heure du contenu
SRT |  | C-B0300 | Contrast agent NOS | Produit de contraste, SAI
SRT |  | T-12282 | Coracoid process of scapula | Apophyse coracoïde
DCM |  | 112105 | Corona radiata | Couronne radiaire
SRT |  | G-A138 | Coronal | Coronal
SRT |  | T-11240 | Costal Cartilage | Cartilage costal
SRT |  | T-11308 | Costal groove | Sillon de la côte
SRT |  | T-46180 | Costocervical trunk | Tronc cervico-thoracique
SRT |  | G-A108 | Cranial-caudal | Tête-pieds
SRT |  | G-A108 | Cranio-caudal | Cranio-caudal
SRT |  | R-10242 | cranio-caudal | Face
SRT |  | Y-X1770 | cranio-caudal exaggerated laterally | Face exagérée externe
SRT |  | Y-X1771 | cranio-caudal exaggerated medially | Face exagérée interne
LN |  | 18747-6 | CT Report | Compte rendu TDM
UCUM |  | cm3 | Cubic centimeter | Centimètre cube
UCUM |  | dm3 | Cubic decimeter | Décimètre cube
UCUM |  | um3 | Cubic micrometer | Micromètre cube
UCUM |  | mm3 | Cubic millimeter | Millimètre cube
SRT |  | F-10410 | curled-up | En chien de fusil
DCM |  | 121064 | Current Procedure Descriptions | Description de la procédure en cours
DCM |  | 112048 | Current Response | Réponse actuelle
SRT |  | D7-90360 | Cyst of breast | Kyste du sein
DCM |  | 111147 | Cytologic analysis | Analyse cytologique
DCM |  | 111193 | Date sticker is missing | L'étiquette de date est absente
UCUM |  | d | Day | Jour
SRT |  | F-01727 | Decrease in number of calcifications | Diminution du nombre de calcifications
SRT |  | M-02530 | Decrease in size | Diminution de taille
SRT |  | G-A140 | Deep | Profond
SRT |  | T-13660 | Deltoid muscle | Muscle deltoïde
DCM |  | 112118 | Density | Densité ### Note Typically used with chest CT
SRT |  | F-01796 | Mammography breast density | Opacité mammaire à la mammographie
DCM |  | 112119 | Dependent opacity | Opacité déclive
SRT |  | G-D785 | Depth | Profondeur
DCM |  | 111020 | Depth | Profondeur
DCM |  | 121401 | Derivation | Méthode de calcul
SRT |  | T-D0765 | Descending aorta | Aorte thoracique descendante
DCM |  | 111021 | Description of Change | Description des modifications
DCM |  | 111022 | Detection Performed | Détection effectuée
DCM |  | 111214 | Detector artifact(s) | Artéfacts du détecteur
DCM |  | 111259 | Diabetic fibrous mastopathy | Mastopathie diabétique
SRT |  | M-02550 | Diameter | Diamètre
SRT |  | G-A198 | Diameter of circumscribed circle | Diamètre du cercle circonscrit
SRT |  | T-D3400 | Diaphragm | Diaphragme
DCM |  | 110011 | Dictation | Dictée
DCM |  | 112166 | Difference in border definition | Modification de la netteté des bords
DCM |  | 112165 | Difference in border shape | Modification de la forme des bords
DCM |  | 112167 | Difference in distribution | Modification de la distribution
SRT |  | F-017B3 | Difference in location | Différence de localisation
SRT |  | F-05179 | Difference in location | Différence de localisation
SRT |  | F-017B7 | Difference in margin | Différence de contours
SRT |  | F-017B5 | Difference in number of calcifications | Différence du nombre de calcifications
SRT |  | F-017B2 | Difference in opacity | Différence d'opacité
SRT |  | F-017B6 | Difference in shape | Différence de forme
DCM |  | 112168 | Difference in site involvement | Modification du siège des lésions
SRT |  | F-017B1 | Difference in size | Différence de taille
SRT |  | F-05173 | Difference in size | Différence de taille
SRT |  | F-017B4 | Difference in spatial proximity | Différence de proximité dans l'espace
SRT |  | F-017B8 | Difference in symmetry | Différence de symétrie
DCM |  | 112170 | Difference in Texture | Modification de texture
DCM |  | 112169 | Difference in Type of Content | Modification du contenu
DCM |  | 111023 | Differential Diagnosis/Impression | Diagnostic différentiel/Impression
SRT |  | G-A321 | Diffuse | Diffus
SRT |  | F-01770 | Diffuse calcification distribution | Calcifications diffuses(disséminées)
SRT |  | M-020FA | Discoid | Discoïde
SRT |  | G-A324 | Disseminated | Disséminé
SRT |  | G-A119 | Distal | Distal
DCM |  | 121206 | Distance | Distance
DCM |  | 112138 | Distinctly defined | Distincts les uns des autres
DCM |  | 112006 | Distribution Descriptor | Descripteur de la distribution
DCM |  | 113011 | Document Title Modifier | Modificateur du titre du document
SRT |  | G-A106 | Dorsal | Dorsal
SRT |  | T-12287 | Dorsal aspect of scapula | Corps de l'omoplate
SRT |  | T-461A0 | Dorsal scapular artery | Artère scapulaire postérieure
DCM |  | 111258 | Ductal adenoma | Adénome ductal
SRT |  | M-72170 | Ductal hyperplasia, Usual | Hyperplasie canalaire
SRT |  | P5-40060 | mammary ductogram | Galactographie
SRT |  | F-01762 | Dystrophic calcification | Dystrophique
SRT |  | D4-48014 | Ectopic (accessory) breast tissue | Tissu mammaire ectopique (glande mammaire accessoire)
SRT |  | M-36300 | Edema | Oedème
SRT |  | G-A174 | Edge | Bord
SRT |  | G-A128 | Efferent | Efférent
SRT |  | F-01763 | Eggshell calcification | En coquille d'oeuf
DCM |  | 111217 | Electrical failure | Défaillance électrique
DCM |  | 112134 | Elliptic | Elliptique
SRT |  | A-25350 | Endotracheal tube | Tube endotrachéal
SRT |  | R-40750 | Enlarged | Augmenté de taille
SRT |  | M-33410 | Epidermal inclusion cyst | Kyste épidermique
SRT |  | F-01752 | Equal density (isodense) lesion | Lésion de densité identique (isodense)
SRT |  | F-10440 | erect | Debout
SRT |  | T-14020 | Erector spinae muscle | Muscles érecteurs du rachis
SRT |  | T-4630D | Esophageal artery | Artère oesophagienne
SRT |  | T-D3412 | Esophageal Hiatus | Hiatus oesophagien
SRT |  | T-56000 | Esophagus | Oesophage
SRT |  | R-10260 | Estimated | Estimé
SRT |  | R-102CF | exaggerated cranio-caudal | Face exagérée
SRT |  | G-A112 | External | Externe
SRT |  | R-40941 | External | Externe
SRT |  | T-1416B | External intercostal muscle | Muscle intercostal externe
SRT |  | M-88211 | Extra abdominal desmoid | Tumeur desmoïde extraabdominale
SRT |  | G-A151 | Extra-articular | Extra-articulaire
SRT |  | F-01714 | Extremely dense | Très dense
DCM |  | 112131 | Extremely small | Extrêmement petit
DCM |  | 111224 | Failed | Echec
DCM |  | 111024 | Failed Analyses | Échec des analyses
DCM |  | 111025 | Failed Detections | Échec des détections
SRT |  | T-D0634 | Fascial layer | Fascia
SRT |  | T-D008A | Fat | Graisse
SRT |  | F-01754 | Fat containing (radiolucent) lesion | Lésion contenant de la graisse (radiotransparent)
SRT |  | D7-90434 | Fat necrosis of breast | Cytostéatonécrose mammaire
DCM |  | 111159 | Feature detected on images from multiple modalities | Caractéristique détectée sur les images provenant de plusieurs modalités
DCM |  | 111158 | Feature detected on multiple images | Caractéristique détectée sur plusieurs images
DCM |  | 111157 | Feature detected on only one of the images | Caractéristique détectée sur une seule des images
DCM |  | 111156 | Feature detected on the only image | Caractéristique détectée sur la seule image
SRT |  | A-26430 | Feeding tube | Sonde d'alimentation
DCM |  | F | female | Femme
DCM |  | 111264 | Fibroadenolipoma | Adénofibrolipome
SRT |  | M-90100 | Fibroadenoma | Fibroadénome
DCM |  | 111263 | Fibroadenomatoid hyperplasia | Hyperplasie fibro-adénomatoïde
DCM |  | 112163 | Fibrocalcific | Fibrocalcique
SRT |  | D7-90310 | Fibrocystic disease of breast | Dysplasie fibrokystique du sein
SRT |  | M-78800 | Fibromatosis | Fibomatose
DCM |  | 112148 | Fibronodular | Fibro-nodulaire
SRT |  | M-88103 | Fibrosarcoma | Fibrosarcome
DCM |  | 112171 | Fiducial mark | Point de repère
DCM |  | 110010 | Film | Film
DCM |  | 121071 | Finding | Résultat
SRT |  | F-01722 | Finding partially removed | Exérèse partielle de l'élément
DCM |  | 121070 | Findings | Résultats
SRT |  | F-0176D | Fine, linear (casting) calcification | Calcification fine linéaire (vermiculaire)
SRT |  | F-0176E | Fine, linear, branching (casting) calcification | Calcification fine linéaire, arborisée (ramifiée)
SRT |  | T-D051D | Fissure of lung | Scissure
DCM |  | 111191 | Flash doesn't include cassette/screen/detector identification | Le marquage n'indique pas l'identifiant de cassette/écran/détecteur
DCM |  | 111188 | Flash doesn't include date of examination | Le marquage n'indique pas la date de l'examen
DCM |  | 111189 | Flash doesn't include facility name and location | Le marquage n'indique ni le nom de l'établissement ni son adresse
DCM |  | 111192 | Flash doesn't include mammography unit identification | Le marquage n'indique pas l'identifiant du mammographe
DCM |  | 111187 | Flash doesn't include patient name and additional patient id | Le marquage n'indique ni le nom du patient ni son identifiant.
DCM |  | 111186 | Flash is illegible, does not fit, or is lopsided | Le marquage est illisible, mal positionné ou de travers
DCM |  | 111185 | Flash is not near edge of film | Le marquage n'est pas au bord du film
DCM |  | 112107 | Fleischner's line(s) | Ligne(s) de Fleischner
DCM |  | 112164 | Flocculent | Floconneux
DCM |  | 112149 | Fluffy | Flou ### Note The word-to-word translation of "Fluffy" is " Duveteux ", but this term is never used. For tissues, the translation must be " Floconneux " but this term is only used for calcifications (Flocculent = Floconneux ) in CID 6132 “Chest Calcification Descriptor”. We retained " Flou " (in English, "Fuzzy") as the most appropriate meaning.
SRT |  | G-A351 | Focal | Localisé
SRT |  | F-01792 | Focal asymmetric breast tissue | Asymétrie focale du tissu mammaire
SRT |  | P5-B3410 | Focal asymmetric density analysis | Analyse de l'asymétrie de densité focale
SRT |  | M-78266 | Focal fibrosis | Fibrose focale
DCM |  | 111142 | Follow-up at short interval (1-11 months) | Surveillance à court terme (1-11 mois)
DCM |  | 113005 | For Conference | Pour une conférence
DCM |  | 113007 | For Patient | Pour le patient
DCM |  | 113008 | For Peer Review | Pour relecture par un pair
DCM |  | 113002 | For Referring Provider | Pour le référent
DCM |  | 113009 | For Research | Pour la recherche
DCM |  | 113003 | For Surgery | Pour la chirurgie
DCM |  | 113004 | For Teaching | Pour l'enseignement
DCM |  | 113006 | For Therapy | Pour la thérapeutique
SRT |  | M-44140 | Foreign body (reaction) | Réaction à corps étranger
SRT |  | M-30400 | Foreign material (iodized oil, mercury,talc) | Corps étranger (lipiodol, mercure,talc)
SRT |  | F-10380 | frog | Position de la grenouille
SRT |  | G-A138 | Frontal | Frontal
SRT |  | D7-90364 | Galactocele | Galactocèle
SRT |  | G-A366 | Generalized | Généralisé
SRT |  | M-90160 | Giant fibroadenoma | Adénofibrome géant
SRT |  | T-1228A | Glenoid cavity of scapula | Cavité glénoïde
SRT |  | M-83153 | Glycogen-rich carcinoma | Carcinome riche en glycogène
SRT |  | M-95800 | Granular cell tumor | Tumeur à cellules granuleuses
DCM |  | 112128 | Granular pattern | Aspect micronodulaire
DCM |  | 111208 | Grid artifact(s) | Artéfact(s) de grille
DCM |  | 112120 | Ground glass opacity | Opacité en verre dépoli
SRT |  | F-01772 | Grouped calcification distribution | Calcification groupées (ou en foyer)
SRT |  | G-A169 | Gutter | Gouttière
SRT |  | D7-90420 | Gynecomastia | Gynécomastie
DCM |  | 112073 | Halo sign | Signe du halo
SRT |  | M-75500 | Hamartoma | Hamartome
SRT |  | T-11301 | Head of rib | Tête de le côte
SRT |  | T-32000 | Heart | Coeur
SRT |  | A-04110 | Heart valve prosthesis | Prothèse valvulaire
SRT |  | M-91200 | Hemangioma | Hémangiome
SRT |  | D3-F0620 | Hemangioma of subcutaneous tissue | Hémangiome des tissues sous-cutané
SRT |  | M-91220 | Hemangioma - venous | Hémangiome veineux
SRT |  | M-91501 | Hemangiopericytoma | Hémangiopéricytome
SRT |  | M-35060 | Hematoma | Hématome
SRT |  | F-0176F | Heterogeneous calcification | Calcification punctiforme irrégulière (polymorphe, hétérogène)
SRT |  | F-01713 | Heterogeneously dense | Dense et hétérogène
DCM |  | 112095 | Hiatus | Hiatus
SRT |  | F-01751 | High density lesion | Lésion de forte densité
DCM |  | 111149 | Highly suggestive of malignancy - take appropriate action | Haute probabilité de malignité - une action appropriée doit être entreprise
SRT |  | G-A170 | Hilar | Hilaire
SRT |  | T-28080 | Hilum of lung | Hile pulmonaire
SRT |  | G-A170 | Hilus | Hile
DCM |  | 111145 | Histology using core biopsy | Histologie par biopsie à l'aiguille
DCM |  | 121060 | History | Antécédents
SRT |  | M-96503 | Hodgkin's disease (lymphoma) | Maladie de Hodgkin
DCM |  | 112160 | Homogeneous (uniform opacity) | Homogène (opacité uniforme)
DCM |  | 112106 | Honeycomb pattern | Aspect en rayon de miel
SRT |  | G-A142 | Horizontal | Horizontal
SRT |  | G-A142 | Horizontal | Horizontal
DCM |  | 111026 | Horizontal Pixel Spacing | Espacement horizontal des pixels
UCUM |  | h | hour | Heure
SRT |  | T-12410 | Humerus | Humérus
DCM |  | 112159 | Hyper-acute | Suraigu
SRT |  | M-72000 | Hyperplasia, usual | Hyperplasie simple
SRT |  | A-16016 | ID Plate | Zone d'identification
SRT |  | T-14030 | Iliocostalis muscle | Muscle ilio-costal
DCM |  | 111027 | Image Laterality | Latéralité de l'image
DCM |  | 111028 | Image Library | Bibliothèque d'images
DCM |  | 110001 | Image Processing | Traitement d'image
DCM |  | 111101 | Image Quality | Qualité image
SRT |  | P5-B3408 | Image quality analysis | Analyse de la qualité d'image
DCM |  | 111029 | Image Quality Rating | Score de qualité image
DCM |  | 111030 | Image Region | Région de l'image
DCM |  | 111031 | Image View | Incidence
DCM |  | 111032 | Image View Modifier | Modificateur de l'incidence
SRT |  | A-04010 | Implant | Prothèse
SRT |  | R-102D5 | Implant Displaced | Prothèse déplacée
SRT |  | F-0172B | Implant revised since previous mammogram | Prothèse révisée depuis la mammographie précédente
DCM |  | 121073 | Impression | Impression
DCM |  | 111033 | Impression Description | Description de l'impression
DCM |  | 121072 | Impressions | Impressions
DCM |  | 111196 | Inadequate compression | Compression inadéquate
DCM |  | 111219 | Inappropriate image processing | Défaillance du processus de traitement d'image
SRT |  | F-01726 | Increase in number of calcifications | Augmentation du nombre de calcifications
SRT |  | M-02520 | Increase in size | Augmentation de taille
DCM |  | 121109 | Indications for Procedure | Indications de la procédure
SRT |  | F-01744 | Indistinct lesion | Lésion indistincte
SRT |  | F-01776 | Individual Calcification | Calcification isolée
DCM |  | 111233 | Individual Impression / Recommendation Analysis | Analyse de l'Impression / recommandation élémentaire
DCM |  | 111034 | Individual Impression/Recommendation | Impression élémentaire/Recommandation
SRT |  | D7-90452 | Infarction of breast | Infarctus mammaire
SRT |  | G-A115 | Inferior | Inférieur
SRT |  | T-116EF | Inferior articular facet of axis | Facette articulaire inférieure de l'axis
SRT |  | T-1153F | Inferior articular process of vertebra | Massif articulaire inférieur
SRT |  | T-46940 | Inferior phrenic artery | Artère phrénique inférieure
SRT |  | T-48710 | Inferior vena cava | Veine cave inférieure
DCM |  | 112121 | Infiltrate | Infiltrat
SRT |  | M-85003 | Infiltrating duct carcinoma | Carcinome canalaire infiltrant
SRT |  | M-40000 | Inflammation | Infection
SRT |  | M-85303 | Inflammatory carcinoma | Carcinome inflammatoire
SRT |  | T-13620 | Infraspinatus muscle | Muscle sous épineux
DCM |  | 112161 | Inhomogeneous | Hétérogène
SRT |  | G-A113 | Inner | En dedans
SRT |  | T-14165 | Innermost intercostal muscles | Muscles intercostaux intimes
DCM |  | 111240 | Institutionally defined quality control standard | Standards de contrôle de qualité définis par l'institution
DCM |  | 111206 | Insufficient implant displacement incorrect | Refoulement de la prothèse insuffisant
SRT |  | T-D305A | Intercostal artery | Artère intercostale
DCM |  | 112082 | Interface | Interface
SRT |  | G-A114 | Intermediate | Intermédiaire
UMLS |  | C1144859 | Intern | Interne
SRT |  | G-A113 | Internal | Interne
SRT |  | R-40819 | Internal | Interne
SRT |  | T-14183 | Internal intercostal muscle | Muscle intercostal interne
SRT |  | T-48170 | Internal jugular vein | Veine jugulaire interne
SRT |  | T-46200 | Internal thoracic artery | Artère thoracique interne
DCM |  | 110005 | Interpretation | Interprétation
SRT |  | T-1A007 | Interstitial tissue | Interstitium
SRT |  | T-32410 | Interventricular septum | Septum interventriculaire
SRT |  | G-A15A | Intra-articular | Intra-articulaire
DCM |  | 111315 | Intracystic papillary carcinoma | Carcinome papillaire intrakystique
SRT |  | M-85040 | Intracystic papilloma | Papillome intrakystique
SRT |  | M-85072 | Intraductal carcinoma micro-papillary | Carcinome intracanalaire de type micropapillaire
DCM |  | 111341 | Intraductal carcinoma, high grade | Carcinome intracanalaire
DCM |  | 111313 | Intraductal carcinoma, low grade | Carcinome intracanalaire de bas grade
DCM |  | 111312 | Intraductal comedocarcinoma with necrosis | Carcinome intracanalaire de type comédo avec nécrose
SRT |  | M-85030 | Intraductal papilloma | Papillome intragalactophorique
DCM |  | 112108 | Intralobular lines | Lignes intra-lobulaires
SRT |  | T-C4351 | Intra-mammary lymph node | Ganglion intramammaire
DCM |  | 111316 | Invasive and in-situ carcinoma | Carcinome infiltrant et in situ
SRT |  | M-82013 | Invasive cribriform carcinoma | Carcinome infiltrant cribriforme
SRT |  | M-85203 | Invasive lobular carcinoma | Carcinome lobulaire infiltrant
SRT |  | F-10349 | inverse Trendelenburg | Trendelenburg inversé
