TID 3910 Flow Quantification
Prev | CT/MR Cardiovascular Analysis Report Templates | Next

### TID 3910 Flow Quantification

Contains the flow quantification measurement results

Type:

  Extensible

Order:

  Significant

Table TID 3910. Flow Quantification

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  | CONTAINS | CONTAINER | EV (121070, DCM, "Findings") | 1 | M |  |
2 | > | HAS CONCEPT MOD | CODE | EV (111004, DCM, "Analysis Performed") | 1 | M |  | EV (122604, DCM, "Flow Quantification")
3 | > | HAS OBS CONTEXT | INCLUDE | DTID 3929 “Cardiovascular Analysis Observation Context” | 1 | U |  |
4 | > | HAS OBS CONTEXT | DATETIME | EV (G-D321, SRT, "Start Time") | 1 | M |  |
5 | > | HAS OBS CONTEXT | DATETIME | EV (G-D320, SRT, "Stop Time") | 1 | M |  |
6 | > | CONTAINS | INCLUDE | DTID 3990 “Two Dimensional Measurement Graph” | 1 | U |  | $MeasurementGraph = EV (122667, DCM, "Blood velocity vs. time of cardiac cycle") $X-Concept = EV (122666, DCM, "Time relative to R-wave peak") $Y-Concept = EV (F-0319E, SRT, "Arterial Velocity") $X-AxisUnits = DT (ms, UCUM, "ms") $Y-AxisUnits = DT (cm/s, UCUM, "cm/s")
7 | > | CONTAINS | NUM | EV (122642, DCM, "Velocity Encoding Minimum Value") | 1 | U |  | UNITS = DT (cm/s, UCUM, "cm/s")
8 | > | CONTAINS | NUM | EV (122643, DCM, "Velocity Encoding Maximum Value") | 1 | U |  | UNITS = DT (cm/s, UCUM, "cm/s")
9 | > | CONTAINS | CONTAINER | EV (125007, DCM, "Measurement Group") | 1-n | M |  |
10 | >> | HAS CONCEPT MOD | TEXT | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | XOR row 11 |
11 | >> | HAS CONCEPT MOD | CODE | EV (G-C0E3, SRT, "Finding Site") | 1 | MC | XOR row 10 |
12 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122207, DCM, "Blood velocity, peak") $Units = DT (cm/s, UCUM, "cm/s")
13 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122205, DCM, "Blood velocity, mean") $Units = DT (cm/s, UCUM, "cm/s")
14 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (F-39200, SRT, "Blood Flow") $Derivation = EV (R-00317, SRT, "Mean") $Units = DT (ml/s, UCUM, "ml/s")
15 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-2 | U |  | $Measurement = EV (F-39200, SRT, "Blood Flow") $ModType = EV (G-C048, SRT, "Direction of flow") $ModValue = DCID 12221 “Flow Direction” $Units = DT (ml/s, UCUM, "ml/s")
16 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122645, DCM, "Net Forward Volume") $Units = DT (ml, UCUM, "ml")
17 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1 | U |  | $Measurement = EV (122645, DCM, "Net Forward Volume") $ModType = EV (121425, DCM, "Index") $ModValue = DT (8277-6, LN, "BSA") $Units = DT (ml/m2, UCUM, "ml/m2")
18 | >> | CONTAINS | INCLUDE | DTID 300 “Measurement” | 1-n | U |  | $Measurement = EV (G-0366, SRT, "Vessel Lumen Cross-Sectional Area") $Derivation = DCID 3488 “Min/Max/Mean” $Units = DT (mm2, UCUM, "mm2")

Prev | Up | Next
TID 3909 Best Illustration of Findings | Home | TID 3911 Plaque Properties