A.30 RT Brachy Treatment Record IOD
Prev | A Composite Information Object Definitions (Normative) | Next

## A.30 RT Brachy Treatment Record IOD

### A.30.1 RT Brachy Treatment Record IOD Description

The focus for this Radiotherapy Brachy Treatment Record IOD (RT Brachy Treatment Record IOD) is to address the requirements for transfer of treatment session reports generated by a treatment verification system during a course of Brachytherapy treatment, with optional cumulative summary information. It may also be used for transfer of treatment information during delivery.

### A.30.2 RT Brachy Treatment Record IOD Entity-Relationship Model

The E-R model for the RT Brachy Treatment Record IOD is illustrated in Figure A.30-1.

Figure A.30-1. DICOM RT Brachy Treatment Record IOD Information Model

### A.30.3 RT Brachy Treatment Record IOD Module Table

Table A.30.3-1. RT Brachy Treatment Record IOD Modules

IE | Module | Reference | Usage
--- | --- | --- | ---
Patient | Patient | C.7.1.1 | M
Clinical Trial Subject | C.7.1.3 | U
Study | General Study | C.7.2.1 | M
Patient Study | C.7.2.2 | U
Clinical Trial Study | C.7.2.3 | U
Series | RT Series | C.8.8.1 | M
Clinical Trial Series | C.7.3.2 | U
Equipment | General Equipment | C.7.5.1 | M
Treatment Record | RT General Treatment Record | C.8.8.17 | M
RT Patient Setup | C.8.8.12 | U
RT Treatment Machine Record | C.8.8.18 | M
Measured Dose Reference Record | C.8.8.19 | U
Calculated Dose Reference Record | C.8.8.20 | U
RT Brachy Session Record | C.8.8.22 | M
RT Treatment Summary Record | C.8.8.23 | U
SOP Common | C.12.1 | M
Common Instance Reference | C.12.2 | U

### Note

The Curve Module was previously included in this IOD but has been retired. See PS3.3-2004.

Prev | Up | Next
A.29 RT Beams Treatment Record IOD | Home | A.31 RT Treatment Summary Record IOD