DCM |  | 113850 | Irradiation Authorizing | Médecin responsable de l'indication
SRT |  | G-A402 | Irregular | Irrégulière
SRT |  | A-1016B | J Wire | Hameçon
SRT |  | A-26434 | Jejunostomy tube | Tube de jéjunostomie
SRT |  | A-61000 | Jewelry | Bijoux
SRT |  | M-90300 | Juvenile fibroadenoma | Fibroadénome juvénile
DCM |  | 111277 | Juvenille papillomatosis | Papillomatose juvénile
DCM |  | 112109 | Kerley A line | Ligne A de Kerley
DCM |  | 112110 | Kerley B line | Ligne B de Kerley
DCM |  | 112111 | Kerley C lines | Lignes C de Kerley
DCM |  | 113012 | Key Object Description | Description de l'objet clé
DCM |  | 112175 | Kidney stent | Stent rénal
SRT |  | F-10336 | knee-chest | Genu pectoral
SRT |  | F-10330 | kneeling | À genou [à genou]
SRT |  | M-82040 | Lactating adenoma | Adénome lactant
DCM |  | 111279 | Lactational change | Lobule sécrétant
SRT |  | T-11514 | Lamina of vertebra | Lame de la vertèbre
SRT |  | G-A405 | Laminated | Lamellaire
SRT |  | R-404AA | Large | Gros
DCM |  | 111281 | Large duct papilloma | Papillome solitaire
SRT |  | F-01764 | Large rod-like calcification | Calcification en bâtonnet
SRT |  | G-A104 | Lateral | Externe
SRT |  | F-10318 | lateral decubitus | Décubitus latéral
SRT |  | G-C171 | Laterality | Latéralité
SRT |  | R-10228 | latero-medial | Profil externe
SRT |  | R-10230 | latero-medial oblique | Latéro-médial oblique
SRT |  | T-14172 | Latissimus dorsi muscle | Muscle grand dorsal
SRT |  | G-A101 | Left | Gauche
SRT |  | T-04030 | Left breast | Sein gauche
SRT |  | G-A101 | Left lateral | Latéral gauche
SRT |  | F-10319 | left lateral decubitus | Décubitus latéral gauche
SRT |  | T-26500 | Left main bronchus | Bronche principale gauche
SRT |  | M-88900 | Leiomyoma | Léiomyome
SRT |  | M-88903 | Leiomyosarcoma | Léiomyosarcome
SRT |  | G-A22A | Length | Longueur
DCM |  | 111035 | Lesion Density | Densité de la lésion
SRT |  | F-01728 | Less defined | Moins bien défini
DCM |  | 111318 | Leukemic infiltration | Infiltration leucémique
SRT |  | T-14150 | Levatores costarum muscles | Muscles élévateurs des côtes
SRT |  | T-42370 | Ligamentum arteriosum | Ligament artériel
DCM |  | 112083 | Line | Ligne
DCM |  | 112150 | Linear | Linéaire
SRT |  | F-01771 | Linear calcification distribution | Distribution linéaire des calcifications
SRT |  | M-83143 | Lipid-rich (lipid-secreting) carcinoma | Carcinome à cellules lipidiques
SRT |  | M-88500 | Lipoma of the breast | Lipome
SRT |  | M-88503 | Liposarcoma | Liposarcome
SRT |  | F-10346 | lithotomy | Lithotomie
DCM |  | 112158 | Lobar | Lobaire
SRT |  | T-28770 | Lobe of lung | Lobe pulmonaire
SRT |  | G-A640 | Lobular | Lobulée
SRT |  | D7-F0A02 | Lobular carcinoma in situ of breast | Carcinome lobulaire in situ mammaire
DCM |  | 112135 | Lobulated | Lobulée
DCM |  | 112013 | Location in Chest | Localisation thoracique
SRT |  | G-A185 | Long Axis | Grand axe
SRT |  | T-14040 | Longissimus muscle | Muscle longissimus du thorax
SRT |  | G-A143 | Longitudinal | Longitudinal
SRT |  | F-01753 | Low density (not containing fat) lesion | Faible densité (sans contenu graisseux)
SRT |  | T-04003 | Lower inner quadrant of breast | Quadrant inféro-interne du sein
SRT |  | T-04003 | Lower inner quadrant of breast, NOS | Quadrant inféro-interne du sein, SAI
SRT |  | T-28830 | Lower lobe of lung | Lobe pulmonaire inférieur
SRT |  | T-04005 | Lower outer quadrant of breast | Quadrant inféro-externe du sein
SRT |  | T-04005 | Lower outer quadrant of breast, NOS | Quadrant inféro-externe du sein, SAI
SRT |  | T-D320A | Lower zone of lung | Zone inférieure du poumon
DCM |  | 112084 | Lucency | Clarté
SRT |  | F-01766 | Lucent-centered calcification | Calcification à centre clair
SRT |  | T-28000 | Lung | Poumon
DCM |  | 111320 | Lympathic vessel invasion | Embole lymphatique
SRT |  | T-C4000 | Lymph node | Ganglion lymphatique
SRT |  | M-95903 | Lymphoma | Lymphome
SRT |  | R-102D6 | Magnification | Agrandissement
SRT |  | R-102D6 | Magnification views | Agrandissements
SRT |  | G-A193 | Major Axis | Axe principal
DCM |  | M | male | Homme
SRT |  | M-88303 | Malignant fibrous histiocytoma | Histiocytofibrome malin
DCM |  | 111334 | Malignant melanoma of nipple | Mélanome malin du mamelon
SRT |  | D7-90370 | Mammary duct ectasia | Galactophorite ectasiante mammaire (ectasie canalaire mammaire)
SRT |  | F-01791 | Mammographic breast mass | Masse du sein à la mammographie
DCM |  | 111036 | Mammography CAD Report | Compte rendu d'analyse mammographique par système de DAO
SRT |  | T-11211 | Manubrium of sternum | Manubrium sternal
SRT |  | G-A177 | Marginal | Marginal
DCM |  | 111037 | Margins | Contours
SRT |  | M-03000 | Mass | Masse
DCM |  | 112057 | Mass scoring method | Appréciation de la charge calcique par la méthode du score de masse
DCM |  | 112180 | Maximum Attenuation Coefficient | Coefficient d'atténuation maximum
DCM |  | 112181 | Mean Attenuation Coefficient | Coefficient d'atténuation moyen
DCM |  | 112051 | Measurement of Response | Quantification de la réponse
DCM |  | 111216 | Mechanical failure | Défaillance mécanique
SRT |  | G-A109 | Medial | Médial
SRT |  | G-A109 | Median | Médian
DCM |  | 112182 | Median Attenuation Coefficient | Médiane des coefficients d'atténuation
SRT |  | T-D3300 | Mediastinum | Médiastin
SRT |  | R-10224 | medio-lateral | Profil interne
SRT |  | R-10226 | medio-lateral oblique | Médiolatéral oblique
SRT |  | R-404A9 | Medium | Moyen
SRT |  | M-85103 | Medullary carcinoma | Carcinome médullaire
DCM |  | 111333 | Metastasis to an intramammary lymph node | Ganglion intramammaire métastatique
DCM |  | 111323 | Metastatic cancer to the breast | Cancer métastatique au sein
DCM |  | 111324 | Metastatic cancer to the breast from the colon | Métastase intramammaire d'un cancer colique
DCM |  | 111325 | Metastatic cancer to the breast from the lung | Métastase intramammaire d'un cancer pulmonaire
DCM |  | 111327 | Metastatic cancer to the breast from the ovary | Métastase intramammaire d'un cancer ovarien
DCM |  | 111330 | Metastatic disease to axillary node | Ganglion axillaire métastatique
DCM |  | 111326 | Metastatic melanoma to the breast | Métastase intramammaire d'un mélanome malin
DCM |  | 111328 | Metastatic sarcoma to the breast | Métastase intramammaire d'un sarcome
DCM |  | 111284 | Microglandular adenosis | Adénose microglandulaire
SRT |  | F-01742 | Microlobulated lesion | Lésion microlobulée
UCUM |  | um | micrometer | Micromètre
DCM |  | 112122 | Micronodule | Micronodule
SRT |  | G-A109 | Middle | Milieu
SRT |  | G-A109 | Middle | Médian
SRT |  | T-28825 | Middle lobe of lung | Lobe moyen du poumon
SRT |  | T-D3209 | Middle zone of lung | Zone moyenne du poumon
DCM |  | 112085 | Midlung window | Fenêtre lobaire moyenne
SRT |  | R-404FA | Mild | faible
DCM |  | 112129 | Miliary pattern | Aspect miliaire
SRT |  | F-01765 | Milk of calcium calcification | Lait calcique
UCUM |  | mm | millimeter | Millimètre
DCM |  | 112179 | Minimum Attenuation Coefficient | Coefficient d'atténuation minimum
SRT |  | G-A194 | Minor Axis | Axe secondaire
UCUM |  | min | minute | Minute
SRT |  | T-35300 | Mitral Valve | Valve atrio-ventriculaire gauche
DCM |  | 111200 | MLO Evidence of motion blur | Oblique externe: présence d'un flou cinétique
DCM |  | 111201 | MLO Inframammary fold is not open | Oblique externe: sillon sous-mammaire non visible
DCM |  | 111197 | MLO Insufficient pectoral muscle | Oblique externe: muscle pectoral insuffisamment visible
DCM |  | 111198 | MLO No fat is visualized posterior to fibroglandular tissues | Oblique externe: lame graisseuse rétroglandulaire non visualisée
DCM |  | 111199 | MLO Poor separation of deep and superficial breast tissues | Oblique externe: mauvaise séparation des tissus superficiels et profonds
SRT |  | G-A002 | Moderate | Modéré
UCUM |  | mo | Month | Mois
SRT |  | F-01729 | More defined | Mieux défini
DCM |  | 112130 | Mosaic pattern | Aspect en mosaïque
DCM |  | 112080 | Mosaic perfusion | Perfusion en mosaïque
DCM |  | 111210 | Motion blur | Flou cinétique
DCM |  | 111210 | Motion blur | Artefact de mouvement
LN |  | 18755-9 | MR Report | Compte rendu IRM
SRT |  | M-84803 | Mucinous adenocarcinoma (Colloid carcinoma) | Carcinome (mucineux) colloïde
SRT |  | G-A443 | Multifocal | Multifocal
DCM |  | 111329 | Multifocal intraductal carcinoma | Carcinome intracanalaire multifocal
DCM |  | 111332 | Multifocal invasive ductal carcinoma | Carcinome canalaire infiltrant multifocal
DCM |  | 111285 | Multiple Intraductal Papillomas | Papillomes multiples
SRT |  | R-420AE | Muscular | Musculaire
SRT |  | M-88250 | Myofibroblastoma | Myofibroblastome
SRT |  | R-41727 | Narrow | Etroit
SRT |  | T-11303 | Neck of rib | Col de la côte
SRT |  | A-30360 | Needle | Aiguille
DCM |  | 111144 | Needle localization and biopsy | Répérage métallique préopératoire et biopsie-exérèse
SRT |  | D0-F035F | Neoplasm of mammary skin | Tumeur de la peau mammaire
SRT |  | M-95400 | Neurofibroma | Neurofibrome
SRT |  | M-95401 | Neurofibromatosis | Neurofibromatose
SRT |  | F-01721 | New finding | Nouvel élément
SRT |  | T-04100 | Nipple | Mamelon
DCM |  | 111297 | Nipple Characteristic | Caractéristiques du mamelon
DCM |  | 111205 | Nipple not in profile | Le mamelon n'est pas de profil
SRT |  | D7-90554 | Nipple retraction | Rétraction mamelonnaire
DCM |  | 112177 | Nipple ring | Cerclage mammelonnaire
DCM |  | 111286 | No abnormality | Pas d'anomalie
DCM |  | 111245 | No algorithms succeeded; without findings | Aucun algorithme n'a réussi; sans élément découvert
DCM |  | 111213 | No image | Pas d'image
SRT |  | F-01723 | No significant changes in the finding | Pas de modification significative de l'élément
DCM |  | 110009 | No subsequent Workitems | Aucun sujet de travail ultérieur
SRT |  | R-403A7 | Nodular | Nodulaire
DCM |  | 112067 | Nodular pattern | Aspect nodulaire
SRT |  | M-03010 | Nodule | Nodule
SRT |  | M-95913 | Non-Hodgkin's lymphoma | Lymphome non hodgkinien
DCM |  | 111102 | Non-lesion | Pas de lésion
DCM |  | 112076 | Non-Lesion at Baseline | Anomalie « non lésion » à T0
DCM |  | 112037 | Non-lesion Modifier | Modificateur lié à une « non lésion »
DCM |  | 112075 | Non-Target Lesion at Baseline | Lésion « non cible » à T0
DCM |  | 112045 | Non-Target Lesion Complete Response | Disparition des lésions « non cibles »
DCM |  | 112046 | Non-Target Lesion Incomplete Response or Stable Disease | Réponse partielle ou maladie stable sur lésions « non cibles »
DCM |  | 112047 | Non-Target Lesion Progressive Disease | Progression sur lésions « non cibles »
DCM |  | 111251 | Normal axillary node | Ganglion axillaire normal
DCM |  | 111287 | Normal breast tissue | Tissu mamaire normal
DCM |  | 111140 | Normal interval follow-up | Intervalle normal de surveillance
SRT |  | M-02000 | Normal shape | Forme normale
DCM |  | 111244 | Not all algorithms succeeded; with findings | Certains algorithmes n'ont pas réussi; avec élément découvert
DCM |  | 111243 | Not all algorithms succeeded; without findings | Certains algorithmes n'ont pas réussi; sans élément découvert
DCM |  | 111225 | Not Attempted | Non traité
DCM |  | 111152 | Not for Presentation: Rendering device expected not to present | Pas de présentation
DCM |  | 111038 | Number of calcifications | Nombre de calcifications
SRT |  | J-07100 | Nurse | Infirmière
DCM |  | 111039 | Object type | Type d'objet
SRT |  | G-A472 | Oblique | Oblique
SRT |  | F-01743 | Obscured lesion | Lésion masquée
DCM |  | 111322 | Occult carcinoma presenting with axillary lymph node metastases | Carcinome occulte révélé par des métastases axillaires
DCM |  | 113000 | Of Interest | Interressant
DCM |  | 111290 | Oil cyst (fat necrosis cyst) | Cytostéatonécrose kystisée
DCM |  | 111138 | Old films for comparison | Clichés antérieurs pour comparaison
DCM |  | 112060 | Oligemia | Oligémie
SRT |  | G-A103 | One-sided | Situé d'un seul côté
DCM |  | 112001 | Opacity | Opacité ### Note Typically used with projection chest X-Ray
DCM |  | 112027 | Opacity Descriptor | Descripteur de l'opacité
SRT |  | A-00D7B | Opaque marker | Marqueur Opaque
DCM |  | 112014 | Orientation Descriptor | Descripteur de l'orientation
DCM |  | 111040 | Original Source | Source originelle
DCM |  | 112053 | Osseous | Osseux
DCM |  | 112038 | Osseous Modifier | Modificateur lié à une structure osseuse
SRT |  | F-12100 | Ossification | Ossification
SRT |  | M-91803 | Osteogenic sarcoma | Ostéosarcome
DCM |  | 121102 | Other sex | Autre sexe
DCM |  | 111220 | Other failure | Autre défaillance
DCM |  | 111175 | Other Marker | Autre marqueur
SRT |  | G-A112 | Outer | En dehors
DCM |  | 111041 | Outline | Contours
DCM |  | 111212 | Over exposed | Sur-exposé
DCM |  | 111234 | Overall Impression / Recommendation Analysis | Analyse de l'Impression / ecommandation globale
SRT |  | M-02120 | Ovoid shape (Oval) | Forme ovale (Ovale)
SRT |  | A-11101 | Cardiac Pacemaker | Stimulateur cardiaque
SRT |  | A-10042 | Compression paddle | Palette de compression
SRT |  | M-85403 | Paget's disease, mammary (of the nipple) | Maladie de Paget du mamelon
DCM |  | 112176 | Pancreatic stent | Stent pancréatique
SRT |  | M-80503 | Papillary carcinoma (invasive) | Carcinome papillaire infiltrant
SRT |  | M-80502 | Papillary carcinoma in-situ | Carcinome papillaire in-situ
SRT |  | M-80500 | Papilloma | Papillome
DCM |  | 112091 | Paraspinal line | Ligne paravertébrale
DCM |  | 112112 | Parenchymal band | Bande parenchymateuse
DCM |  | 111223 | Partially Succeeded | Succès partiel
DCM |  | 121055 | Path | Tracé
DCM |  | 121211 | Path length | Longueur du tracé
DCM |  | 111042 | Pathology | Pathologie
DCM |  | 111043 | Patient Orientation Column | Colonne concernant l'orientation du patient
DCM |  | 111044 | Patient Orientation Row | Ligne concernant l'orientation du patient
SRT |  | T-D2236 | Pectoral girdle | Ceinture pectorale
DCM |  | 111045 | Pectoral Muscle Outline | Contour du muscle pectoral
SRT |  | T-14110 | Pectoralis major muscle | Muscle grand pectoral
SRT |  | T-14120 | Pectoralis minor muscle | Muscle petit pectoral
SRT |  | T-11515 | Pedicle of vertebra | Pédicule de la vertèbre
DCM |  | 111046 | Percent Glandular Tissue | Pourcentage de tissu glandulaire
DCM |  | 112185 | Performance of CT for Detection of Pulmonary Embolism in Adults | Le scanner dans les embolies pulmonaires de l'adulte, ACR
DCM |  | 112186 | Performance of High-Resolution CT of the Lungs in Adults | Le scanner thoracique haute résolution de l'adulte, ACR
DCM |  | 112035 | Performance of Pediatric and Adult Chest Radiography, ACR | Les radiographies thoraciques de l'enfant et de l'adulte, ACR
DCM |  | 112184 | Performance of Pediatric and Adult Thoracic CT | Le scanner thoracique de l'enfant et de l'adulte, ACR
DCM |  | 121094 | Performing | Réalisateur de l'examen
SRT |  | T-46210 | Pericardiophrenic Artery | Artère péricardo-phrénique
SRT |  | G-A197 | Perimeter | Périmètre
DCM |  | 121057 | Perimeter Outline | Délimitation du périmètre
SRT |  | G-A111 | Peripheral | Périphérique
SRT |  | G-A111 | Peripheral | Périphérique
DCM |  | 111299 | Peripheral duct papillomas | Papillomes périphériques
SRT |  | G-A195 | Perpendicular Axis | Axe orthogonal
DCM |  | 112123 | Phantom tumor (pseudotumor) | Image pseudo-tumorale
SRT |  | M-90201 | Phyllodes tumor | Tumeur phyllode
SRT |  | M-90203 | Phyllodes tumor, malignant | Sarcome phyllode (Cystosarcome phyllode malin)
SRT |  | J-004E8 | Physician | Médecin
SRT |  | A-12024 | Pin | Epingle
SRT |  | M-97313 | Plasmacytoma | Plasmocytome
SRT |  | D2-60302 | Plate-like atelectasis | Atélectasie plane
SRT |  | M-89400 | Pleomorphic adenoma | Adénome pléomorphe
DCM |  | 112081 | Pleonemia | Hypervascularisation
SRT |  | T-29000 | Pleural structure | Plèvres
SRT |  | D2-81180 | Pneumomediastinum | Pneumomédiastin
SRT |  | D2-80300 | Pneumothorax | Pneumothorax
SRT |  | R-428E7 | Poorly defined | Mal définies
DCM |  | 112141 | Poorly demarcated | Mal délimité
DCM |  | 112172 | Portacath | Chambre de perfusion implantable
DCM |  | 112011 | Positioner Primary Angle | Angle de positionnement primaire
DCM |  | 112012 | Positioner Secondary Angle | Angle de positionnement secondaire
DCM |  | 111209 | Positioning | Positionnement
DCM |  | 111291 | Post reduction mammoplasty | Mammoplastie après réduction
SRT |  | G-A120 | Postaxial | Postaxial
SRT |  | G-A106 | Posterior | Postérieur
SRT |  | G-A106 | Posterior | Postérieur
DCM |  | 112089 | Posterior junction line | Ligne médiastinale postérieure
SRT |  | T-28220 | Posterior segment of right upper lobe | Segment postérieur du lobe supérieur droit
DCM |  | 112092 | Posterior tracheal stripe | Bande trachéale postérieure
SRT |  | G-A182 | Posterolateral | Postéro-latéral
SRT |  | G-A121 | Preaxial | Pré-axial
DCM |  | 111532 | Pregnancy Status | Grossesse
DCM |  | 111151 | Presentation Optional: Rendering device may present | Présentation optionnelle
DCM |  | 111150 | Presentation Required: Rendering device is expected to present | Présentation requise
DCM |  | 121069 | Previous Finding | Résultat antérieur
DCM |  | 121068 | Previous Findings | Résultats antérieurs
DCM |  | 112059 | Primary complex | Complexe primaire
DCM |  | 110008 | Print | Imprimer
DCM |  | 121066 | Prior Procedure Descriptions | Description de la procédure précédente
DCM |  | 111047 | Probability of cancer | Probabilité de cancer
DCM |  | 121065 | Procedure Description | Description de la procédure
