TID 3301 Stress Test Procedure Description
Prev | Stress Testing Report Templates | Next

### TID 3301 Stress Test Procedure Description

Type:

  Extensible

Order:

  Significant

Table TID 3301. Stress Test Procedure Description

| NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint
--- | --- | --- | --- | --- | --- | --- | --- | ---
1 |  |  | CONTAINER | DT (121064, DCM, "Current Procedure Descriptions") | 1 | M |  |
2 | > | CONTAINS | CODE | DT (109056, DCM, "Stress Protocol") | 1 | U |  | BCID 3261 “Stress Protocols”
3 | > | CONTAINS | TEXT | DT (109056, DCM, "Stress Protocol") | 1 | U |  |
4 | > | CONTAINS | CODE | DT (10:11345, MDC, "Lead System") | 1 | U |  | BCID 3263 “Electrode Placement Values”
5 | > | CONTAINS | CODE | DT (A-17200, SRT, "Exerciser Device") | 1 | U |  | BCID 3203 “Exerciser Device”
6 | > | CONTAINS | CODE | DT (G-C11C, SRT, "Pharmacological Stress Agent") | 1 | MC | IFF Pharmacological Stress used | BCID 3204 “Stress Agents”
7 | > | CONTAINS | CONTAINER | EV (122700, DCM, "Indications for Pharmacological Stress") | 1 | MC | IFF Pharmacological Stress used |
8 | >> | CONTAINS | CODE | EV (121071, DCM, "Finding") | 1-n | U |  | DCID 3205 “Indications for Pharmacological Stress Test”
9 | > | CONTAINS | CODE | DT (P0-0099A, SRT, "Imaging procedure") | 1 | MC | IFF imaging used in procedure | DCID 3206 “Non-invasive Cardiac Imaging Procedures”
10 | > | CONTAINS | CODE | DT (125203, DCM, "Acquisition Protocol") | 1 | UC | IFF Nuclear imaging | DCID 3110 “Nuclear Cardiology Protocols”
11 | > | CONTAINS | CODE | DT (125203, DCM, "Acquisition Protocol") | 1 | UC | IFF PET imaging | DCID 3106 “PET Cardiology Protocols”
12 | > | CONTAINS | TEXT | DT (121141, DCM, "Image Type") | 1 | UC | IFF Nuclear or PET imaging | STATIC, DYNAMIC, or GATED. See note.
13 | > | CONTAINS | CODE | DT (RID11248, RADLEX, "Cardiac Gating") | 1 | U |  | DCID 3104 “Cardiac Synchronization Technique”
14 | > | CONTAINS | CODE | DT (125203, DCM, "Acquisition Protocol") | 1 | UC | IFF Contrast echocardiography | DT (P5-B3008, SRT, "Contrast echocardiography")
15 | > | CONTAINS | CODE | DT (113743, DCM, "Patient Orientation") | 1 | U |  | DCID 19 “Patient Orientation”
16 | >> | HAS CONCEPT MOD | CODE | EV (113744, DCM, "Patient Orientation Modifier") | 1 | U |  | DCID 20 “Patient Orientation Modifier”
17 | > | CONTAINS | TEXT | DT (121065, DCM, "Procedure Description") | 1 | U |  |
18 | > | CONTAINS | DATETIME | DT (122701, DCM, "Procedure Time Base") | 1 | U |  | See note.

Content Item Descriptions

Row 12 | Image Type may be copied from the NM Image SOP Instance attribute Image Type (0008,0008) value 3, or from the PET Image SOP Instance attribute Series Type (0054,1000).
Row 18 | The Procedure Time Base is the time from which elapsed times are measured. The Study Time (0008,0030) may include the patient prep period, while this Procedure Time Base is typically established when baseline data collection begins.

Prev | Up | Next
Stress Testing Report Templates | Home | TID 3303 Stress Test Phase Data