SRT |  | G-A140 | Profundis | Profondeur
DCM |  | 112151 | Profusion | Profusion
SRT |  | F-10310 | prone | Procubitus
SRT |  | A-04000 | Prosthesis | Prothèse
SRT |  | G-A118 | Proximal | Proximal
DCM |  | 111292 | Pseudoangiomatous stromal hyperplasia | Hyperplasie stromale pseudo-angiomateuse
DCM |  | 112068 | Pseudoplaque | Pseudo-plaque
SRT |  | T-44000 | Pulmonary artery | Artère pulmonaire
SRT |  | D3-40230 | Pulmonary embolism | Embolie pulmonaire
SRT |  | T-44100 | Pulmonary trunk | Tronc artériel pulmonaire
SRT |  | T-48500 | Pulmonary vein | Veine pulmonaire
SRT |  | F-01767 | Punctate calcification | Calcification punctiforme régulière
DCM |  | 111048 | Quadrant location | Localisation du quadrant
DCM |  | 111049 | Qualitative Difference | Différence qualitative
DCM |  | 111050 | Quality Assessment | Évaluation de la qualité
DCM |  | 110002 | Quality Control | Contrôle de qualité
DCM |  | 111051 | Quality Control Standard | Standard de contrôle de qualité
DCM |  | 111052 | Quality Finding | Critère de qualité
DCM |  | 113010 | Quality Issue | Problème de qualité
SRT |  | M-78731 | Radial scar | Cicatrice radiaire
DCM |  | 113921 | Radiation Exposure | Exposition aux rayonnements
DCM |  | 113923 | Radiation Exposure and Protection Information | Exposition aux rayonnements et informations de radiopro-tection
DCM |  | 113922 | Radioactive Substance Administered | Substance radioactive administrée
SRT |  | J-00187 | Radiographer | Manipulateur (rice)
DCM |  | 112005 | Radiographic anatomy | Radio-anatomie
LN |  | 11528-7 | Radiology Report | Compte rendu radiologique
SRT |  | G-A196 | Radius | Rayon
DCM |  | 112022 | RECIST | Critères d'évaluation de la réponse tumorale (tumeurs solides)
DCM |  | 121075 | Recommendation | Recommandation
DCM |  | 121074 | Recommendations | Recommandations
DCM |  | 111053 | Recommended Follow-up | Surveillance recommandée
DCM |  | 111054 | Recommended Follow-up Date | Date recommandée de surveillance
DCM |  | 111055 | Recommended Follow-up Interval | Intervalle recommandé de surveillance
DCM |  | 121097 | Recording | Qui fait le compte rendu
SRT |  | F-10450 | recumbent | Couché
DCM |  | 111338 | Recurrent malignancy | Cancer récidivant
UMLS |  | C1709880 | Referring | Médecin référent
SRT |  | F-01773 | Regional calcification distribution | Distribution régionale des calcifications
SRT |  | J-00172 | Registrar | Secrétaire
DCM |  | 113001 | Rejected for Quality Reasons | Rejetées pour des motifs de qualité
SRT |  | F-0172A | Removal of implant since previous mammogram | Exérèse de la prothèse mammaire depuis la mammographie précédente
DCM |  | 111056 | Rendering Intent | Intention d'insertion
DCM |  | 110007 | Report Verification | Vérification du compte rendu
DCM |  | 121062 | Request | Demande
DCM |  | 121096 | Requesting | Médecin demandeur
SRT |  | J-005E6 | Resident | Résident
DCM |  | 112020 | Response Evaluation | Evaluation de la réponse
DCM |  | 112021 | Response Evaluation Method | Méthode d'évaluation de la réponse
DCM |  | 112113 | Reticular pattern | Aspect réticulaire
DCM |  | 112065 | Reticulonodular pattern | Aspect réticulo-nodulaire
SRT |  | T-11300 | Rib | Côte
DCM |  | 112096 | Rib Scalene Tubercle | Tubercule scalénique de la première côte
SRT |  | G-A100 | Right | Droit
SRT |  | G-A102 | Right and left | Droit et gauche
SRT |  | T-04020 | Right breast | Sein droit
SRT |  | G-A100 | Right lateral | Latéral droit
SRT |  | F-10317 | right lateral decubitus | Décubitus latéral droit
SRT |  | T-26100 | Right main bronchus | Bronche principale droite
DCM |  | 112093 | Right tracheal stripe | Bande paratrachéale droite
SRT |  | R-102D3 | Rolled Lateral | Roulé externe
SRT |  | R-102D4 | Rolled Medial | Roulé interne
SRT |  | M-02100 | Round shape | Ronde
SRT |  | F-01768 | Round shaped calcification | Calcification ronde
SRT |  | G-A145 | Sagittal | Sagittal
SRT |  | T-13450 | Scalenous anterior muscle | Muscle scalène antérieur
SRT |  | T-12280 | Scapula | Scapula
DCM |  | 112101 | Scapular Infraspinatus Fossa | Fosse sous épineuse
DCM |  | 112099 | Scapular Spine | Epine de l'omoplate
DCM |  | 112100 | Scapular Supraspinatus Fossa | Fosse sus épineuse
SRT |  | M-78060 | Scar tissue | Tissu cicatriciel
SRT |  | F-01712 | Scattered fibroglandular densities | Opacités fibro-glandulaires éparses
SRT |  | M-74220 | Sclerosing adenosis | Adénose sclérosante
DCM |  | 111057 | Scope of Feature | Champ des caractéristiques
DCM |  | 112054 | Secondary pulmonary lobule | Lobule pulmonaire secondaire
SRT |  | M-85023 | Secretory (juvenile) carcinoma of the breast | Carcinome mammaire sécrétoire (juvénile)
SRT |  | T-280D0 | Segment of lung | Segmentdu poumon
SRT |  | G-A137 | Segmental | Segmentaire
SRT |  | F-01774 | Segmental calcification distribution | Segmentaires
DCM |  | 111099 | Selected region | Région sélectionnée
DCM |  | 111058 | Selected Region Description | Description de la région sélectionnée
SRT |  | F-10460 | semi-erect | Semi-couché
SRT |  | F-10316 | semi-prone | Semi-procubitus
DCM |  | 112114 | Septal line(s) | Ligne(s) septale(s)
DCM |  | 112002 | Series Instance UID | Identificateur unique d'instance de série
SRT |  | M-36050 | Seroma | Lymphocèle
SRT |  | T-14140 | Serratus anterior muscle | Muscle dentelé antérieur
SRT |  | G-A003 | Severe | Sévère
SRT |  | G-C197 | Severity | Gravité
DCM |  | 112124 | Shadow | Image
SRT |  | T-11309 | Shaft of rib | Corps de la côte
SRT |  | M-020F9 | Shape | Forme
DCM |  | 112137 | Sharply defined | A limites nettes
DCM |  | 112140 | Sharply demarcated | Très nettement délimité
SRT |  | G-A186 | Short Axis | Petit axe
SRT |  | M-84903 | Signet ring cell carcinoma | Carcinome à cellules en bague à chaton
DCM |  | 112069 | Signet-ring sign | Signe de la bague à châton
DCM |  | 112152 | Silhouette sign | Signe de la silhouette
DCM |  | 111296 | Silicone granuloma | Granulome au silicone
DCM |  | 111059 | Single Image Finding | Elément présent sur une seule image
DCM |  | 112024 | Single Image Finding Modifier | Modificateur lié à une anomalie visible sur une seule image
DCM |  | 112008 | Site involvement | Site atteint
SRT |  | F-103A0 | sitting | Assis
DCM |  | 112025 | Size Descriptor | Descripteur de la taille
SRT |  | D0-00050 | Skin lesion | Lésion cutanée
SRT |  | F-01799 | Skin retraction of breast | Rétraction cutanée du sein
SRT |  | F-0179A | Skin thickening of breast | Épaississement cutané du sein
SRT |  | R-404A8 | Small | Petit
DCM |  | 112125 | Small irregular opacities | Petites opacités irrégulières
DCM |  | 112126 | Small rounded opacities | Micro-nodules
DCM |  | 112144 | Soft tissue | Tissus mous
DCM |  | 111218 | Software failure | Défaillance logicielle
SRT |  | P5-B3402 | Spatial collocation analysis | Analyse de colocalisation spatiale
SRT |  | P5-B3404 | Spatial proximity analysis | Analyse de proximité spatiale
DCM |  | 112136 | Spiculated | Spiculée
SRT |  | F-01745 | Spiculated lesion | Lésion spiculée
SRT |  | T-14050 | Spinalis muscle | Muscles spinaux
SRT |  | M-78190 | Spindle cell nodule (tumor) | Nodule (tumeur) à cellules fusiformes
SRT |  | T-11500 | Spine | Rachis
SRT |  | T-11512 | Spinous process of vertebra | Apophyse épineuse de la vertèbre
SRT |  | R-102D7 | Spot Compression | Compression localisée
SRT |  | R-102D7 | Spot compression | Compression localisée
DCM |  | 111136 | Spot magnification view(s) | Agrandissement localisé
SRT |  | M-80703 | Squamous cell carcinoma | Carcinome épidermoïde
DCM |  | 111340 | Squamous cell carcinoma of the nipple | Carcinome épidermoïde du mamelon
UCUM |  | cm2 | Square centimeter | Centimètre carré
UCUM |  | um2 | Square micrometer | Micromètre carré
UCUM |  | mm2 | Square millimeter | Millimètre carré
DCM |  | 112183 | Standard Deviation of Attenuation Coefficient | Ecart-type des coefficients d'atténuation
SRT |  | F-10320 | standing | En position verticale
SRT |  | A-13600 | Staple | Agrafe
SRT |  | T-11221 | Sternal angle | Angle sternal
SRT |  | T-13310 | Sternocleidomastoid muscle | Muscle sterno-cleïdo-mastoïdien
SRT |  | T-11210 | Sternum | Sternum
SRT |  | F-10390 | stooped-over | Penché en avant
DCM |  | 112094 | Stripe | Bande
DCM |  | 111060 | Study Date | Date de l'étude
DCM |  | 111061 | Study Time | Heure de l'étude
SRT |  | G-A561 | Subacute | Subaigu
SRT |  | F-0178D | Subareolar position | Situation rétroaréolaire
SRT |  | G-A172 | Subcapsular | Sous-capsulaire
SRT |  | T-46100 | Subclavian artery | Artère subclavière
SRT |  | T-48330 | Subclavian vein | Veine subclavière
SRT |  | T-14166 | Subcostal muscle | Muscle subcostal
DCM |  | 112153 | Subpleural | Sous-pleural
DCM |  | 112115 | Subpleural line | Ligne sous-pleurale
DCM |  | 112098 | Subscapular Fossa | Fosse subscapulaire
SRT |  | T-13650 | Subscapularis muscle | Muscle subscapulaire
DCM |  | 111222 | Succeeded | Succès
DCM |  | 111062 | Successful Analyses | Analyses réussies
DCM |  | 111063 | Successful Detections | Procédures de détection réussies
DCM |  | 111146 | Suggestive of malignancy - take appropriate action | Évocateur de malignité, une action appropriée doit être entreprise
DCM |  | 111065 | Summary of Analyses | Résumé des analyses
DCM |  | 111064 | Summary of Detections | Résumé des procédures de détections
SRT |  | G-A139 | Superficial | Superficiel
SRT |  | G-A116 | Superior | Supérieur
SRT |  | T-116EE | Superior articular facet of axis | Facette articulaire supérieure de l'axis
SRT |  | T-1153E | Superior articular process of vertebra | Massif articulaire supérieur
SRT |  | T-46350 | Superior phrenic artery | Artère phrénique supérieure
SRT |  | T-48610 | Superior vena cava | Veine cave supérieure
SRT |  | R-102D0 | superolateral to inferomedial oblique | Supérolatéral vers inféromédial oblique
SRT |  | F-10340 | supine | Décubitus
SRT |  | T-13610 | Supraspinatus muscle | Muscle supraépineux
SRT |  | T-11218 | Suprasternal notch | Creux sus-sternal
SRT |  | G-A168 | Surface | Surface
SRT |  | A-13510 | Suture material | Matériel de suture
SRT |  | G-A572 | Systemic | Systémique
SRT |  | T-4000E | Systemic vascular structure | Structure vasculaire systémique
SRT |  | R-102C2 | Tangential | Tangentiel
DCM |  | 112162 | Target | « cible »
DCM |  | 111155 | Target content items are related contra-laterally | Les items de contenu sont situés de façon controlatérale
DCM |  | 111154 | Target content items are related spatially | Les items de contenu sont reliés spatialement
DCM |  | 111153 | Target content items are related temporally | Les items de contenu sont reliés temporellement
DCM |  | 112074 | Target Lesion at Baseline | Lésion « cible » à T0
DCM |  | 112041 | Target Lesion Complete Response | Réponse complète sur lésions « cibles »
DCM |  | 112042 | Target Lesion Partial Response | Réponse partielle sur lésions « cibles »
DCM |  | 112043 | Target Lesion Progressive Disease | Progression de la maladie sur lésions « cibles »
DCM |  | 112044 | Target Lesion Stable Disease | Maladie stable sur ésions « cibles»
DCM |  | 123014 | Target Region | Région cible
DCM |  | 111194 | Technical factors missing | Paramètres techniques absents
SRT |  | J-00187 | Technologist | Technicien
SRT |  | P5-B3406 | Temporal correlation | Corrélation temporelle
SRT |  | T-13640 | Teres major muscle | Muscle grand rond
SRT |  | T-13630 | Teres minor muscle | Muscle petit rond
DCM |  | 112010 | Texture Descriptor | Descripteur de la texture
SRT |  | T-C6510 | Thoracic Duct | Canal thoracique
DCM |  | 112032 | Threshold Attenuation Coefficient | Valeur de coefficient d'atténuation seuil
SRT |  | D3-87780 | Thrombophlebitis of breast (Mondor's disease) | Thrombophlébite du sein (maladie de Mondor)
SRT |  | T-C8000 | Thymus Gland | Thymus
SRT |  | T-46130 | Thyrocervical trunk | Tronc thyro-bicervico-scapulaire
SRT |  | T-B6000 | Thyroid | Thyroïde
DCM |  | 112133 | Too small | Trop petit
SRT |  | T-32423 | Trabeculae carnae | Piliers du ventricule
SRT |  | F-01798 | Trabecular thickening of breast | Épaississement trabéculaire du sein
SRT |  | T-25000 | Trachea | Trachée
SRT |  | P1-26100 | Tracheotomy | Trachéotomie
DCM |  | 112039 | Tracking Identifier | Identifiant d'anomalie
DCM |  | 112040 | Tracking Unique Identifier | Identifiant unique d'anomalie
DCM |  | 112116 | Tramline shadow | Image en rail
DCM |  | 110006 | Transcription (task) | Transcription
DCM |  | 110012 | Transcription (type of output) | Transcription
SRT |  | G-A117 | Transverse | Transverse
SRT |  | T-11513 | Transverse process of vertebra | Apophyse transverse de la vertèbre
SRT |  | T-141A5 | Transversus thoracis | Muscle transverse du thorax
SRT |  | T-14171 | Trapezius muscle | Muscle trapèze
DCM |  | 112127 | Tree-in-bud sign | Signe de l'arbre en bourgeons
SRT |  | F-10348 | Trendelenburg | Trendelenburg
SRT |  | T-35100 | Tricuspid Valve | Valve atrioventriculaire droite
SRT |  | T-11304 | Tubercle of rib | Tubercule de la côte
SRT |  | M-82113 | Tubular adenocarcinoma | Carcinome tubuleux
SRT |  | M-82110 | Tubular adenoma | Adénome tubuleux
SRT |  | F-01797 | Tubular density | Opacité tubulaire
DCM |  | 112117 | Tubular shadow | Image tubulée
DCM |  | 112009 | Type of Content | Type de contenu
SRT |  | P5-B0099 | Ultrasound procedure | Procédure échographique
LN |  | 18760-9 | Ultrasound Report | Compte rendu d'échographie
DCM |  | 111211 | Under exposed | Sous-exposé
SRT |  | G-A103 | Unilateral | Unilatéral
SRT |  | G-A103 | Unilateral | Unilatéral
DCM |  | 111221 | Unknown failure | Défaillance inconnue
DCM |  | 111176 | Unspecified | Non spécifié
DCM |  | 112187 | Unspecified method of calculation | Méthode de calcul non spécifiée
DCM |  | 111235 | Unusable - Quality renders image unusable | Inexploitable - La qualité rend l'image inexploitable
SRT |  | G-A116 | Upper | En haut
SRT |  | T-D4001 | Upper abdomen | Abdomen supérieur
SRT |  | T-04002 | Upper inner quadrant of breast | Quadrant supéro-interne du sein
SRT |  | T-04002 | Upper inner quadrant of breast, NOS | Quadrant supéro-interne du sein, SAI
SRT |  | T-28820 | Upper lobe of lung | Lobe supérieur du poumon
SRT |  | T-04004 | Upper outer quadrant of breast | Quadrant supéro-externe du sein
SRT |  | T-04004 | Upper outer quadrant of breast, NOS | Quadrant supéro-externe du sein, SAI
SRT |  | D-3208 | Upper zone of lung | Zone supérieure du poumon
SRT |  | A-11C08 | Ureteric stent | Stent urétral
DCM |  | 111236 | Usable - Does not meet the quality control standard | Exploitable - Ne répond pas aux standards de contrôle de qualité
DCM |  | 111237 | Usable - Meets the quality control standard | Exploitable - Répond aux standards de contrôle de qualité
SRT |  | F-0176B | Vascular calcification | Calcification vasculaire
DCM |  | 112077 | Vasoconstriction | Vasoconstriction
DCM |  | 112078 | Vasodilation | Vasodilatation
SRT |  | A-14611 | Vena cava filter | Filtre cave
SRT |  | G-A105 | Ventral | Ventral
SRT |  | T-32400 | Ventricle | Ventricule
DCM |  | 121098 | Verifying | Qui vérifie
SRT |  | T-11510 | Vertebra | Vertèbre
SRT |  | T-45700 | Vertebral artery | Artère vertébrale
SRT |  | T-1151F | Vertebral canal | Canal vertébral
SRT |  | T-11531 | Vertebral foramen | Foramen intervertébral
DCM |  | 112097 | Vertebral Intervertebral Notch | Trou des apophyses transverses cervicales
SRT |  | G-A144 | Vertical | Vertical
SRT |  | G-A144 | Vertical | Vertical
DCM |  | 111066 | Vertical Pixel Spacing | Espacement vertical des pixels
DCM |  | 112132 | Very small | Très petit
DCM |  | 111178 | View and Laterality Marker does not have both view and laterality | Le marquage n'indique ni l'incidence ni le côté
DCM |  | 111183 | View and Laterality Marker is incorrect | Le marquage est incorrect
DCM |  | 111177 | View and Laterality Marker is missing | Marquage absent
DCM |  | 111180 | View and Laterality Marker is not near the axilla | Le marquage n'est pas près de l'aisselle
DCM |  | 111184 | View and Laterality Marker is off image | Le marquage est en dehors du film
DCM |  | 111182 | View and Laterality Marker is partially obscured | Le marquage est partiellement masqué
DCM |  | 111181 | View and Laterality Marker overlaps breast tissue | Le marquage chevauche le sein
DCM |  | 111298 | Virginal hyperplasia | Hypertrophie juvénile
SRT |  | G-D705 | Volume | Volume
DCM |  | 121216 | Volume estimated from single 2D region | Volume estimé à partir d'une seule région 2D
DCM |  | 121217 | Volume estimated from three or more non-coplanar 2D regions | Volume estimé à partir de trois régions 2D non coplanaires ou plus
DCM |  | 121218 | Volume estimated from two non-coplanar 2D regions | Volume estimé à partir de deux régions 2D non coplanaires
DCM |  | 121219 | Volume of bounding three dimensional region | Volume d'une région tridimensionnelle de forme quelconque
DCM |  | 121220 | Volume of circumscribed sphere | Volume de la sphère circonscrite
DCM |  | 121221 | Volume of ellipsoid | Volume d'un éllipsoïde
DCM |  | 121222 | Volume of sphere | Volume d'une sphère
DCM |  | 112056 | Volume scoring method | Score de calcification coronaire basé sur le volume de chaque calcification
UCUM |  | wk | Week | Semaine
SRT |  | R-40771 | Well defined | Bien définie
DCM |  | 112139 | Well demarcated | Bien délimité
DCM |  | 112029 | WHO | OMS
SRT |  | G-A220 | Width | Largeur
DCM |  | 112026 | Width Descriptor | Descripteur de la largeur
SRT |  | T-11227 | Xiphoid process of sternum | Appendice xiphoïde
UCUM |  | a | Year | Année

### Note

1. DAO = Détection Assistée par Ordinateur

2. In (113006, DCM, "For Therapy"), therapy could be translated as " thérapeutique " as well as " traitement ". There is an issue with the word " traitement " because it is the same word used for image processing. To avoid any ambiguity we have chosen the word " thérapeutique ", which is less used in common language.

Table E-2 provides a mapping of pathology codes used in DICOM, to ADICAP ( L'association pour le Développement de l'Informatique en Anatomie et Cytologie Pathologiques ).

Table E-2. Mapping of Pathology Codes used in DICOM to ADICAP

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning French Language | Equivalent ADICAP Code
--- | --- | --- | --- | ---
SRT |  | M-55160 | (Tumeur) amyloïde | 5310
SRT |  | M-83240 | Adénolipome | A0L2
DCM |  | 111258 | Adénome ductal | A0B2
SRT |  | M-82040 | Adénome lactant | A0M2
SRT |  | M-89400 | Adénome pléomorphe | A0R8
SRT |  | M-82110 | Adénome tubuleux | A0P1
DCM |  | 111250 | Adénomyoépithéliome | A0A0
SRT |  | M-74200 | Adénose | 6772
DCM |  | 111284 | Adénose microglandulaire | 6772
SRT |  | M-74220 | Adénose sclérosante | 6772
SRT |  | M-88610 | Angiolipome | L0P1
SRT |  | M-76100 | Angiomatose | V0C0
SRT |  | M-91203 | Angiosarcome (hémangiosarcome) | V7A0
SRT |  | M-84803 | Carcinome (mucineux) colloïde | A7N4
SRT |  | M-82003 | Carcinome adénoïde kystique (cylindrome) | A7X6
SRT |  | M-84013 | Carcinome apocrine | A7K6
DCM |  | 111307 | Carcinome basocellulaire du mamelon | B7A0
SRT |  | M-85003 | Carcinome canalaire infiltrant | A7A0
DCM |  | 111340 | Carcinome épidermoïde du mamelon | E7A0
DCM |  | 111341 | Carcinome intracanalaire | A5B2
SRT |  | D7-F0A02 | Carcinome lobulaire in situ mammaire | A5B0
SRT |  | M-85203 | Carcinome lobulaire infiltrant | A7B1
SRT |  | M-85023 | Carcinome mammaire sécrétoire (juvénile) | A7N7
SRT |  | M-85103 | Carcinome médullaire | A7X2
SRT |  | M-85733 | Carcinome métaplasique | A7W0
SRT |  | M-80503 | Carcinome papillaire infiltrant | A7C6
SRT |  | M-82113 | Carcinome tubuleux | A7F0
SRT |  | M-92200 | Chondrome | C0A0
SRT |  | M-92203 | Chondrosarcome | C7A0
SRT |  | M-78731 | Cicatrice radiaire | 6773
SRT |  | D7-90434 | Cytostéatonécrose mammaire | 5230
SRT |  | M-78800 | Fibomatose | F0F0
SRT |  | M-90100 | Fibroadénome | A0P2
SRT |  | M-90300 | Fibroadénome juvénile | A0P2
SRT |  | M-88103 | Fibrosarcome | F7A0
SRT |  | D7-90370 | Galactophorite ectasiante mammaire (ectasie canalaire mammaire) | 6546
SRT |  | D7-90420 | Gynécomastie | 6551
SRT |  | M-75500 | Hamartome | D0S0
SRT |  | M-91200 | Hémangiome | V0A0
SRT |  | D3-F0620 | Hémangiome sous-cutané non parenchymateux | V0A0
SRT |  | M-91220 | Hémangiome veineux | VOA8
SRT |  | M-91501 | Hémangiopéricytome | V0K0
SRT |  | M-72170 | Hyperplasie canalaire | 6712
SRT |  | M-72175 | Hyperplasie intracanalaire atypique | 6830
SRT |  | M-72105 | Hyperplasie lobulaire atypique | 6840
SRT |  | D7-90428 | Hyperplasie lobulaire mammaire | 6721
DCM |  | 111298 | Hypertrophie juvénile | 6080
SRT |  | D7-90452 | Infarctus mammaire | 4710
SRT |  | M-40000 | Infection | 7140
SRT |  | D7-90360 | Kyste du sein | 6544
SRT |  | M-88900 | Léiomyome | L0A0
SRT |  | M-88903 | Léiomyosarcome | L7A0
SRT |  | M-88500 | Lipome | L0L0
SRT |  | M-95913 | Lymphome non hodgkinien | K7G0
SRT |  | M-96503 | Maladie de Hodgkin | K7A0
SRT |  | M-85403 | Maladie de Paget du mamelon | A7B7
DCM |  | 111259 | Mastopathie diabétique | 5010
DCM |  | 111334 | Mélanome malin du mamelon | M7A0
SRT |  | M-95400 | Neurofibrome | N0L0
SRT |  | M-91803 | Ostéosarcome | Q7A0
SRT |  | M-80500 | Papillome | A0P4 (unique), A0S4 (multiple)
SRT |  | M-97313 | Plasmocytome | K7M0
SRT |  | M-44140 | Réaction à corps étranger | 7440
SRT |  | M-90203 | Sarcome phyllode (Cystosarcome phyllode malin) | A7P6
SRT |  | M-95800 | Tumeur à cellules granuleuses | X0H4
SRT |  | M-90201 | Tumeur phyllode | A0P6

# F Japanese Language Meanings of Selected Codes Used in The DCMR (Normative)

Table F-1. Japanese Language Meanings of Selected Codes

Coding Scheme Designator | Coding Scheme Version | Code Value | Code Meaning English Language | Code Meaning Japanese Language
--- | --- | --- | --- | ---
BI | 3.0 | II.AC.a | 0 - Need additional imaging evaluation | ０－追加撮影が必要
BI | 3.0 | II.AC.b.1 | 1 - Negative | １－異常なし
SRT |  | F-01781 | 1 o'clock position | 1時
SRT |  | F-0178A | 10 o'clock position | １０時
SRT |  | F-0178B | 11 o'clock position | １１時
SRT |  | F-0178C | 12 o'clock position | １２時
BI | 3.0 | II.AC.b.2 | 2 - Benign Finding | ２－良性所見
SRT |  | F-01782 | 2 o'clock position | 2時
BI | 3.0 | II.AC.b.3 | 3 - Probably Benign Finding - short interval follow-up | ３－良性―しかし悪性を否定できず所見―短い間隔での経過観察が必要
SRT |  | F-01783 | 3 o'clock position | 3時
BI | 3.0 | II.AC.b.4 | 4 - Suspicious abnormality, biopsy should be considered | ４－悪性の疑い、生検を考慮
SRT |  | F-01784 | 4 o'clock position | 4時
BI | 3.0 | II.AC.b.5 | 5 - Highly suggestive of malignancy, take appropriate action | ５－悪性、適切な処置が必要
SRT |  | F-01785 | 5 o'clock position | 5時
SRT |  | F-01786 | 6 o'clock position | ６時
SRT |  | F-01787 | 7 o'clock position | ７時
SRT |  | F-01788 | 8 o'clock position | ８時
SRT |  | F-01789 | 9 o'clock position | ９時
DCM |  | 111135 | Additional projections | 追加撮影（P）
SRT |  | M-82003 | Adenoid cystic carcinoma | 嚢胞腺癌
SRT |  | M-83240 | Adenolipoma | 腺脂肪腫
SRT |  | M-89830 | Adenomyoepithelioma | 腺筋上皮腫
SRT |  | M-74200 | Adenosis | 腺症
DCM |  | 111001 | Algorithm Name | アルゴリズム 名
DCM |  | 111002 | Algorithm Parameters | アルゴリズム・パラメータ
DCM |  | 111003 | Algorithm Version | アルゴリズム・バージョン（版番号）
DCM |  | 111242 | All algorithms succeeded; with findings | 全てのアルゴリズムが成功；所見あり
DCM |  | 111241 | All algorithms succeeded; without findings | 全てのアルゴリズムが成功；所見なし
SRT |  | F-01711 | Almost entirely fat | 脂肪性
SRT |  | F-0176C | Amorphous calcification | 淡く不明瞭な
SRT |  | M-55160 | Amyloid (tumor) | アミロイド腫瘍
DCM |  | 111004 | Analysis Performed | 解析済みの
SRT |  | M-88610 | Angiolipoma | 血管脂肪腫
SRT |  | M-76100 | Angiomatosis | 血管腫症
SRT |  | M-91203 | Angiosarcoma | 血管肉腫
SRT |  | G-A105 | Anterior | 前方の
DCM |  | 111141 | Any decision to biopsy should be based on clinical assessment | 臨床評価に基づいた生検の適応決定（D）
SRT |  | M-84013 | Apocrine adenocarcinoma | アポクリン癌
SRT |  | F-01795 | Architectural distortion of breast | 乳房の構築の乱れ
DCM |  | 111215 | Artifact(s) other than grid or detector artifact | 検出器のアーチファクト以外のアーチファクト
DCM |  | 111005 | Assessment Category | カテゴリー評価
SRT |  | F-01793 | Asymmetric breast tissue | 非対称性乳房組織
SRT |  | P5-B3412 | Asymmetric breast tissue analysis | 非対称性乳房組織解析
SRT |  | F-8A063 | Asynchronous involution of breast | 乳房の非同期性退縮
SRT |  | M-72175 | Atypical intraductal hyperplasia | 異型性乳管過形成；異型性乳管内過形成
SRT |  | M-72105 | Atypical lobular hyperplasia | 異型性小葉過形成
BI | 3.0 | I.E.6 | Axillary adenopathy | 腋窩リンパ節腫大
SRT |  | F-0178E | Axillary tail position | 腋窩稜：乳腺の腋窩稜（C'領域）
DCM |  | 111307 | Basal cell carcinoma of the nipple | 乳頭の基底細胞癌
SRT |  | A-32475 | BB shot (Lead Pellet) | 鉛小球；BBマーカー
DCM |  | 111143 | Biopsy should be considered | 要生検（B）
SRT |  | T-04080 | Both breasts | 両側：両側乳房
SRT |  | F-01710 | Breast composition | 乳房の構成
SRT |  | P5-B3414 | Breast composition analysis | 乳房の構成の解析
DCM |  | 111100 | Breast geometry | 乳房の形状
SRT |  | D7-90428 | Breast lobular hyperplasia | 小葉過形成：乳腺小葉過形成
DCM |  | 111007 | Breast Outline including Pectoral Muscle Tissue | 胸筋組織を含む乳房の輪郭
SRT |  | A-32110 | Bullet | マーカー
DCM |  | 111017 | CAD Processing and Findings Summary | CAD処理と所見の要約
SRT |  | F-01775 | Calcification Cluster | 石灰化の集簇
DCM |  | 111008 | Calcification Distribution | 石灰化の分布
DCM |  | 111009 | Calcification Type | 石灰化のタイプ
SRT |  | F-01769 | Calcified skin of breast | 皮膚；乳房の皮膚
DCM |  | 111304 | Carcinoma in children | 小児乳癌
DCM |  | 111305 | Carcinoma in ectopic breast | 副乳の乳癌
DCM |  | 111310 | Carcinoma in pregnancy and lactation | 妊娠・授乳期乳癌
SRT |  | D7-F0902 | Carcinoma in situ of male breast | 男性乳癌
DCM |  | 111306 | Carcinoma with endocrine differentiation | 内分泌分化を伴う癌
SRT |  | M-85733 | Carcinoma with metaplasia | 化生を伴う癌
SRT |  | A-26800 | Catheter | カテーテル
DCM |  | 111203 | CC Nipple not centered on image | 頭尾方向撮影　乳頭が画像の中央にない
DCM |  | 111202 | CC Not all medial tissue visualized | 頭尾方向撮影　内側組織が十分見えていない
DCM |  | 111204 | CC Posterior nipple line does not measure within 1 cm of MLO | 頭尾方向撮影　乳頭後方線が内外斜位方向の１ｃｍ以内に計測できない
DCM |  | 111010 | Center | 中心部
SRT |  | F-0178F | Central portion of breast position | 中央部：乳腺の中央部
DCM |  | 111011 | Certainty of Feature | 特徴の確信度
DCM |  | 111012 | Certainty of Finding | 所見の確信度
DCM |  | 111013 | Certainty of Impression | インプレッションの確信度
SRT |  | M-92200 | Chondroma | 軟骨腫
SRT |  | M-92203 | Chondrosarcoma | 軟骨肉腫
SRT |  | F-01741 | Circumscribed lesion | 境界明瞭平滑
SRT |  | A-12062 | Clip | クリップ
DCM |  | 111014 | Clockface or region | 時計表示あるいは領域
SRT |  | F-01761 | Coarse (popcorn-like) calcification | 粗大（ポップコーン状）
DCM |  | 111195 | Collimation too close to breast | コリメーションが乳房に近すぎる
SRT |  | A-1044 | Collimator | コリメータ
DCM |  | 111015 | Composite Feature | 乳房の構成の特徴
DCM |  | 111016 | Composite type | 乳房の構成のタイプ
DCM |  | 111018 | Content Date | 記録日
DCM |  | 111019 | Content Time | 記録時間
SRT |  | C-B0300 | Contrast agent NOS | 造影剤
SRT |  | D7-90360 | Cyst of breast | 嚢胞：乳腺嚢胞
DCM |  | 111147 | Cytologic analysis | 細胞診（Y）
DCM |  | 111193 | Date sticker is missing | 日付けステッカーがない
UCUM |  | d | Day | 日
SRT |  | F-01727 | Decrease in number of calcifications | 石灰化の数の減少
SRT |  | M-02530 | Decrease in size | サイズの縮小
SRT |  | F-01796 | Mammography breast density | 乳房画像の濃度
DCM |  | 111020 | Depth | 深さ（三次元表示の奥行き）
DCM |  | 111021 | Description of Change | 変化の記載
DCM |  | 111022 | Detection Performed | 検出済みの
DCM |  | 111214 | Detector artifact(s) | 検出器のアーチファクト
DCM |  | 111259 | Diabetic fibrous mastopathy | 糖尿病性乳腺症
SRT |  | F-017B3 | Difference in location | 部位
SRT |  | F-05179 | Difference in location | 部位
SRT |  | F-017B7 | Difference in margin | 辺縁
SRT |  | F-017B5 | Difference in number of calcifications | 石灰化の数
SRT |  | F-017B2 | Difference in opacity | 濃度
SRT |  | F-017B6 | Difference in shape | 形状
SRT |  | F-017B1 | Difference in size | 大きさ
SRT |  | F-05173 | Difference in size | 大きさ
SRT |  | F-017B4 | Difference in spatial proximity | 空間的近接判定
SRT |  | F-017B8 | Difference in symmetry | 対称性
DCM |  | 111023 | Differential Diagnosis/Impression | 鑑別診断/インプレッション
SRT |  | F-01770 | Diffuse calcification distribution | びまん性／散在性
DCM |  | 111258 | Ductal adenoma | 乳管腺腫
SRT |  | M-72170 | Ductal hyperplasia, Usual | 乳管過形成；乳管内過形成
SRT |  | P5-40060 | Mammary ductogram | 乳房造影 （G）
SRT |  | F-01762 | Dystrophic calcification | 異栄養性；異栄養性石灰化
SRT |  | F-01763 | Eggshell calcification | 卵殻状
DCM |  | 111217 | Electrical failure | 電気系の故障
SRT |  | F-01752 | Equal density (isodense) lesion | 等濃度
SRT |  | F-01714 | Extremely dense | 高濃度
DCM |  | 111224 | Failed | 失敗
DCM |  | 111024 | Failed Analyses | 解析の失敗
DCM |  | 111025 | Failed Detections | 検出の失敗
SRT |  | F-01754 | Fat containing (radiolucent) lesion | 脂肪濃度を含む（X線透亮性）
SRT |  | D7-90434 | Fat necrosis of breast | 脂肪壊死：乳房の脂肪壊死
DCM |  | 111159 | Feature detected on images from multiple modalities | 多数の検査法で検出される特徴
DCM |  | 111158 | Feature detected on multiple images | 多数の画像で検出される特徴
DCM |  | 111157 | Feature detected on only one of the images | １画像でのみ検出される特徴
DCM |  | 111156 | Feature detected on the only image | 画像のみで検出される特徴
SRT |  | M-90100 | Fibroadenoma | 線維腺腫
DCM |  | 111263 | Fibroadenomatoid hyperplasia | 線維腺腫様過形成：腺線維筋腫様過形成
SRT |  | M-78800 | Fibromatosis | 線維腫症
SRT |  | M-88103 | Fibrosarcoma | 線維肉腫
DCM |  | 111072 | Finding partially removed | 部分的に消失した所見
SRT |  | F-0176D | Fine, linear (casting) calcification | 微細線状
SRT |  | F-0176E | Fine, linear, branching (casting) calcification | 微細線状分枝状
DCM |  | 111191 | Flash doesn't include cassette/screen/detector identification | 患者情報等欄にカセッテ／スクリーン/検出器名がない
DCM |  | 111188 | Flash doesn't include date of examination | 患者情報等欄に検査日がない
DCM |  | 111189 | Flash doesn't include facility name and location | 患者情報等欄に施設名と所在地がない
DCM |  | 111192 | Flash doesn't include mammography unit identification | 患者情報等欄に乳房撮影装置名がない
DCM |  | 111187 | Flash doesn't include patient name and additional patient id | 患者情報等欄に患者の氏名および追加情報がない
DCM |  | 111190 | Flash doesn't include technologist identification | 患者情報等欄に技師名がない
DCM |  | 111186 | Flash is illegible, does not fit, or is lopsided | 患者情報等欄が読みにくい、大きさがあっていない、あるいは傾いている
DCM |  | 111185 | Flash is not near edge of film | 患者情報等欄がフィルムの端にない
SRT |  | F-01792 | Focal asymmetric breast tissue | 局所性非対称性乳房組織
SRT |  | P5-B3410 | Focal asymmetric density analysis | 局所性非対称性陰影
DCM |  | 111142 | Follow-up at short interval (1-11 months) | 短期間での経過観察（１－１１ヶ月）（F）
SRT |  | M-44140 | Foreign body (reaction) | 異物反応
SRT |  | M-83153 | Glycogen-rich carcinoma | グリコーゲンに富む癌
SRT |  | M-95800 | Granular cell tumor | 顆粒細胞腫
DCM |  | 111208 | Grid artifact(s) | グリッドのアーチファクト
SRT |  | F-01772 | Grouped calcification distribution | 集簇性
SRT |  | D7-90420 | Gynecomastia | 女性化乳房
SRT |  | M-75500 | Hamartoma | 過誤腫
SRT |  | M-91200 | Hemangioma | 血管腫
SRT |  | D3-F0620 | Hemangioma of subcutaneous tissue | 非実質性皮下組織血管腫
SRT |  | M-91220 | Hemangioma - venous | 静脈性血管腫
SRT |  | M-91501 | Hemangiopericytoma | 血管周皮腫
SRT |  | F-0176F | Heterogeneous calcification | 不均一なあるいは多形性の
SRT |  | F-01713 | Heterogeneously dense | 不均一高濃度
SRT |  | F-01751 | High density lesion | 高濃度
DCM |  | 111145 | Histology using core biopsy | コア針生検（H）
SRT |  | M-96503 | Hodgkin's disease (lymphoma) | ホジキン病
DCM |  | 111026 | Horizontal Pixel Spacing | 水平方向ピクセル間隔
SRT |  | A-16016 | ID Plate | IDプレート
DCM |  | 111027 | Image Laterality | 画像の左右差
DCM |  | 111028 | Image Library | 画像ライブラリ
DCM |  | 111101 | Image Quality | 画像の品質
SRT |  | P5-B3408 | Image quality analysis | 画像の品質解析
DCM |  | 111029 | Image Quality Rating | 画質のランク付
DCM |  | 111030 | Image Region | 画像領域
DCM |  | 111031 | Image View | 画像表示用符号変換系列
DCM |  | 111032 | Image View Modifier | 画像表示用符号系列
SRT |  | A-04010 | Implant | インプラント
SRT |  | F-0172B | Implant revised since previous mammogram | インプラントの修正
DCM |  | 111033 | Impression Description | インプレッションの記載
DCM |  | 111196 | Inadequate compression | 圧迫不良
DCM |  | 111219 | Inappropriate image processing | 現像機の故障
SRT |  | F-01726 | Increase in number of calcifications | 石灰化の数の増加
SRT |  | M-02520 | Increase in size | サイズの増大
SRT |  | F-01744 | Indistinct lesion | 境界不明瞭
SRT |  | F-01776 | Individual Calcification | 個々の石灰化
DCM |  | 111233 | Individual Impression / Recommendation Analysis | 個々のインプレッション／推奨の解析
DCM |  | 111034 | Individual Impression/Recommendation | 個々のインプレッション／推奨
SRT |  | D7-90452 | Infarction of breast | 梗塞：乳腺の梗塞
SRT |  | M-40000 | Inflammation | 感染
SRT |  | M-85303 | Inflammatory carcinoma | 炎症性乳癌
DCM |  | 111206 | Insufficient implant displacement incorrect | インプラントの圧排不十分
DCM |  | 111341 | Intraductal carcinoma, high grade | 非浸潤性乳管癌：ＤＣＩＳ
SRT |  | T-C4351 | Intra-mammary lymph node | 乳房内リンパ節
SRT |  | M-82013 | Invasive cribriform carcinoma | 浸潤性篩状癌
SRT |  | M-85003 | Infiltrating duct carcinoma | 浸潤性乳管癌
SRT |  | M-85203 | Invasive lobular carcinoma | 浸潤性小葉癌
SRT |  | G-A402 | Irregular | 不整形
SRT |  | A-1016B | J Wire | Jワイヤー
SRT |  | M-90300 | Juvenile fibroadenoma | 若年性線維腺腫
DCM |  | 111277 | Juvenille papillomatosis | 若年性乳頭腫症
SRT |  | M-82040 | Lactating adenoma | 授乳性腺腫
SRT |  | F-01764 | Large rod-like calcification | 大きな桿状
SRT |  | T-04030 | Left breast | 左：左乳房
SRT |  | M-88900 | Leiomyoma | 平滑筋腫
SRT |  | M-88903 | Leiomyosarcoma | 平滑筋肉腫
DCM |  | 111035 | Lesion Density | 病変の濃度
SRT |  | F-01728 | Less defined | より不明瞭になってきた
DCM |  | 111318 | Leukemic infiltration | 白血病浸潤
SRT |  | F-01771 | Linear calcification distribution | 線状
SRT |  | M-83143 | Lipid-rich (lipid-secreting) carcinoma | 脂肪に富む（脂質分泌）癌
SRT |  | M-88500 | Lipoma of the breast | 脂肪腫
SRT |  | G-A640 | Lobular | 分葉状
SRT |  | D7-F0A02 | Lobular carcinoma in situ of breast | 非浸潤性小葉癌：ＬＣＩＳ
SRT |  | F-01753 | Low density (not containing fat) lesion | 低濃度（脂肪を含まない）
SRT |  | T-04003 | Lower inner quadrant of breast | 内下部：乳房の内下部１／４（B領域）
SRT |  | T-04005 | Lower outer quadrant of breast | 外下部：乳房の外下部１／４（D領域）
SRT |  | F-01766 | Lucent-centered calcification | 中心透亮性
SRT |  | R-102D6 | Magnification views | 拡大撮影（M）
DCM |  | 111334 | Malignant melanoma of nipple | 乳頭の悪性黒色腫
SRT |  | D7-90370 | Mammary duct ectasia | 乳管拡張症
SRT |  | F-01791 | Mammographic breast mass | 腫瘤
DCM |  | 111036 | Mammography CAD Report | マンモグラフィCADのレポート
DCM |  | 111238 | Mammography Quality Control Manual 1999, ACR | マンモグラフィ品質管理マニュアル1999, ACR
DCM |  | 111037 | Margins | 辺縁
DCM |  | 111216 | Mechanical failure | 機械の故障
SRT |  | M-85103 | Medullary carcinoma | 髄様癌
DCM |  | 111284 | Microglandular adenosis | 微小腺管腺症
SRT |  | F-01742 | Microlobulated lesion | 微細分葉状
SRT |  | G-A109 | Middle | 中央の
SRT |  | F-01765 | Milk of calcium calcification | 石灰乳
DCM |  | 111200 | MLO Evidence of motion blur | 内外斜位方向撮影　体動によるブレがある
DCM |  | 111201 | MLO Inframammary fold is not open | 内外斜位方向撮影　乳房下溝が開いていない
DCM |  | 111197 | MLO Insufficient pectoral muscle | 内外斜位方向撮影　胸筋の描出が不十分
DCM |  | 111198 | MLO No fat is visualized posterior to fibroglandular tissues | 内外斜位方向撮影　乳腺後隙の脂肪が見られない
DCM |  | 111199 | MLO Poor separation of deep and superficial breast tissues | 内外斜位方向撮影　乳房組織の深部および表在乳腺の分離が不良である
UCUM |  | mo | Month | 月
SRT |  | F-01729 | More defined | より明瞭になってきた
DCM |  | 111210 | Motion blur | 患者の体動
SRT |  | M-84803 | Mucinous adenocarcinoma (Colloid carcinoma) | 粘液癌
DCM |  | 111283 | Myofibroblastoma | 筋線維芽腫
DCM |  | 111144 | Needle localization and biopsy | 針留置による位置決めと生検（L）
SRT |  | D0-F035F | Neoplasm of mammary skin | 乳房皮膚の新生物
SRT |  | M-95400 | Neurofibroma | 神経線維腫
SRT |  | F-01721 | New finding | 新しい所見
SRT |  | T-04100 | Nipple | 乳頭
SRT |  | D7-90554 | Nipple retraction | 乳頭陥凹
DCM |  | 111245 | No algorithms succeeded; without findings | 全てのアルゴリズムが失敗；所見なし
DCM |  | 111213 | No image | 画像なし
SRT |  | F-01723 | No significant changes in the finding | 所見上、著変なし
SRT |  | M-95913 | Non-Hodgkin's lymphoma | 非ホジキンリンパ腫
DCM |  | 111102 | Non-lesion | 病変がない
DCM |  | 111140 | Normal interval follow-up | 通常間隔での経過観察（N）
SRT |  | M-02000 | Normal shape | 正常乳頭
DCM |  | 111244 | Not all algorithms succeeded; with findings | 全てのアルゴリズムが成功した訳ではない；所見あり
DCM |  | 111243 | Not all algorithms succeeded; without findings | 全てのアルゴリズムが成功した訳ではない；所見なし
DCM |  | 111225 | Not Attempted | 未施行
DCM |  | 111152 | Not for Presentation: Rendering device expected not to present | 提示の必要なし：表示装置提示の必要なし
DCM |  | 111038 | Number of calcifications | 石灰化の数
DCM |  | 111039 | Object type | 対象のタイプ
SRT |  | F-01743 | Obscured lesion | 評価困難
DCM |  | 111322 | Occult carcinoma presenting with axillary lymph node metastases | 腋窩リンパ節転移を伴う潜伏癌
DCM |  | 111138 | Old films for comparison | 比較のための以前のフィルム（O）
SRT |  | A-00D7B | Opaque marker | 不透明マーカー
DCM |  | 111040 | Original Source | 情報源
SRT |  | M-91803 | Osteogenic sarcoma | 骨肉腫
DCM |  | 111220 | Other failure | 他の故障
DCM |  | 111175 | Other Marker | 他のマーカー
DCM |  | 111041 | Outline | 輪郭
DCM |  | 111212 | Over exposed | 露光過多
DCM |  | 111234 | Overall Impression / Recommendation Analysis | 全体のインプレッション／推奨の解析
SRT |  | M-02120 | Ovoid shape (Oval) | 楕円形
SRT |  | A-11101 | Cardiac Pacemaker | ペースメーカー
SRT |  | A-10042 | Compression paddle | 圧縮パドル
SRT |  | M-85403 | Paget's disease, mammary (of the nipple) | 乳頭のパジェット病
SRT |  | M-80503 | Papillary carcinoma (invasive) | 浸潤性乳頭癌
SRT |  | M-80500 | Papilloma | 乳頭腫
DCM |  | 111223 | Partially Succeeded | 部分的成功
DCM |  | 111042 | Pathology | 病理
DCM |  | 111043 | Patient Orientation Column | 患者情報　行
DCM |  | 111044 | Patient Orientation Row | 患者情報　列
DCM |  | 111045 | Pectoral Muscle Outline | 胸筋輪郭
DCM |  | 111046 | Percent Glandular Tissue | 乳腺組織の割合（％）
SRT |  | M-90201 | Phyllodes tumor | 良性葉状腫瘍
SRT |  | M-90203 | Phyllodes tumor, malignant | 悪性葉状腫瘍
SRT |  | M-97313 | Plasmacytoma | 形質細胞腫
SRT |  | M-89400 | Pleomorphic adenoma | 混合腫瘍（多形腺腫）
DCM |  | 111209 | Positioning | ポジショニング
SRT |  | G-A106 | Posterior | 後方の
DCM |  | 111151 | Presentation Optional: Rendering device may present | 提示はオプションである：表示装置の提示は自由
DCM |  | 111150 | Presentation Required: Rendering device is expected to present | 提示が必要である：表示装置の提示必要
DCM |  | 111047 | Probability of cancer | 癌の可能性
DCM |  | 111292 | Pseudoangiomatous stromal hyperplasia | 偽血管腫様間質過形成
SRT |  | F-01767 | Punctate calcification | 点状
DCM |  | 111048 | Quadrant location | 位置表示（四分の一円）
DCM |  | 111049 | Qualitative Difference | 質的相違
DCM |  | 111050 | Quality Assessment | 品質評価
DCM |  | 111051 | Quality Control Standard | 品質管理の基準
DCM |  | 111052 | Quality Finding | 品質に関する所見
SRT |  | M-78731 | Radial scar | 放射状硬化性病変（放射状瘢痕）
DCM |  | 111053 | Recommended Follow-up | 経過観察の推奨
DCM |  | 111054 | Recommended Follow-up Date | 推奨される経過観察日
DCM |  | 111055 | Recommended Follow-up Interval | 推奨される経過観察間隔
SRT |  | F-01773 | Regional calcification distribution | 領域性
SRT |  | F-0172A | Removal of implant since previous mammogram | インプラントの除去
DCM |  | 111056 | Rendering Intent | 結果表示するかどうか
SRT |  | T-04020 | Right breast | 右：右乳房
SRT |  | M-02100 | Round shape | 円形
SRT |  | M-78060 | Scar tissue | 瘢痕組織
SRT |  | F-01712 | Scattered fibroglandular densities | 乳腺散在
SRT |  | M-74220 | Sclerosing adenosis | 硬化性腺症
DCM |  | 111057 | Scope of Feature | 特徴の範囲
SRT |  | M-85023 | Secretory (juvenile) carcinoma of the breast | 分泌癌（若年性癌）：分泌性乳癌（若年性乳癌）
SRT |  | F-01774 | Segmental calcification distribution | 区域性
DCM |  | 111099 | Selected region | 選択された領域
DCM |  | 111058 | Selected Region Description | 選択領域の記述
SRT |  | M-020F9 | Shape | 形状
DCM |  | 111059 | Single Image Finding | 1画像の所見
SRT |  | D0-00050 | Skin lesion | 皮膚病変
SRT |  | F-01799 | Skin retraction of breast | 乳房の皮膚陥凹
SRT |  | F-0179A | Skin thickening of breast | 乳房の皮膚肥厚
DCM |  | 111218 | Software failure | ソフトウエアの故障
SRT |  | P5-B3402 | Spatial collocation analysis | 空間的なデータ対応付け解析
SRT |  | P5-B3404 | Spatial proximity analysis | 空間的なデータ近接判定解析
SRT |  | F-01745 | Spiculated lesion | スピキュラを伴う
SRT |  | R-102D7 | Spot compression | スポット圧迫撮影（S）
DCM |  | 111136 | Spot magnification view(s) | 拡大スポット撮影（V）
DCM |  | 111340 | Squamous cell carcinoma of the nipple | 乳頭の扁平上皮癌
SRT |  | A-13600 | Staple | ステープル
DCM |  | 111060 | Study Date | 検査日
DCM |  | 111061 | Study Time | 検査時刻
SRT |  | F-0178D | Subareolar position | 乳輪下
DCM |  | 111222 | Succeeded | 成功
DCM |  | 111062 | Successful Analyses | 解析の成功
DCM |  | 111063 | Successful Detections | 検出の成功
DCM |  | 111146 | Suggestive of malignancy - take appropriate action | 悪性―適切な処置が必要（T）
DCM |  | 111065 | Summary of Analyses | 解析の要約
DCM |  | 111064 | Summary of Detections | 検出の要約
SRT |  | A-13510 | Suture material | 縫合；縫合材料
DCM |  | 111155 | Target content items are related contra-laterally | Target content itemsは対側のそれらに関連している
DCM |  | 111154 | Target content items are related spatially | Target content itemsは空間的に関連している
DCM |  | 111153 | Target content items are related temporally | Target content itemsは時間的に関連している
DCM |  | 111194 | Technical factors missing | 撮影条件がない
SRT |  | P5-B3406 | Temporal correlation | 経時的相関
SRT |  | F-01798 | Trabecular thickening of breast | 乳房の梁柱の肥厚
SRT |  | M-82113 | Tubular adenocarcinoma | 管状癌
SRT |  | M-82110 | Tubular adenoma | 管状腺腫
SRT |  | F-01797 | Tubular density | 管状影
SRT |  | P5-B0099 | Ultrasound procedure | 超音波検査手技（U）
DCM |  | 111211 | Under exposed | 露光不足
DCM |  | 111221 | Unknown failure | 原因不詳の故障
DCM |  | 111176 | Unspecified | 非特定の物質
DCM |  | 111235 | Unusable - Quality renders image unusable | 使用不可―画像構成の品質は使用不可である
SRT |  | T-04002 | Upper inner quadrant of breast | 内上部：乳房の内上部１／４（A領域）
SRT |  | T-04004 | Upper outer quadrant of breast | 外上部：乳房の外上部１／４（C領域）
DCM |  | 111236 | Usable - Does not meet the quality control standard | 使用可―品質管理の基準に達していない
DCM |  | 111237 | Usable - Meets the quality control standard | 使用可―品質管理の基準に達している
SRT |  | F-0176B | Vascular calcification | 血管
DCM |  | 111066 | Vertical Pixel Spacing | 垂直方向のピクセル間隔
DCM |  | 111179 | View and Laterality Marker does not have approved codes | 鉛マーカーはFDAのコードがない
DCM |  | 111178 | View and Laterality Marker does not have both view and laterality | 鉛マーカーは撮影方向と左右の表示がない
DCM |  | 111183 | View and Laterality Marker is incorrect | 鉛マーカーは正しい位置にない
DCM |  | 111177 | View and Laterality Marker is missing | 鉛マーカーがみられない
DCM |  | 111180 | View and Laterality Marker is not near the axilla | 鉛マーカーは腋窩の近くにない
DCM |  | 111184 | View and Laterality Marker is off image | 鉛マーカーがフィルム外である
DCM |  | 111182 | View and Laterality Marker is partially obscured | 鉛マーカーは一部覆い隠されている
DCM |  | 111298 | Virginal hyperplasia | 若年性過形成
UCUM |  | wk | Week | 週
UCUM |  | a | Year | 年

# G English Code Meanings of Selected Codes (Normative)

Table G-1. English Code Meanings of Selected Codes

Coding Scheme Designator | Code Value | Code Meaning
--- | --- | ---
UCUM | 1 | unary
no units
UCUM | {ratio} | ratio
SRT | C-10520 | Carbon dioxide, NOS
Carbon dioxide gas
SRT | C-21005 | Ethanol
Ethyl alcohol
SRT | C-81100 | Hypotensive agent, NOS
Antihypertensive agent, NOS
Antihypertensive drug, NOS
SRT | C-A7400 | Thrombolytic agent, NOS
Fibrinolytic agent, NOS
SRT | C-A7440 | Injectable fibrinolysin
Injectable plasmin
SRT | C-B0300 | Contrast agent, NOS
Radiographic contrast agent, NOS
SRT | C-B1091 | Iodohippurate I^131^ sodium
Iodine^131^ hippuran
SRT | C-B1109 | Iodine^131^ polyvinylpyrrolidone
Iodine^131^ PVP
SRT | C-B1225 | Technetium Tc^99^ N-substituted iminodiacetate
Tc^99^ labeled HIDA
SRT | D3-40208 | Congenital pulmonary arteriovenous fistula
Congenital coronary artery fistula to pulmonary artery
SRT | D4-33142 | Pulmonary artery conduit
Congenital pulmonary artery conduit
SRT | D4-33512 | Pulmonary vein confluence
Congenital pulmonary vein confluence
SRT | D4-33514 | Pulmonary venous atrium
Congenital pulmonary venous atrium
SRT | D4-33516 | Systemic venous atrium
Congenital systemic venous atrium
SRT | R-10206 | Antero-posterior
AP
SRT | R-10214 | Postero-anterior
PA
SRT | R-10246 | Oblique axial
Oblique caudo-cranial
Oblique cranio-caudal
Oblique transaxial
Off-axial
Off-axial projection
SRT | R-10224 | Medial-lateral
Medio-lateral
SRT | R-10230 | Lateral-medial
Latero-medial
SRT | R-10232 | Right lateral projection
Left to right beam projection
SRT | R-10236 | Left lateral projection
Right to left beam projection
SRT | R-10242 | caudad
caudal projection
cranio-caudal projection
SRT | R-10244 | cephalad
cranial projection
caudo-cranial projection
from below
SRT | R-4087B | transforamenal
optic foramen projection
SRT | G-A100 | Right
Right lateral
SRT | G-A101 | Left
Left lateral
SRT | G-A102 | Bilateral
Right and left
SRT | G-A103 | Unilateral
One-sided
SRT | G-A105 | Anterior
Ventral
SRT | G-A106 | Posterior
Dorsal
SRT | G-A107 | Cephalic
Cephalad
Rostral
Cranial
SRT | G-A108 | Caudal
Caudad
SRT | G-A109 | Medial
Median
Middle
SRT | G-A10A | Mediolateral
Midline
SRT | G-A112 | External
Outer
SRT | G-A113 | Internal
Inner
SRT | G-A115 | Inferior
Lower
SRT | G-A116 | Superior
Upper
SRT | G-A138 | Coronal
Frontal
SRT | G-A140 | Deep
Profundis
SRT | R-102CD | Sagittal Projection
Lateral Projection
SRT | G-4022 | Contact with
Direct contact
SRT | G-A170 | Hilar
Hilus
SRT | G-A174 | Edge
Along edge
SRT | G-D105 | Intracutaneous route
Intradermal route
SRT | G-D140 | Oral route
Peroral route
SRT | G-D164 | Vaginal route
Per vagina
SRT | P1-05535 | Catheterization
Insertion of catheter
SRT | P1-30350 | Atherectomy, NOS
Removal of atherosclerotic plaque from artery, NOS
SRT | T-15460 | Wrist joint, NOS
Joint of Wrist, NOS
SRT | T-32000 | Endo-cardiac
Intra-cardiac
SRT | T-41000 | Endo-arterial
Intra-arterial
SRT | T-46010 | Innominate artery
Brachiocephalic artery
Brachiocephalic trunk
SRT | T-48170 | Internal jugular vein
Vena jugularis interna
SRT | T-48620 | Innominate vein
Brachiocephalic vein
SRT | T-48810 | Portal vein
Vena portae
SRT | T-D4450 | Omental bursa
Lesser peritoneal sac
LN | 33068-8 | Thoracic Area
FTA
LN | 33070-4 | Inner Orbital Diameter
IOD
LN | 11727-5 | Estimated Weight
EFW
LN | 11948-7 | Fetal Heart Rate
HR
LN | 11778-8 | Estimated Date of Delivery
EDD
LN | 11955-2 | Last Menstrual Period
LMP
LN | 11979-2 | Abdominal Circumference
AC
LN | 11818-2 | Anterior-Posterior Abdominal Diameter
APAD
LN | 11820-8 | Biparietal Diameter
BPD
LN | 11824-0 | BPD area corrected
BPDa
LN | 11963-6 | Femur Length
FL
LN | 11984-2 | Head Circumference
HC
LN | 11851-3 | Occipital-Frontal Diameter
OFD
LN | 11988-3 | Thoracic Circumference
TC
LN | 11862-0 | Tranverse Abdominal Diameter
TAD
LN | 11863-8 | Trans Cerebellar Diameter
TCD
TDC
LN | 11864-6 | Transverse Thoracic Diameter
TTD
LN | 11629-3 | Outer Orbital Diameter
OOD
LN | 11726-7 | Peak Velocity
Peak Systolic Velocity
SRT | G-A188 | Mid-Longitudinal
Mid
SRT | T-45170 | Carotid Bulb
Carotid Sinus
LN | 8277-6 | Body Surface Area
BSA
LN | 29462-9 | Pulmonary-to-Systemic Shunt Flow Ratio
Qp/Qs
SRT | R-42047 | Antegrade Direction
Antegrade Flow
SRT | R-42E61 | Retrograde Direction
Regurgitant Flow
LN | 11957-8 | Crown Rump Length
CRL
SRT | P1-48501 | Breast implantation
Implant procedure
SRT | P1-48520 | Removal of breast implant
Explantation
SRT | D0-00165 | Weal
Hives
SRT | D7-90010 | Disorder of breast implant
Breast implant problem
SRT | D7-90530 | Breast lump
Lump or thickening
SRT | D7-90560 | Peau d'orange surface of breast
Peau d'orange
SRT | D7-90565 | Bloody nipple discharge
Bloody discharge
SRT | DD-66A67 | Hemorrhage postprocedure
Abnormal bleeding
SRT | DD-67700 | Infection as complication of medical care
Infection
SRT | F-01BF8 | Ultrasound scan normal
Normal; the finding is not seen sonographically
SRT | F-01E06 | Indeterminate result
Inconclusive
SRT | F-02B9B | Nottingham Combined Grade cannot be determined
GX - grade cannot be assessed
SRT | F-8A057 | Calcification of breast
Calcifications
SRT | F-8A074 | Discoloration of skin of breast
Redness of skin
SRT | F-8A09C | Nipple problem
Nipple abnormality
SRT | F-A2632 | Persistent pain following procedure
Unusual pain
SRT | F-A5581 | Vasovagal attack
Vasovagal reaction
SRT | G-F616 | Nottingham Combined Grade I: 3-5 points
G1 - Low combined histologic grade (favorable)
SRT | G-F617 | Nottingham Combined Grade II: 6-7 points
G2 - Intermediate combined histo grade (moderately favorable)
SRT | G-F618 | Nottingham Combined Grade III: 8-9 points
G3 - High combined histologic grade (unfavorable)
SRT | M-78280 | Surgical scar
Post-surgical scar
SRT | P1-03106 | Computed tomography guided biopsy
CT guided
SRT | P1-03107 | Magnetic resonance imaging guided biopsy
MRI guided
SRT | P1-03115 | Ultrasound guided biopsy
Ultrasound guided
SRT | P1-48011 | Pre-biopsy localization of breast lesion
Localization for surgical biopsy
SRT | P1-48142 | Diagnostic aspiration of breast cyst
Cyst aspiration
SRT | P1-48145 | Fine needle aspiration of breast
FNA - Fine needle aspiration
SRT | P1-48304 | Core needle biopsy of breast
Core biopsy
SRT | P1-4830F | Breast - surgical biopsy
Surgical biopsy
SRT | P2-4A000 | Examination of breast
Clinical breast exam
SRT | P5-00032 | Diagnostic radiography, stereotactic localization
Stereotactic
SRT | P5-40030 | Specimen radiography of breast
Specimen imaging
SRT | P5-D0042 | Radionuclide localization of tumor, limited area
Scintimammography
SRT | R-20099 | O/E - axillary lymphadenopathy
Large axillary lymph nodes
SRT | R-207D7 | O/E - Breast lump palpated
Palpable abnormality
SRT | R-40FB9 | Before procedure
Pre-
SRT | R-41DDC | High risk tumor
High risk
SRT | R-422A4 | After procedure
Follow-up
SRT | R-101BA | vessel lumen cross sectional area reduction
lumen area stenosis
SRT | R-101BB | vessel lumen diameter reduction
lumen diameter stenosis
SRT | P5-B0700 | Ultrasonic guidance procedure
Ultrasound guided
SRT | F-01711 | Almost entirely fat
Almost entirely fat (< = 10% fibroglandular)
SRT | F-01712 | Scattered fibroglandular densities
Scattered fibroglandular tissue (11% - 50% fibroglandular)
SRT | F-01713 | Heterogeneously dense
Heterogeneously dense (51% - 75% fibroglandular)
SRT | F-01714 | Extremely dense
Extremely dense (greater than 75% fibroglandular)
SRT | F-0176F | Heterogeneous calcification
Coarse heterogeneous calcification
SRT | F-01792 | Focal asymmetric breast tissue
Focal asymmetry
SRT | F-01793 | Asymmetric breast tissue
Global asymmetry
SRT | F-01797 | Tubular density
Asymmetric tubular structure/solitary dilated duct
SRT | M-85002 | Intraductal carcinoma, non-infiltrating
DCIS
SRT | P0-009B4 | Evaluation procedure
Clinical evaluation
SRT | P5-D0061 | Radioisotope scan of lymphatic system
Lymphoscintigraphy
SRT | A-25612 | Embolization coil
Gianturco coil
SRT | T-46010 | Brachiocephalic artery
Brachiocephalic trunk
Innominate artery
DCM | 111046 | Percent Fibroglandular Tissue
Percent Glandular Tissue

# H Code Meanings of LOINC Codes in DCMR

Table H-1. Code Meanings of LOINC Codes

Code Value | Code Meaning
--- | ---
10160-0 | History of Medication Use
11329-0 | History
11450-4 | Problem List
11522-0 | Echocardiography Report
11525-3 | Ultrasound Obstetric and Gyn Report
11528-7 | Radiology Report
11538-6 | CT Chest Report
11539-4 | CT Head Report
11540-2 | CT Abdomen Report
11541-0 | MRI Head Report
11612-9 | Aborta
11623-6 | Fourth Quadrant Diameter
11624-4 | First Quadrant Diameter
11625-1 | Third Quadrant Diameter
11626-9 | Second Quadrant Diameter
11629-3 | Outer Orbital Diameter
11636-8 | Live Births
11653-3 | End Diastolic Velocity
11665-7 | Minimum Diastolic Velocity
11692-1 | Time averaged peak velocity
11726-7 | Peak Velocity
11726-7 | Peak Velocity
11727-5 | Estimated Weight
11732-5 | EFW by AC, BPD, FL, HC, Hadlock 1985
11734-1 | EFW by AC, BPD, FL, Hadlock 1984
11735-8 | EFW by AC, BPD, FL, Hadlock 1985
11738-2 | EFW by AC, BPD, Hadlock 1984
11739-0 | EFW by AC and BPD, Shepard 1982
11746-5 | EFW by AC, FL, HC, Hadlock 1985
11750-7 | EFW by AC, FL, Hadlock 1984
11751-5 | EFW by AC, FL, Hadlcok 1985
11754-9 | EFW by AC, HC Hadlock 1984
11756-4 | EFW by AC, Campbell 1975
11767-1 | EFW percentile rank
11778-8 | Estimated Date of Delivery
11779-6 | EDD from LMP
11780-4 | EDD from ovulation date
11781-2 | EDD from average ultrasound age
11793-7 | Follicle Diameter
11816-6 | Yolk Sac length
11818-2 | Anterior-Posterior Abdominal Diameter
11819-0 | Anterior-Posterior Trunk Diameter
11820-8 | Biparietal Diameter
11823-2 | Cephalic Index
11824-0 | BPD area corrected
11825-7 | Left Kidney width
11827-3 | Right Kidney width
11829-9 | Left Ovary Width
11830-7 | Right Ovary Width
11834-9 | Left Kidney length
11836-4 | Right Kidney length
11840-6 | Left Ovary Length
11841-4 | Right Ovary Length
11850-5 | Gestational Sac Diameter
11851-3 | Occipital-Frontal Diameter
11853-9 | Left Kidney thickness
11855-4 | Right Kidney thickness
11857-0 | Left Ovary Height
11858-8 | Right Ovary Height
11860-4 | Cisterna Magna
11860-4 | Cisterna Magna length
11862-0 | Tranverse Abdominal Diameter
11863-8 | Trans Cerebellar Diameter
11863-8 | Trans Cerebellar Diameter
11864-6 | Transverse Thoracic Diameter
11871-1 | FL/AC
11872-9 | FL/BPD
11873-7 | FL/HC
11878-6 | Number of Fetuses by US
11884-4 | Average Ultrasound Age
11885-1 | Gestational Age by LMP
11885-1 | Gestational Age by LMP
11886-9 | Gestational Age by ovulation date
11888-5 | Composite Ultrasound Age
11889-3 | AC, Campbell 1975
11892-7 | AC, Hadlock 1984
11893-5 | AC, Jeanty 1984
11900-8 | BPD, Doubilet 1993
11901-6 | BPDa, Hadlock 1982
11902-4 | BPD, Hadlock 1984
11903-2 | BPD, Hansmann 1985
11905-7 | BPD, Jeanty 1984
11906-5 | BPD, Kurtz 1980
11907-3 | BPD, Sabbagha 1978
11910-7 | CRL, Hadlock 1992
11911-5 | CRL, Hansmann 1985
11913-1 | CRL, Nelson 1981
11914-9 | CRL, Robinson 1975
11917-2 | CRL, Jeanty 1984
11918-0 | Fibula, Merz 1987
11920-6 | FL, Hadlock 1984
11921-4 | FL, Hansmann 1985
11922-2 | FL, Hohler 1982
11923-0 | FL, Jeanty 1984
11924-8 | FL, Merz 1987
11926-3 | Foot Length, Mercer 1987
11928-9 | GS, Hellman 1969
11929-7 | GS, Rempen 1991
11932-1 | HC, Hadlock 1984
11934-7 | HC, Jeanty 1984
11936-2 | Humerus, Jeanty 1984
11937-0 | Humerus, Merz 1987
11939-6 | Radius, Merz 1987
11941-2 | Tibia, Jeanty 1984
11944-6 | Ulna, Jeanty 1984
11945-3 | Ulna, Merz 1987
11947-9 | HC/AC
11948-7 | Fetal Heart Rate
11955-2 | Last Menstrual Period
11957-8 | Crown Rump Length
11961-0 | Cervix Length
11962-8 | Clavicle length
11963-6 | Femur Length
11964-4 | Fibula length
11965-1 | Foot length
11966-9 | Humerus length
11967-7 | Radius length
11968-5 | Tibia length
11969-3 | Ulna length
11976-8 | Ovulation date
11977-6 | Para
11979-2 | Abdominal Circumference
11984-2 | Head Circumference
11988-3 | Thoracic Circumference
11996-6 | Gravida
12008-9 | Pulsatility Index
12023-8 | Resistivity Index
12144-2 | Systolic to Diastolic Velocity Ratio
12145-9 | Endometrium Thickness
12146-7 | Nuchal Fold thickness
12164-0 | Left Ovary Volume
12165-7 | Right Ovary Volume
12170-7 | Width of Hemisphere
12171-5 | Lateral Ventrical width
17977-0 | Left Atrium Area A4C view
17978-8 | Mitral Valve A-Wave Peak Velocity
17985-3 | Left Atrium to Aortic Root Ratio
17988-7 | Right Atrium Area A4C view
17995-2 | Thoracic Aorta Coarctation Systolic Peak Instantaneous Gradient
17996-0 | Aortic Valve Cusp Separation
17998-6 | Aortic Valve Regurgitant Diastolic Deceleration Time
18006-7 | Inferior Vena Cava Diameter
18011-7 | Aortic Arch Diameter
18012-5 | Ascending Aortic Diameter
18013-3 | Descending Aortic Diameter
18015-8 | Aortic Root Diameter
18019-0 | Left Pulmonary Artery Diameter
18020-8 | Main Pulmonary Artery Diameter
18021-6 | Right Pulmonary Artery Diameter
18026-5 | Left Ventricular End Diastolic Volume
18030-7 | Tricuspid Valve A Wave Peak Velocity
18031-5 | Tricuspid Valve E Wave Peak Velocity
18035-6 | Mitral Regurgitation dP/dt derived from Mitral Reg. velocity
18037-2 | Mitral Valve E-Wave Peak Velocity
18038-0 | Mitral Valve E to A Ratio
18040-6 | Mitral Valve E-F Slope by M-Mode
18041-4 | Aortic Valve Ejection Time
18043-0 | Left Ventricular Ejection Fraction by US
18050-5 | Inferior Vena Cava % Collapse
18051-3 | Left Ventricular Fractional Shortening
18053-9 | Left Ventricle Posterior Wall % Thickening
18054-7 | Interventricular Septum % Thickening
18070-3 | Right Atrium Systolic Pressure
18071-1 | Left Ventricular Isovolumic Relaxation Time
18076-0 | Left Ventricle Systolic Major Axis
18077-8 | Left Ventricle Diastolic Major Axis
18087-7 | Left Ventricle Mass
18096-8 | Pulmonic valve Area by continuity
18118-0 | LV Wall Motion Segmental Findings
18139-6 | Stage
18148-7 | Left Ventricular End Systolic Volume
18152-9 | Left Ventricle Posterior Wall Diastolic Thickness
18153-7 | Right Ventricle Anterior Wall Diastolic Thickness
18154-5 | Interventricular Septum Diastolic Thickness
18155-2 | Interventricular Septum to Posterior Wall Thickness Ratio
18156-0 | Left Ventricle Posterior Wall Systolic Thickness
18157-8 | Right Ventricular Anterior Wall Systolic Thickness
18158-6 | Interventricular Septum Systolic Thickness
18179-2 | Wall Segment
18185-9 | Gestational Age
18745-0 | Cardiac Catheterization Report
18747-6 | CT Report
18748-4 | Diagnostic Imaging Report
18755-9 | MRI Report
18756-7 | MRI Spine Report
18757-5 | Nuclear Medicine Report
18758-3 | PET Scan Report
18760-9 | Ultrasound Report
18782-3 | Findings
18783-1 | Recommendations
18785-6 | Indications for Procedure
18834-2 | Previous Findings
19005-8 | Impressions
20167-3 | Acceleration Index
20168-1 | Acceleration time
20217-6 | Deceleration time
20247-3 | Peak Gradient
20295-2 | Time from Q wave to Pulmonic Valve Closes
20352-1 | Time averaged mean velocity
29436-3 | Left Ventricle Internal End Diastolic Dimension
29438-9 | Left Ventricle Internal Systolic Dimension
29449-6 | Mitral Valve Regurgitant Volume by Proximal Isovelocity Surface Area Method
29450-4 | Pulmonary Vein Systolic Peak Velocity
29451-2 | Pulmonary Vein Diastolic Peak Velocity
29452-0 | Pulmonary Vein Systolic to Diastolic Ratio
29453-8 | Pulmonary Vein Atrial Contraction Reversal Peak Velocity
29460-3 | Thoracic Aorta Coarctation Systolic Peak Velocity
29462-9 | Pulmonary-to-Systemic Shunt Flow Ratio
29463-7 | Patient Weight
29463-7 | Patient weight
29469-4 | Left Atrium Antero-posterior Systolic Dimension
29471-0 | Hepatic Vein Systolic Peak Velocity
29472-8 | Hepatic Vein Diastolic Peak Velocity
29473-6 | Hepatic Vein Systolic to Diastolic Ratio
29474-4 | Hepatic Vein Atrial Contraction Reversal Peak Velocity
29486-8 | Left Atrial Appendage Peak Velocity
29549-3 | Medications Administered
33065-4 | Ectopic Pregnancies
33066-2 | Estimated LMP by EDD
33067-0 | Conception Date
33068-8 | Thoracic Area
33069-6 | Nuchal Translucency
33070-4 | Inner Orbital Diameter
33071-2 | Spine Length
33072-0 | AC, ASUM 2000
33073-8 | AC, Hansmann1985
33074-6 | AC, Lessoway 1998
33075-3 | AC, Mertz 1988
33076-1 | AC, Shinozuka 1996
33077-9 | A-P Abdominal Diameter, Lessoway 1998
33078-7 | AxT, Shinozuka 1996
33079-5 | BPD, ASUM 1989
33080-3 | BPD, Lessoway 1998
33081-1 | BPD, Mertz 1988
33082-9 | BPD, Osaka 1989
33083-7 | BPD, Rempen 1991
33084-5 | BPD, Shinozuka 1996
33085-2 | BPD, Tokyo 1986
33086-0 | BPD-oi, Chitty 1997
33087-8 | BPD-oo, Chitty 1997
33088-6 | Clavicle length, Yarkoni 1985
33089-4 | CRL, ASUM 1991
33090-2 | CRL, ASUM 2000
33091-0 | CRL, Daya 1993
33092-8 | CRL, Jeanty 1982
33093-6 | CRL, Osaka 1989
33094-4 | CRL, Rempen 1991
33095-1 | CRL, Shinozuka 1996
33096-9 | CRL, Tokyo 1986
33097-7 | Fibula, Jeanty 1983
33098-5 | FL, Chitty 1997
33099-3 | FL, Jeanty 1982
33100-9 | FL, Lessoway 1998
33101-7 | FL, Osaka 1989
33102-5 | FL, Shinozuka 1996
33103-3 | FL, Tokyo 1986
33104-1 | GS, Daya 1991
33105-8 | GS, Hansmann 1979
33106-6 | GS, Hansmann 1982
33107-4 | GS, Nyberg 1992
33108-2 | GS, Tokyo 1986
33109-0 | HC, ASUM 2000
33110-8 | HC measured, Chitty 1997
33111-6 | HC derived, Chitty 1997
33112-4 | HC, Hansmann 1985
33113-2 | HC, Jeanty 1982
33114-0 | HC, Lessoway 1998
33115-7 | HC Merz, 1988
33116-5 | Humerus Length, ASUM 2000
33117-3 | Humerus Length, Osaka 1989
33118-1 | Length of Vertebra, Tokyo 1986
33119-9 | OFD, ASUM 2000
33120-7 | OFD, Hansmann 1986
33121-5 | OFD, Lessoway 1998
33122-3 | IOD, Mayden 1982
33123-1 | IOD, Trout 1994
33124-9 | OOD, Mayden, 1982
33125-6 | OOD, Trout 1994
33126-4 | Radius, Jeanty 1983
33127-2 | Spine Length, Tokyo, 1989
33128-0 | TAD, Eriksen 1985
33129-8 | TAD Hansmann, 1979
33130-6 | TAD, Tokyo 1986
33131-4 | ThC, Chitkara 1987
33132-2 | TCD, Chitty 1994
33133-0 | TCD, Goldstein 1987
33134-8 | TCD, Hill 1990
33135-5 | TCD, Nimrod 1986
33136-3 | Transverse Thoracic Diameter, Hansmann 1985
33137-1 | Transverse Thoracic Diameter, Lessoway 1998
33138-9 | Fetal Trunk Cross-Sectional Area, Osaka 1989
33139-7 | EFW by BPD, TTD, Hansmann 1986
33140-5 | EFW by BPD, FTA, FL, Osaka 1990
33141-3 | EFW1 by Shinozuka 1996
33142-1 | EFW2 by Shinozuka 1996
33143-9 | EFW3 by Shinozuka 1996
33144-7 | EFW by BPD, APAD, TAD, FL, Tokyo 1987
33145-4 | AC by GA, ASUM 2000
33146-2 | AC by GA, Hadlock 1984
33147-0 | AC (measured) by GA, Chitty 1994
33147-0 | AC (measured) by GA, Chitty 1994
33148-8 | AC by GA, Merz 1988
33149-6 | AC by GA, Shinozuka 1996
33150-4 | AxT by GA, Shinozuka 1996
33151-2 | BPD by GA, ASUM 2000
33152-0 | BPD outer-outer by GA, Chitty 1994
33153-8 | BPD by GA, Jeanty 1982
33154-6 | BPD by GA, Merz 1988
33155-3 | BPD by GA, Rempen 1991
33156-1 | BPD by GA, Shinozuka 1996
33157-9 | Cephalic Index, by GA Chitty 1994
33158-7 | Cephalic Index by GA, Hadlock 1981
33159-5 | CRL by GA, ASUM 2000
33160-3 | CRL by GA, Rempen1991
33161-1 | CRL, by GA, Shinozuka 1996
33162-9 | EFW by GA, Hadlock 1991
33163-7 | EFW by GA, Hansmann 1986
33164-5 | Fibula by GA, by GA Jeanty 1983
33165-2 | FL by GA, ASUM 2000
33166-0 | FL by GA, Hadlock 1984
33167-8 | FL by GA, Chitty 1994
33168-6 | FL by GA, Jeanty 1982
33169-4 | FL by GA, Merz 1988
33170-2 | FL by GA, Shinozuka 1996
33171-0 | GS by GA, Rempen 1991
33172-8 | HC by GA, ASUM 2000
33173-6 | HC by GA, Hadlock 1984
33174-4 | HC derived by GA, Chitty 1994
33175-1 | HC by GA, Jeanty 1982
33176-9 | HC by GA, Merz 1988
33177-7 | Humerus Length by GA, ASUM 2000
33178-5 | OFD by GA, ASUM 2000
33179-3 | OFD by GA, Chitty 1994
33180-1 | Radius, by GA, Jeanty 1983
33181-9 | TCD by GA, Goldstein 1987
33182-7 | HC/AC by GA, Campbell 1977
33183-5 | FWP by GA, Hadlock 1991
33184-3 | FWP by GA, Williams, 1982
33185-0 | FWP by GA, Alexander, 1996
33186-8 | Male Singleton BWP by GA, Arbuckle 1993
33187-6 | Female Singleton BWP by GA, Arbuckle 1993
33188-4 | Female Twins BWP by GA, Arbuckle 1993
33189-2 | FWP by GA, Brenner 1976
33190-0 | FWP by MA, Hadlock 1985
33191-8 | APAD * TAD
33192-6 | Uterus Volume
33196-7 | Posterior Horn Lateral ventricular width
33197-5 | Anterior Horn Lateral ventricular width
33198-3 | BPD by GA, Hadlock 1984
33199-1 | Male Twins BWP by GA, Arbuckle 1993
33537-2 | AC, Jeanty 1982
33538-0 | BPD, Hansmann 1986
33539-8 | BPD, Jeanty 1982
33540-6 | CRL, Hansmann 1986
33541-4 | FL, Hansmann 1986
33542-2 | FL, Merz 1988
33543-0 | HC, Hansmann 1986
33544-8 | OFD, Hansmann 1985
33545-5 | BD, Jeanty 1982
33546-3 | AC (derived), Chitty 1994
33556-2 | BPD outer-inner by GA, Chitty 1994
33867-3 | Velocity ratio
33868-1 | ICA/CCA velocity ratio
33869-9 | Renal Artery/Aorta velocity ratio
55107-7 | Addendum
55108-5 | Patient Presentation
55109-3 | Complications
55110-1 | Conclusions
55111-9 | Current Procedure Descriptions
55112-7 | Summary
55113-5 | Key Images
55114-3 | Prior Procedure Descriptions
55115-0 | Request
55281-0 | Number of Fetuses
55752-0 | Clinical Information
73568-8 | Communication of Critical Results
73569-6 | Radiation Exposure and Protection Information
8277-6 | Body Surface Area
8302-2 | Patient Height

# I Relationship of Endoscopy Procedures to Anatomic Regions (Informative)

Table I-1 provides examples of the common nomenclature for the type of endoscopy performed, and the code value suggested for use for anatomic region in CID 4040 “Endoscopy Anatomic Regions”.

Table I-1. Examples of the Common Nomenclature for the Type of Endoscopy Performed

Coding Scheme Designator | Code Value | Code Meaning | Example of the type of endoscopy for which this region is applicable (Informative)
--- | --- | --- | ---
SRT | T-D4000 | Abdomen | Laparoscopy
SRT | T-59490 | Anus, rectum and sigmoid colon | Rectosigmoidoscopy
SRT | T-60610 | Bile duct |
SRT | T-74000 | Bladder | Cystoscopy
SRT | T-DD123 | Bladder and urethra | Panendoscopy (urethrocystoscopy)
SRT | T-26000 | Bronchus | Bronchoscopy
SRT | T-83200 | Cervix | Colposcopy
SRT | T-D3000 | Chest | Thoracoscopy
SRT | T-DD163 | Esophagus, stomach and duodenum | Upper gastrointestinaI endoscopy
SRT | T-AB200 | External auditory canal | Otoscopy
SRT | T-63000 | Gallbladder | Laparoscopic cholecystectomy
SRT | T-D7000 | Inguinal region | Endoscopic inguinal hernia repair
SRT | T-15001 | Joint | Arthroscopy
SRT | T-71000 | Kidney | Percutaneous renal endoscopy
SRT | T-D9200 | Knee | Arthroscopy of knee
SRT | T-59000 | Large intestine | Colonoscopy
SRT | T-24100 | Larynx | Laryngoscopy
SRT | T-40230 | Lumen of blood vessel | Endoluminal (intravascular) endoscopy
SRT | T-D3300 | Mediastinum | Mediastinoscopy
SRT | T-2300C | Naso pharynx | Naso pharyngoscopy
SRT | T-22000 | Paranasal sinus | Endoscopic sinus surgery
SRT | T-55002 | Pharynx | Pharyngoscopy
SRT | T-20101 | Pharynx and larynx | Laryngopharyngoscopy
SRT | T-59600 | Rectum | Proctoscopy
SRT | T-D2220 | Shoulder | Arthroscopy of shoulder
SRT | T-59470 | Sigmoid colon | Sigmoidoscopy
SRT | T-D0146 | Spine | Spinal endoscopy
SRT | T-DD006 | Trachea and bronchus | Tracheobronchoscopy
SRT | T-70010 | Upper urinary tract | Percutaneous or retrograde ureteric and renal endoscopy
SRT | T-73800 | Ureter | Percutaneous or retrograde ureteric endoscopy
SRT | T-88920 | Uterus and fallopian tubes | Culdoscopy

# J SNOMED DICOM Microglossary Retired Codes (Normative)

This Annex identifies coded terms specified in earlier versions of the Standard that were included in the SNOMED DICOM Microglossary. The Microglossary and these coded terms are retired. Some of the codes conflict with codes defined in SNOMED. Additionally, some SNOMED coded terms specified in earlier versions of the Standard have been retired and replaced by SNOMED to avoid ambiguities in concept, and are noted here as well.

Implementers of the Standard are cautioned that:

- some of the codes noted as retired are still valid SNOMED codes, but with different meanings; it is thus the combination of code and meaning that is retired

- some applications may continue to send retired codes with the meaning defined in this Annex

- the retired codes may be associated with coding scheme designator 99SDM or SNM3

- retired codes may be encountered in existing SOP Instances stored in archives

- applications receiving SOP Instances should continue to support retired codes with the meaning defined in this Annex

- some applications may not trigger expected behavior (e.g., hanging protocols, image processing) when receiving SOP Instances with the replacement codes

- DICOM applications and SOP Instances shall never use the retired codes with a meaning other than that defined in this Annex

Table J-1. SNOMED DICOM Microglossary Retired Codes

Retired Code Value | Code Meaning | Replacement Code | Notes
--- | --- | --- | ---
G-5190 | Headfirst | F-10470 |
G-5191 | Feet-first | F-10480 |
G-A11A | Mid-longitudinal | G-A188 |
G-A11B | Parasagittal | G-A189 |
G-A12A | Intraluminal | R-42142 |
G-A16A | Capsule | G-A171 | Replacement code has meaning "Capsular" G-A16A remains in use as "Area of defined region"
G-A16B | Lumen | T-D0048 |
G-A16C | Contact | G-4022 | Replacement code has meaning "Contact with" G-A16C remains in use as "Part of tooth"
G-A16D | Parenchyma | T-D0062 |
J-83250 | Metal (Lead) Marker | A-00D7B |
R-102C9 | Transthoracic | R-40885 |
R-102CA | Lordotic | R-40799 |
R-102CB | Transforamenal | R-4087B |
R-102CC | Transoral | G-D00B |
R-102CE | Transorbital | R-40554 |
R-11300 | Transverse | G-A117 |
Y-X1770 | Cranio-caudal exaggerated laterally | R-1024A |
Y-X1771 | Cranio-caudal exaggerated medially | R-1024B |
T-D1217 | Maxilla and mandible | T-D1213 |
T-D1480 | Orbit | T-D14AE |
T-D6151 | Uterus and fallopian tubes | T-88920 |
G-0371 | % Area Reduction | R-101BA |
G-0372 | % Diameter Reduction | R-101BB |
G-C295 | Route of Administration | G-C340 |
G-D100 | Route of Administration | G-C340 |
T-42501 | Abdominal Aorta | T-42500 |
T-42303 | Aortic Arch | T-42300 |
T-45011 | Carotid Artery | T-45010 |
T-A600A | Cerebellum | T-A6000 |
T-D00CC | Entire Spine | T-D0146 |
T-48500 | Pulmonary Vein | T-48581 |
T-D8300 | Elbow | T-15430 |
T-12402 | Forearm | T-D8500 |
T-D2500 | Hip | T-15710 |
T-D4909 | Kidney | T-71000 |
T-62002 | Liver | T-62000 |
T-D4034 | Pancreas | T-65000 |
T-55002 | Pharynx | T-55000 |
T-11500 | Spine | T-D0146 |
T-D4035 | Spleen | T-C3000 |
T-9400F | Testis | T-94000 |
T-4600A | Thoracic aorta | T-42070 |
T-C8001 | Thymus | T-C8000 |
T-D6151 | Uterus and fallopian tubes | T-88920 |
T-73800 | Ureter | T-73000 |
T-83009 | Uterus | T-83000 |
T-D8600 | Wrist | T-15460 |
T-11167 | Zygoma | T-11166 |
P5-B3003 | Transthoracic echocardiography | P5-B3012 |
P5-B3004 | Epicardial echocardiography | P0-05F95 |
P5-B3082 | Pediatric echocardiography | P5-B300F |
P5-B3083 | Intraoperative echocardiography | P5-B300C |
P5-01000 | Image acquisition procedure |  |
P5-01101 | Image acquisition after administration of contrast agent |  |
P5-01103 | Image acquisition during cardiac pacing | P2-35000 |
P5-01104 | Image acquisition at user-defined cardiac pacing rate | P2-35000 |
P5-01111 | Image acquisition during hand grip maneuver | P2-71306 |
P5-01112 | Image acquisition during Valsalva | R-40928 |
P5-01113 | Image acquisition during postural maneuver |  |
P5-01120 | Pre-procedure image acquisition | R-40FB9 |
P5-01121 | Preoperative image acquisition | R-40FB9 |
P5-01130 | Intra-procedure image acquisition | R-40FBA |
P5-01131 | Intra-operative image acquisition | R-40FBA |
P5-01140 | Post-procedure image acquisition | R-422A4 |
P5-01141 | Post-operative image acquisition | R-422A4 |
P5-01142 | Image acquisition following first cardiopulmonary bypass | R-422A4 |
P5-01143 | Image acquisition following second cardiopulmonary bypass | R-422A4 |
P5-01144 | Image acquisition following third cardiopulmonary bypass | R-422A4 |
P5-01200 | Image acquisition during stress procedure | R-40FBA |
P5-01201 | Image acquisition at baseline | F-01602 |
P5-01202 | Pre-stress image acquisition | F-01602 |
P5-01203 | Mid-stress image acquisition | F-05019 |
P5-01204 | Peak-stress image acquisition | F-05028 |
P5-01205 | Image acquisition during recovery | F-05018 |
P5-01300 | Image acquisition after drug administration | F-05019 |
P5-01310 | Image acquisition at user-defined dobutamine dose | F-05019 |
P5-01311 | Image acquisition at low-dose dobutamine | F-05019 |
P5-01312 | Image acquisition at mid-dose dobutamine | F-05019 |
P5-01313 | Image acquisition at peak dose dobutamine | F-05019 |
P5-01314 | Image acquisition at dobutamine 5 mcg/kg/min | F-05019 |
P5-01315 | Image acquisition at dobutamine 10 mcg/kg/min | F-05019 |
P5-01316 | Image acquisition at dobutamine 20 mcg/kg/min | F-05019 |
P5-01317 | Image acquisition at dobutamine 30 mcg/kg/min | F-05019 |
P5-01318 | Image acquisition at dobutamine 40 mcg/kg/min | F-05019 |
P5-01319 | Image acquisition at dobutamine 50 mcg/kg/min | F-05019 |
P5-0131A | Image at dobutamine 40 mcg/kg/min plus atropine | F-05019 |
P5-0131B | Image acquisition at dobutamine 50 mcg/kg/min plus atropine | F-05019 |
P5-01323 | Image acquisition at peak Arbutamine dose | F-05028 |
P5-01333 | Image acquisition at peak dipyridamole | F-05028 |
P5-01341 | Image acquisition after nitroglycerin | F-05019 |
P5-01342 | Image acquisition after amyl nitrite | F-05019 |
P5-01343 | Image acquisition after adenosine | F-05019 |
P5-B301F | Limited M-mode only echocardiography | P5-B3000 |
P5-B303F | Limited Doppler only echocardiography | P5-B3000 |
P5-B3051 | Maximal stress echocardiography | P5-B3050 |
P5-B3052 | Submaximal stress echocardiography | P5-B3050 |
P5-B3053 | Treadmill exercise stress echocardiography | P5-B3050 |
P5-B3054 | Bruce treadmill stress echocardiography | P2-7131A |
P5-B3055 | Modified Bruce treadmill stress echocardiography | P2-7131B |
P5-B3056 | Naughton treadmill stress echocardiography | P2-713A0 |
P5-B3058 | Bicycle exercise stress echocardiography | P2-31102 |
P5-B3060 | Echocardiography with administered drug stress | P2-31107 |
P5-B3061 | Dobutamine stress echocardiography | P2-31108 |
P5-B3062 | High dose dobutamine stress echocardiography | P2-31108 |
P5-B3063 | Low dose dobutamine stress echocardiography | P2-31108 |
P5-B3065 | Arbutamine stress echocardiography | P2-31107 |
P5-B3066 | Dipyridamole stress echocardiography | P2-3110A |
P5-B3070 | Cardiac pacing echocardiography | P2-3110B |
P5-B3081 | Adult echocardiography | P5-B3004 |
P5-B3084 | Upright echocardiography | P5-B3004 |
P5-B3085 | Supine echocardiography | P5-B3004 |
P5-B3091 | Contrast left ventricular opacification echocardiography | P5-B3090 |
P5-B3092 | Contrast perfusion echocardiography | P5-B3090 |
P5-B3093 | Contrast Doppler enhancement echocardiography | P5-B3090 |
P5-B3191 | 2D complete echocardiography | P5-B3004 |
P5-B3192 | Limited 2D only echocardiography | P5-B3004 |
F-F7102 | Valsalva maneuver | R-40928 |
L-8061A | Sterling pig breed | L-8063D |
L-8061F | Black Slavonian pig breed | L-8B151 |
L-807E1 | Bizanian Hound dog breed | L-807E3 |
L-80B03 | Rideau Arcott sheep breed | L-80B24 |
L-8BC43 | Beefalo bison X cattle breed | L-8B949 |
L-8BC44 | Beefalo bison X cattle breed | L-801E8 |
R-4041B | Hypokinesis | F-32056 |
F-32056 | Mild hypokinesis | R-00327 |
P5-B3009 | Exercise stress echocardiography | P5-B3050 |
R-10218 | right anterior oblique | R-40985 |
R-10222 | sagittal | G-A145 |

# K Relevant Patient Information Templates (Normative)

The following templates are appropriate to use as Root templates for the Relevant Patient Information Query Service Class:

- TID 9007 “General Relevant Patient Information”

- TID 9000 “Relevant Patient Information for Breast Imaging”

- TID 9001 “Gynecological History”

- TID 9002 “Medication, Substance, Environmental Exposure”

- TID 9003 “Previous Procedure”

- TID 9004 “Indicated Problem”

- TID 9005 “Risk Factor”

- TID 9006 “Obstetric History”

- TID 3802 “Cardiovascular Patient History”

# L Correspondence of Anatomic Region Codes and Body Part Examined Defined Terms

This Annex defines a correspondence between the SNOMED codes used in context groups for Anatomic Region Sequence (0008,2218) and Body Part Examined (0018,0015), as well as providing a list of the Defined Terms for Body Part Examined (0018,0015), for human use in Table L-1 and for animal use in Table L-2.

Table L-1. Corresponding SNOMED Terms for Human Use

SNOMED Code Value | Code Meaning | Body Part Examined
--- | --- | ---
T-D4000 | Abdomen | ABDOMEN
R-FAB57 | Abdomen and Pelvis | ABDOMENPELVIS
T-42500 | Abdominal aorta |
T-41070 | Abdominal aorta and its branches |
T-15420 | Acromioclavicular joint |
T-B3000 | Adrenal gland | ADRENAL
T-15750 | Ankle joint | ANKLE
T-48503 | Anomalous pulmonary vein |
T-49215 | Antecubital vein |
T-48440 | Anterior cardiac vein |
T-45530 | Anterior communicating artery |
T-45730 | Anterior spinal artery |
T-47700 | Anterior tibial artery |
T-59490 | Anus, rectum and sigmoid colon |
T-42000 | Aorta | AORTA
T-42300 | Aortic arch |
D3-81922 | Aortic fistula |
T-32602 | Apex of left ventricle |
T-280A0 | Apex of Lung |
T-32502 | Apex of right ventricle |
T-41000 | Artery |
T-42100 | Ascending aorta |
T-32100 | Atrium |
T-D8100 | Axilla | AXILLA
T-47100 | Axillary Artery |
T-49110 | Axillary vein |
T-48340 | Azygos vein |
T-D2100 | Back | BACK
A-00203 | Baffle |
T-45800 | Basilar artery |
T-60610 | Bile duct |
T-74000 | Bladder | BLADDER
T-DD123 | Bladder and urethra |
T-D00AB | Body conduit |
T-49424 | Boyd's perforating vein |
T-47160 | Brachial artery |
T-49350 | Brachial vein |
T-A0100 | Brain | BRAIN
T-04000 | Breast | BREAST
T-D6500 | Broad ligament |
T-26000 | Bronchus | BRONCHUS
T-D1206 | Buccal region of face |
T-D2600 | Buttock | BUTTOCK
T-12770 | Calcaneus | CALCANEUS
T-D9440 | Calf of leg | CALF
T-72100 | Calyx |
T-45010 | Carotid Artery | CAROTID
T-49240 | Cephalic vein |
T-A6000 | Cerebellum | CEREBELLUM
T-45510 | Cerebral artery |
T-11501 | Cervical spine | CSPINE
T-D00F7 | Cervico-thoracic spine | CTSPINE
T-83200 | Cervix | CERVIX
T-D1206 | Cheek | CHEEK
T-D3000 | Chest | CHEST
R-FAB55 | Chest and Abdomen | CHESTABDOMEN
R-FAB56 | Chest, Abdomen and Pelvis | CHESTABDPELVIS
T-45526 | Circle of Willis | CIRCLEOFWILLIS
T-12310 | Clavicle | CLAVICLE
T-11BF0 | Coccyx | COCCYX
T-59300 | Colon | COLON
D4-31320 | Common atrium |
T-45100 | Common carotid artery |
T-46710 | Common iliac artery |
T-48920 | Common iliac vein |
D4-31120 | Common ventricle |
D4-32504 | Congenital coronary artery fistula to left atrium |
D4-32506 | Congenital coronary artery fistula to left ventricle |
D4-32509 | Congenital coronary artery fistula to right atrium |
D4-32510 | Congenital coronary artery fistula to right ventricle |
D3-40208 | Congenital pulmonary arteriovenous fistula |
T-AA200 | Cornea | CORNEA
T-43000 | Coronary artery | CORONARYARTERY
T-48410 | Coronary sinus |
T-A0193 | Cranial venous system |
T-42400 | Descending aorta |
T-49429 | Dodd's perforating vein |
T-58200 | Duodenum | DUODENUM
T-AB000 | Ear | EAR
T-15430 | Elbow joint | ELBOW
T-41000 | Endo-arterial |
T-32000 | Endo-cardiac |
T-56000 | Endo-esophageal |
T-21300 | Endo-nasal |
T-23050 | Endo-nasopharyngeal |
T-59600 | Endo-rectal |
T-71000 | Endo-renal |
T-73000 | Endo-ureteric |
T-75000 | Endo-urethral |
T-82000 | Endo-vaginal |
T-40000 | Endo-vascular |
T-48000 | Endo-venous |
T-74250 | Endo-vesical |
T-D0010 | Entire body | WHOLEBODY
T-D4200 | Epigastric region |
T-56000 | Esophagus | ESOPHAGUS
T-DD163 | Esophagus, stomach and duodenum |
T-AB200 | External auditory canal |
T-45200 | External carotid artery |
T-46910 | External iliac artery |
T-48930 | External iliac vein |
T-D0300 | Extremity | EXTREMITY
T-AA000 | Eye | EYE
T-AA810 | Eyelid | EYELID
T-D1200 | Face | FACE
T-45240 | Facial artery |
T-11196 | Facial bones |
T-47400 | Femoral artery |
T-49410 | Femoral vein |
T-12710 | Femur | FEMUR
T-D8800 | Finger | FINGER
T-D2310 | Flank |
T-15200 | Fontanel of skull |
T-D9700 | Foot | FOOT
T-D8500 | Forearm |
T-63000 | Gallbladder | GALLBLADDER
T-48820 | Gastric vein |
T-47490 | Genicular artery |
T-D2600 | Gluteal region |
T-48420 | Great cardiac vein |
T-D8700 | Hand | HAND
T-D1100 | Head | HEAD
T-D1000 | Head and Neck | HEADNECK
T-32000 | Heart | HEART
T-46420 | Hepatic artery |
T-48720 | Hepatic vein |
T-15710 | Hip joint | HIP
T-12410 | Humerus | HUMERUS
T-4942A | Hunterian perforating vein |
T-D4240 | Hypogastric region |
T-D1212 | Hypoglossal |
T-55300 | Hypopharynx |
T-58600 | Ileum | ILEUM
T-41040 | Iliac arterial system |
T-46700 | Iliac artery |
T-12340 | Ilium | ILIUM
T-48470 | Inferior cardiac vein |
T-48540 | Inferior left pulmonary vein |
T-46520 | Inferior mesenteric artery |
T-48520 | Inferior right pulmonary vein |
T-48710 | Inferior vena cava |
T-D7000 | Inguinal region |
T-46010 | Innominate artery |
T-48620 | Innominate vein |
T-AB959 | Internal Auditory Canal | IAC
T-45300 | Internal carotid artery |
T-46740 | Internal iliac artery |
T-48170 | Internal jugular vein |
T-46200 | Internal mammary artery |
T-D4010 | Intra-abdominal |
G-A15A | Intra-articular |
T-56000 | Intra-esophageal |
T-D6221 | Intra-pelvic |
T-D3000 | Intra-thoracic |
T-D1400 | Intracranial |
T-D1213 | Jaw region | JAW
T-58400 | Jejunum | JEJUNUM
T-15001 | Joint |
D4-31052 | Juxtaposed atrial appendage |
T-71000 | Kidney | KIDNEY
T-D9200 | Knee | KNEE
T-45410 | Lacrimal artery |
T-45416 | Lacrimal artery of right eye |
T-59000 | Large intestine |
T-24100 | Larynx | LARYNX
T-32300 | Left atrium |
T-32310 | Left auricular appendage |
T-47420 | Left femoral artery |
T-D4211 | Left hypochondriac region |
T-D7020 | Left inguinal region |
T-D4140 | Left lower quadrant of abdomen |
T-D2340 | Left lumbar region |
T-44400 | Left pulmonary artery |
T-D4130 | Left upper quadrant of abdomen |
T-32600 | Left ventricle |
T-32640 | Left ventricle inflow |
D4-31022 | Left ventricle outflow chamber |
T-32650 | Left ventricle outflow tract |
T-45230 | Lingual artery |
T-62000 | Liver | LIVER
T-04003 | Lower inner quadrant of breast |
T-D9400 | Lower leg | LEG
T-04005 | Lower outer quadrant of breast |
T-46960 | Lumbar artery |
T-D2300 | Lumbar region |
T-11503 | Lumbar spine | LSPINE
T-D00F9 | Lumbo-sacral spine | LSSPINE
T-40230 | Lumen of blood vessel |
T-28000 | Lung | LUNG
T-11180 | Mandible | JAW
T-11133 | Mastoid bone |
T-11170 | Maxilla | MAXILLA
T-D3300 | Mediastinum | MEDIASTINUM
T-46500 | Mesenteric artery |
T-4884A | Mesenteric vein |
T-51000 | Mouth | MOUTH
T-11149 | Nasal bone |
T-2300C | Naso pharynx |
T-D1600 | Neck | NECK
R-FAB52 | Neck and Chest | NECKCHEST
R-FAB53 | Neck, Chest and Abdomen | NECKCHESTABDOMEN
R-FAB54 | Neck, Chest, Abdomen and Pelvis | NECKCHESTABDPELV
T-21000 | Nose | NOSE
T-45250 | Occipital artery |
T-48214 | Occipital vein |
T-D4450 | Omental bursa |
T-D4600 | Omentum |
T-45400 | Ophthalmic artery |
T-11102 | Optic canal |
T-D14AE | Orbital structure | ORBIT
T-87000 | Ovary | OVARY
T-65000 | Pancreas | PANCREAS
T-65010 | Pancreatic duct |
T-65600 | Pancreatic duct and bile duct systems |
T-22000 | Paranasal sinus |
T-D3136 | Parasternal |
T-B7000 | Parathyroid |
T-61100 | Parotid gland | PAROTID
T-12730 | Patella | PATELLA
D4-32012 | Patent ductus arteriosus |
T-D6000 | Pelvis | PELVIS
R-FAB58 | Pelvis and lower extremities |
T-91000 | Penis | PENIS
T-D2700 | Perineum |
T-47630 | Peroneal artery |
T-55000 | Pharynx | PHARYNX
T-20101 | Pharynx and larynx |
T-47500 | Popliteal artery |
T-D9310 | Popliteal fossa |
T-48810 | Portal vein |
T-45320 | Posterior communicating artery |
T-49535 | Posterior medial tributary |
T-47600 | Posterior tibial artery |
T-F7001 | Primitive aorta |
T-F7040 | Primitive pulmonary artery |
T-9200B | Prostate | PROSTATE
T-44000 | Pulmonary artery |
D4-33142 | Pulmonary artery conduit |
T-32190 | Pulmonary chamber of cor triatriatum |
T-48581 | Pulmonary vein |
D4-33512 | Pulmonary vein confluence |
D4-33514 | Pulmonary venous atrium |
T-47300 | Radial artery |
T-59600 | Rectum | RECTUM
T-46600 | Renal artery |
T-72000 | Renal pelvis |
T-48740 | Renal vein |
T-D4900 | Retroperitoneum |
T-11300 | Rib | RIB
T-32200 | Right atrium |
T-32210 | Right auricular appendage |
T-47410 | Right femoral artery |
T-D4212 | Right hypochondriac region |
T-D7010 | Right inguinal region |
T-D4120 | Right lower quadrant of abdomen |
T-D2342 | Right lumbar region |
T-44200 | Right pulmonary artery |
T-D4110 | Right upper quadrant of abdomen |
T-32500 | Right ventricle |
T-32540 | Right ventricle inflow |
D4-31032 | Right ventricle outflow chamber |
T-32550 | Right ventricle outflow tract |
T-15680 | Sacroiliac joint |
T-11AD0 | Sacrum | SSPINE
T-D930A | Saphenofemoral junction |
T-49530 | Saphenous vein |
T-D1160 | Scalp | SCALP
T-12280 | Scapula | SCAPULA
T-AA110 | Sclera | SCLERA
T-98000 | Scrotum | SCROTUM
T-D1460 | Sella turcica |
T-12980 | Sesamoid bones of foot |
T-D2220 | Shoulder | SHOULDER
T-59470 | Sigmoid colon |
T-11100 | Skull | SKULL
T-58000 | Small intestine |
T-A7010 | Spinal cord |
T-D0146 | Spine | SPINE
T-C3000 | Spleen | SPLEEN
T-46460 | Splenic artery |
T-48890 | Splenic vein |
T-15610 | Sternoclavicular joint |
T-11210 | Sternum | STERNUM
T-57000 | Stomach | STOMACH
T-46100 | Subclavian artery |
T-48330 | Subclavian vein |
T-D4210 | Subcostal |
T-D1603 | Submandibular area |
T-61300 | Submandibular gland | SUBMANDIBULAR
T-D3213 | Subxiphoid |
T-45270 | Superficial temporal artery |
T-48530 | Superior left pulmonary vein |
T-46510 | Superior mesenteric artery |
T-48510 | Superior right pulmonary vein |
T-45210 | Superior thyroid artery |
T-48610 | Superior vena cava |
T-D1620 | Supraclavicular region of neck |
T-D4240 | Suprapubic region |
T-11218 | Suprasternal notch |
T-44007 | Systemic collateral artery to lung |
D4-33516 | Systemic venous atrium |
T-15770 | Tarsal joint |
T-15290 | Temporomandibular joint | TMJ
T-94000 | Testis | TESTIS
T-D9100 | Thigh | THIGH
T-42070 | Thoracic aorta |
T-11502 | Thoracic spine | TSPINE
T-D00F8 | Thoraco-lumbar spine | TLSPINE
T-D3000 | Thorax |
T-D8810 | Thumb | THUMB
T-C8000 | Thymus | THYMUS
T-B6000 | Thyroid | THYROID
T-12740 | Tibia | TIBIA
T-12701 | Tibia and fibula | TIBIAFIBULA
T-D9800 | Toe | TOE
T-53000 | Tongue | TONGUE
T-25000 | Trachea | TRACHEA
T-DD006 | Trachea and bronchus |
D4-31400 | Truncus arteriosus communis |
T-46400 | Truncus coeliacus |
T-12430 | Ulna | ULNA
T-47200 | Ulnar artery |
T-F1810 | Umbilical artery |
T-D4230 | Umbilical region |
T-48817 | Umbilical vein |
T-D8200 | Upper arm | ARM
T-04002 | Upper inner quadrant of breast |
T-04004 | Upper outer quadrant of breast |
T-70010 | Upper urinary tract |
T-73000 | Ureter | URETER
T-75000 | Urethra | URETHRA
T-83000 | Uterus | UTERUS
T-88920 | Uterus and fallopian tubes |
T-82000 | Vagina | VAGINA
A-04140 | Vascular graft |
T-48000 | Vein |
T-48003 | Venous network |
T-32400 | Ventricle |
T-45700 | Vertebral artery |
T-11011 | Vertebral column and cranium |
T-81000 | Vulva | VULVA
T-15460 | Wrist joint | WRIST
T-11166 | Zygoma | ZYGOMA

### Note

In prior versions of this table, different codes were used for some concepts; see PS3.16-2011.

Table L-2. Corresponding SNOMED Terms for Animals

SNOMED Code Value | Code Meaning | Body Part Examined
T-D4000 | Abdomen | ABDOMEN
T-D8030 | All legs | LEGS
T-1531B | Atlantal-axial joint | ATLANTOAXIAL
T-15311 | Atlanto-occipital joint | ATLANTOOCCIPITAL
T-74000 | Bladder | BLADDER
T-12771 | Calcaneal tubercle |
T-D0788 | Carpus | CARPUS
T-11501 | Cervical spine | CSPINE
T-D00F7 | Cervico-thoracic spine | CTSPINE
T-D3000 | Chest | CHEST
R-FAB55 | Chest and Abdomen | CHESTABDOMEN
T-11B02 | Coccygeal vertrebrae | TAIL
T-59300 | Colon | COLON
T-D0310 | Digit | DIGIT
T-110A2 | Distal phalanx | DISTALPHALANX
T-15430 | Elbow joint | ELBOW
T-D0010 | Entire body | WHOLEBODY
T-56000 | Esophagus | ESOPHAGUS
T-12710 | Femur | FEMUR
T-D8640 | Fetlock of forelimb | FOREFETLOCK
T-D9540 | Fetlock of hindlimb | HINDFETLOCK
T-12750 | Fibula | FIBULA
T-D04F2 | Forefoot | FOREFOOT
T-22200 | Frontal sinus | FRONTALSINUS
T-D9713 | Hindfoot | HINDFOOT
T-15710 | Hip joint | HIP
T-12410 | Humerus | HUMERUS
T-11503 | Lumbar spine | LSPINE
T-D00F9 | Lumbo-sacral spine | LSSPINE
T-11180 | Mandible | JAW
T-54170 | Mandibular dental arch |
T-540EE | Mandibular incisor teeth |
T-54160 | Maxillary dental arch |
T-540ED | Maxillary incisor teeth |
T-1254D | Metacarpus | METACARPUS
T-12847 | Metatarsus | METATARSUS
T-22000 | Nasal sinus |
T-12450 | Navicular of forefoot | FORENAVICULAR
T-127EC | Navicular of hindfoot | HINDNAVICULAR
T-D14AD | Orbital region |
T-D8650 | Pastern of forefoot | FOREPASTERN
T-D9550 | Pastern of hindfoot | HINDPASTERN
T-12730 | Patella | PATELLA
T-D6000 | Pelvis | PELVIS
T-12420 | Radius | RADIUS
T-12403 | Radius and ulna | RADIUSULNA
T-11AD0 | Sacrum | SSPINE
T-D2220 | Shoulder | SHOULDER
T-11100 | Skull | SKULL
T-15728 | Stiffle | STIFLE
T-11096 | Tarsus | TARSUS
T-11502 | Thoracic spine | TSPINE
T-D00F8 | Thoraco-lumbar spine | TLSPINE
T-12740 | Tibia | TIBIA
T-12701 | Tibia and fibula | TIBIAFIBULA
T-50110 | Upper gastro-intestinal tract | UGITRACT
T-12430 | Ulna | ULNA
T-75000 | Urethra | URETHRA
T-70000 | Urinary tract | URINARYTRACT
T-D8040 | Wing | WING

### Note

In prior versions of this table, different codes were used for some concepts; see PS3.16-2011.

# M German Language Meanings of Selected Codes Used in The DCMR (Normative)

Table M-1. German Language Meanings of Selected Codes

Coding Scheme Designator | Code Value | Code Meaning English Language | Code Meaning German Language
--- | --- | --- | ---
LN | 11528-7 | Radiology Report | Radiologischer Befundbericht
DCM | 121066 | Prior Procedure Descriptions | Frühere Untersuchungen
DCM | 111532 | Pregnancy Status | Schwangerschaft
DCM | 121109 | Indications for Procedure | Indikationen für die Untersuchung
DCM | 123014 | Target Region | Körperregion
DCM | 121064 | Current Procedure Descriptions | Untersuchungstechnik
DCM | 111060 | Study Date | Datum der Untersuchung
DCM | 111061 | Study Time | Zeitpunkt der Untersuchung
DCM | 110180 | Study Instance UID | Study Instance UID
DCM | 121060 | History | Krankengeschichte
DCM | 121062 | Request | Fragestellung
DCM | 121071 | Finding | Beschreibung
DCM | 121072 | Impressions | Wertungen
DCM | 121075 | Recommendation | Empfehlung
DCM | 113850 | Irradiation Authorizing | Indikationsstellender Arzt
DCM | 113921 | Radiation Exposure | Strahlenexposition
DCM | 113922 | Radioactive Substance Administered | Verabreichter radioaktiver Stoff
DCM | 113923 | Radiation Exposure and Protection Information | Informationen zum Strahlenschutz